# Experiment 7: Familiarization with the Network Analyzer

 To view this video please enable JavaScript, and consider upgrading to a web browser that supports HTML5 video
Save this PDF as:

Size: px
Start display at page:

Download "Experiment 7: Familiarization with the Network Analyzer"

## Transcription

1 Experiment 7: Familiarization with the Network Analyzer Measurements to characterize networks at high frequencies (RF and microwave frequencies) are usually done in terms of scattering parameters (S parameters). This is because practical network characterization can no longer be done in terms of simply open- or short-circuit measurements that lead to impedance and admittance parameters. Practical measurement of S parameters is often performed using a network analyzer. Network analyzers typically function in the 9 khz to 110 GHz range. First let us consider how S parameters are defined. S Parameters S parameters are descriptors that permit us to define the input-output relations of a network in terms of incident and reflected power waves. In Figure 1 an is the incident normalized power wave and bn is the reflected normalized power wave, defined as follows where the index n refers to the port number 1 or 2. Here, Vn and In are the peak voltage and current values at the nth port (n=1,2) and impedance Zo is the characteristic impedance of the connecting lines at each port. a1 a2 b1 b2 The physical meaning of (1) and (2) become clear when it is noted that the net power flowing into the nth port is The S parameters for a two-port network are now defined as

2 Equations (4) and (5) can be written in matrix form as The meaning of each S parameter is as follows Impedance, admittance, hybrid, and ABCD parameters can be determined for a network, once S parameters are known. This can be done in the following way. Impedance parameters can be obtained from S parameters as follows. Note: The S parameters used in (11) are typically complex and require both amplitude and phase information versus frequency. The table on the next page can be used to obtain the other traditional network parameters. Network Analyzers Network analyzers measure S parameters for two port networks. The measurement of S parameters requires reflection and transmission evaluations of traveling waves at both ports. There are two types of network analyzers: scalar and vector network analyzers. Scalar network analyzers evaluate only amplitudes of these signals. Vector network analyzers measure both amplitude and phase of these signals. Usually analyzers have an output port that provides the RF signal from either an external or internal signal generator, and three measured channels, which are denoted as R, A, and B (see Figure 2).

3 The RF source is usually swept over a specified frequency range. The measurement channel R is employed for measuring the incident wave. Channel R also serves as a reference port. Channels A and B usually measure the reflected and transmitted waves. As shown in Figure 2, S11 and S21 can be measured. S11 can be obtained by evaluating the ratio A/R and S21 through computing B/R. S12 and S22 can be evaluated for the DUT (device under test) by reversing the DUT ports in this configuration. It is apparent that a practical test arrangement is more complicated when compared to the ideal S parameter measurement concept that employs perfectly matched transmission lines and ideal directional couplers with perfect isolation and impedance match. Network Analyzer Calibration To overcome the effect of these non-ideal elements in the S parameter measurement process, a calibration procedure must be performed. The main goal of the calibration procedure is to characterize the non-ideal elements, provide this

4 information to a computer internal to the network analyzer to evaluate the errorfree S parameters of the actual DUT relative to a desired reference plane for phase measurements. Calibration involves using known standards. Convenient standards are perfect shorts, opens, loads, and a zero length transmission line (connecting the input to the Figure 2 Measurement system for S11 and S21 parameters using a network analyzer output with no DUT). Such standards have simple reflection and transmission coefficients, namely: A short circuit at the reference plane gives = -1 A load at the reference plane gives An open circuit at the reference plane gives Connecting input to the output gives S21 = S12 = 1 Connecting a load at the input and output gives S21 = S12 = 0 After making these measurements, the network analyzer can compute some correction values to produce the expected answer. For answers that are supposed to be zero, the analyzer can subtract the residual. For non-zero values, the analyzer could calculate complex factors that will compensate for both phase and amplitude

5 errors. It is not easy to make ideal standards. Network analyzers will account for imperfections in standards. 1 A calibration using a mechanical calibration kit may take a significant amount of time. Not only must the operator sweep through all the frequencies of interest, but also the operator must disconnect and reconnect the various standards. To avoid that work, network analyzers can employ automated calibration standards. The operator connects one box to the network analyzer. The box contains a set of standards and switches that have already been characterized. The network analyzer can read the characterization and control the configuration using a digital bus such as USB. Agilent calls such boxes used with their network analyzers ECal (electronic calibration) modules. Laboratory Procedure 1. Calibrate the vector network analyzer over the 1 GHz to 6 GHz frequency range. A new calibration procedure should be performed each time the cables and/or type of connectors (e.g. N-type or SMA) feeding the device under test are changed. 2. Over the 1 to 6 GHz range, measure and record all appropriate S parameters for each of the following devices after the calibration procedure is completed: a. An open circuit b. A short circuit c. A 50 ohm termination d. The input connected directly to the output (or connected to each other using a type adapter, if necessary) Make sure that data is taken at enough frequencies to record the major features of the S parameters versus frequency behavior. For example, data taken at intervals of 100 MHz is probably reasonable. 3. Measure and record all appropriate S parameters (both amplitude and phase data at each frequency for all four S parameters) for both the commercially available 10 db directional coupler and the quadrature hybrids over the frequency ranges stipulated on the pertinent specification sheet. Remember, these are inherently 4- port devices. Measure amplitude of the signal reflected from the input port. Also, measure the amplitude and phase of the coupled and the transmitted signal ports. Measure the amplitude of the signal emerging from the isolated port. 4. Over the 1 to 6 GHz frequency range, measure and record all appropriate S parameters for a straight-through microstrip transmission line on a PCB. Make the length of this line as close as practical to the length of the line to be constructed in part 5. Describe and photograph the resulting straight line geometry for use in the lab report. (S parameters here will be compared to those obtained in the part 5.)

6 5. Over the 1 to 6 GHz frequency range, measure and record all appropriate S parameters for the right-angle microstrip bend on a PCB. You will create a right angle microstrip bend by attaching and trimming two pieces of conducting tape of the appropriate width. Experiment with modifications to the sharp corner in this bend to obtain the optimum performance in terms of impedance match and insertion loss through the bend. Do your best to sketch and photograph the resulting bend geometry for use in the lab report. Compare the performance of this line to that of part 4.

7 6. Lab Report a. For the 1 to 6 GHz frequency range show plots of the appropriate S parameters from experimental parts 2a through 2d. For part 2d, convert the S21 parameter values at each frequency to Z parameters using equation (11) and then to ABCD parameters. The ratio of V2/V1 = 1/A. For part 2d, plot the amplitude and phase of V2/V1 and reflection coefficient versus frequency for the 1 to 6 GHz frequency range. Discuss how these values correlate with what is expected based upon the calibration of the equipment. b. For the appropriate frequency bands show plots for the directional coupler and quadrature hybrids of the coupled signal, transmitted signal, isolated signal, and VSWR versus frequency. Note that Be aware that the network analyzer in the laboratory can read VSWR directly. On one graph for each of these devices, plot the phase of the coupled and transmitted signals. Compare and discuss how well these characteristics agree with the manufacturer provided specifications and with what would be expected for an ideal coupler. Also, calculate the amount of (power in db) dissipated in each type of coupler at its midband frequency. This can be determined by adding up the power reflected from the input port, the power emerging from the transmitted, coupled, and isolated ports and subtracting this sum from the total input power. c. For the frequency range from 1 to 6 GHz show plots of transmission loss and reflection coefficient for the straight-through transmission line. d. For the frequency range from 1 to 6 GHz show plots of transmission loss and reflection coefficient for the right-angle microstrip bend. Look in the literature for technical papers on how to mitre (shape) the corner for a 90 o microstrip bend. Show the optimum shape you found in the lab and compare it to what the paper describes for such a bend. Compare performance with that obtained in part c. References 1. R. Ludwig and P. Bretchko, RF Circuit Evaluation, Theory and Application, Prentice-Hall, Upper Saddle River, NY, 2000,

### Impedance 50 (75 connectors via adapters)

VECTOR NETWORK ANALYZER PLANAR TR1300/1 DATA SHEET Frequency range: 300 khz to 1.3 GHz Measured parameters: S11, S21 Dynamic range of transmission measurement magnitude: 130 db Measurement time per point:

### Agilent De-embedding and Embedding S-Parameter Networks Using a Vector Network Analyzer. Application Note 1364-1

Agilent De-embedding and Embedding S-Parameter Networks Using a Vector Network Analyzer Application Note 1364-1 Introduction Traditionally RF and microwave components have been designed in packages with

### 2. The Vector Network Analyzer

ECE 584 Laboratory Experiments 2. The Vector Network Analyzer Introduction: In this experiment we will learn to use a Vector Network Analyzer to measure the magnitude and phase of reflection and transmission

### ECE 435 INTRODUCTION TO THE MICROWAVE NETWORK ANALYZER

ECE 435 INTRODUCTION TO THE MICROWAVE NETWORK ANALYZER Latest revision: October 1999 Introduction A vector network analyzer (VNA) is a device capable of measuring both the magnitude and phase of a sinusoidal

### RF Network Analyzer Basics

RF Network Analyzer Basics A tutorial, information and overview about the basics of the RF Network Analyzer. What is a Network Analyzer and how to use them, to include the Scalar Network Analyzer (SNA),

### Vector Network Analyzer Techniques to Measure WR340 Waveguide Windows

LS-296 Vector Network Analyzer Techniques to Measure WR340 Waveguide Windows T. L. Smith ASD / RF Group Advanced Photon Source Argonne National Laboratory June 26, 2002 Table of Contents 1) Introduction

### S-Parameters and Related Quantities Sam Wetterlin 10/20/09

S-Parameters and Related Quantities Sam Wetterlin 10/20/09 Basic Concept of S-Parameters S-Parameters are a type of network parameter, based on the concept of scattering. The more familiar network parameters

### Keysight Technologies Understanding the Fundamental Principles of Vector Network Analysis. Application Note

Keysight Technologies Understanding the Fundamental Principles of Vector Network Analysis Application Note Introduction Network analysis is the process by which designers and manufacturers measure the

### Engineering Sciences 151. Electromagnetic Communication Laboratory Assignment 3 Fall Term 1998-99

Engineering Sciences 151 Electromagnetic Communication Laboratory Assignment 3 Fall Term 1998-99 WAVE PROPAGATION II: HIGH FREQUENCY SLOTTED LINE AND REFLECTOMETER MEASUREMENTS OBJECTIVES: To build greater

### Performing Amplifier Measurements with the Vector Network Analyzer ZVB

Product: Vector Network Analyzer R&S ZVB Performing Amplifier Measurements with the Vector Network Analyzer ZVB Application Note This document describes typical measurements that are required to be made

### MEASUREMENT UNCERTAINTY IN VECTOR NETWORK ANALYZER

MEASUREMENT UNCERTAINTY IN VECTOR NETWORK ANALYZER W. Li, J. Vandewege Department of Information Technology (INTEC) University of Gent, St.Pietersnieuwstaat 41, B-9000, Gent, Belgium Abstract: Precision

### Agilent 8510-13 Measuring Noninsertable Devices

Agilent 8510-13 Measuring Noninsertable Devices Product Note A new technique for measuring components using the 8510C Network Analyzer Introduction The majority of devices used in real-world microwave

### RF-Microwaves formulas - 1-port systems

RF-Microwaves formulas - -port systems s-parameters: Considering a voltage source feeding into the DUT with a source impedance of. E i E r DUT The voltage into the DUT is composed of 2 parts, an incident

### APPLICATION NOTES POWER DIVIDERS. Things to consider

Internet Copy Rev A Overview Various RF applications require power to be distributed among various paths. The simplest way this can be done is by using a power splitter/divider. Power dividers are reciprocal

### Technical Datasheet Scalar Network Analyzer Model 8003-10 MHz to 40 GHz

Technical Datasheet Scalar Network Analyzer Model 8003-10 MHz to 40 GHz The Giga-tronics Model 8003 Precision Scalar Network Analyzer combines a 90 db wide dynamic range with the accuracy and linearity

### Application Note Noise Frequently Asked Questions

: What is? is a random signal inherent in all physical components. It directly limits the detection and processing of all information. The common form of noise is white Gaussian due to the many random

### 1. The Slotted Line. ECE 584 Microwave Engineering Laboratory Experiments. Introduction:

ECE 584 Microwave Engineering Laboratory Experiments 1. The Slotted Line Introduction: In this experiment we will use a waveguide slotted line to study the basic behavior of standing waves and to measure

### Designing the NEWCARD Connector Interface to Extend PCI Express Serial Architecture to the PC Card Modular Form Factor

Designing the NEWCARD Connector Interface to Extend PCI Express Serial Architecture to the PC Card Modular Form Factor Abstract This paper provides information about the NEWCARD connector and board design

### Understanding the Fundamental Principles of Vector Network Analysis. Application Note 1287-1. Table of Contents. Page

Understanding the Fundamental Principles of Vector Network Analysis Application Note 1287-1 Table of Contents Page Introduction 2 Measurements in Communications Systems 2 Importance of Vector Measurements

### Balun Parameter Definitions & Measurement May 2004

Balun Parameter Definitions & Measurement May 2004 Differential circuits are becoming more widely used in RF circuits for the same reason that they have been used for years in lower frequency circuits.

### R3765/67 CG Network Analyzer

R3765/67 CG Network Analyzer RSE 05.03.02 1 R376XG Series Overview R3765 300kHz ~ 3.8 GHz R3767 300kHz ~ 8 GHz AG BG Basic model Built-in Bridge A/R & B/R Transmission Reflection CG Built-in S-parameter

### Power Amplifier Gain Compression Measurements

Technical Brief Power Amplifier Gain Compression Measurements GPIB Private Bus Sweep Out Sweep In Pulse In AC Mod Out Blank/Marker Out Blanking In Overview The 1 db gain compression of an amplifier describes

### Lecture 22: Measurement Errors. TRL Calibration of a VNA.

Whites, EE 48/58 Lecture Page of Lecture : Measurement Errors. TRL Calibration of a VNA. A VNA measures both the magnitude and phase of S parameters. However, there will invariably be significant errors

### Equation 2 details the relationship between the S-parameters and impedance: Stephane Wloczysiak ED Online ID #22407 February 2010

Match The Ports Of Differential Devices This straightforward approach shows how to match the impedances of high-frequency, differential devices both with discrete components and microstrip circuit elements.

### University of Alberta Department of Electrical and Computer Engineering. EE 250 Laboratory Experiment #5 Diodes

University of Alberta Department of Electrical and Computer Engineering EE 250 Laboratory Experiment #5 Diodes Objective: To introduce basic diode concepts. Introduction: The diode is the most fundamental

### Agilent PN 8753-1 RF Component Measurements: Amplifier Measurements Using the Agilent 8753 Network Analyzer. Product Note

Agilent PN 8753-1 RF Component Measurements: Amplifier Measurements Using the Agilent 8753 Network Analyzer Product Note 2 3 4 4 4 4 6 7 8 8 10 10 11 12 12 12 13 15 15 Introduction Table of contents Introduction

### 0HDVXULQJWKHHOHFWULFDOSHUIRUPDQFH FKDUDFWHULVWLFVRI5),)DQGPLFURZDYHVLJQDO SURFHVVLQJFRPSRQHQWV

0HDVXULQJWKHHOHFWULFDOSHUIRUPDQFH FKDUDFWHULVWLFVRI5),)DQGPLFURZDYHVLJQDO SURFHVVLQJFRPSRQHQWV The treatment given here is introductory, and will assist the reader who wishes to consult the standard texts

### ENGR 210 Lab 11 Frequency Response of Passive RC Filters

ENGR 210 Lab 11 Response of Passive RC Filters The objective of this lab is to introduce you to the frequency-dependent nature of the impedance of a capacitor and the impact of that frequency dependence

### Shielding Effectiveness Test Method. Harbour s LL, SB, and SS Coaxial Cables. Designs for Improved Shielding Effectiveness

Shielding Effectiveness Test Method Harbour s LL, SB, and SS Coaxial Cables Designs for Improved Shielding Effectiveness Harbour Industries 4744 Shelburne Road Shelburne Vermont 05482 USA 802-985-3311

### S-Parameter Matrices

The objective of microwave circuit analysis is to move from the requirement to solve for all the fields and waves of a structure to an equivalent circuit that is amenable to all the tools of the circuit

### spinner Measurement & Calibration equipment for network analyzers

spinner Measurement & Calibration equipment for network analyzers Edition B 2011 High Frequency Performance Worldwide www.spinner-group.com SPINNER test & calibration equipment for best measurement results

### Probes and Setup for Measuring Power-Plane Impedances with Vector Network Analyzer

Probes and Setup for Measuring Power-Plane Impedances with Vector Network Analyzer Plane impedance measurement with VNA 1 Outline Introduction, Y, and S parameters Self and transfer impedances VNA One-port

### Comparison of Vector Network Analyzer and TDA Systems IConnect Generated S-Parameters

Comparison of Vector Network Analyzer and TDA Systems IConnect Generated S-Parameters Purpose: This technical note presents single-ended insertion loss ( SE IL) and return loss ( SE RL) data generated

### Category 8 Cable Transmission Measurements Comparative Study between 4-port single wire measurements and 2-port balun measurements

Category 8 Cable Transmission Measurements Comparative Study between 4-port single wire measurements and 2-port balun measurements Stefan Estevanovich Rafael Herrera, Nadim Kafati Hitachi Cable USA NDC

### THREE-BEAD-BALUN REFLECTION BRIDGE 250 KHz to 1.5 GHz Sam Wetterlin 9/22/08

THREE-BEAD-BALUN REFLECTION BRIDGE 250 KHz to 1.5 GHz Sam Wetterlin swetterlin@comcast.net 9/22/08 This paper presents a reflection bridge useful from 250 KHz to the neighborhood of 1.5 GHz. This bridge

### Millimeter Wave Components, 18-40 GHz

Millimeter Wave Components, 18-40 GHz D7050 F9014 F9021 SPST & SP2T F90 SERIES SWITCHES General Microwave millimeter wave switches are available in SPST and SP2T models in a variety of topologies and configurations,

### Optimizing IP3 and ACPR Measurements

Optimizing IP3 and ACPR Measurements Table of Contents 1. Overview... 2 2. Theory of Intermodulation Distortion... 2 3. Optimizing IP3 Measurements... 4 4. Theory of Adjacent Channel Power Ratio... 9 5.

### Agilent Test Solutions for Multiport and Balanced Devices

Agilent Test Solutions for Multiport and Balanced Devices Duplexer test solutions 8753ES option H39/006 During design and final alignment of duplexers, it is often necessary to see both the transmit-antenna

### Paricon Technologies

Paricon Technologies Flex connector Measurement Results prepared by Gert Hohenwarter 1/24/07 1 Table of Contents TABLE OF CONTENTS... 2 OBJECTIVE... 3 METHODOLOGY... 3 Test procedures... 4 Setup... 4 MEASUREMENTS...

### Agilent 10 Hints for Making Better Network Analyzer Measurements. Application Note 1291-1B

Agilent 10 Hints for Making Better Network Analyzer Measurements Application Note 1291-1B Contents HINT 1. Measuring high-power amplifiers HINT 2. Compensating for time delay in cable HINT 3. Improving

### Applying Error Correction to Network Analyzer Measurements. Application Note 1287-3. Table of Contents. Page

Applying Error Correction to Network Analyzer Measurements Application Note 287-3 Table of Contents Page Introduction 2 Sources of Errors and Types of Errors 3 Types of Error Correction 4 One-Port 4 The

### Using Simple Calibration Load Models to Improve Accuracy of Vector Network Analyzer Measurements

Using Simple Calibration Load Models to Improve Accuracy of Vector Network Analyzer Measurements Nick M. Ridler 1 and Nils Nazoa 2 1 National Physical Laboratory, UK (www.npl.co.uk) 2 LA Techniques Ltd,

### Microwave Circuit Design and Measurements Lab. INTRODUCTION TO MICROWAVE MEASUREMENTS: DETECTION OF RF POWER AND STANDING WAVES Lab #2

EE 458/558 Microwave Circuit Design and Measurements Lab INTRODUCTION TO MICROWAVE MEASUREMENTS: DETECTION OF RF POWER AND STANDING WAVES Lab #2 The purpose of this lab is to gain a basic understanding

### 2 Port Parameters I 1. ECE145A/218A Notes Set #4 1. Two-ways of describing device: A. Equivalent - Circuit-Model

ECE45A/8A Notes et #4 Port Parameters Two-ways of describing device: A. Equivalent - Circuit-Model Physically based Includes bias dependence Includes frequency dependence Includes size dependence - scalability

### The N2PK Vector Network Analyzer (N2PK VNA) Original by Paul Kiciak, N2PK, 2007 * (*Pages edited or added by VE7WRS, 2009)

The N2PK Vector Network Analyzer (N2PK VNA) Original by Paul Kiciak, N2PK, 2007 * (*Pages edited or added by VE7WRS, 2009) What Does Vector Mean?! Vector: magnitude & direction (angle) 01/22/07 N2PK *

### TC-3000C Bluetooth Tester

TC-3000C Bluetooth Tester Product Overview TC-3000C Bluetooth Tester Data Sheet TC-3000C Bluetooth Tester is able to analyze the data of each packet that is transmitted to the upper application protocol

### a 1 a 2 2 Port b 2 b 1 Multi-Port Handset Switch S-Parameters Application Note AN20 Seven-Port S-Parameter Definition Introduction Summary:

AN2 Multi-Port Handset Switch S-Parameters Introduction High-power UltraCMOS switches are the nextgeneration solution for wireless handset power amplifiers and antenna switch modules. Most multi-throw

### One Port Network Analyzer

99 Washington Street Melrose, MA 02176 Phone 781-665-1400 Toll Free 1-800-517-8431 Visit us at www.testequipmentdepot.com One Port Network Analyzer 5.4GHz Impendance : 50Ω(75Ωconnectors via adapters) Test

### Measuring RF Parameters of Networks Bill Leonard NØCU

Measuring RF Parameters of Networks Bill Leonard NØCU NAØTC - 285 TechConnect Radio Club http://www.naøtc.org/ What is a Network? A Network is a group of electrical components connected is a specific way

### Agilent Electronic Calibration (ECal) Modules for Vector Network Analyzers

Agilent Electronic Calibration (ECal) Modules for Vector Network Analyzers N4690 Series, 2-port Microwave ECal 85090 Series, 2-port RF ECal N4430 Series, 4-port ECal Technical Overview Control ECal directly

### INTEGRATED ASSEMBLIES TELEDYNE MICROWAVE SOLUTIONS

INTEGRATED ASSEMBLIES TELEDYNE MICROWAVE SOLUTIONS INTEGRATED ASSEMBLIES FROM TELEDYNE MICROWAVE SOLUTIONS Teledyne Microwave Solutions (TMS) offers full first-level integration capabilities, providing

### EE 186 LAB 2 FALL 2004. Network Analyzer Fundamentals and Two Tone Linearity

Network Analyzer Fundamentals and Two Tone Linearity Name: Name: Name: Objective: To become familiar with the basic operation of a network analyzer To use the network analyzer to characterize the in-band

### Users Guide -TEZ SNA Software Scalar Network Analyzer

TeleApp #007 Users Guide -TEZ SNA Software Scalar Network Analyzer Synthesized Signal Generators TEG250-15 TEG1000-10 TEG2700-6 TEG3500-8 TEG4000-1 TEG6200-8 RF Power Sensors TED6000-50 TED8000-40 RF Digital

### APPLICATION NOTE - 016

APPLICATION NOTE - 016 Testing RFI Line Filters Frequency Response Analysis Testing RFI line filters Radio frequency interference (RFI) is unwanted electromagnetic noise within a radio communications frequency

### ELECTRICAL CHARACTERISATION OF SEMI-RIGID COAXIAL CABLES WITH SMA AND K CONNECTORS. Carmen Diez Rafael García Juan Daniel Gallego.

ELECTRICAL CHARACTERISATION OF SEMI-RIGID COAXIAL CABLES WITH SMA AND K CONNECTORS March 2005 TECHNICAL REPORT C.A.Y. 2005-4 ABSTRACT Semi-rigid coaxial transitions are normally used in closed cycle refrigerators

### RF Measurements Using a Modular Digitizer

RF Measurements Using a Modular Digitizer Modern modular digitizers, like the Spectrum M4i series PCIe digitizers, offer greater bandwidth and higher resolution at any given bandwidth than ever before.

### S-parameter Simulation and Optimization

S-parameter Simulation and Optimization Slide 5-1 S-parameters are Ratios Usually given in db as 20 log of the voltage ratios of the waves at the ports: incident, reflected, or transmitted. S-parameter

### Impedance Matching and Matching Networks. Valentin Todorow, December, 2009

Impedance Matching and Matching Networks Valentin Todorow, December, 2009 RF for Plasma Processing - Definition of RF What is RF? The IEEE Standard Dictionary of Electrical and Electronics Terms defines

### DSA800 Series Spectrum Analyzer

DSA800 Series Spectrum Analyzer Configuration Guide This guide is used to help users to configure DSA800 series spectrum analyzer according to their requirements. You can get an overall understanding of

### Six-Port Reflectometer: an Alternative Network Analyzer for THz Region. Guoguang Wu

Six-Port Reflectometer: an Alternative Network Analyzer for THz Region Guoguang Wu Outline General Background of Network Analyzer Principles of Six-Port Reflectometer WR-15 Six-port Reflectometer Design,

### Antenna Trainer EAN. www.edibon.com. Technical Teaching Equipment INTRODUCTION

Antenna Trainer EAN Technical Teaching Equipment Products Products range Units 3.-Communications INTRODUCTION Antennas are the main element of aerial communications. They are the transition between a transmission

### Tuning a Monopole Antenna Using a Network Analyzer

11/21/11 Tuning a Monopole Antenna Using a Network Analyzer Chris Leonard Executive Summary: When designing a monopole antenna it is important to know at which frequency the antenna will be operating at.

### Antenna Properties and their impact on Wireless System Performance. Dr. Steven R. Best. Cushcraft Corporation 48 Perimeter Road Manchester, NH 03013

Antenna Properties and their impact on Wireless System Performance Dr. Steven R. Best Cushcraft Corporation 48 Perimeter Road Manchester, NH 03013 Phone (603) 627-7877 FAX: (603) 627-1764 Email: sbest@cushcraft.com

### Network Analyzer Basics- EE142 Fall 07

- EE142 Fall 07 Lightwave Analogy to RF Energy Incident Transmitted Reflected Lightwave DUT RF Why Do We Need to Test Components? Verify specifications of building blocks for more complex RF systems Create

### Electrical Resonance

Electrical Resonance (R-L-C series circuit) APPARATUS 1. R-L-C Circuit board 2. Signal generator 3. Oscilloscope Tektronix TDS1002 with two sets of leads (see Introduction to the Oscilloscope ) INTRODUCTION

### Antenna Selection for Optimum Wireless LAN Performance

Antenna Selection for Optimum Wireless LAN Performance Dr. Steven R. Best Cushcraft Corporation 48 Perimeter Road Manchester, NH 03108 (603) 627-7877 PURPOSE Overview of antenna properties and performance

### Cable Impedance and Structural Return Loss Measurement Methodologies

Cable Impedance and Structural Return Loss Measurement Methodologies Introduction Joe Rowell Joel Dunsmore and Les Brabetz Hewlett Packard Company Santa Rosa, California Two critical electrical specifications

### Calibration Procedure for Measuring S-Parameters in Balun Applications on 150-Ω High-Speed Cables

NASA/TM 2012-217296 Calibration Procedure for Measuring S-Parameters in Balun Applications on 150-Ω High-Speed Cables Onoufrios Theofylaktos and Joseph D. Warner Glenn Research Center, Cleveland, Ohio

### Agilent 8753ET/8753ES Network Analyzers

Agilent 8753ET/8753ES Network Analyzers 8753ET, 300 khz to 3 or 6 GHz 8753ES, 30 khz to 3 or 6 GHz Configuration Guide System configuration summary The following summary lists the main components required

### Techniques for Precise Cable and Antenna Measurements in the Field

Techniques for Precise Cable and Antenna Measurements in the Field Using FieldFox handheld analyzers Application Note This application note introduces the practical aspects of cable and antenna testing,

### MANUAL RETURN LOSS MEASUREMENTS Sam Wetterlin 10/5/08

MANUAL RETURN LOSS MEASUREMENTS Sam Wetterlin swetterlin@comcast.net 10/5/08 This paper describes the procedure for using a reflection bridge (a/k/a return loss bridge ) to manually measure return loss

### How to make a Quick Turn PCB that modern RF parts will actually fit on!

How to make a Quick Turn PCB that modern RF parts will actually fit on! By: Steve Hageman www.analoghome.com I like to use those low cost, no frills or Bare Bones [1] type of PCB for prototyping as they

### Calculating Antenna System Return Loss As Viewed Through The RF Path

Calculating Antenna System Return Loss As Viewed Through The RF Path Lou Meyer, Director of Technical Marketing December 2009 1. Introduction Return loss (RL), reflection coefficient (Γ) and voltage standing

### Understanding Power Splitters

Understanding Power Splitters How they work, what parameters are critical, and how to select the best value for your application. Basically, a 0 splitter is a passive device which accepts an input signal

### Introduction to Network Analyzer Measurements

Introduction to Network Analyzer Measurements Introduction to Network Analyzer Measurements 1 Table of Contents 1. Introduction to Network Analyzer Measurements... 3 VNA Basics...3 Applications for Network

### ELECTRICAL ENGINEERING DEPARTMENT. California Polytechnic State University SIGNAL TRANSMISSION LABORATORY EE 353

ELECTRICAL ENGINEERING DEPARTMENT California Polytechnic State University SIGNAL TRANSMISSION LABORATORY EE 353 Fall 2003 ELECTRICAL ENGINEERING DEPARTMENT California Polytechnic State University EE 353

### Laboratory #5: RF Filter Design

EEE 194 RF Laboratory Exercise 5 1 Laboratory #5: RF Filter Design I. OBJECTIVES A. Design a third order low-pass Chebyshev filter with a cutoff frequency of 330 MHz and 3 db ripple with equal terminations

### Equipment: Power Supply, DAI, Variable resistance (8311), Variable inductance (8321)

Lab 4: 3-phase circuits. Objective: to study voltage-current relationships in 3-phase circuits; to learn to make delta and Y connections; to calculate and measure real, apparent, and reactive powers. Equipment:

### Agilent AN 1287-9 In-Fixture Measurements Using Vector Network Analyzers

Agilent AN 1287-9 In-Fixture Measurements Using Vector Network Analyzers Application Note Agilent Network Analysis Solutions Table of Contents 3 3 4 4 5 5 6 7 8 12 13 13 13 15 16 17 17 17 17 18 18 19 19

### R&S ZVA Vector Network Analyzer Specifications

www.atecorp.com 8-44-ATEC (2832) R&S ZVA Vector Network Analyzer Specifications ZVA_dat-sw_en_523-568-22_v_cover.indd Data Sheet. E stablished 98 Advanced Test Equipment Rentals Test & Measurement 4.5.22

### Making Accurate Voltage Noise and Current Noise Measurements on Operational Amplifiers Down to 0.1Hz

Author: Don LaFontaine Making Accurate Voltage Noise and Current Noise Measurements on Operational Amplifiers Down to 0.1Hz Abstract Making accurate voltage and current noise measurements on op amps in

### Field Calibration Software

SIGNAL HOUND Field Calibration Software User s Manual Version 1.1.0 7/8/2016 This information is being released into the public domain in accordance with the Export Administration Regulations 15 CFR 734

### Time and Frequency Domain Analysis for Right Angle Corners on Printed Circuit Board Traces

Time and Frequency Domain Analysis for Right Angle Corners on Printed Circuit Board Traces Mark I. Montrose Montrose Compliance Services 2353 Mission Glen Dr. Santa Clara, CA 95051-1214 Abstract: For years,

### National Radio Astronomy Observatory ELECTRONICS DIVISION TECHNICAL NOTE 210

National Radio Astronomy Observatory ELECTRONICS DIVISION TECHNICAL NOTE 210 Measurements of 2320-2345 MHz Bandstop Filters Fabricated using a High-Temperature Superconductor Roger Norrod Ken Ward Michael

### RF and Microwave Accessories. CD-ROM Catalog. Find the right component for your Rohde & Schwarz test & measurement equipment

RF and Microwave Accessories CD-ROM Catalog Find the right component for your Rohde & Schwarz test & measurement equipment Product group Typical applications Adapters Interchanging of various connector

### A Small, Simple, USB-Powered Vector Network Analyzer Covering 1 khz to 1.3 GHz

Prof. Dr. Thomas C. Baier, DG8SAQ University of Applied Sciences, Prittwitzstrasse 10, 89075 Ulm, Germany; baier@hs-ulm.de A Small, Simple, USB-Powered Vector Network Analyzer Covering 1 khz to 1.3 GHz

### Mixed-mode S-parameters and Conversion Techniques

Mixed-mode S-parameters and Conversion Techniques 1 x 1 Mixed-mode S-parameters and Conversion Techniques Allan Huynh, Magnus Karlsson and Shaofang Gong Linköping University Sweden 1. Introduction Differential

### Applications in EMC testing. Outline. Antennas for EMC Testing. Terminology

Antennas for EMC Testing Zhong Chen ETS-Lindgren 1301 Arrow Point Drive Cedar Park, TX 78613 Zhong.Chen@ets-lindgren.com Outline EMC Terms and Definitions Typical EMC Antennas Calibration of EMC Antennas

### Agilent Advanced impedance measurement capability of the RF I-V method compared to the network analysis method. Application Note 1369-2

Agilent Advanced impedance measurement capability of the RF I-V method compared to the network analysis method Application Note 1369-2 1. Introduction Explosive demand for the electronic devices used in

### Network Analyzer Operation

Network Analyzer Operation 2004 ITTC Summer Lecture Series John Paden Purposes of a Network Analyzer Network analyzers are not about computer networks! Purposes of a Network Analyzer Measures S-parameters

### Antenna Deployment Technical Brief

ProCurve Networking Antenna Deployment Technical Brief Introduction... 2 Antenna types... 2 Omni directional antennas... 2 Directional antennas... 2 Diversity antennas... 3 High gain directional antennas...

### R&S FSW signal and spectrum analyzer: best in class now up to 50 GHz

R&S FSW signal and spectrum analyzer: best in class now up to 50 GHz The new R&S FSW 43 and R&S FSW 50 signal and spectrum analyzers make the outstanding features of the R&S FSW family available now also

### Agilent AN 1316 Optimizing Spectrum Analyzer Amplitude Accuracy

Agilent AN 1316 Optimizing Spectrum Analyzer Amplitude Accuracy Application Note RF & Microwave Spectrum Analyzers Table of Contents 3 3 4 4 5 7 8 8 13 13 14 16 16 Introduction Absolute versus relative

### Experiments Using the HP8714 RF Network Analyzer

Experiments Using the HP8714 RF Network Analyzer Purpose: The purpose of this set of experiments is two folded: to get familiar with the basic operation of a RF network analyzer, and to gain a physical

### Agilent AN 154 S-Parameter Design Application Note

Agilent AN 154 S-Parameter Design Application Note Introduction The need for new high-frequency, solid-state circuit design techniques has been recognized both by microwave engineers and circuit designers.

### You will need the following pieces of equipment to complete this experiment:

UNIVERSITY OF TORONTO FACULTY OF APPLIED SCIENCE AND ENGINEERING The Edward S. Rogers Sr. Department of Electrical and Computer Engineering ECE422H1S: RADIO AND MICROWAVE WIRELESS SYSTEMS EXPERIMENT 3:

### The Effect of Network Cabling on Bit Error Rate Performance. By Paul Kish NORDX/CDT

The Effect of Network Cabling on Bit Error Rate Performance By Paul Kish NORDX/CDT Table of Contents Introduction... 2 Probability of Causing Errors... 3 Noise Sources Contributing to Errors... 4 Bit Error

### Standex-Meder Electronics. Custom Engineered Solutions for Tomorrow

Standex-Meder Electronics Custom Engineered Solutions for Tomorrow RF Reed Relays Part II Product Training Copyright 2013 Standex-Meder Electronics. All rights reserved. Introduction Purpose Designing