# Understanding the Fundamental Principles of Vector Network Analysis. Application Note Table of Contents. Page

Save this PDF as:

Size: px
Start display at page:

## Transcription

1 Understanding the Fundamental Principles of Vector Network Analysis Application Note Table of Contents Page Introduction 2 Measurements in Communications Systems 2 Importance of Vector Measurements 4 The Basis of and Reflected Power 5 The Smith Chart 5 Power Transfer Conditions 6 Network Analysis Terminology 9 Measuring Group Delay 11 Network Characterization 12

2 2 Introduction Network analysis is the process by which designers and manufacturers measure the electrical performance of the components and circuits used in more complex systems. When these systems are conveying signals with information content, we are most concerned with getting the signal from one point to another with maximum efficiency and minimum distortion. Vector network analysis is a method of accurately characterizing such components by measuring their effect on the amplitude and phase of swept-frequency and swept-power test signals. In this application note, the fundamental principles of vector network analysis will be reviewed. The discussion includes the common parameters that can be measured, including the concept of scattering parameters (S-parameters). RF fundamentals such as transmission lines and the Smith chart will also be reviewed. Hewlett-Packard Company offers a wide range of both scalar and vector network analyzers for characterizing components from DC to 110 GHz. These instruments are available with a wide range of options to simplify testing in both laboratory and production environments. Measurements in Communications Systems In any communications system, the effect of signal distortion must be considered. While we generally think of the distortion caused by nonlinear effects (for example, when intermodulation products are produced from desired carrier signals), purely linear systems can also introduce signal distortion. Linear systems can change the time waveform of signals passing through them by altering the amplitude or phase relationships of the spectral components that make up the signal. Let s examine the difference between linear and nonlinear behavior more closely. Linear devices impose magnitude and phase changes on input signals (Figure 1). Any sinusoid appearing at the input will also appear at the output, and at the same frequency. No new signals are created. Both active and passive nonlinear devices can shift an input signal in frequency or add other frequency components, such as harmonic and spurious signals. Large input signals can drive normally linear devices into compression or saturation, causing nonlinear operation. Figure 1. Linear versus Nonlinear Behavior Sin 360 * f * t Input DUT A t o A * Sin 360 * f ( t t ) A phase shift = to * 360 * f f 1 Output Linear behavior: input and output frequencies are the same (no additional frequencies created) output frequency only undergoes magnitude and phase change f 1 Nonlinear behavior: output frequency may undergo frequency shift (e.g. with mixers) additional frequencies created (harmonics, intermodulation) f 1

3 3 For linear distortion-free transmission, the amplitude response of the device under test (DUT) must be flat and the phase response must be linear over the desired bandwidth. As an example, consider a square-wave signal rich in high-frequency components passing through a bandpass filter that passes selected frequencies with little attenuation while attenuating frequencies outside of the passband by varying amounts. Even if the filter has linear phase performance, the out-of-band components of the square wave will be attenuated, leaving an output signal that, in this example, is more sinusoidal in nature (Figure 2). If the same square-wave input signal is passed through a filter that only inverts the phase of the third harmonic, but leaves the harmonic amplitudes the same, the output will be more impulse-like in nature (Figure 3). While this is true for the example filter, in general, the output waveform will appear with arbitrary distortion, depending on the amplitude and phase nonlinearities. Figure 2. Magnitude Variation with F(t) = sin wt + 1 /3 sin 3wt + 1 /5 sin 5wt Linear Network Magnitude Figure 3. Phase Variation with F(t) = sin wt + 1 /3 sin 3wt + 1 /5 sin 5wt Linear Network Magnitude

4 4 Figure 4. Nonlinear Induced Distortion Nonlinear Networks Saturation, crossover, intermodulation, and other nonlinear effects can cause signal distortion Nonlinear devices also introduce distortion (Figure 4). For example, if an amplifier is overdriven, the output signal clips because the amplifier is saturated. The output signal is no longer a pure sinusoid, and harmonics are present at multiples of the input frequency. Passive devices may also exhibit nonlinear behavior at high power levels, a good example of which is an L-C filter that uses inductors with magnetic cores. Magnetic materials often exhibit hysteresis effects that are highly nonlinear. Efficient transfer of power is another fundamental concern in communications systems. In order to efficiently convey, transmit or receive RF power, devices such as transmissions lines, antennas and amplifiers must present the proper impedance match to the signal source. Impedance mismatches occur when the real and imaginary parts of input and output impedances are not ideal between two connecting devices. Importance of Vector Measurements Measuring both magnitude and phase of components is important for several reasons. First, both measurements are required to fully characterize a linear network and ensure distortion-free transmission. To design efficient matching networks, complex impedance must be measured. Engineers developing models for computer-aided-engineering (CAE) circuit simulation programs require magnitude and phase data for accurate models. In addition, time-domain characterization requires magnitude and phase information in order to perform an inverse-fourier transform. Vector error correction, which improves measurement accuracy by removing the effects of inherent measurement-system errors, requires both magnitude and phase data to build an effective error model. Phase-measurement capability is very important even for scalar measurements such as return loss, in order to achieve a high level of accuracy (see Applying Error Correction to Network Analyzer Measurements, Hewlett-Packard Application Note ).

5 5 The Basis of and Reflected Power In its fundamental form, network analysis involves the measurement of incident, reflected, and transmitted waves that travel along transmission lines. Using optical wavelengths as an analogy, when light strikes a clear lens (the incident energy), some of the light is reflected from the lens surface, but most of it continues through the lens (the transmitted energy) (Figure 5). If the lens has mirrored surfaces, most of the light will be reflected and little or none will pass through it. While the wavelengths are different for RF and microwave signals, the principle is the same. Network analyzers accurately measure the incident, reflected, and transmitted energy, e.g., the energy that is launched onto a transmission line, reflected back down the transmission line toward the source (due to impedence mismatch), and successfully transmitted to the terminating device (such as an antenna). Figure 5. Lightwave Analogy to High- Device Characterization Lightwave Analogy Reflected Transmitted The Smith Chart The amount of reflection that occurs when characterizing a device depends on the impedance that the incident signal sees. Since any impedance can be represented with real and imaginary parts (R+jX or G+jB), they can be plotted on a rectilinear grid known as the complex impedance plane. Unfortunately, an open circuit (a common RF impedence) appears at infinity on the real axis, and therefore cannot be shown. The polar plot is useful because the entire impedance plane is covered. However, instead of plotting impedance directly, the complex reflection coefficient is displayed in vector form. The magnitude of the vector is the distance from the center of the display, and phase is displayed as the angle of vector referenced to a flat line from the center to the right-most edge. The drawback of polar plots is that impedance values cannot be read directly from the display.

6 6 Since there is a one-to-one correspondence between complex impedance and reflection coefficient, the positive real half of the complex impedance plane can be mapped onto the polar display. The result is the Smith chart. All values of reactance and all positive values of resistance from 0 to infinity fall within the outer circle of the Smith chart (Figure 6). On the Smith chart, loci of constant resistance appear as circles, while loci of constant reactance appear as arcs. Impedances on the Smith chart are always normalized to the characteristic impedance of the component or system of interest, usually 50 ohms for RF and microwave systems and 75 ohms for broadcast and cable-television systems. A perfect termination appears in the center of the Smith chart. Figure 6. Smith Chart Review +jx Polar plane 90 o R o o jx Rectilinear impedance plane Smith chart maps rectilinear impedance plane onto polar plane 0 Z L = Zo Γ = 0 Z Γ L = 0(short) = 1 ±180 O 90 o Constant X Constant R Z L= (open) = 1 0 O Γ Smith chart Power Transfer Conditions A perfectly matched condition must exist at a connection between two devices for maximum power transfer into a load, given a source resistance of R S and a load resistance of R L. This condition occurs when R L = R S, and is true whether the stimulus is a DC voltage source or a source of RF sine waves (Figure 7). When the source impedance is not purely resistive, maximum power transfer occurs when the load impedance is equal to the complex conjugate of the source impedance. This condition is met by reversing the sign of the imaginary part of the impedance. For example, if R S = j 0.3, then the complex conjugate is R S * = 0.6 j 0.3. The need for efficient power transfer is one of the main reasons for the use of transmission lines at higher frequencies. At very low frequencies (with much larger wavelengths), a simple wire is adequate for conducting power. The resistance of the wire is relatively low and has little effect on low-frequency signals. The voltage and current are the same no matter where a measurement is made on the wire.

7 7 At higher frequencies, wavelengths are comparable to or smaller than the length of the conductors in a high-frequency circuit, and power transmission can be thought of in terms of traveling waves. When the transmission line is terminated in its characteristic impedance, maximum power is transferred to the load. When the termination is not equal to the characteristic impedance, that part of the signal that is not absorbed by the load is reflected back to the source. If a transmission line is terminated in its characteristic impedance, no reflected signal occurs since all of the transmitted power is absorbed by the load (Figure 8). Looking at the envelope of the RF signal versus distance along the transmission line shows no standing waves because without reflections, energy flows in only one direction. Figure 7. Power Transfer R S Load Power (normalized) R L For complex impedances, maximum power transfer occurs when Z L = Z S * (conjugate match) Zs = R + jx R L / R S Maximum power is transferred when R L = R S Z L = Zs * = R jx Figure 8. Transmission Line Terminated with Z o Zs = Zo Zo = characteristic impedance of transmission line Zo V inc V refl = 0 (all the incident power is absorbed in the load) For reflection, a transmission line terminated in Zo behaves like an infinitely long transmission line

8 8 When the transmission line is terminated in a short circuit (which can sustain no voltage and therefore dissipates zero power), a reflected wave is launched back along the line toward the source (Figure 9). The reflected voltage wave must be equal in magnitude to the incident voltage wave and be 180 degrees out of phase with it at the plane of the load. The reflected and incident waves are equal in magnitude but traveling in the opposite directions. If the transmission line is terminated in an open-circuit condition (which can sustain no current), the reflected current wave will be 180 degrees out of phase with the incident current wave, while the reflected voltage wave will be in phase with the incident voltage wave at the plane of the load. This guarantees that the current at the open will be zero. The reflected and incident current waves are equal in magnitude, but traveling in the opposite directions. For both the short and open cases, a standing wave pattern is set up on the transmission line. The voltage valleys will be zero and the voltage peaks will be twice the incident voltage level. If the transmission line is terminated with say a 25-ohm resistor, resulting in a condition between full absorption and full reflection, part of the incident power is absorbed and part is reflected. The amplitude of the reflected voltage wave will be one-third that of the incident wave, and the two waves will be 180 degrees out of phase at the plane of the load. The valleys of the standing-wave pattern will no longer be zero, and the peaks will be less than those of the short and open cases. The ratio of the peaks to valleys will be 2:1. The traditional way of determining RF impedance was to measure VSWR using an RF probe/detector, a length of slotted transmission line, and a VSWR meter. As the probe was moved along the transmission line, the relative position and values of the peaks and valleys were noted on the meter. From these measurements, impedance could be derived. The procedure was repeated at different frequencies. Modern network analyzers measure the incident and reflected waves directly during a frequency sweep, and impedance results can be displayed in any number of formats (including VSWR). Figure 9. Transmission Line Terminated with Short, Open Zs = Zo V inc V refl o In phase (0 ) for open o Out of phase (180 ) for short For reflection, a transmission line terminated in a short or open reflects all power back to source

9 9 Network Analysis Terminology Now that we understand the fundamentals of electromagnetic waves, we must learn the common terms used for measuring them. Network analyzer terminology generally denotes measurements of the incident wave with the R or reference channel. The reflected wave is measured with the A channel, and the transmitted wave is measured with the B channel (Figure 10). With the amplitude and phase information in these waves, it is possible to quantify the reflection and transmission characteristics of a DUT. The reflection and transmission characteristics can be expressed as vector (magnitude and phase), scalar (magnitude only), or phase-only quantities. For example, return loss is a scalar measurement of reflection, while impedance is a vector reflection measurement. Ratioed measurements allow us to make reflection and transmission measurements that are independent of both absolute power and variations in source power versus frequency. Ratioed reflection is often shown as A/R and ratioed transmission as B/R, relating to the measurement channels in the instrument. Figure 10. Common Terms for High- Device Characterization R REFLECTION Reflected Reflected = A A R Transmitted B TRANSMISSION Transmitted = B R SWR S-Parameters S11,S22 Reflection Coefficient Γ, ρ Return Loss Impedance, Admittance R+jX, G+jB Gain / Loss S-Parameters S21,S12 Transmission Coefficient Τ,τ Group Delay Insertion Phase The most general term for ratioed reflection is the complex reflection coefficient, Γ or gamma (Figure 11). The magnitude portion of Γ is called ρ or rho. The reflection coefficient is the ratio of the reflected signal voltage level to the incident signal voltage level. For example, a transmission line terminated in its characteristic impedance Z o, will have all energy transferred to the load so V refl = 0 and ρ = 0. When the impedance of the load, Z L is not equal to the characteristic impedance, energy is reflected and ρ is greater than zero. When the load impedance is equal to a short or open circuit, all energy is reflected and ρ = 1. As a result, the range of possible values for ρ is 0 to 1.

10 10 Figure 11. Reflection Parameters Reflection Coefficient = V reflected Γ = ρ V incident Φ Return loss = 20 log(ρ), ρ = Γ = Z L Z O Z L + Z O Emax Emin Voltage Standing Wave Ratio VSWR = Emax Emin = 1 + ρ 1 ρ No reflection (Z L = Zo) ρ Full reflection (Z L = open, short) 0 1 db RL 0 db 1 VSWR Return loss is a way to express the reflection coefficient in logarithmic terms (decibels). Return loss is the number of decibels that the reflected signal is below the incident signal. Return loss is always expressed as a positive number and varies between infinity for a load at the characteristic impedance and 0 db for an open or short circuit. Another common term used to express reflection is voltage standing wave ratio (VSWR), which is defined as the maximum value of the RF envelope over the minimum value of the RF envelope. It is related to ρ as (1 + ρ)/(1 ρ). VSWR ranges from 1 (no reflection) to infinity (full reflection). The transmission coefficient is defined as the transmitted voltage divided by the incident voltage (Figure 12). If the absolute value of the transmitted voltage is greater than the absolute value of the incident voltage, a DUT or system is said to have gain. If the absolute value of the transmitted voltage is less than the absolute value of the incident voltage, the DUT or system is said to have attenuation or insertion loss. The phase portion of the transmission coefficient is called insertion phase. Figure 12. Transmission Parameters V DUT V Transmitted Transmission Coefficient = Τ = V Transmitted V = τ φ Insertion Loss (db) = 20 Log V Trans V Inc = 20 log τ Gain (db) = 20 Log V Trans V Inc = 20 log τ

11 11 Direct examination of insertion phase usually does not provide useful information. This is because the insertion phase has a large (negative) slope with respect to frequency due to the electrical length of the DUT. The slope is proportional to the length of the DUT. Since it is only deviation from linear phase that causes distortion in communications systems, it is desirable to remove the linear portion of the phase response to analyze the remaining nonlinear portion. This can be done by using the electrical delay feature of a network analyzer to mathematically cancel the average electrical length of the DUT. The result is a high-resolution display of phase distortion or deviation from linear phase (Figure 13). Figure 13. Deviation from Linear Phase Use electrical delay to remove linear portion of phase response o Phase 45 /Div RF filter response Linear electrical length added (Electrical delay function) + yields Deviation from linear phase o Phase 1 /Div Low resolution High resolution Measuring Group Delay Another useful measure of phase distortion is group delay (Figure 14). This parameter is a measure of the transit time of a signal through a DUT versus frequency. Group delay can be calculated by differentiating the DUT s phase response versus frequency. It reduces the linear portion of the phase response to a constant value, and transforms the deviations from linear phase into deviations from constant group delay, (which causes phase distortion in communications systems). The average delay represents the average signal transit time through a DUT. Figure 14. What Is Group Delay? Phaseφ φ Group Delay (t ) g ω ω = δ φ d ω = o in radians in radians/sec in degrees in Hz * φ ω φ f ω 2π f ( = ) d φ d f tg Group Delay t o Group Delay Average Delay Deviation from constant group delay indicates distortion Average delay indicates transit time

12 12 Depending on the device, both deviation from linear phase and group delay may be measured, since both can be important. Specifying a maximum peak-to-peak phase ripple in a device may not be sufficient to completely characterize it, since the slope of the phase ripple depends on the number of ripples that occur per unit of frequency. Group delay takes this into account because it is the differentiated phase response. Group delay is often a more easily interpreted indication of phase distortion (Figure 15). Figure 15. Why Measure Group Delay? Phase Phase dφ dω f dφ dω f Group Delay Group Delay f f Same peak-to-peak phase ripple can result in different group delay Network Characterization In order to completely characterize an unknown linear two-port device, we must make measurements under various conditions and compute a set of parameters. These parameters can be used to completely describe the electrical behavior of our device (or network), even under source and load conditions other than when we made our measurements. Low-frequency device or network characterization is usually based on measurement of H, Y, and Z parameters. To do this, the total voltage and current at the input or output ports of a device or nodes of a network must be measured. Furthermore, measurements must be made with open-circuit and short-circuit conditions. Since it is difficult to measure total current or voltage at higher frequencies, S-parameters are generally measured instead (Figure 16). These parameters relate to familiar measurements such as gain, loss, and reflection coefficient. They are relatively simple to measure, and do not require connection of undesirable loads to the DUT. The measured S-parameters of multiple devices can be cascaded to predict overall system performance. S-parameters are readily used in both linear and nonlinear CAE circuit simulation tools, and H, Y, and Z parameters can be derived from S-parameters when necessary. The number of S-parameters for a given device is equal to the square of the number of ports. For example, a two-port device has four S-parameters. The numbering convention for S-parameters is that the first number following the S is the port at which energy emerges, and the second number is the port at which energy enters. So S 21 is a measure of power emerging from Port 2 as a result of applying an RF stimulus to Port 1. When the numbers are the same (e.g. S 11 ), a reflection measurement is indicated.

13 13 Figure 16. Limitations of H, Y, and Z Parameters (Why Use S-parameters?) H,Y, and Z parameters Hard to measure total voltage and current at device ports at high frequencies Active devices may oscillate or self-destruct with shorts or opens S-parameters Relate to familiar measurements S (gain, loss, reflection coefficient, etc.) 21 Transmitted a 1 Relatively easy to measure S 11 b2 Reflected DUT S 22 Can cascade S-parameters of multiple Port 1 Port 2 b Reflected devices to predict system performance 1 a 2 Transmitted S 12 Analytically convenient b 1 = S 11 a CAD programs 1 + S 12 a 2 b 2 = S 21 a 1 + S 22 a 2 Flow-graph analysis Can compute H, Y, or Z parameters from S-parameters if desired Figure 17. Measuring S-Parameters S 21 b Transmitted 2 a 1 Forward S 11 Z 0 b Reflected DUT Load 1 a 2 =0 S 11 = Reflected = b 1 a 1 a 2 =0 S 21 = Transmitted = b 2 a a 2 =0 1 S 22 = Reflected S 12 = Transmitted = b a = b a2 a =0 1 a =0 1 Z 0 Load a 1 =0 b 1 DUT Transmitted S 12 S 22 Reflected b 2 a 2 Reverse Forward S-parameters are determined by measuring the magnitude and phase of the incident, reflected, and transmitted signals when the output is terminated in a load that is precisely equal to the characteristic impedance of the test system. In the case of a simple two-port network, S 11 is equivalent to the input complex reflection coefficient or impedance of the DUT, while S 21 is the forward complex transmission coefficient. By placing the source at the output port of the DUT and terminating the input port in a perfect load, it is possible to measure the other two (reverse) S-parameters. Parameter S 22 is equivalent to the output complex reflection coefficient or output impedance of the DUT while S 12 is the reverse complex transmission coefficient (Figure 17). Suggested Reading Exploring the Architectures of Network Analyzers, Hewlett-Packard Application Note Applying Error Correction to Network Analyzer Measurements, Hewlett-Packard Application Note Network Analyzer Measurements: Filter and Amplifier Examples, Hewlett-Packard Application Note

14 For more information about Hewlett-Packard test and measurement products, applications, services, and for a current sales office listing, visit our web site, You can also contact one of the following centers and ask for a test and measurement sales representative. United States: Hewlett-Packard Company Test and Measurement Call Center P.O. Box 4026 Englewood, CO Canada: Hewlett-Packard Canada Ltd Spectrum Way Mississauga, Ontario L4W 5G1 (905) Europe: Hewlett-Packard European Marketing Centre P.O. Box AZ Amstelveen The Netherlands (31 20) Japan: Hewlett-Packard Japan Ltd. Measurement Assistance Center 9-1, Takakura-Cho, Hachioji-Shi, Tokyo 192, Japan Tel: (81-426) Fax: (81-426) Latin America: Hewlett-Packard Latin American Region Headquarters 5200 Blue Lagoon Drive, 9th Floor Miami, Florida 33126, U.S.A. (305) /4220 Australia/New Zealand: Hewlett-Packard Australia Ltd Joseph Street Blackburn, Victoria 3130, Australia Asia Pacific: Hewlett-Packard Asia Pacific Ltd /F Shell Tower, s Square, 1 Matheson Street, Causeway Bay, Hong Kong Tel: (852) Fax: (852) Data Subject to Change Copyright 1997 Hewlett-Packard Company Printed in U.S.A. 5/ E

### Keysight Technologies Understanding the Fundamental Principles of Vector Network Analysis. Application Note

Keysight Technologies Understanding the Fundamental Principles of Vector Network Analysis Application Note Introduction Network analysis is the process by which designers and manufacturers measure the

### HP 71500A Microwave Transition Analyzer

HP 71500A Microwave Transition Analyzer Non-linear component testing dc to 40 GHz Two channels Measurements Phase settling Rise/fall times On/off ratios Delay Switching time Peak/average power Group delay

### Applying Error Correction to Network Analyzer Measurements. Application Note 1287-3. Table of Contents. Page

Applying Error Correction to Network Analyzer Measurements Application Note 287-3 Table of Contents Page Introduction 2 Sources of Errors and Types of Errors 3 Types of Error Correction 4 One-Port 4 The

### RF Network Analyzer Basics

RF Network Analyzer Basics A tutorial, information and overview about the basics of the RF Network Analyzer. What is a Network Analyzer and how to use them, to include the Scalar Network Analyzer (SNA),

### Impedance 50 (75 connectors via adapters)

VECTOR NETWORK ANALYZER PLANAR TR1300/1 DATA SHEET Frequency range: 300 khz to 1.3 GHz Measured parameters: S11, S21 Dynamic range of transmission measurement magnitude: 130 db Measurement time per point:

### Fundamentals of Vector Signal Analysis

Hiroshi Kikuyama Verigy Japan 1. Introduction In RF/Microwave device testing that involves higher load board density, more complex impedance matching approaches need to be understood. Vector Network nalysis

### Agilent Network Analyzer Basics

Agilent Network Analyzer Basics 2 Abstract This presentation covers the principles of measuring high-frequency electrical networks with network analyzers. You will learn what kinds of measurements are

### The purpose of this paper is as follows: Examine the architectures of modern vector network analyzers (VNAs) Provide insight into nonlinear

1 The purpose of this paper is as follows: Examine the architectures of modern vector network analyzers (VNAs) Provide insight into nonlinear characterizations of amplifiers, mixers, and converters using

### S-Parameters and Related Quantities Sam Wetterlin 10/20/09

S-Parameters and Related Quantities Sam Wetterlin 10/20/09 Basic Concept of S-Parameters S-Parameters are a type of network parameter, based on the concept of scattering. The more familiar network parameters

### Network Analyzer Basics

David Ballo ewlett-packard Company Microwave Instruments Division 1400 Fountaingrove Parkway Santa Rosa, California 95403 U.S.A. 1997 Back to Basics Seminar (c) ewlett-packard Company Abstract This presentation

### Experiment 7: Familiarization with the Network Analyzer

Experiment 7: Familiarization with the Network Analyzer Measurements to characterize networks at high frequencies (RF and microwave frequencies) are usually done in terms of scattering parameters (S parameters).

### Performing Amplifier Measurements with the Vector Network Analyzer ZVB

Product: Vector Network Analyzer R&S ZVB Performing Amplifier Measurements with the Vector Network Analyzer ZVB Application Note This document describes typical measurements that are required to be made

### The Smith chart. Smith chart components. The normalized impedance line CHAPTER

26 CHAPTER The Smith chart The mathematics of transmission lines, and certain other devices, becomes cumbersome at times, especially when dealing with complex impedances and nonstandard situations. In

### Agilent PN 8753-1 RF Component Measurements: Amplifier Measurements Using the Agilent 8753 Network Analyzer. Product Note

Agilent PN 8753-1 RF Component Measurements: Amplifier Measurements Using the Agilent 8753 Network Analyzer Product Note 2 3 4 4 4 4 6 7 8 8 10 10 11 12 12 12 13 15 15 Introduction Table of contents Introduction

### Agilent AN 1316 Optimizing Spectrum Analyzer Amplitude Accuracy

Agilent AN 1316 Optimizing Spectrum Analyzer Amplitude Accuracy Application Note RF & Microwave Spectrum Analyzers Table of Contents 3 3 4 4 5 7 8 8 13 13 14 16 16 Introduction Absolute versus relative

### HP 70950B OPTICAL SPECTRUM ANALYZER

HP 71450B, 71451B, and 71452B Optical Spectrum Analyzers Technical Specifications Spectral Measurements from 600 to 1700 nm HP 70950B OPTICAL SPECTRUM ANALYZER OPTICAL INPUT The HP 71450B, 71451B, and

### RF-Microwaves formulas - 1-port systems

RF-Microwaves formulas - -port systems s-parameters: Considering a voltage source feeding into the DUT with a source impedance of. E i E r DUT The voltage into the DUT is composed of 2 parts, an incident

### Network Analyzer Basics- EE142 Fall 07

- EE142 Fall 07 Lightwave Analogy to RF Energy Incident Transmitted Reflected Lightwave DUT RF Why Do We Need to Test Components? Verify specifications of building blocks for more complex RF systems Create

### Modern Definition of Terms

Filters In the operation of electronic systems and circuits, the basic function of a filter is to selectively pass, by frequency, desired signals and to suppress undesired signals. The amount of insertion

### Interface. Displaying Transactions

H HP E2492S for the Intel Pentium II Processor with MMX Technology at 333 MHz Product Overview Requires both the HP 16500B/C and the HP 16505A HP 16505A Prototype Analyzer Probe Adapter Interface HP 16500C

### Laboratory #5: RF Filter Design

EEE 194 RF Laboratory Exercise 5 1 Laboratory #5: RF Filter Design I. OBJECTIVES A. Design a third order low-pass Chebyshev filter with a cutoff frequency of 330 MHz and 3 db ripple with equal terminations

### Engineering Sciences 151. Electromagnetic Communication Laboratory Assignment 3 Fall Term 1998-99

Engineering Sciences 151 Electromagnetic Communication Laboratory Assignment 3 Fall Term 1998-99 WAVE PROPAGATION II: HIGH FREQUENCY SLOTTED LINE AND REFLECTOMETER MEASUREMENTS OBJECTIVES: To build greater

### MEASUREMENT UNCERTAINTY IN VECTOR NETWORK ANALYZER

MEASUREMENT UNCERTAINTY IN VECTOR NETWORK ANALYZER W. Li, J. Vandewege Department of Information Technology (INTEC) University of Gent, St.Pietersnieuwstaat 41, B-9000, Gent, Belgium Abstract: Precision

### POWER DIVIDERS AND DIRECTIONAL COUPLERS

POWER DIVIDERS AND DIRECTIONAL COUPLERS A directional coupler is a passive device which couples part of the transmission power by a known amount out through another port, often by using two transmission

### Agilent AN 1315 Optimizing RF and Microwave Spectrum Analyzer Dynamic Range. Application Note

Agilent AN 1315 Optimizing RF and Microwave Spectrum Analyzer Dynamic Range Application Note Table of Contents 3 3 3 4 4 4 5 6 7 7 7 7 9 10 10 11 11 12 12 13 13 14 15 1. Introduction What is dynamic range?

### Reflection Coefficient Applications in Test Measurements Bernhart A. Gebs Senior Product Development Engineer Belden Electronics Division. Rho.

Reflection Coefficient Applications in Test Measurements Bernhart A. Gebs Senior Product Development Engineer Belden Electronics Division Reflection coefficient is the basis of several measurements common

### VCO Phase noise. Characterizing Phase Noise

VCO Phase noise Characterizing Phase Noise The term phase noise is widely used for describing short term random frequency fluctuations of a signal. Frequency stability is a measure of the degree to which

### 0HDVXULQJWKHHOHFWULFDOSHUIRUPDQFH FKDUDFWHULVWLFVRI5),)DQGPLFURZDYHVLJQDO SURFHVVLQJFRPSRQHQWV

0HDVXULQJWKHHOHFWULFDOSHUIRUPDQFH FKDUDFWHULVWLFVRI5),)DQGPLFURZDYHVLJQDO SURFHVVLQJFRPSRQHQWV The treatment given here is introductory, and will assist the reader who wishes to consult the standard texts

### April 8. Physics 272. Spring Prof. Philip von Doetinchem

Physics 272 April 8 Spring 2014 http://www.phys.hawaii.edu/~philipvd/pvd_14_spring_272_uhm.html Prof. Philip von Doetinchem philipvd@hawaii.edu Phys272 - Spring 14 - von Doetinchem - 218 L-C in parallel

### 1. The Slotted Line. ECE 584 Microwave Engineering Laboratory Experiments. Introduction:

ECE 584 Microwave Engineering Laboratory Experiments 1. The Slotted Line Introduction: In this experiment we will use a waveguide slotted line to study the basic behavior of standing waves and to measure

### T = 1 f. Phase. Measure of relative position in time within a single period of a signal For a periodic signal f(t), phase is fractional part t p

Data Transmission Concepts and terminology Transmission terminology Transmission from transmitter to receiver goes over some transmission medium using electromagnetic waves Guided media. Waves are guided

### Phase Shifters, Vector Modulators, Delay-Lines and Frequency Translators Application Notes

Phase Shifters, Vector Modulators, Delay-Lines and Frequency Translators Application Notes G.T. Microwave has amassed a wide product line of passive, analog and digital RF and Microwave Integrated Components.

### ENGR 210 Lab 11 Frequency Response of Passive RC Filters

ENGR 210 Lab 11 Response of Passive RC Filters The objective of this lab is to introduce you to the frequency-dependent nature of the impedance of a capacitor and the impact of that frequency dependence

### Agilent 8510-13 Measuring Noninsertable Devices

Agilent 8510-13 Measuring Noninsertable Devices Product Note A new technique for measuring components using the 8510C Network Analyzer Introduction The majority of devices used in real-world microwave

### Antenna Properties and their impact on Wireless System Performance. Dr. Steven R. Best. Cushcraft Corporation 48 Perimeter Road Manchester, NH 03013

Antenna Properties and their impact on Wireless System Performance Dr. Steven R. Best Cushcraft Corporation 48 Perimeter Road Manchester, NH 03013 Phone (603) 627-7877 FAX: (603) 627-1764 Email: sbest@cushcraft.com

### RF Engineering Basic Concepts: The Smith Chart

RF Engineering Basic Concepts: The Smith Chart Fritz Caspers Contents Motivation Definition of the Smith Chart Navigation in the Smith Chart Application Examples Rulers RF Basic Concepts, Caspers, McIntosh,

### Current Probes, More Useful Than You Think

Current Probes, More Useful Than You Think Training and design help in most areas of Electrical Engineering Copyright 1998 Institute of Electrical and Electronics Engineers. Reprinted from the IEEE 1998

### Yesterday s discrete. Ordinary Vector Network Analyzers Get Differential Port Measurement Capability DIFFERENTIAL MEASUREMENTS

From November 2003 High Frequency Electronics Copyright 2003 Summit Technical Media, LLC Ordinary Vector Network Analyzers Get Differential Port Measurement Capability By Dale D. Henkes Applied Computational

### Optimizing IP3 and ACPR Measurements

Optimizing IP3 and ACPR Measurements Table of Contents 1. Overview... 2 2. Theory of Intermodulation Distortion... 2 3. Optimizing IP3 Measurements... 4 4. Theory of Adjacent Channel Power Ratio... 9 5.

### Conditioning and Correction of Arbitrary Waveforms Part 2: Other Impairments

From September 2005 High Frequency Electronics Copyright 2005 Summit Technical Media Conditioning and Correction of Arbitrary Waveforms Part 2: Other Impairments By Mike Griffin and John Hansen Agilent

### Agilent AN 154 S-Parameter Design Application Note

Agilent AN 154 S-Parameter Design Application Note Introduction The need for new high-frequency, solid-state circuit design techniques has been recognized both by microwave engineers and circuit designers.

### Introduction to Network Analyzer Measurements

Introduction to Network Analyzer Measurements Introduction to Network Analyzer Measurements 1 Table of Contents 1. Introduction to Network Analyzer Measurements... 3 VNA Basics...3 Applications for Network

### Chapter 8. Introduction to Alternating Current and Voltage. Objectives

Chapter 8 Introduction to Alternating Current and Voltage Objectives Identify a sinusoidal waveform and measure its characteristics Describe how sine waves are generated Determine the various voltage and

### A Network Analyzer For Active Components

A Network Analyzer For Active Components EEEfCom 29-30 Juni ULM Marc Vanden Bossche, NMDG Engineering Remi Tuijtelaars, BSW Copyright 2005 NMDG Engineering Version 2 Outline Review of S-parameters Theory

### RF measurements, tools and equipment E. B. Boskamp, A. Nabetani, J. Tropp (eddy.boskamp@med.ge.com)

RF measurements, tools and equipment E. B. Boskamp, A. Nabetani, J. Tropp (eddy.boskamp@med.ge.com) INTRODUCTION I am often asked by researchers what kind of equipment is needed to set up an RF lab. The

### 9 Measurements on Transmission Lines

Measurements on Transmission Lines Power and Attenuation Measurements Although a variety of instruments measure power, the most accurate instrument is a power meter and a power sensor. The sensor is an

### Critical thin-film processes such as deposition and etching take place in a vacuum

WHITEPAPER INTRODUCING POWER SUPPLIES AND PLASMA Critical thin-film processes such as deposition and etching take place in a vacuum SYSTEMS chamber in the presence of a plasma. A plasma is an electrically

### Bharathwaj Muthuswamy EE100 Active Filters

Bharathwaj Muthuswamy EE100 mbharat@cory.eecs.berkeley.edu 1. Introduction Active Filters In this chapter, we will deal with active filter circuits. Why even bother with active filters? Answer: Audio.

### Scalar Network Analysis with the HP 8590 Series Spectrum Analyzers Product Overview

This literature was published years prior to the establishment of Agilent Technologies as a company independent from Hewlett-Packard and describes products or services now available through Agilent. It

### FILTER CIRCUITS. A filter is a circuit whose transfer function, that is the ratio of its output to its input, depends upon frequency.

FILTER CIRCUITS Introduction Circuits with a response that depends upon the frequency of the input voltage are known as filters. Filter circuits can be used to perform a number of important functions in

### Part 2: Receiver and Demodulator

University of Pennsylvania Department of Electrical and Systems Engineering ESE06: Electrical Circuits and Systems II Lab Amplitude Modulated Radio Frequency Transmission System Mini-Project Part : Receiver

### Application Note Noise Frequently Asked Questions

: What is? is a random signal inherent in all physical components. It directly limits the detection and processing of all information. The common form of noise is white Gaussian due to the many random

### EE246 Microwave Engineering 29 Oct Leeson H. O. #31 Autumn Impedance Matching

EE246 Microwave Engineering 29 Oct. 1999 Leeson H. O. #31 Autumn 1999 Impedance Matching Why Impedance Match? A question often asked by people new to the microwave field is, "what is so important about

### Video Feedthrough. A common source of RF distortion in pulse-modulated RF sources

Video Feedthrough El Toro 68C/69B Microwave Synthesizers Application Note A common source of RF distortion in pulse-modulated RF sources Introduction Video feedthrough is a very misunderstood term, which

### Basic Electrical Theory

Basic Electrical Theory Impedance PJM State & Member Training Dept. PJM 2014 10/24/2013 Objectives Identify the components of Impedance in AC Circuits Calculate the total Impedance in AC Circuits Identify

### Filters and Waveform Shaping

Physics 333 Experiment #3 Fall 211 Filters and Waveform Shaping Purpose The aim of this experiment is to study the frequency filtering properties of passive (R, C, and L) circuits for sine waves, and the

### EE 311: Electrical Engineering Junior Lab Active Filter Design (Sallen-Key Filter)

EE 311: Electrical Engineering Junior Lab Active Filter Design (Sallen-Key Filter) Objective The purpose of this experiment is to design a set of second-order Sallen-Key active filters and to investigate

### Eðlisfræði 2, vor 2007

[ Assignment View ] [ Pri Eðlisfræði 2, vor 2007 31. Alternating Current Circuits Assignment is due at 2:00am on Wednesday, March 21, 2007 Credit for problems submitted late will decrease to 0% after the

### S-Parameter Matrices

The objective of microwave circuit analysis is to move from the requirement to solve for all the fields and waves of a structure to an equivalent circuit that is amenable to all the tools of the circuit

### Jeff Thomas Tom Holmes Terri Hightower. Learn RF Spectrum Analysis Basics

Jeff Thomas Tom Holmes Terri Hightower Learn RF Spectrum Analysis Basics Learning Objectives Name the major measurement strengths of a swept-tuned spectrum analyzer Explain the importance of frequency

### R f. V i. ET 438a Automatic Control Systems Technology Laboratory 4 Practical Differentiator Response

ET 438a Automatic Control Systems Technology Laboratory 4 Practical Differentiator Response Objective: Design a practical differentiator circuit using common OP AMP circuits. Test the frequency response

### RLC Resonant Circuits

C esonant Circuits Andrew McHutchon April 20, 203 Capacitors and Inductors There is a lot of inconsistency when it comes to dealing with reactances of complex components. The format followed in this document

### SIGNAL GENERATORS and OSCILLOSCOPE CALIBRATION

1 SIGNAL GENERATORS and OSCILLOSCOPE CALIBRATION By Lannes S. Purnell FLUKE CORPORATION 2 This paper shows how standard signal generators can be used as leveled sine wave sources for calibrating oscilloscopes.

### Filters & Wave Shaping

Module 8 AC Theory Filters & Wave Shaping Passive Filters & Wave Shaping What you'll learn in Module 8. Module 8 Introduction Recognise passive filters with reference to their response curves. High pass,

### Understanding Mixers Terms Defined, and Measuring Performance

Understanding Mixers Terms Defined, and Measuring Performance Mixer Terms Defined Statistical Processing Applied to Mixers Today's stringent demands for precise electronic systems place a heavy burden

### MANUAL RETURN LOSS MEASUREMENTS Sam Wetterlin 10/5/08

MANUAL RETURN LOSS MEASUREMENTS Sam Wetterlin swetterlin@comcast.net 10/5/08 This paper describes the procedure for using a reflection bridge (a/k/a return loss bridge ) to manually measure return loss

### Understanding SWR by Example

Understanding SWR by Example Take the mystery and mystique out of standing wave ratio. Darrin Walraven, K5DVW It sometimes seems that one of the most mysterious creatures in the world of Amateur Radio

### Understanding Power Splitters

Understanding Power Splitters How they work, what parameters are critical, and how to select the best value for your application. Basically, a 0 splitter is a passive device which accepts an input signal

### Applying Faraday s law to both the primary and the secondary (noting the possibility of sources applied to either winding), yields

TRANSFORMERS Transformers are key elements in power systems. In order to effectively transmit power over long distances without prohibitive line losses, the voltage from the generator (a maximum of output

### S-Band Low Noise Amplifier Using the ATF-10136. Application Note G004

S-Band Low Noise Amplifier Using the ATF-10136 Application Note G004 Introduction This application note documents the results of using the ATF-10136 in low noise amplifier applications at S band. The ATF-10136

### Agilent Time Domain Analysis Using a Network Analyzer

Agilent Time Domain Analysis Using a Network Analyzer Application Note 1287-12 0.0 0.045 0.6 0.035 Cable S(1,1) 0.4 0.2 Cable S(1,1) 0.025 0.015 0.005 0.0 1.0 1.5 2.0 2.5 3.0 3.5 4.0 Frequency (GHz) 0.005

### CRP718 RF and Microwave Measurements Laboratory

CRP718 RF and Microwave Measurements Laboratory Experiment- MW2 Handout. Updated Jan 13, 2011 SPECTRUM ANALYZER BASED MEASUREMENTS (groups of 2 may omit part 2.1, but are advised to study the procedures

### RF Generator. figure 2

Why is impedance matching important: In general sputtering or etching applications, power generators are used to transfer either DC, Medium Frequency (typically in the several khz range) or RF (usually

### 2. The Vector Network Analyzer

ECE 584 Laboratory Experiments 2. The Vector Network Analyzer Introduction: In this experiment we will learn to use a Vector Network Analyzer to measure the magnitude and phase of reflection and transmission

### ECE 435 INTRODUCTION TO THE MICROWAVE NETWORK ANALYZER

ECE 435 INTRODUCTION TO THE MICROWAVE NETWORK ANALYZER Latest revision: October 1999 Introduction A vector network analyzer (VNA) is a device capable of measuring both the magnitude and phase of a sinusoidal

### Chapt ha e pt r e r 12 RL Circuits

Chapter 12 RL Circuits Sinusoidal Response of RL Circuits The inductor voltage leads the source voltage Inductance causes a phase shift between voltage and current that depends on the relative values of

### Network Analyzer Operation

Network Analyzer Operation 2004 ITTC Summer Lecture Series John Paden Purposes of a Network Analyzer Network analyzers are not about computer networks! Purposes of a Network Analyzer Measures S-parameters

### Understanding Power Splitters

Understanding Power Splitters how they work, what parameters are critical, and how to select the best value for your application. Basically, a 0 splitter is a passive device which accepts an input signal

### Measuring RF Parameters of Networks Bill Leonard NØCU

Measuring RF Parameters of Networks Bill Leonard NØCU NAØTC - 285 TechConnect Radio Club http://www.naøtc.org/ What is a Network? A Network is a group of electrical components connected is a specific way

### FREQUENCY RESPONSE AND PASSIVE FILTERS LABORATORY. We start with examples of a few filter circuits to illustrate the concept.

FREQUENCY RESPONSE AND PASSIVE FILTERS LABORATORY In this experiment we will analytically determine and measure the frequency response of networks containing resistors, AC source/sources, and energy storage

### Differences in Application Between Power Dividers and Power Splitters. Application Note

Differences in Application Between Dividers and Splitters Application Note Introduction dividers are an RF microwave accessory constructed with equivalent 50Ω resistance at each port. These accessories

### S-parameter Simulation and Optimization

S-parameter Simulation and Optimization Slide 5-1 S-parameters are Ratios Usually given in db as 20 log of the voltage ratios of the waves at the ports: incident, reflected, or transmitted. S-parameter

### EMC STANDARDS STANDARDS AND STANDARD MAKING BODIES. International. International Electrotechnical Commission (IEC) http://www.iec.

EMC STANDARDS The EMC standards that a particular electronic product must meet depend on the product application (commercial or military) and the country in which the product is to be used. These EMC regulatory

### Chapter 12. RL Circuits. Objectives

Chapter 12 RL Circuits Objectives Describe the relationship between current and voltage in an RL circuit Determine impedance and phase angle in a series RL circuit Analyze a series RL circuit Determine

### Circuits with inductors and alternating currents. Chapter 20 #45, 46, 47, 49

Circuits with inductors and alternating currents Chapter 20 #45, 46, 47, 49 RL circuits Ch. 20 (last section) Symbol for inductor looks like a spring. An inductor is a circuit element that has a large

### First and Second Order Filters

First and Second Order Filters These functions are useful for the design of simple filters or they can be cascaded to form high-order filter functions First Order Filters General first order bilinear transfer

### Agilent De-embedding and Embedding S-Parameter Networks Using a Vector Network Analyzer. Application Note 1364-1

Agilent De-embedding and Embedding S-Parameter Networks Using a Vector Network Analyzer Application Note 1364-1 Introduction Traditionally RF and microwave components have been designed in packages with

### The W5JCK Guide to the Mathematic Equations Required for the Amateur Extra Class Exam

The W5JCK Guide to the Mathematic Equations Required for the Amateur Extra Class Exam This document contains every question from the Extra Class (Element 4) Question Pool* that requires one or more mathematical

### Loop Bandwidth and Clock Data Recovery (CDR) in Oscilloscope Measurements. Application Note 1304-6

Loop Bandwidth and Clock Data Recovery (CDR) in Oscilloscope Measurements Application Note 1304-6 Abstract Time domain measurements are only as accurate as the trigger signal used to acquire them. Often

### DRAFT. University of Pennsylvania Moore School of Electrical Engineering ESE319 Electronic Circuits - Modeling and Measurement Techniques

University of Pennsylvania Moore School of Electrical Engineering ESE319 Electronic Circuits - Modeling and Measurement Techniques 1. Introduction. Students are often frustrated in their attempts to execute

### TCOM 370 NOTES 99-4 BANDWIDTH, FREQUENCY RESPONSE, AND CAPACITY OF COMMUNICATION LINKS

TCOM 370 NOTES 99-4 BANDWIDTH, FREQUENCY RESPONSE, AND CAPACITY OF COMMUNICATION LINKS 1. Bandwidth: The bandwidth of a communication link, or in general any system, was loosely defined as the width of

### S5103 Handheld Microwave Tester

Typical Applications Filter Performance Test Time Domain Measurement Radar Communication Systematic Performance Test RFID Domain Test Bluetooth Domain Test Components Test and whole set equipment Test

### Vector Network Analyzer Techniques to Measure WR340 Waveguide Windows

LS-296 Vector Network Analyzer Techniques to Measure WR340 Waveguide Windows T. L. Smith ASD / RF Group Advanced Photon Source Argonne National Laboratory June 26, 2002 Table of Contents 1) Introduction

### TESTS OF 1 MHZ SIGNAL SOURCE FOR SPECTRUM ANALYZER CALIBRATION 7/8/08 Sam Wetterlin

TESTS OF 1 MHZ SIGNAL SOURCE FOR SPECTRUM ANALYZER CALIBRATION 7/8/08 Sam Wetterlin (Updated 7/19/08 to delete sine wave output) I constructed the 1 MHz square wave generator shown in the Appendix. This

### Filters for Data Transmission

Application Report SLOA063 July 2001 Filters for Data Transmission Bruce Carter ABSTRACT The process of filtering an analog signal removes all frequencies above or below a corner frequency, passing only

### Capacitance and Inductance Measurements Using an Oscilloscope and a Function Generator

Capacitance and Inductance Measurements Using an Oscilloscope and a Function Generator Application Note Most labs have an ample supply of DMM's for measuring DC resistance, but when it comes to measuring

### Introduction to the Smith Chart for the MSA Sam Wetterlin 10/12/09 Z +

Introduction to the Smith Chart for the MSA Sam Wetterlin 10/12/09 Quick Review of Reflection Coefficient The Smith chart is a method of graphing reflection coefficients and impedance, and is often useful

### AN-837 APPLICATION NOTE

APPLICATION NOTE One Technology Way P.O. Box 916 Norwood, MA 262-916, U.S.A. Tel: 781.329.47 Fax: 781.461.3113 www.analog.com DDS-Based Clock Jitter Performance vs. DAC Reconstruction Filter Performance