HCL Dynamic Spiking Protocol


 Hector McKinney
 2 years ago
 Views:
Transcription
1 ELI LILLY AND COMPANY TIPPECANOE LABORATORIES LAFAYETTE, IN Revisio 2.0
2 TABLE OF CONTENTS REVISION HISTORY REVISION REVISION OVERVIEW DEFINITIONS EQUIPMENT EQUIPMENT CALIBRATION AND STANDARDS HCL CEMS ELEMENT OF OPPORTUNITY CEMS FLOW MEASUREMENT SYSTEM Lamiar flow elemet Flow by dilutio GAS MANIFOLD HCL CALIBRATION GAS FOR DYNAMIC SPIKING DYNAMIC SPIKING RANGE TYPICAL DYNAMIC SPIKING SETUP DYNAMIC SPIKING PROCEDURE COLLECTION OF BASELINE DATA COLLECTION OF TEST DATA DATA CALCULATIONS Baselie data Dyamic Spikig Test ru data Referece HCL cocetratio calculatios Liear Regressio CRITERIA FOR ACCEPTANCE BIAS CORRECTION TABLES REFERENCES APPENDIXES...2
3 REVISION HISTORY. REVISION.0 Revisio.0, Tippecaoe Laboratories NONGMP Stadard Operatig Procedure, Eli Lilly ad Compay, December 8, REVISION 2.0 This revisio is a update of the origial Dyamic Spikig protocol, December The followig are key updates i revisio 2.0:. Recommedatio that all HCL QA gases be certified with a miimum tolerace of 5% 2. Discussio for the use of quatitative itroductio of the HCL calibratio gas ad subsequet total flow measuremet as method for determiig the referece HCL cocetratio durig each test ru. 3. Updated the dyamic spikig procedure descriptio for easier flow of iformatio 4. Updated the equatios to iclude percet relative stadard deviatio ad the calculatio for the referece HCL cocetratio whe usig quatitative itroductio of the HCL calibratio gas with subsequet total flow measuremet. 2
4 2 OVERVIEW Dyamic Spikig is procedure to establish the accuracy, precisio, ad bias of a moitorig system. Durig the dyamic spikig procedure, a certified calibratio gas (+/ 5%) is spiked ito the moitorig system while the system is samplig flue gas. It is importat that the spiked calibratio gas flow through the etire samplig system. The target ratio of flue gas to HCL calibratio gas is 9:, with a miimum requiremet of :. The HCL calibratio gas is spiked i three rages (low, mid, high) through the rage of HCL detector. A miimum of 30, oe miute averaged data poits should be collected for each of the three test rus. It is recommeded that the percet relative stadard deviatio of the data set, for each test level, be less tha or equal to 20%. If the flue gas has sigificat amouts of HCL preset, the data may be corrected for the baselie HCL cocetratio. This is performed by collectig at least 0, oe miute average HCL data poits prior to each test ru. The average of the te baselie HCL values will be used to compesate the subsequet test ru data for backgroud HCL cocetratios. It is importat that the cotrol device be ruig at steady state; alleviatig large fluctuatios i the backgroud HCL cocetratios. The HCL CEMS will provide the measured HCL values. These values are compared, usig liear regressio, to the kow or spiked HCL cocetratios. There are two methods for calculatig the kow HCL cocetratios: () Use of a elemet of opportuity ad (2) Quatitative itroductio of the HCL calibratio gas with total flow through the system. A elemet of opportuity ca be ay elemet ormally preset, i steady cocetratios, i the flue gas stream. Examples are O2, CO, NOx, SO2, CO2, ad moisture. Total flow through the system is also a example of a elemet of opportuity. The elemet of opportuity is measured durig the baselie data collectio, prior to the test ru. The average value durig this baselie collectio period is used to establish the baselie cocetratio, or flow value, without ay spikig of the calibratio gas. The elemet of opportuity is measured durig each test ru while the HCL calibratio gas is beig spiked. The elemets of opportuity values, collected durig each test ru are averaged. Kowig the cocetratio of the HCL calibratio gas, ad the dilutio of the elemet of opportuity due to the spikig, the kow value of the HCL calibratio gas i the measuremet stream ca be calculated. The kow value of the spiked HCL calibratio gas i the measuremet stream ca also be calculated by quatitatively itroducig the HCL calibratio gas, ad, measurig the total flow through the system. I this case, the HCL calibratio gas cocetratio, the flow rate of spiked HCL calibratio gas ito the measuremet stream, ad the total flow through the system are used to calculate the kow value of the spiked calibratio gas. As with the HCL CEMS data sets, it is recommeded that the percet relative stadard deviatio for the elemet of opportuity data sets be less tha or equal to 20%. Oce the data is collected, liear regressio is performed usig the kow HCL spike values ad the correlatig HCL CEMS measured values. Criteria for the HCL dyamic spikig are:. The correlatio coefficiet must be greater tha or equal to the slope of the regressio lie must be.0 +/
5 3. The itercept, of the regressio lie, must be less tha or equal to 5% of the istrumet spa. The HCL CEMS must meet the correlatio criteria. If either the slope ad/or itercept criteria are ot met, a bias may be applied to the HCL CEMS oe miute value. Performace Specificatio Z: Specificatios ad Test Procedures for the HCL Cotiuous Emissio Moitorig Systems at Statioary Sources should be as a referece documet for dyamic spikig requiremets ad calculatios. 4
6 3 DEFINITIONS Bias: The differece betwee the referece value ad the measured value. Calibratio Drift: The differece i the CEMS output readigs from the established referece value after a stated period of operatio durig which o uscheduled maiteace or adjustmet took place. Cotiuous Emissio Moitorig System: meas the total equipmet required for the determiatio of a gas cocetratio or emissio rate. The sample iterface, pollutat aalyzer, diluet aalyzer, ad data recorder are the major subsystems of the CEMS. Correlatio Coefficiet: determies the extet of a liear relatioship betwee two fields over a give period of time. Data Recorder: The portio of the CEMS that provides a record of aalyzer output, flags which idicate ormal operatio, ad flags idicatig abormal operatio. The data recorder may record other pertiet data such as effluet flow rates, ad various istrumet temperatures. Dyamic Spikig: a procedure used to documet the accuracy, precisio, ad bias of the moitorig system by quatitatively spikig a certified gas ito the pollutat gas stream. Elemet of Opportuity: Ay elemet, preset i the flue gas matrix (or total flow) that has little fluctuatio ad ca be used to determie the referece HCL cocetratio durig the dyamic spikig test rus. Gas Chromatograph: A aalytical istrumet which uses a carrier gas to trasport gas/vapor phase samples to a solid matrix for separatio, ad subsequet detectio by a aalyte specific detector. Gas maifold: Idividual, or series of flow cotrollers that allow cotrolled flow of a gas. Calibrated flow cotrollers ca be used to quatitatively cotrol the flow of gases. 5
7 Istrumet Measuremet Rage: The rage of HCl cocetratios the istrumet ca reliably measure from the lowest cocetratio to the highest. Itercept: value of the Y variable whe the X variable is equal to zero. Lamiar Flow Elemet: A device used to measure the flow gas through a system by measurig the pressure drop of the gas over a restrictio (forces parallel, or lamiar flow). Liear Regressio: a methodology used to fid a formula that ca be used to relate two variables that are liearly related. Respose Time: The time iterval betwee the start of a step chage i the system iput ad whe the pollutat aalyzer output reached 95% of the fial value. Sample Iterface: The portio of the CEMS used for oe or more of the followig: sample acquisitio, sample trasport, sample coditioig, or protectio of the moitor from the effects of stack gas. Slope: The rate of chage of Y relative to the chage i X. Upscale drift: meas the absolute differece betwee a highlevel calibratio gas ad the moitor respose, i uits of the applicable stadard. Zero drift: meas the absolute differece betwee a highlevel calibratio gas ad the moitor respose, i uits of the applicable stadard. 6
8 4 EQUIPMENT The followig list cotais typical equipmet required to perform the HCL dyamic spikig.. HCL CEMS 2. Referece Aalyzer(s) for the elemet of opportuity (examples are NOx, SO2, CO, CO2, O2) 3. Flow Measuremet Systems a. Lamiar flow elemet b. Gas chromatograph (for dilutio flow with a hydrocarbo) 4. Calibratio Equipmet a. Bubble Flow meter, maometer, or equivalet for primary flow calibratio b. Certified HCL Calibratio gas for spikig c. HCL calibratio gases for zero ad upscale drift check of the HCL CEMS d. Calibratio gas for the elemet(s) of opportuity CEMS calibratio e. Hydrocarbo gas for flow measuremet via dilutio samplig 5. Quatitative gas maifold, or equivalet, for itroductio of the HCL calibratio gas 6. Quatitative gas maifold, or equivalet, for the itroductio of the hydrocarbo for dilutio flow (if used) 7. Various fittigs ad tubig 7
9 5 EQUIPMENT CALIBRATION AND STANDARDS Prior to performig the HCL dyamic spikig, it is importat that all equipmet ad/or aalyzers are calibrated. This sectio provides guidace o the calibratio requiremets for all equipmet used durig the HCL dyamic spikig procedure. 5. HCL CEMS Prior to performig the dyamic spikig procedure, the HCL CEMS must pass the QA requiremets specified i performace specificatio Z: Specificatios ad Test Procedures For HCL Cotiuous Emissio Moitorig Systems at Statioary Sources. Typically, this would iclude passig the daily zero ad upscale calibratio drift check, ad for aual audits passig both the zero ad upscale drift check as well as a seve day drift test. All calibratio gases must be certified gases, with a recommeded tolerace ot to exceed +/ 5%. 5.2 ELEMENT OF OPPORTUNITY CEMS The moitorig system used to measure the elemet(s) of opportuity, other tha total flow, must pass a calibratio directly prior to use. After dyamic spikig ru, the zero ad upscale calibratio drift are checked. The drift requiremet, for both zero ad upscale, is 3% of the referece value, uless stated otherwise i the applicable referece method. If either the zero or upscale calibratio check fails the drift limit after a dyamic spikig ru, the ru must be repeated. 5.3 FLOW MEASUREMENT SYSTEM Total flow may be used as a elemet of opportuity, or, as part of measurig the quatitative itroductio of the HCL calibratio gas for the dyamic spikig tests. There are several methods that ca be used to provide a total flow measuremet. This protocol outlies two approaches, lamiar flow elemet ad flow by dilutio. Other total flow measuremet systems ca be used; however, specific protocol describig the methodology must be used Lamiar flow elemet Prior to the dyamic spikig procedure, the lamiar flow elemet is calibrated agaist a primary flow stadard, such as a bubble flow meter. The calibratio must iclude at least three poits throughout the expected measuremet rage. After each dyamic spikig test ru, the calibratio drift is checked at each of the three poits. The lamiar flow elemet must be recalibrated if ay of the drift check poits exceed 0% of the expected value (established durig the pretest calibratio). 8
10 5.3.2 Flow by dilutio Flow by dilutio is a method where a kow cocetratio of gas is quatitatively itroduced to the flue gas stream ad measured dowstream by a calibrated istrumet. For example, a kow cocetratio of propae ca be quatitatively itroduced to the flue gas stream ad the diluted propae measured dowstream. Prior to the dyamic spikig procedure, the gas chromatograph is calibrated, usig at least four poits through the expected measuremet rage. After the dyamic spikig procedure is complete, the gas chromatograph is checked for bias at each of the four calibratio levels. The bias should be less tha 5% of the expected value for each poit. Method 8 provides guidace for the calibratio procedure ad requiremets. Prior to the dyamic spikig procedure, the gas maifold (used to quatitatively itroduce propae, or similar hydrocarbo), which is typically a mass flow cotroller, is calibrated to a primary stadard. After each test ru, the gas maifold is checked for drift, ad adjusted if the drift exceeds 0% of the expected value (calibrated value prior to dyamic spikig procedure). Method 205 provides guidace for this procedure. 5.4 GAS MANIFOLD HCL may be quatitatively itroduced ito the flue gas stream, ad whe used i cojuctio with a total flow measuremet, ca provide the referece HCL spikig cocetratios for each dyamic spikig ru. Prior to the dyamic spikig procedure, the gas maifold (used to quatitatively itroduce HCL), which is typically a mass flow cotroller, is calibrated to a primary stadard. After each test ru, the gas maifold is checked for drift, ad adjusted if the drift exceeds 0% of the expected value (calibrated value prior to dyamic spikig procedure). Method 205 provides guidace for this procedure. 5.5 HCL CALIBRATION GAS FOR DYNAMIC SPIKING The HCL calibratio gas used for dyamic spikig must be a certified gas with a tolerace less tha or equal to 5%. 9
11 6 DYNAMIC SPIKING RANGE The flue gas stream must be spiked at a miimum of three separate levels (low, mid, high) throughout the rage of the HCL CEMS. The low level should be betwee 0 50%, the mid rage should be betwee 25 75%, ad the high rage should be betwee 50 00% of the spa of the istrumet. The dyamic spiked HCL cocetratios ca oly be used for oe rage (Table ). 7 TYPICAL DYNAMIC SPIKING SETUP Figure. Typical Dyamic Spikig Apparatus Dyamic Spikig Apparatus Sample Probe Heated lie iside sample umbillical Flue Gas Heated sample lie Heated Filter Pressure Cotrol Lamiar Flow Elemet Calibratio Maifold (Mass flow meters) ppm HCL i N2 HCL CEMS Gas Coditioer Referece Measuremet(s) 0
12 8 DYNAMIC SPIKING PROCEDURE This sectio describes the procedure used to perform HCL dyamic spikig. Prior to testig, it is importat that all aalyzers are calibrated accordig to sectio 6 of this documet. I additio, the cotrol device should be at a steady state (small fluctuatios of HCL or elemets of opportuity). 8. COLLECTION OF BASELINE DATA The baselie data will be used to compesate the measured HCl, durig the dyamic spikig ru, for backgroud cocetratios of HCL which may be preset durig ormal operatios. It is importat to collect all elemet of opportuity data as well as HCL backgroud data durig the baselie collectio period.. Collect a miimum of te, miute averaged HCL data poits prior to each dyamic spikig test ru. 2. Collect te, miute averaged data poits for the elemet(s) of opportuity to be used for calculatig the referece HCL cocetratio. The elemet of opportuity data set is collected simultaeous to the HCL baselie data, ad must be collected prior to each dyamic spikig test ru. 3. If usig total flow, collect a miimum of oe flow readig durig the baselie data collectio. 4. Prior to begiig the dyamic spike, esure that all baselie data (HCL, elemet of opportuity, flow) does ot exhibit large fluctuatios i cocetratio (steady state). If large fluctuatios exist, the baselie data collectio procedure must be repeated. 8.2 COLLECTION OF TEST DATA The HCL CEMS must be tested at a miimum of three levels through the measuremet rage of the istrumet (low, mid, high), Table. Prior to each dyamic spikig test ru, it is importat that the appropriate baselie data is collected (sectio 7.) ad that all equipmet is calibrated ad appropriate bias checks are performed (sectio 6).. Itroduce the HCL calibratio gas to the measuremet system ad allow the HCL CEMS values to reach a steady state. This may take up to 5 miutes.
13 2. Collect a miimum of 30, oe miute average HCL data poits. 3. Collect a miimum of 30, oe miute average elemet(s) of opportuity data poits. 4. If usig total flow through the system, collect at least five data poits durig the dyamic spikig test ru. 5. If usig quatitative itroductio of the HCL calibratio gas, collect at least five data poits for the calibratio gas flow rate ito the system durig the dyamic spikig test ru. 6. Before stoppig the dyamic spikig test ru, esure that the data sets are complete. It is recommeded that the percet stadard deviatio, for all data sets, be less tha or equal to 20%. 7. Stop the HCL calibratio gas dyamic spikig ad allow the system to retur to baselie coditios. 8. Perform a bias check for the elemet of opportuity referece CEMS, the flow measuremet system, ad the gas maifold for the HCL calibratio gas ad hydrocarbo (if dilutio flow is used to calculate the referece HCL cocetratio). Sectio 6 provides guidace for the bias checks. 9. Repeat the baselie data collectio, ad the dyamic spikig data collectio, for subsequet HCL test levels. 0. If dilutio flow is used to calculate the HCl referece values, after the HCL dyamic spikig test rus are complete, perform a bias check of the dilutio aalyzer. 8.3 DATA CALCULATIONS This sectio provides the calculatios required to determie the accuracy, precisio, ad bias of the HCL CEMS Baselie data HCL CEMS data The baselie HCL cocetratio is measured prior to each dyamic spikig ru. For each baselie HCL data set, calculate the average, i ppm, usig equatio. 2
14 Calculate the arithmetic mea of a data set as follows: x = xi (Equatio ) where : x = Arithmetic mea = Number of data poits x i = Value of each data poit Elemet of Opportuity Data The elemet of opportuity data set is acquired prior to each dyamic spikig test ru, simultaeous to the baselie HCL cocetratio data set. For each elemet of opportuity, calculate the average value usig equatio Flow via dilutio Whe usig dilutio of a measured aalyte (such as a hydrocarbo with subsequet GC measuremet) as a idicator of total flow, calculate the total flow, i liters per miute usig equatio 2. Calculate the total flow as follows: Tflow = ( Aaltye spike ) * Tflow ( Aalyte measured) (Equatio 2) Tflow = Total flow through the system i lpm AalyteSpike = Cocetratio of Spiked Aalyte (ppm) Aalytemeasured = Measuremet, dowstream, of spiked Aalyte (ppm) Calculate the average flow value, for the data set, usig equatio. 3
15 8.3.2 Dyamic Spikig Test ru data HCL CEMS data For each dyamic spikig test ru, calculate the average ad percet relative stadard deviatio of the data set usig equatios ad 3, respectively. Calculate the percet stadard deviatio as follows: PRS d = i = / d i = i d i (Equatio 3) x Where: PRS = Percet Relative Stadard Deviatio of the data set d x = Arithmetic mea Elemet of Opportuity data For each dyamic spikig test ru, calculate the average ad percet relative stadard deviatio of the data set usig equatios ad 3, respectively. If usig total flow as a meas to determie the referece HCL calibratio gas cocetratio, use equatio 2 to calculate the total flow prior to determiig the average ad percet relative stadard deviatio of the data set. 4
16 8.3.3 Referece HCL cocetratio calculatios Elemet of opportuity to calculate referece HCL cocetratio Equatio 4 is used to calculate the referece HCL cocetratio for a give dyamic spikig test ru. Opp post HCl ref = * CalGas + BaselieHCl (Equatio 4) Opp pre avg avg where: Opppost avg = average referece elemet of opportuity value durig dyamic spikig test ru Opppre avg = average referece elemet of opportuity value durig baselie test ru CalGas = HCl calibratio gas cocetratio Baselie HCl = CEMS HCl average baselie value Quatitative HCL calibratio gas itroductio ad total flow to calculate referece HCL cocetratio Whe usig the quatitative itroductio of calibratio gas ad total system flow to calculatio the referece HCL cocetratio for a give dyamic spikig test ru, use equatio 5. Equatio 2 is used to calculate the total system flow. HCl ref = Cflow Tflow avg avg * CalGas + BaselieHCl (Equatio 5) Where: HCl ref = Referece HCl value for ru X Cflow = Average calibratio HCL gas flow rate ito system (lpm) 5
17 Tflow = Average Total System Flow durig ru (lpm) CalGas = HCl calibratio gas cocetratio Baselie HCl = CEMS HCl average baselie value Liear Regressio Liear regressio is used to calculate the accuracy, precisio, ad bias of the HCL CEMS. For each dyamic spikig test ru, plot the HCL CEMS referece value, Xaxis, versus the HCL measured value, Y axis. Determie the correlatio coefficiet (r ), the slope, ad the itercept usig equatios 69. ˆ = b + b x (Equatio 6) y o where: b o = The y itercept b = The slope The itercept is calculated accordig to the followig equatio: b o = y b x (Equatio 7) where: x = y = x i i= y i i= The slope of the lie is calculated accordig to equatio below: 6
18 ( xi x )( yi y ) i= b = (Equatio 8) 2 ( x x ) i= i The liear correlatio coefficiet is calculated accordig to the followig equatio. r 2 ( x x)( y i i i= = 2 ( xi x) ( i= i= y) y y) i 2 2 (Equatio 9) 8.4 CRITERIA FOR ACCEPTANCE The criteria for the HCl dyamic spikig is: The correlatio coefficiet, r, must be greater tha or equal to 0.90 The slope must be.0, +/ 0.5 The Itercept must be equal to or less tha 5% of the istrumet spa. At a miimum, the data set must meet the correlatio coefficiet. If this is ot met, the dyamic spikig procedure must be repeated util the correlatio coefficiet criteria are met. If the HCl CEMS has met the correlatio coefficiet requiremet, but does ot meet either the slope, or itercept criteria, a bias exists ad a correctio factor must be applied to the HCl CEMS for data collectio. The equatios to determie the corrected value are provided i Sectio 8.5. The correctio factor must be applied to the HCl CEMS oemiute average data util a ew dyamic spikig procedure idicates the specific bias o loger exists, or, a differet correctio factor (bias) is idicated. 7
19 8.5 BIAS CORRECTION If the HCl CEMS fails to meet both the slope ad itercept criteria, the followig correctio factor must be applied to the oemiute average HCl data: C C y b i 0 = (Equatio 0) b Where: C C = corrected CEMS HCL cocetratio y i = CEMS reported HCL cocetratio b 0 = the itercept of the least squares liear regressio lie b = the slope of the least squares liear regressio lie b). If the HCl CEMS fails to meet the slope criteria, but meets the itercept criteria, the followig correctio factor must be applied to the oemiute average HCl data: yi C C = (Equatio ) b c). If the HCl CEMS fails to meet the itercept criteria, but meets the slope criteria, the followig correctio factor must be applied to the oemiute average HCl data: C C = y (Equatio 2) i b 0 8
20 9 TABLES Table. Ogoig Quality Assurace/Quality Cotrol Calibratio Gas Rages HCl Calibratio Gas Cocetratios a Test Uits Zerolevel MidLevel HighLevel Daily Calibratio Drift % of Spa 030 NA Absolute Audit Calibratio % of Spa Dyamic Spikig % of Spa a A copy of the supplier s certificate of aalysis must be provided for each gas cylider. Calibratio gases do ot eed to be Protocol gases. It is recommeded that the calibratio gases, for all QA, have a miimum tolerace of 5%. 9
21 0 REFERENCES. Uited States Evirometal Protectio Agecy (EPA). Techology Trasfer Network, Emissio Measuremet Ceter, Promulgated Methods, Test Method 205: Verificatio of Gas Dilutio Systems for Field Istrumet Calibratios. Available: pdf via the iteret. 2. Uited States Evirometal Protectio Agecy (EPA). Techology Trasfer Network, Emissio Measuremet Ceter, Promulgated Methods, Test Method 8: Measuremet of Gaseous Orgaic Compoud Emissios by Gas Chromatography. Available: via the iteret. 3. Performace Specificatio Z: Specificatios ad Test Procedures for Hydrochloric Acid Cotiuous Emissio Moitorig Systems at Statioary Sources. Eli Lilly ad Compay, Procedure DD: Quality Cotrol ad Quality Assurace Requiremets for Hydrochloric Acid Cotiuous Emissio Moitorig Systems at Statioary Sources. Eli Lilly ad Compay, Dilutio Flow Measuremet Protocol: Measuremet of Total flow usig a tracer gas with subsequet dow stream quatitatio. Eli Lilly ad Compay,
22 APPENDICES 2
Environmental Protection Agency Pt. 60, App. B, Spec. 1
Evirometal Protectio Agecy Pt. 60, App. B, Spec. APPENDIX B TO PART 60 PERFORMANCE SPECIFICATIONS Performace Specificatio Specificatios ad test procedures for cotiuous opacity moitorig systems i statioary
More informationConfidence Intervals for One Mean
Chapter 420 Cofidece Itervals for Oe Mea Itroductio This routie calculates the sample size ecessary to achieve a specified distace from the mea to the cofidece limit(s) at a stated cofidece level for a
More informationConfidence Intervals for One Mean with Tolerance Probability
Chapter 421 Cofidece Itervals for Oe Mea with Tolerace Probability Itroductio This procedure calculates the sample size ecessary to achieve a specified distace from the mea to the cofidece limit(s) with
More information1 Correlation and Regression Analysis
1 Correlatio ad Regressio Aalysis I this sectio we will be ivestigatig the relatioship betwee two cotiuous variable, such as height ad weight, the cocetratio of a ijected drug ad heart rate, or the cosumptio
More informationME 101 Measurement Demonstration (MD 1) DEFINITIONS Precision  A measure of agreement between repeated measurements (repeatability).
INTRODUCTION This laboratory ivestigatio ivolves makig both legth ad mass measuremets of a populatio, ad the assessig statistical parameters to describe that populatio. For example, oe may wat to determie
More informationSimple Linear Regression
Simple Liear Regressio W. Robert Stepheso Departmet of Statistics Iowa State Uiversity Oe of the most widely used statistical techiques is simple liear regressio. This techique is used to relate a measured
More informationNow here is the important step
LINEST i Excel The Excel spreadsheet fuctio "liest" is a complete liear least squares curve fittig routie that produces ucertaity estimates for the fit values. There are two ways to access the "liest"
More informationTECHNIQUES FOR IMPROVING THE ACCURACY OF CALIBRATION IN THE ENVIRONMENTAL LABORATORY
TECHNIQUES FOR IMPROVING THE ACCURACY OF CALIBRATION IN THE ENVIRONMENTAL LABORATORY Deis A Edgerley Quaterra Evirometal Services, 880 Riverside Parkway, West Sacrameto, Califoria 95605 Abstract Cosistet
More informationDomain 1: Designing a SQL Server Instance and a Database Solution
Maual SQL Server 2008 Desig, Optimize ad Maitai (70450) 18004186789 Domai 1: Desigig a SQL Server Istace ad a Database Solutio Desigig for CPU, Memory ad Storage Capacity Requiremets Whe desigig a
More informationAnalyzing Longitudinal Data from Complex Surveys Using SUDAAN
Aalyzig Logitudial Data from Complex Surveys Usig SUDAAN Darryl Creel Statistics ad Epidemiology, RTI Iteratioal, 312 Trotter Farm Drive, Rockville, MD, 20850 Abstract SUDAAN: Software for the Statistical
More informationSection 73 Estimating a Population. Requirements
Sectio 73 Estimatig a Populatio Mea: σ Kow Key Cocept This sectio presets methods for usig sample data to fid a poit estimate ad cofidece iterval estimate of a populatio mea. A key requiremet i this sectio
More informationConfidence Intervals. CI for a population mean (σ is known and n > 30 or the variable is normally distributed in the.
Cofidece Itervals A cofidece iterval is a iterval whose purpose is to estimate a parameter (a umber that could, i theory, be calculated from the populatio, if measuremets were available for the whole populatio).
More informationCHAPTER 3 THE TIME VALUE OF MONEY
CHAPTER 3 THE TIME VALUE OF MONEY OVERVIEW A dollar i the had today is worth more tha a dollar to be received i the future because, if you had it ow, you could ivest that dollar ad ear iterest. Of all
More informationAlternatives To Pearson s and Spearman s Correlation Coefficients
Alteratives To Pearso s ad Spearma s Correlatio Coefficiets Floreti Smaradache Chair of Math & Scieces Departmet Uiversity of New Mexico Gallup, NM 8730, USA Abstract. This article presets several alteratives
More informationCantilever Beam Experiment
Mechaical Egieerig Departmet Uiversity of Massachusetts Lowell Catilever Beam Experimet Backgroud A disk drive maufacturer is redesigig several disk drive armature mechaisms. This is the result of evaluatio
More informationThis document contains a collection of formulas and constants useful for SPC chart construction. It assumes you are already familiar with SPC.
SPC Formulas ad Tables 1 This documet cotais a collectio of formulas ad costats useful for SPC chart costructio. It assumes you are already familiar with SPC. Termiology Geerally, a bar draw over a symbol
More informationBiology 171L Environment and Ecology Lab Lab 2: Descriptive Statistics, Presenting Data and Graphing Relationships
Biology 171L Eviromet ad Ecology Lab Lab : Descriptive Statistics, Presetig Data ad Graphig Relatioships Itroductio Log lists of data are ofte ot very useful for idetifyig geeral treds i the data or the
More informationConfidence Intervals for Linear Regression Slope
Chapter 856 Cofidece Iterval for Liear Regreio Slope Itroductio Thi routie calculate the ample ize eceary to achieve a pecified ditace from the lope to the cofidece limit at a tated cofidece level for
More informationStat 104 Lecture 16. Statistics 104 Lecture 16 (IPS 6.1) Confidence intervals  the general concept
Statistics 104 Lecture 16 (IPS 6.1) Outlie for today Cofidece itervals Cofidece itervals for a mea, µ (kow σ) Cofidece itervals for a proportio, p Margi of error ad sample size Review of mai topics for
More informationLesson 17 Pearson s Correlation Coefficient
Outlie Measures of Relatioships Pearso s Correlatio Coefficiet (r) types of data scatter plots measure of directio measure of stregth Computatio covariatio of X ad Y uique variatio i X ad Y measurig
More informationINVESTMENT PERFORMANCE COUNCIL (IPC)
INVESTMENT PEFOMANCE COUNCIL (IPC) INVITATION TO COMMENT: Global Ivestmet Performace Stadards (GIPS ) Guidace Statemet o Calculatio Methodology The Associatio for Ivestmet Maagemet ad esearch (AIM) seeks
More informationChapter 9: Correlation and Regression: Solutions
Chapter 9: Correlatio ad Regressio: Solutios 9.1 Correlatio I this sectio, we aim to aswer the questio: Is there a relatioship betwee A ad B? Is there a relatioship betwee the umber of emploee traiig hours
More informationFACTSHEET 1: DEVELOPING A STRATEGIC MANAGEMENT PLAN
FACTSHEET 1: DEVELOPING A STRATEGIC MANAGEMENT PLAN THIS FACTSHEET RELATES TO QUESTION 2.3 OF THE MSPI (MUSEUM STANDARDS PROGRAMME FOR IRELAND) 1. Itroductio Writig a Strategic Maagemet Pla ca provide
More informationCHAPTER 3 DIGITAL CODING OF SIGNALS
CHAPTER 3 DIGITAL CODING OF SIGNALS Computers are ofte used to automate the recordig of measuremets. The trasducers ad sigal coditioig circuits produce a voltage sigal that is proportioal to a quatity
More informationNPTEL STRUCTURAL RELIABILITY
NPTEL Course O STRUCTURAL RELIABILITY Module # 0 Lecture 1 Course Format: Web Istructor: Dr. Aruasis Chakraborty Departmet of Civil Egieerig Idia Istitute of Techology Guwahati 1. Lecture 01: Basic Statistics
More information5: Introduction to Estimation
5: Itroductio to Estimatio Cotets Acroyms ad symbols... 1 Statistical iferece... Estimatig µ with cofidece... 3 Samplig distributio of the mea... 3 Cofidece Iterval for μ whe σ is kow before had... 4 Sample
More informationWindWise Education. 2 nd. T ransforming the Energy of Wind into Powerful Minds. editi. A Curriculum for Grades 6 12
WidWise Educatio T rasformig the Eergy of Wid ito Powerful Mids A Curriculum for Grades 6 12 Notice Except for educatioal use by a idividual teacher i a classroom settig this work may ot be reproduced
More informationBaan Service Master Data Management
Baa Service Master Data Maagemet Module Procedure UP069A US Documetiformatio Documet Documet code : UP069A US Documet group : User Documetatio Documet title : Master Data Maagemet Applicatio/Package :
More informationChapter 5 Unit 1. IET 350 Engineering Economics. Learning Objectives Chapter 5. Learning Objectives Unit 1. Annual Amount and Gradient Functions
Chapter 5 Uit Aual Amout ad Gradiet Fuctios IET 350 Egieerig Ecoomics Learig Objectives Chapter 5 Upo completio of this chapter you should uderstad: Calculatig future values from aual amouts. Calculatig
More informationGCSE STATISTICS. 4) How to calculate the range: The difference between the biggest number and the smallest number.
GCSE STATISTICS You should kow: 1) How to draw a frequecy diagram: e.g. NUMBER TALLY FREQUENCY 1 3 5 ) How to draw a bar chart, a pictogram, ad a pie chart. 3) How to use averages: a) Mea  add up all
More informationModified Line Search Method for Global Optimization
Modified Lie Search Method for Global Optimizatio Cria Grosa ad Ajith Abraham Ceter of Excellece for Quatifiable Quality of Service Norwegia Uiversity of Sciece ad Techology Trodheim, Norway {cria, ajith}@q2s.tu.o
More informationAQA STATISTICS 1 REVISION NOTES
AQA STATISTICS 1 REVISION NOTES AVERAGES AND MEASURES OF SPREAD www.mathsbox.org.uk Mode : the most commo or most popular data value the oly average that ca be used for qualitative data ot suitable if
More informationDefinition. Definition. 72 Estimating a Population Proportion. Definition. Definition
7 stimatig a Populatio Proportio I this sectio we preset methods for usig a sample proportio to estimate the value of a populatio proportio. The sample proportio is the best poit estimate of the populatio
More informationNonlife insurance mathematics. Nils F. Haavardsson, University of Oslo and DNB Skadeforsikring
Nolife isurace mathematics Nils F. Haavardsso, Uiversity of Oslo ad DNB Skadeforsikrig Mai issues so far Why does isurace work? How is risk premium defied ad why is it importat? How ca claim frequecy
More informationIncremental calculation of weighted mean and variance
Icremetal calculatio of weighted mea ad variace Toy Fich faf@cam.ac.uk dot@dotat.at Uiversity of Cambridge Computig Service February 009 Abstract I these otes I eplai how to derive formulae for umerically
More informationChapter 7: Confidence Interval and Sample Size
Chapter 7: Cofidece Iterval ad Sample Size Learig Objectives Upo successful completio of Chapter 7, you will be able to: Fid the cofidece iterval for the mea, proportio, ad variace. Determie the miimum
More information428 CHAPTER 12 MULTIPLE LINEAR REGRESSION
48 CHAPTER 1 MULTIPLE LINEAR REGRESSION Table 18 Team Wis Pts GF GA PPG PPcT SHG PPGA PKPcT SHGA Chicago 47 104 338 68 86 7. 4 71 76.6 6 Miesota 40 96 31 90 91 6.4 17 67 80.7 0 Toroto 8 68 3 330 79.3
More informationStudy on the application of the software phaselocked loop in tracking and filtering of pulse signal
Advaced Sciece ad Techology Letters, pp.3135 http://dx.doi.org/10.14257/astl.2014.78.06 Study o the applicatio of the software phaselocked loop i trackig ad filterig of pulse sigal Sog Wei Xia 1 (College
More informationXFM DIGITAL MASS FLOW METER. XFM with Profibus Interface. XFM without. Readout. XFM with. Readout. Option
DIGITAL MASS FLOW METER Desig Features Supports up to 23 Egieerig Uits (icludig User Defied). Stores calibratio data for up to 10 gases. Programmable Totalizer idicates total gas quatity. High ad low gas
More informationCHAPTER 3 The Simple Surface Area Measurement Module
CHAPTER 3 The Simple Surface Area Measuremet Module I chapter 2, the quality of charcoal i each batch might chage due to traditioal operatio. The quality test shall be performed before usig it as a adsorbet.
More information*The most important feature of MRP as compared with ordinary inventory control analysis is its time phasing feature.
Itegrated Productio ad Ivetory Cotrol System MRP ad MRP II Framework of Maufacturig System Ivetory cotrol, productio schedulig, capacity plaig ad fiacial ad busiess decisios i a productio system are iterrelated.
More informationFIRE PROTECTION SYSTEM INSPECTION, TESTING AND MAINTENANCE PROGRAMS
STRATEGIC OUTCOMES PRACTICE TECHNICAL ADVISORY BULLETIN February 2011 FIRE PROTECTION SYSTEM INSPECTION, TESTING AND MAINTENANCE PROGRAMS www.willis.com Natioal Fire Protectio Associatio (NFPA) #25 a mai
More informationPoisson Statistics. Objectives
by Dr. James E. Parks Departmet of Physics ad Astroomy 401 ielse Physics Buildig The Uiversity of Teessee Koxville, Teessee 379961200 Copyright March, 2001 by James Edgar Parks* *All rights are reserved.
More informationOutput Analysis (2, Chapters 10 &11 Law)
B. Maddah ENMG 6 Simulatio 05/0/07 Output Aalysis (, Chapters 10 &11 Law) Comparig alterative system cofiguratio Sice the output of a simulatio is radom, the comparig differet systems via simulatio should
More informationODBC. Getting Started With Sage Timberline Office ODBC
ODBC Gettig Started With Sage Timberlie Office ODBC NOTICE This documet ad the Sage Timberlie Office software may be used oly i accordace with the accompayig Sage Timberlie Office Ed User Licese Agreemet.
More information7. Sample Covariance and Correlation
1 of 8 7/16/2009 6:06 AM Virtual Laboratories > 6. Radom Samples > 1 2 3 4 5 6 7 7. Sample Covariace ad Correlatio The Bivariate Model Suppose agai that we have a basic radom experimet, ad that X ad Y
More informationZTEST / ZSTATISTIC: used to test hypotheses about. µ when the population standard deviation is unknown
ZTEST / ZSTATISTIC: used to test hypotheses about µ whe the populatio stadard deviatio is kow ad populatio distributio is ormal or sample size is large TTEST / TSTATISTIC: used to test hypotheses about
More informationA GUIDE TO LEVEL 3 VALUE ADDED IN 2013 SCHOOL AND COLLEGE PERFORMANCE TABLES
A GUIDE TO LEVEL 3 VALUE ADDED IN 2013 SCHOOL AND COLLEGE PERFORMANCE TABLES Cotets Page No. Summary Iterpretig School ad College Value Added Scores 2 What is Value Added? 3 The Learer Achievemet Tracker
More informationChapter Gaussian Elimination
Chapter 04.06 Gaussia Elimiatio After readig this chapter, you should be able to:. solve a set of simultaeous liear equatios usig Naïve Gauss elimiatio,. lear the pitfalls of the Naïve Gauss elimiatio
More informationDescriptive statistics deals with the description or simple analysis of population or sample data.
Descriptive statistics Some basic cocepts A populatio is a fiite or ifiite collectio of idividuals or objects. Ofte it is impossible or impractical to get data o all the members of the populatio ad a small
More informationCCH Accountants Starter Pack
CCH Accoutats Starter Pack We may be a bit smaller, but fudametally we re o differet to ay other accoutig practice. Util ow, smaller firms have faced a stark choice: Buy cheaply, kowig that the practice
More informationA Guide to the Pricing Conventions of SFE Interest Rate Products
A Guide to the Pricig Covetios of SFE Iterest Rate Products SFE 30 Day Iterbak Cash Rate Futures Physical 90 Day Bak Bills SFE 90 Day Bak Bill Futures SFE 90 Day Bak Bill Futures Tick Value Calculatios
More informationEquation of a line. Line in coordinate geometry. Slopeintercept form ( 斜 截 式 ) Intercept form ( 截 距 式 ) Pointslope form ( 點 斜 式 )
Chapter : Liear Equatios Chapter Liear Equatios Lie i coordiate geometr I Cartesia coordiate sstems ( 卡 笛 兒 坐 標 系 統 ), a lie ca be represeted b a liear equatio, i.e., a polomial with degree. But before
More informationOccupational Safety and Health Act of 1970
Occupatioal Safety ad Health Act of 1970 To assure safe ad healthful workig coditios for workig me ad wome; by authorizig eforcemet of the stadards developed uder the Act; by assistig ad ecouragig the
More information(VCP310) 18004186789
Maual VMware Lesso 1: Uderstadig the VMware Product Lie I this lesso, you will first lear what virtualizatio is. Next, you ll explore the products offered by VMware that provide virtualizatio services.
More informationMESSAGE TO TEACHERS: NOTE TO EDUCATORS:
MESSAGE TO TEACHERS: NOTE TO EDUCATORS: Attached herewith, please fid suggested lesso plas for term 1 of MATHEMATICS Grade 12. Please ote that these lesso plas are to be used oly as a guide ad teachers
More informationBasic Measurement Issues. Sampling Theory and AnalogtoDigital Conversion
Theory ad AalogtoDigital Coversio Itroductio/Defiitios Aalogtodigital coversio Rate Frequecy Aalysis Basic Measuremet Issues Reliability the extet to which a measuremet procedure yields the same results
More informationQuadrat Sampling in Population Ecology
Quadrat Samplig i Populatio Ecology Backgroud Estimatig the abudace of orgaisms. Ecology is ofte referred to as the "study of distributio ad abudace". This beig true, we would ofte like to kow how may
More informationThe following example will help us understand The Sampling Distribution of the Mean. C1 C2 C3 C4 C5 50 miles 84 miles 38 miles 120 miles 48 miles
The followig eample will help us uderstad The Samplig Distributio of the Mea Review: The populatio is the etire collectio of all idividuals or objects of iterest The sample is the portio of the populatio
More informationINVESTMENT PERFORMANCE COUNCIL (IPC) Guidance Statement on Calculation Methodology
Adoptio Date: 4 March 2004 Effective Date: 1 Jue 2004 Retroactive Applicatio: No Public Commet Period: Aug Nov 2002 INVESTMENT PERFORMANCE COUNCIL (IPC) Preface Guidace Statemet o Calculatio Methodology
More informationREFURBISHMENTS AND AUGMENTATIONS
INTRODUCTION TIER WORKING PAPER No. 0 REFURBISHMENTS AND AUGMENTATIONS Workig Paper No. How Water Prices are Set provided a overview of how water prices are set o the basis of lower boud costs. As oted
More informationRegression and Calibration
LC GC Europe Olie Supplemet statistics ad data aalysis 13 Regressio ad Calibratio Shau Burke, RHM Techology Ltd, High Wycombe, Buckighamshire, UK. Oe of the most frequetly used statistical methods i calibratio
More informationTHE REALISATION OF SYSTEM OF INDIRECT MEASUREMENT TEMPERATURE IN LD CONVERTER
Iteratioal Carpathia Cotrol Coferece ICCC 00 MALENOVICE, CZECH REPUBLIC May 730, 00 THE REALISATION OF SYSTEM OF INDIRECT MEASUREMENT TEMPERATURE IN LD CONVERTER Gabriel TRÉFA 1 ad Karol KOSTÚR 1 Departmet
More informationRealtime Metrics for Process Performance By focusing on realtime measures, plant operators can improve bottomline results
Feature Egieerig Report Practice Realtime Metrics for Process Performace By focusig o realtime measures, plat operators ca improve bottomlie results George Buckbee ExperTue, Ic. Process performace, quality
More informationYour grandmother and her financial counselor
Sectio 10. Arithmetic Sequeces 963 Objectives Sectio 10. Fid the commo differece for a arithmetic sequece. Write s of a arithmetic sequece. Use the formula for the geeral of a arithmetic sequece. Use the
More informationPowerView. Model AP9215. User s Manual
PowerView Model AP9215 User s Maual Thak You! Thak you for selectig the America Power Coversio PowerView. It has bee desiged for may years of reliable, maiteacefree service i combiatio with your America
More informationPSYCHOLOGICAL STATISTICS
UNIVERSITY OF CALICUT SCHOOL OF DISTANCE EDUCATION B Sc. Cousellig Psychology (0 Adm.) IV SEMESTER COMPLEMENTARY COURSE PSYCHOLOGICAL STATISTICS QUESTION BANK. Iferetial statistics is the brach of statistics
More informationhp calculators HP 12C Platinum Statistics  correlation coefficient The correlation coefficient HP12C Platinum correlation coefficient
HP 1C Platium Statistics  correlatio coefficiet The correlatio coefficiet HP1C Platium correlatio coefficiet Practice fidig correlatio coefficiets ad forecastig HP 1C Platium Statistics  correlatio coefficiet
More information5.4 Amortization. Question 1: How do you find the present value of an annuity? Question 2: How is a loan amortized?
5.4 Amortizatio Questio 1: How do you fid the preset value of a auity? Questio 2: How is a loa amortized? Questio 3: How do you make a amortizatio table? Oe of the most commo fiacial istrumets a perso
More informationConfidence Intervals for the Population Mean
Cofidece Itervals Math 283 Cofidece Itervals for the Populatio Mea Recall that from the empirical rule that the iterval of the mea plus/mius 2 times the stadard deviatio will cotai about 95% of the observatios.
More information9.8: THE POWER OF A TEST
9.8: The Power of a Test CD91 9.8: THE POWER OF A TEST I the iitial discussio of statistical hypothesis testig, the two types of risks that are take whe decisios are made about populatio parameters based
More informationFlood Emergency Response Plan
Flood Emergecy Respose Pla This reprit is made available for iformatioal purposes oly i support of the isurace relatioship betwee FM Global ad its cliets. This iformatio does ot chage or supplemet policy
More informationtechnical guide Uncertainty of Measurement, Precision and Limits of Detection in Chemical and Microbiological Testing Laboratories
techical guide Ucertaity of Measuremet, Precisio ad Limits of Detectio i Chemical ad Microbiological Testig Laboratories Published by: Iteratioal Accreditatio New Zealad 66 Great South Road, Greelae, Aucklad
More informationIntroducing Your New Wells Fargo Trust and Investment Statement. Your Account Information Simply Stated.
Itroducig Your New Wells Fargo Trust ad Ivestmet Statemet. Your Accout Iformatio Simply Stated. We are pleased to itroduce your ew easytoread statemet. It provides a overview of your accout ad a complete
More informationMEI Structured Mathematics. Module Summary Sheets. Statistics 2 (Version B: reference to new book)
MEI Mathematics i Educatio ad Idustry MEI Structured Mathematics Module Summary Sheets Statistics (Versio B: referece to ew book) Topic : The Poisso Distributio Topic : The Normal Distributio Topic 3:
More informationSECTION 1.5 : SUMMATION NOTATION + WORK WITH SEQUENCES
SECTION 1.5 : SUMMATION NOTATION + WORK WITH SEQUENCES Read Sectio 1.5 (pages 5 9) Overview I Sectio 1.5 we lear to work with summatio otatio ad formulas. We will also itroduce a brief overview of sequeces,
More informationPower Factor in Electrical Power Systems with NonLinear Loads
Power Factor i Electrical Power Systems with NoLiear Loads By: Gozalo Sadoval, ARTECHE / INELAP S.A. de C.V. Abstract. Traditioal methods of Power Factor Correctio typically focus o displacemet power
More informationGregory Carey, 1998 Linear Transformations & Composites  1. Linear Transformations and Linear Composites
Gregory Carey, 1998 Liear Trasformatios & Composites  1 Liear Trasformatios ad Liear Composites I Liear Trasformatios of Variables Meas ad Stadard Deviatios of Liear Trasformatios A liear trasformatio
More informationMR. STEIN S WORDS OF WISDOM
MR. STEIN S WORDS OF WISDOM P a g e 1 I am writig this review essay for two tests the AP Stat exam ad the Applied Stat Fial exam. The topics are more or less the same, so reviewig for the two tests should
More informationCHAPTER 7: Central Limit Theorem: CLT for Averages (Means)
CHAPTER 7: Cetral Limit Theorem: CLT for Averages (Meas) X = the umber obtaied whe rollig oe six sided die oce. If we roll a six sided die oce, the mea of the probability distributio is X P(X = x) Simulatio:
More informationMeasures of Spread and Boxplots Discrete Math, Section 9.4
Measures of Spread ad Boxplots Discrete Math, Sectio 9.4 We start with a example: Example 1: Comparig Mea ad Media Compute the mea ad media of each data set: S 1 = {4, 6, 8, 10, 1, 14, 16} S = {4, 7, 9,
More informationA Study on the False Alarm Rates of X, EWMA and CUSUM Control Charts when Parameters are Estimated
11 Iteratioal Coferece o Circuits, System ad Simulatio IPCSIT vol.7 (11) (11) IACSIT Press, Sigapore A Study o the False Alarm Rates of X, EWMA ad CUSUM Cotrol Charts whe Parameters are Estimated Michael
More informationStatistical Methods. Chapter 1: Overview and Descriptive Statistics
Geeral Itroductio Statistical Methods Chapter 1: Overview ad Descriptive Statistics Statistics studies data, populatio, ad samples. Descriptive Statistics vs Iferetial Statistics. Descriptive Statistics
More informationMeasures of Central Tendency
Measures of Cetral Tedecy A studet s grade will be determied by exam grades ( each exam couts twice ad there are three exams, HW average (couts oce, fial exam ( couts three times. Fid the average if the
More informationCREATIVE MARKETING PROJECT 2016
CREATIVE MARKETING PROJECT 2016 The Creative Marketig Project is a chapter project that develops i chapter members a aalytical ad creative approach to the marketig process, actively egages chapter members
More informationSimple Linear Regression
Simple Liear Regressio We have bee itroduced to the otio that a categorical variable could deped o differet levels of aother variable whe we discussed cotigecy tables. We ll exted this idea to the case
More informationSystems Design Project: Indoor Location of Wireless Devices
Systems Desig Project: Idoor Locatio of Wireless Devices Prepared By: Bria Murphy Seior Systems Sciece ad Egieerig Washigto Uiversity i St. Louis Phoe: (805) 6985295 Email: bcm1@cec.wustl.edu Supervised
More informationEstimating Probability Distributions by Observing Betting Practices
5th Iteratioal Symposium o Imprecise Probability: Theories ad Applicatios, Prague, Czech Republic, 007 Estimatig Probability Distributios by Observig Bettig Practices Dr C Lych Natioal Uiversity of Irelad,
More informationStatistics Lecture 14. Introduction to Inference. Administrative Notes. Hypothesis Tests. Last Class: Confidence Intervals
Statistics 111  Lecture 14 Itroductio to Iferece Hypothesis Tests Admiistrative Notes Sprig Break! No lectures o Tuesday, March 8 th ad Thursday March 10 th Exteded Sprig Break! There is o Stat 111 recitatio
More informationConfidence Intervals and Sample Size
8/7/015 C H A P T E R S E V E N Cofidece Itervals ad Copyright 015 The McGrawHill Compaies, Ic. Permissio required for reproductio or display. 1 Cofidece Itervals ad Outlie 71 Cofidece Itervals for the
More informationOverview. Learning Objectives. Point Estimate. Estimation. Estimating the Value of a Parameter Using Confidence Intervals
Overview Estimatig the Value of a Parameter Usig Cofidece Itervals We apply the results about the sample mea the problem of estimatio Estimatio is the process of usig sample data estimate the value of
More informationEthernet Option Board
Etheret Optio Board Assembly ad Iitializatio Guide for Addig Etheret Commuicatios to a ADP etime Timeclock Documet Part Number: 47055200 Documet Revisio: B The iformatio i this documet is subject to chage
More informationUsing Four Types Of Notches For Comparison Between Chezy s Constant(C) And Manning s Constant (N)
INTERNATIONAL JOURNAL OF SCIENTIFIC & TECHNOLOGY RESEARCH OLUME, ISSUE, OCTOBER ISSN  Usig Four Types Of Notches For Compariso Betwee Chezy s Costat(C) Ad Maig s Costat (N) Joyce Edwi Bategeleza, Deepak
More informationLEASEPURCHASE DECISION
Public Procuremet Practice STANDARD The decisio to lease or purchase should be cosidered o a caseby case evaluatio of comparative costs ad other factors. 1 Procuremet should coduct a cost/ beefit aalysis
More informationCS100: Introduction to Computer Science
Review: History of Computers CS100: Itroductio to Computer Sciece Maiframes Miicomputers Lecture 2: Data Storage  Bits, their storage ad mai memory Persoal Computers & Workstatios Review: The Role of
More informationNATIONAL SENIOR CERTIFICATE GRADE 11
NATIONAL SENIOR CERTIFICATE GRADE MATHEMATICS P NOVEMBER 007 MARKS: 50 TIME: 3 hours This questio paper cosists of 9 pages, diagram sheet ad a page formula sheet. Please tur over Mathematics/P DoE/November
More information3rd International Conference on Material, Mechanical and Manufacturing Engineering (IC3ME 2015)
3rd Iteratioal Coferece o Material, Mechaical ad Maufacturig Egieerig (IC3ME 205) A Sesor Parameter Calibratio Method of Tire Pressure Moitor System Based o Liear Regressio Techology Meilig Li,a, Xiaopeg
More informationHypothesis Tests Applied to Means
The Samplig Distributio of the Mea Hypothesis Tests Applied to Meas Recall that the samplig distributio of the mea is the distributio of sample meas that would be obtaied from a particular populatio (with
More informationDefinition. A variable X that takes on values X 1, X 2, X 3,...X k with respective frequencies f 1, f 2, f 3,...f k has mean
1 Social Studies 201 October 13, 2004 Note: The examples i these otes may be differet tha used i class. However, the examples are similar ad the methods used are idetical to what was preseted i class.
More informationPUBLIC RELATIONS PROJECT 2016
PUBLIC RELATIONS PROJECT 2016 The purpose of the Public Relatios Project is to provide a opportuity for the chapter members to demostrate the kowledge ad skills eeded i plaig, orgaizig, implemetig ad evaluatig
More information