1 Keysight Technologies Understanding the Fundamental Principles of Vector Network Analysis Application Note
2 Introduction Network analysis is the process by which designers and manufacturers measure the electrical performance of the components and circuits used in more complex systems. When these systems are conveying signals with information content, we are most concerned with getting the signal from one point to another with maximum efficiency and minimum distortion. Vector network analysis is a method of accurately characterizing such components by measuring their effect on the amplitude and phase of swept-frequency and swept-power test signals. In this application note, the fundamental principles of vector network analysis will be reviewed. The discussion includes the common parameters that can be measured, including the concept of scattering parameters (S-parameters). RF fundamentals such as transmission lines and the Smith chart will also be reviewed. Keysight Technologies, Inc. offers a wide range of portable and benchtop vector network analyzers for characterizing components from DC to 110 GHz. These instruments are available with a wide range of options to simplify testing in the field, laboratory, and production environments.
3 03 Keysight Understanding the Fundamental Principles of Vector Network Analysis Application Note Measurements in Communications Systems In any communications system, the effect of signal distortion must be considered. While we generally think of the distortion caused by nonlinear effects (for example, when intermodulation products are produced from desired carrier signals), purely linear systems can also introduce signal distortion. Linear systems can change the time waveform of signals passing through them by altering the amplitude or phase relationships of the spectral components that make up the signal. Let s examine the difference between linear and nonlinear behavior more closely. Linear devices impose magnitude and phase changes on input signals (Figure 1). Any sinusoid appearing at the input will also appear at the output, and at the same frequency. No new signals are created. Both active and passive nonlinear devices can shift an input signal in frequency or add other frequency components, such as harmonic and spurious signals. Large input signals can drive normally linear devices into compression or saturation, causing nonlinear operation. Sin 360º * f * t Input DUT A t o A * Sin 360º * f (t t o ) A phase shift = to * 360º * f f 1 Output Linear behavior Input and output frequencies are the same (no additional frequencies created) Output frequency only undergoes magnitude and phase change f 1 Nonlinear behavior Output frequency may undergo frequency shift (e.g. with mixers) Additional frequencies created (harmonics, intermodulation) f 1 Figure 1. Linear versus nonlinear behavior For linear distortion-free transmission, the amplitude response of the device under test (DUT) must be flat and the phase response must be linear over the desired bandwidth. As an example, consider a square-wave signal rich in high-frequency components passing through a bandpass filter that passes selected frequencies with little attenuation while attenuating frequencies outside of the passband by varying amounts. Even if the filter has linear phase performance, the out-of-band components of the square wave will be attenuated, leaving an output signal that, in this example, is more sinusoidal in nature (Figure 2). If the same square-wave input signal is passed through a filter that only inverts the phase of the third harmonic, but leaves the harmonic amplitudes the same, the output will be more impulse-like in nature (Figure 3). While this is true for the example filter, in general, the output waveform will appear with arbitrary distortion, depending on the amplitude and phase nonlinearities.
4 04 Keysight Understanding the Fundamental Principles of Vector Network Analysis Application Note Measurements in Communications Systems (Continued) F(t) = sin wt + 1 /3 sin 3wt + 1 /5 sin 5wt Linear network Magnitude Figure 2. Magnitude variation with frequency F(t) = sin wt + 1 /3 sin 3wt + 1 /5 sin 5wt Linear network Magnitude 0º 180º 360º Figure 3. Phase variation with frequency
5 05 Keysight Understanding the Fundamental Principles of Vector Network Analysis Application Note Measurements in Communications Systems (Continued) Nonlinear networks Saturation, crossover, intermodulation, and other nonlinear effects can cause signal distortion Figure 4. Nonlinear induced distortion Nonlinear devices also introduce distortion (Figure 4). For example, if an ampli fier is overdriven, the output signal clips because the amplifier is saturated. The output signal is no longer a pure sinusoid, and harmonics are present at multiples of the input frequency. Passive devices may also exhibit nonlinear behavior at high power levels, a good example of which is an L-C filter that uses inductors with magnetic cores. Magnetic materials often exhibit hysteresis effects that are highly nonlinear. Efficient transfer of power is another fundamental concern in communications systems. In order to efficiently convey, transmit or receive RF power, devices such as transmissions lines, antennas and amplifiers must present the proper impedance match to the signal source. Impedance mismatches occur when the real and imaginary parts of input and output impedances are not ideal between two connecting devices. Importance of Vector Measurements Measuring both magnitude and phase of components is important for several reasons. First, both measurements are required to fully characterize a linear network and ensure distortion-free transmission. To design efficient matching networks, complex impedance must be measured. Engineers developing models for computer-aided-engineering (CAE) circuit simulation programs require magnitude and phase data for accurate models. In addition, time-domain characterization requires magnitude and phase information in order to perform an inverse-fourier transform. Vector error correction, which improves measurement accuracy by removing the effects of inherent measurement-system errors, requires both magnitude and phase data to build an effective error model. Phasemeasurement capability is very important even for scalar measurements such as return loss, in order to achieve a high level of accuracy (see Keysight application note Applying Error Correction to Network Analyzer Measurements, literature number E).
6 06 Keysight Understanding the Fundamental Principles of Vector Network Analysis Application Note The Basis of and Reflected Power In its fundamental form, network analysis involves the measurement of incident, reflected, and transmitted waves that travel along transmission lines. Using optical wavelengths as an analogy, when light strikes a clear lens (the incident energy), some of the light is reflected from the lens surface, but most of it continues through the lens (the transmitted energy) (Figure 5). If the lens has mirrored surfaces, most of the light will be reflected and little or none will pass through it. While the wavelengths are different for RF and microwave signals, the principle is the same. Network analyzers accurately measure the incident, reflected, and transmitted energy, e.g., the energy that is launched onto a transmission line, reflected back down the transmission line toward the source (due to impedence mismatch), and successfully transmitted to the terminating device (such as an antenna). Lightwave analogy Transmitted Reflected Figure 5. Lightwave analogy to high-frequency device characterization
7 07 Keysight Understanding the Fundamental Principles of Vector Network Analysis Application Note The Smith Chart The amount of reflection that occurs when characterizing a device depends on the impedance that the incident signal sees. Since any impedance can be represented with real and imaginary parts (R + jx or G + jb), they can be plotted on a rectilinear grid known as the complex impedance plane. Unfortunately, an open circuit (a common RF impedence) appears at infinity on the real axis, and therefore cannot be shown. The polar plot is useful because the entire impedance plane is covered. However, instead of plotting impedance directly, the complex reflection coefficient is displayed in vector form. The magnitude of the vector is the distance from the center of the display, and phase is displayed as the angle of vector referenced to a flat line from the center to the right-most edge. The drawback of polar plots is that impedance values cannot be read directly from the display. Since there is a one-to-one correspondence between complex impedance and reflection coefficient, the positive real half of the complex impedance plane can be mapped onto the polar display. The result is the Smith chart. All values of reactance and all positive values of resistance from 0 to infinity fall within the outer circle of the Smith chart (Figure 6). On the Smith chart, loci of constant resistance appear as circles, while loci of constant reactance appear as arcs. Impedances on the Smith chart are always normalized to the characteristic impedance of the component or system of interest, usually 50 ohms for RF and microwave systems and 75 ohms for broadcast and cable-television systems. A perfect termination appears in the center of the Smith chart. +jx 0 +R Polar plane ±180º jx Rectilinear impedance plane 0 90º Smith chart maps rectilinear impedance plane onto polar plane Z L = Z O Γ = 0 Z L = 0 (short) Γ = 1 180º Constant X Constant R Z L = (open) Γ = 1 0º Smith chart Figure 6. Smith chart review
8 08 Keysight Understanding the Fundamental Principles of Vector Network Analysis Application Note Power Transfer Conditions A perfectly matched condition must exist at a connection between two devices for maximum power transfer into a load, given a source resistance of R S and a load resistance of R L. This condition occurs when R L = R S, and is true whether the stimulus is a DC voltage source or a source of RF sine waves (Figure 7). When the source impedance is not purely resistive, maximum power transfer occurs when the load impedance is equal to the complex conjugate of the source impedance. This condition is met by reversing the sign of the imaginary part of the impedance. For example, if R S = j 0.3, then the complex conjugate is R S * = 0.6 j 0.3. The need for efficient power transfer is one of the main reasons for the use of transmission lines at higher frequencies. At very low frequencies (with much larger wavelengths), a simple wire is adequate for conducting power. The resistance of the wire is relatively low and has little effect on low-frequency signals. The voltage and current are the same no matter where a measurement is made on the wire. At higher frequencies, wavelengths are comparable to or smaller than the length of the conductors in a high-frequency circuit, and power transmission can be thought of in terms of traveling waves. When the transmission line is terminated in its characteristic impedance, maximum power is transferred to the load. When the termination is not equal to the characteristic impedance, that part of the signal that is not absorbed by the load is reflected back to the source. If a transmission line is terminated in its characteristic impedance, no reflected signal occurs since all of the transmitted power is absorbed by the load (Figure 8). Looking at the envelope of the RF signal versus distance along the transmission line shows no standing waves because without reflections, energy flows in only one direction. R S 1.2 R L For complex impedances, maximum power transfer occurs when Z L = Z S * (conjugate match) Load power (normalized) R L / R S Z S = R + jx Z L = Z S * = R jx Maximum power is transferred when R L = R S Figure 7. Power transfer
9 09 Keysight Understanding the Fundamental Principles of Vector Network Analysis Application Note Power Transfer Conditions (Continued) Zs = Zo Zo = characteristic impedance of transmission line Zo V inc V refl = 0 (all the incident power is absorbed in the load) For reflection, a transmission line terminated in Zo behaves like an infinitely long transmission line Figure 8. Transmission line terminated with Z 0 When the transmission line is terminated in a short circuit (which can sustain no voltage and therefore dissipates zero power), a reflected wave is launched back along the line toward the source (Figure 9). The reflected voltage wave must be equal in magnitude to the incident voltage wave and be 180 degrees out of phase with it at the plane of the load. The reflected and incident waves are equal in magnitude but traveling in the opposite directions. If the transmission line is terminated in an open-circuit condition (which can sustain no current), the reflected current wave will be 180 degrees out of phase with the incident current wave, while the reflected voltage wave will be in phase with the incident voltage wave at the plane of the load. This guarantees that the current at the open will be zero. The reflected and incident current waves are equal in magnitude, but traveling in the opposite directions. For both the short and open cases, a standing wave pattern is set up on the transmission line. The voltage valleys will be zero and the voltage peaks will be twice the incident voltage level. If the transmission line is terminated with say a 25-ohm resistor, resulting in a condition between full absorption and full reflection, part of the incident power is absorbed and part is reflected. The amplitude of the reflected voltage wave will be one-third that of the incident wave, and the two waves will be 180 degrees out of phase at the plane of the load. The valleys of the standing-wave pattern will no longer be zero, and the peaks will be less than those of the short and open cases. The ratio of the peaks to valleys will be 2:1. The traditional way of determining RF impedance was to measure VSWR using an RF probe/detector, a length of slotted transmission line, and a VSWR meter. As the probe was moved along the transmission line, the relative position and values of the peaks and valleys were noted on the meter. From these measurements, impedance could be derived. The procedure was repeated at different frequencies. Modern network analyzers measure the incident and reflected waves directly during a frequency sweep, and impedance results can be displayed in any number of formats (including VSWR).
10 10 Keysight Understanding the Fundamental Principles of Vector Network Analysis Application Note Power Transfer Conditions (Continued) Zs = Zo V inc V refl In phase (0º) for open Out of phase (180º) for short For reflection, a transmission line terminated in a short or open reflects all power back to source Figure 9. Transmission line terminated with short, open
11 11 Keysight Understanding the Fundamental Principles of Vector Network Analysis Application Note Network Analysis Terminology Now that we understand the fundamentals of electromagnetic waves, we must learn the common terms used for measuring them. Network analyzer terminology generally denotes measurements of the incident wave with the R or reference channel. The reflected wave is measured with the A channel, and the transmitted wave is measured with the B channel (Figure 10). With the amplitude and phase information in these waves, it is possible to quantify the reflection and transmission characteristics of a DUT. The reflection and transmission characteristics can be expressed as vector (magnitude and phase), scalar (magnitude only), or phase-only quantities. For example, return loss is a scalar measurement of reflection, while impedance is a vector reflection measurement. Ratioed measurements allow us to make reflection and transmission measurements that are independent of both absolute power and variations in source power versus frequency. Ratioed reflection is often shown as A/R and ratioed transmission as B/R, relating to the measurement channels in the instrument. R Reflected A Reflection Transmitted B Transmission Reflected = A R Transmitted = B R SWR Return loss Gain/loss Group delay Insertion phase S-parameters S11, S22 Reflection coefficient G, r Impedance, admittance R+jX, G+jB S-parameters S21, S12 Transmission coefficient T, t Figure 10. Common terms for high-frequency device characterization The most general term for ratioed reflection is the complex reflection coefficient, G or gamma (Figure 11). The magnitude portion of G is called r or rho. The reflection coefficient is the ratio of the reflected signal voltage level to the incident signal voltage level. For example, a transmission line terminated in its characteristic impedance Z o,will have all energy transferred to the load so V refl = 0 and r = 0. When the impedance of the load, Z L is not equal to the characteristic impedance, energy is reflected and r is greater than zero. When the load impedance is equal to a short or open circuit, all energy is reflected and r = 1. As a result, the range of possible values for r is 0 to 1.
12 12 Keysight Understanding the Fundamental Principles of Vector Network Analysis Application Note Network Analysis Terminology (Continued) Reflection coefficient Γ = V reflected V incident Return loss = 20 log(ρ),ρ = r F = Γ = Z L - Z O Z L + Z O Emax Emin Voltage standing wave ratio VSWR = Emax Emin = 1 + r 1 r No reflection (Z L = Z o ) 0 ρ 1 db RL 0 db 1 VSWR Figure 11. Reflection parameters Full reflection (Z L = open, short) Return loss is a way to express the reflection coefficient in logarithmic terms (decibels). Return loss is the number of decibels that the reflected signal is below the incident signal. Return loss is always expressed as a positive number and varies between infinity for a load at the characteristic impedance and 0 db for an open or short circuit. Another common term used to express reflection is voltage standing wave ratio (VSWR), which is defined as the maximum value of the RF envelope over the minimum value of the RF envelope. It is related to r as (1 + r)/(1 r). VSWR ranges from 1 (no reflection) to infinity (full reflection). The transmission coefficient is defined as the transmitted voltage divided by the incident voltage (Figure 12). If the absolute value of the transmitted voltage is greater than the absolute value of the incident voltage, a DUT or system is said to have gain. If the absolute value of the transmitted voltage is less than the absolute value of the incident voltage, the DUT or system is said to have attenuation or insertion loss. The phase portion of the transmission coefficient is called insertion phase.
13 13 Keysight Understanding the Fundamental Principles of Vector Network Analysis Application Note Network Analysis Terminology (Continued) V DUT V Transmitted Transmission coefficient = T = V Transmitted V = τ φ Insertion loss (db) = 20 Log V Trans V Inc = 20 log τ Gain (db) = 20 Log V Trans V Inc = 20 log τ Figure 12. Transmission parameters Direct examination of insertion phase usually does not provide useful information. This is because the insertion phase has a large (negative) slope with respect to frequency due to the electrical length of the DUT. The slope is proportional to the length of the DUT. Since it is only deviation from linear phase that causes distortion in communications systems, it is desirable to remove the linear portion of the phase response to analyze the remaining nonlinear portion. This can be done by using the electrical delay feature of a network analyzer to mathematically cancel the average electrical length of the DUT. The result is a high-resolution display of phase distortion or deviation from linear phase (Figure 13). Use electrical delay to remove linear portion of phase response Phase 45º/div RF filter response Low resolution Linear electrical length added (Electrical delay function) + Yields Deviation from linear phase High resolution Phase 1º/div Figure 13. Deviation from linear phase
14 14 Keysight Understanding the Fundamental Principles of Vector Network Analysis Application Note Measuring Group Delay Another useful measure of phase distortion is group delay (Figure 14). This parameter is a measure of the transit time of a signal through a DUT versus frequency. Group delay can be calculated by differentiating the DUT s phase response versus frequency. It reduces the linear portion of the phase response to a constant value, and transforms the deviations from linear phase into deviations from constant group delay, (which causes phase distortion in communications systems). The average delay represents the average signal transit time through a DUT. ω ω t g Group delay Group delay φ t o Average delay Phase φ Group delay (tg) = δ φ d ω = * d φ d f Deviation from constant group delay indicates distortion Average delay indicates transit time φ in radians ω in radians/sec φ in degrees f in Hz (ω = 2πf) Figure 14. What is group delay? Depending on the device, both deviation from linear phase and group delay may be measured, since both can be important. Specifying a maximum peak-to-peak phase ripple in a device may not be sufficient to completely characterize it, since the slope of the phase ripple depends on the number of ripples that occur per unit of frequency. Group delay takes this into account because it is the differentiated phase response. Group delay is often a more easily interpreted indication of phase distortion (Figure 15). Group delay Phase Phase dφ dω f dφ dω f Group delay f f Same peak-to-peak phase ripple can result in different group delay Figure 15. Why measure group delay?
15 15 Keysight Understanding the Fundamental Principles of Vector Network Analysis Application Note Network Characterization In order to completely characterize an unknown linear two-port device, we must make measurements under various conditions and compute a set of parameters. These parameters can be used to completely describe the electrical behavior of our device (or network), even under source and load conditions other than when we made our measurements. Low-frequency device or network characterization is usually based on measurement of H, Y, and Z parameters. To do this, the total voltage and current at the input or output ports of a device or nodes of a network must be measured. Furthermore, measurements must be made with open-circuit and short-circuit conditions. Since it is difficult to measure total current or voltage at higher frequencies, S-parameters are generally measured instead (Figure 16). These parameters relate to familiar measurements such as gain, loss, and reflection coefficient. They are relatively simple to measure, and do not require connection of undesirable loads to the DUT. The measured S-parameters of multiple devices can be cascaded to predict overall system performance. S-parameters are readily used in both linear and nonlinear CAE circuit simulation tools, and H, Y, and Z parameters can be derived from S-parameters when necessary. The number of S-parameters for a given device is equal to the square of the number of ports. For example, a two-port device has four S-parameters. The numbering convention for S-parameters is that the first number following the S is the port at which energy emerges, and the second number is the port at which energy enters. So S 21 is a measure of power emerging from Port 2 as a result of applying an RF stimulus to Port 1. When the numbers are the same (e.g. S 11 ), a reflection measurement is indicated. H, Y, and Z parameters Hard to measure total voltage and current at device ports at high frequencies Active devices may oscillate or self-destruct with shorts or opens S-parameters Relate to familiar measurements (gain, loss, reflection coefficient, etc.) Relatively easy to measure Can cascade S-parameters of multiple devices to predict system performance Analytically convenient CAD programs Flow-graph analysis Can compute H, Y, or Z parameters from S-parameters if desired S 21 Transmitted a 1 S b 11 2 DUT Reflected S 22 b Port 1 Port 2 Reflected 1 a 2 Transmitted S 12 b 1 = S 11 a 1 + S 12 a 2 b 2 = S 21 a 1 + S 22 a 2 Figure 16. Limitations of H, Y, and Z parameters (Why use S-parameters?)
16 16 Keysight Understanding the Fundamental Principles of Vector Network Analysis Application Note Network Characterization (Continued) Forward b 2 a 1 S 11 Z 0 b Reflected DUT load 1 a 2 = 0 S 21 Transmitted S 11 = Reflected S 21 = Transmitted = b 1 a 1 a 2 = 0 = b 2 a 1 a 2 = 0 S 22 = Reflected S 12 = Transmitted 2 = b a 2 1 = b a 2 a 1 = 0 a 1 = 0 Z 0 load a 1 = 0 b 1 DUT Transmitted S 12 b 2 S 22 Reflected a 2 Reverse Figure 17. Measuring S-parameters Forward S-parameters are determined by measuring the magnitude and phase of the incident, reflected, and transmitted signals when the output is terminated in a load that is precisely equal to the characteristic impedance of the test system. In the case of a simple two-port network, S 11 is equivalent to the input complex reflection coefficient or impedance of the DUT, while S 21 is the forward complex transmission coefficient. By placing the source at the output port of the DUT and terminating the input port in a perfect load, it is possible to measure the other two (reverse) S-parameters. Parameter S 22 is equivalent to the output complex reflection coefficient or output impedance of the DUT while S 12 is the reverse complex transmission coefficient (Figure 17).
17 17 Keysight Understanding the Fundamental Principles of Vector Network Analysis Application Note Related Literature Exploring the Architectures of Network Analyzers, Application Note, literature number E Applying Error Correction to Network Analyzer Measurements, Application Note, literature number E Network Analyzer Measurements: Filter and Amplifier Examples, Application Note, literature number E Keysight Services Flexible service solutions to minimize downtime and reduce the lifetime cost of ownership. Keysight Infoline Keysight s insight to best in class information management. Free access to your Keysight equipment company reports and e-library. This information is subject to change without notice. Keysight Technologies, Published in USA, May 15, E
Understanding the Fundamental Principles of Vector Network Analysis Application Note 1287-1 Table of Contents Page Introduction 2 Measurements in Communications Systems 2 Importance of Vector Measurements
RF Network Analyzer Basics A tutorial, information and overview about the basics of the RF Network Analyzer. What is a Network Analyzer and how to use them, to include the Scalar Network Analyzer (SNA),
VECTOR NETWORK ANALYZER PLANAR TR1300/1 DATA SHEET Frequency range: 300 khz to 1.3 GHz Measured parameters: S11, S21 Dynamic range of transmission measurement magnitude: 130 db Measurement time per point:
S-Parameters and Related Quantities Sam Wetterlin 10/20/09 Basic Concept of S-Parameters S-Parameters are a type of network parameter, based on the concept of scattering. The more familiar network parameters
1 The purpose of this paper is as follows: Examine the architectures of modern vector network analyzers (VNAs) Provide insight into nonlinear characterizations of amplifiers, mixers, and converters using
Hiroshi Kikuyama Verigy Japan 1. Introduction In RF/Microwave device testing that involves higher load board density, more complex impedance matching approaches need to be understood. Vector Network nalysis
Experiment 7: Familiarization with the Network Analyzer Measurements to characterize networks at high frequencies (RF and microwave frequencies) are usually done in terms of scattering parameters (S parameters).
David Ballo ewlett-packard Company Microwave Instruments Division 1400 Fountaingrove Parkway Santa Rosa, California 95403 U.S.A. 1997 Back to Basics Seminar (c) ewlett-packard Company Abstract This presentation
Product: Vector Network Analyzer R&S ZVB Performing Amplifier Measurements with the Vector Network Analyzer ZVB Application Note This document describes typical measurements that are required to be made
RF-Microwaves formulas - -port systems s-parameters: Considering a voltage source feeding into the DUT with a source impedance of. E i E r DUT The voltage into the DUT is composed of 2 parts, an incident
EEE 194 RF Laboratory Exercise 5 1 Laboratory #5: RF Filter Design I. OBJECTIVES A. Design a third order low-pass Chebyshev filter with a cutoff frequency of 330 MHz and 3 db ripple with equal terminations
Filters In the operation of electronic systems and circuits, the basic function of a filter is to selectively pass, by frequency, desired signals and to suppress undesired signals. The amount of insertion
POWER DIVIDERS AND DIRECTIONAL COUPLERS A directional coupler is a passive device which couples part of the transmission power by a known amount out through another port, often by using two transmission
Reflection Coefficient Applications in Test Measurements Bernhart A. Gebs Senior Product Development Engineer Belden Electronics Division Reflection coefficient is the basis of several measurements common
Engineering Sciences 151 Electromagnetic Communication Laboratory Assignment 3 Fall Term 1998-99 WAVE PROPAGATION II: HIGH FREQUENCY SLOTTED LINE AND REFLECTOMETER MEASUREMENTS OBJECTIVES: To build greater
- EE142 Fall 07 Lightwave Analogy to RF Energy Incident Transmitted Reflected Lightwave DUT RF Why Do We Need to Test Components? Verify specifications of building blocks for more complex RF systems Create
0HDVXULQJWKHHOHFWULFDOSHUIRUPDQFH FKDUDFWHULVWLFVRI5),)DQGPLFURZDYHVLJQDO SURFHVVLQJFRPSRQHQWV The treatment given here is introductory, and will assist the reader who wishes to consult the standard texts
Phase Shifters, Vector Modulators, Delay-Lines and Frequency Translators Application Notes G.T. Microwave has amassed a wide product line of passive, analog and digital RF and Microwave Integrated Components.
VCO Phase noise Characterizing Phase Noise The term phase noise is widely used for describing short term random frequency fluctuations of a signal. Frequency stability is a measure of the degree to which
ENGR 210 Lab 11 Response of Passive RC Filters The objective of this lab is to introduce you to the frequency-dependent nature of the impedance of a capacitor and the impact of that frequency dependence
Physics 272 April 8 Spring 2014 http://www.phys.hawaii.edu/~philipvd/pvd_14_spring_272_uhm.html Prof. Philip von Doetinchem firstname.lastname@example.org Phys272 - Spring 14 - von Doetinchem - 218 L-C in parallel
ECE 584 Microwave Engineering Laboratory Experiments 1. The Slotted Line Introduction: In this experiment we will use a waveguide slotted line to study the basic behavior of standing waves and to measure
Data Transmission Concepts and terminology Transmission terminology Transmission from transmitter to receiver goes over some transmission medium using electromagnetic waves Guided media. Waves are guided
MEASUREMENT UNCERTAINTY IN VECTOR NETWORK ANALYZER W. Li, J. Vandewege Department of Information Technology (INTEC) University of Gent, St.Pietersnieuwstaat 41, B-9000, Gent, Belgium Abstract: Precision
Agilent AN 154 S-Parameter Design Application Note Introduction The need for new high-frequency, solid-state circuit design techniques has been recognized both by microwave engineers and circuit designers.
RF measurements, tools and equipment E. B. Boskamp, A. Nabetani, J. Tropp (email@example.com) INTRODUCTION I am often asked by researchers what kind of equipment is needed to set up an RF lab. The
Chapter 8 Introduction to Alternating Current and Voltage Objectives Identify a sinusoidal waveform and measure its characteristics Describe how sine waves are generated Determine the various voltage and
RF Engineering Basic Concepts: The Smith Chart Fritz Caspers Contents Motivation Definition of the Smith Chart Navigation in the Smith Chart Application Examples Rulers RF Basic Concepts, Caspers, McIntosh,
Current Probes, More Useful Than You Think Training and design help in most areas of Electrical Engineering Copyright 1998 Institute of Electrical and Electronics Engineers. Reprinted from the IEEE 1998
WHITEPAPER INTRODUCING POWER SUPPLIES AND PLASMA Critical thin-film processes such as deposition and etching take place in a vacuum SYSTEMS chamber in the presence of a plasma. A plasma is an electrically
From November 2003 High Frequency Electronics Copyright 2003 Summit Technical Media, LLC Ordinary Vector Network Analyzers Get Differential Port Measurement Capability By Dale D. Henkes Applied Computational
C esonant Circuits Andrew McHutchon April 20, 203 Capacitors and Inductors There is a lot of inconsistency when it comes to dealing with reactances of complex components. The format followed in this document
From September 2005 High Frequency Electronics Copyright 2005 Summit Technical Media Conditioning and Correction of Arbitrary Waveforms Part 2: Other Impairments By Mike Griffin and John Hansen Agilent
Antenna Properties and their impact on Wireless System Performance Dr. Steven R. Best Cushcraft Corporation 48 Perimeter Road Manchester, NH 03013 Phone (603) 627-7877 FAX: (603) 627-1764 Email: firstname.lastname@example.org
ET 438a Automatic Control Systems Technology Laboratory 4 Practical Differentiator Response Objective: Design a practical differentiator circuit using common OP AMP circuits. Test the frequency response
Measurements on Transmission Lines Power and Attenuation Measurements Although a variety of instruments measure power, the most accurate instrument is a power meter and a power sensor. The sensor is an
FILTER CIRCUITS Introduction Circuits with a response that depends upon the frequency of the input voltage are known as filters. Filter circuits can be used to perform a number of important functions in
Basic Electrical Theory Impedance PJM State & Member Training Dept. PJM 2014 10/24/2013 Objectives Identify the components of Impedance in AC Circuits Calculate the total Impedance in AC Circuits Identify
Optimizing IP3 and ACPR Measurements Table of Contents 1. Overview... 2 2. Theory of Intermodulation Distortion... 2 3. Optimizing IP3 Measurements... 4 4. Theory of Adjacent Channel Power Ratio... 9 5.
EE246 Microwave Engineering 29 Oct. 1999 Leeson H. O. #31 Autumn 1999 Impedance Matching Why Impedance Match? A question often asked by people new to the microwave field is, "what is so important about
Physics 333 Experiment #3 Fall 211 Filters and Waveform Shaping Purpose The aim of this experiment is to study the frequency filtering properties of passive (R, C, and L) circuits for sine waves, and the
EE 311: Electrical Engineering Junior Lab Active Filter Design (Sallen-Key Filter) Objective The purpose of this experiment is to design a set of second-order Sallen-Key active filters and to investigate
[ Assignment View ] [ Pri Eðlisfræði 2, vor 2007 31. Alternating Current Circuits Assignment is due at 2:00am on Wednesday, March 21, 2007 Credit for problems submitted late will decrease to 0% after the
TCOM 370 NOTES 99-4 BANDWIDTH, FREQUENCY RESPONSE, AND CAPACITY OF COMMUNICATION LINKS 1. Bandwidth: The bandwidth of a communication link, or in general any system, was loosely defined as the width of
Module 8 AC Theory Filters & Wave Shaping Passive Filters & Wave Shaping What you'll learn in Module 8. Module 8 Introduction Recognise passive filters with reference to their response curves. High pass,
University of Pennsylvania Department of Electrical and Systems Engineering ESE06: Electrical Circuits and Systems II Lab Amplitude Modulated Radio Frequency Transmission System Mini-Project Part : Receiver
A Network Analyzer For Active Components EEEfCom 29-30 Juni ULM Marc Vanden Bossche, NMDG Engineering Remi Tuijtelaars, BSW Copyright 2005 NMDG Engineering Version 2 Outline Review of S-parameters Theory
TRANSFORMERS Transformers are key elements in power systems. In order to effectively transmit power over long distances without prohibitive line losses, the voltage from the generator (a maximum of output
ECE 584 Laboratory Experiments 2. The Vector Network Analyzer Introduction: In this experiment we will learn to use a Vector Network Analyzer to measure the magnitude and phase of reflection and transmission
ECE 435 INTRODUCTION TO THE MICROWAVE NETWORK ANALYZER Latest revision: October 1999 Introduction A vector network analyzer (VNA) is a device capable of measuring both the magnitude and phase of a sinusoidal
The objective of microwave circuit analysis is to move from the requirement to solve for all the fields and waves of a structure to an equivalent circuit that is amenable to all the tools of the circuit
1 SIGNAL GENERATORS and OSCILLOSCOPE CALIBRATION By Lannes S. Purnell FLUKE CORPORATION 2 This paper shows how standard signal generators can be used as leveled sine wave sources for calibrating oscilloscopes.
MANUAL RETURN LOSS MEASUREMENTS Sam Wetterlin email@example.com 10/5/08 This paper describes the procedure for using a reflection bridge (a/k/a return loss bridge ) to manually measure return loss
Agilent De-embedding and Embedding S-Parameter Networks Using a Vector Network Analyzer Application Note 1364-1 Introduction Traditionally RF and microwave components have been designed in packages with
EE 458/558 Microwave Circuit Design and Measurements Lab INTRODUCTION TO MICROWAVE MEASUREMENTS: DETECTION OF RF POWER AND STANDING WAVES Lab #2 The purpose of this lab is to gain a basic understanding
Understanding Power Splitters How they work, what parameters are critical, and how to select the best value for your application. Basically, a 0 splitter is a passive device which accepts an input signal
LS-296 Vector Network Analyzer Techniques to Measure WR340 Waveguide Windows T. L. Smith ASD / RF Group Advanced Photon Source Argonne National Laboratory June 26, 2002 Table of Contents 1) Introduction
Understanding Power Splitters how they work, what parameters are critical, and how to select the best value for your application. Basically, a 0 splitter is a passive device which accepts an input signal
FREQUENCY RESPONSE AND PASSIVE FILTERS LABORATORY In this experiment we will analytically determine and measure the frequency response of networks containing resistors, AC source/sources, and energy storage
Lab 7 AC Impedance and High-Pass Filters In this lab you will become familiar with the concept of AC impedance and apply it to the frequency response of a high-pass filter. 7.1 AC Impedance Just as Ohm
The W5JCK Guide to the Mathematic Equations Required for the Amateur Extra Class Exam This document contains every question from the Extra Class (Element 4) Question Pool* that requires one or more mathematical
First and Second Order Filters These functions are useful for the design of simple filters or they can be cascaded to form high-order filter functions First Order Filters General first order bilinear transfer
Chapter 12 RL Circuits Sinusoidal Response of RL Circuits The inductor voltage leads the source voltage Inductance causes a phase shift between voltage and current that depends on the relative values of
Why is impedance matching important: In general sputtering or etching applications, power generators are used to transfer either DC, Medium Frequency (typically in the several khz range) or RF (usually
Antennas and Propagation : Antenna Parameters Introduction Purpose Introduce standard terms and definitions for antennas Need a common language to specify performance Two types of parameters 1. Radiation
Chapter 12 RL Circuits Objectives Describe the relationship between current and voltage in an RL circuit Determine impedance and phase angle in a series RL circuit Analyze a series RL circuit Determine
CRP718 RF and Microwave Measurements Laboratory Experiment- MW2 Handout. Updated Jan 13, 2011 SPECTRUM ANALYZER BASED MEASUREMENTS (groups of 2 may omit part 2.1, but are advised to study the procedures
Welcome to this presentation which helps to highlight the capability of Genesys in the design of power amplifiers for LTE applications 1 2 3 4 5 Linearity is compromised due to the obvious output waveform.
S-parameter Simulation and Optimization Slide 5-1 S-parameters are Ratios Usually given in db as 20 log of the voltage ratios of the waves at the ports: incident, reflected, or transmitted. S-parameter
Introduction to the Smith Chart for the MSA Sam Wetterlin 10/12/09 Quick Review of Reflection Coefficient The Smith chart is a method of graphing reflection coefficients and impedance, and is often useful
Electromagnetic Waves Maxwell s equations predict the propagation of electromagnetic energy away from time-varying sources (current and charge) in the form of waves. Consider a linear, homogeneous, isotropic
University of Pennsylvania Moore School of Electrical Engineering ESE319 Electronic Circuits - Modeling and Measurement Techniques 1. Introduction. Students are often frustrated in their attempts to execute
TESTS OF 1 MHZ SIGNAL SOURCE FOR SPECTRUM ANALYZER CALIBRATION 7/8/08 Sam Wetterlin (Updated 7/19/08 to delete sine wave output) I constructed the 1 MHz square wave generator shown in the Appendix. This
The Class-D Amplifier (From the book Introduction to Electroacoustics and Audio Amplifier Design, Second Edition - Revised Printing, by W. Marshall Leach, Jr., published by Kendall/Hunt, c 2001.) A class-d
Measuring RF Parameters of Networks Bill Leonard NØCU NAØTC - 285 TechConnect Radio Club http://www.naøtc.org/ What is a Network? A Network is a group of electrical components connected is a specific way
Kathmandu University Department of Electrical and Electronics Engineering BASIC ELECTRICAL LAB (ENGG 103) EXPERIMENT 4:- MEASUREMENT OF REACTANCE OFFERED BY CAPACITOR IN DIFFERENT FREQUENCY FOR R-C CIRCUIT
Phil Sherrod W4PHS Direct versus Alternating Current Things We Can Measure Direct Current (DC) Alternating Current (AC) Voltage Voltage (peak, RMS) Current Current (peak, effective) Power True power, Apparent
EMC STANDARDS The EMC standards that a particular electronic product must meet depend on the product application (commercial or military) and the country in which the product is to be used. These EMC regulatory
CIRCUITS LABORATORY EXPERIMENT 3 AC Circuit Analysis 3.1 Introduction The steady-state behavior of circuits energized by sinusoidal sources is an important area of study for several reasons. First, the
Michael Hiebel Fundamentals of Vector Network Analysis TABIH OF CONTENTS Table of contents 1 Introduction 12 1.1 What is a network analyzer? 12 1.2 Wave quantities and S-parameters 13 1.3 Why vector network
LC Resonant Circuits Dr. Roger King June 01 Introduction Second-order systems are important in a wide range of applications including transformerless impedance-matching networks, frequency-selective networks,
Your consent to our cookies if you continue to use this website.