1 AC : MEASUREMENT OF OP-AMP PARAMETERS USING VEC- TOR SIGNAL ANALYZERS IN UNDERGRADUATE LINEAR CIRCUITS LABORATORY Dr. Tooran Emami, U.S. Coast Guard Academy Tooran Emami received her M.S. and Ph.D. degrees in electrical engineering from Wichita State University (WSU) in 26 and 29, respectively. She is an Assistant Professor in Department of Engineering at the U.S. Coast Guard Academy. Her research interests are in Proportional Integral Derivative (PID) controllers, robust control, time delay, and compensator design for continuous-time, and discrete-time systems. Prof. Richard J. Hartnett, U.S. Coast Guard Academy Richard J. Hartnett is a professor of electrical engineering at the U.S. Coast Guard Academy in New London, Conn. He received his B.S.E.E. degree from the U.S. Coast Guard Academy, the M.S.E.E. degree from Purdue University, and his Ph.D. in E.E. from the University of Rhode Island. His research interests include efficient digital filtering methods, improved receiver signal processing techniques for electronic navigation systems, and autonomous vehicle design. c American Society for Engineering Education, 212
2 Measurement of Op-Amp Parameters Using Vector Signal Analyzers in an Undergraduate Linear Circuits Laboratory Abstract Measurements of op-amp circuit parameters such as open or closed-loop frequency response, or output impedance as a function of frequency, over a reasonably broad range of frequencies, can be tedious and very challenging for undergraduate students to accomplish using signal generators and oscilloscopes. In fact, measurements of total harmonic distortion (measure of op-amp linearity), with and without resistive load, are almost impossible to make in the time domain with conventional oscilloscopes. Of greater concern is that the time-domain methods for measuring frequency domain characteristics of an op-amp do not present a real-time visualization for students. This problem can even mask important operational limitations such as op-amp slewrate nonlinearities for higher frequency sinusoidal inputs. This paper presents some of the successful measurement methodologies that our students use at the U.S. Coast Guard Academy in a junior-level Linear Circuits laboratory. As part of this lab, students use the Agilent 3567 Dynamic Signal Analyzer (DSA) to measure some of the specifications of an inverting amplifier op-amp (µa741) circuit. They use averaging on the measurement data to minimize the impact of noise in the measurements of the µa741circuit. Here we present typical measurement results, along with informal student feedback that suggests to us that the real-time nature of a DSA frequency domain presentation (that looks almost exactly like Matlab TM and MultiSim TM predictions) really does reinforce student learning. The lab procedure consists of four steps: (1) Students set the DSA to measure the open-loop frequency response of the op-amp over two different frequency ranges. (2) Students then measure the closed-loop frequency response of an inverting amplifier for two different gain settings. (3) Students measure the output impedance of an inverting amplifier circuit as a function of frequency. (4) Finally, they compare the Total Harmonic Distortion (THD) at the output and differential input, for unloaded and resistive load conditions. Frequency response measurements are compared with theoretical expectations from Matlab TM and with MultiSim TM AC analysis simulations, thereby solidifying the frequency domain presentation of real-world opamp characteristics. Introduction Measuring Operational Amplifier (op-amp) parameters can represent one of the most challenging laboratories for undergraduate students. For the most part, many of these challenges stem from the fact that classical time-domain methods for measuring op-amp frequency domain
3 parameters can be cumbersome, and typically do not present a real-time visualization for the student. Parameters such as gain-bandwidth product, open and closed loop frequency responses, and Total Harmonic Distortion (THD) at the op-amp output and differential input (under loaded and unloaded conditions) are inherently easier to interpret, compute, and visualize in the frequency domain. Some of the early ideas for considering the effects of design with real-world op-amps originated with Peterson, Hartnett, and Gross at the U.S. Coast Guard Academy 1, however the authors focused primarily on the theoretical analysis and did not discuss measurement methodologies. Classic references such as Schaumann and Van Valkenburg 2 and Van Valkenberg 3 also discuss the idea of one-pole roll-off and two-pole roll-off real-world opamp models, but present no measurements or methodologies. In this paper we introduce successful measurement of op-amp parameters that our undergraduate students achieve at the U.S. Coast Guard Academy in a junior-level Linear Circuits laboratory. Students use Agilent 3567A Dynamic Signal Analyzers (DSA s) to measure the open-loop and closed-loop frequency responses, output resistance, and THD (at output and inverting input) of an inverting amplifier circuit using a µa741 op-amp. From their DSA measurements of frequency response, THD, and output resistance, students learn that even a low-cost µa741 opamp can behave as an ideal amplifier in a circuit, if one operates the amplifier within the gainbandwidth product, slew rate, and output current limitations of the selected op-amp. All frequency response measurements are then verified with theoretical expectations from Matlab TM and MultiSim TM AC analysis simulations. Laboratory Procedures The lab procedure consists of four steps: (1) Students set the DSA to measure the open-loop frequency response of the op-amp over two different frequency ranges. (2) Students then measure the closed-loop frequency response of an inverting amplifier for two different gain settings. (3) Students measure the output impedance of an inverting amplifier circuit as a function of frequency. (4) Finally, students measure the total harmonic distortion at V - and V out for a (clean) sinusoidal input, for loaded and unloaded inverting amplifier configurations. In the first step, students are instructed to build the circuit shown in Figure 1 using a A741 opamp. After verifying proper operation, they begin measuring the A741 op-amp open loop frequency response; that is the transfer function of the op-amp itself from the inverting terminal V - to V out. The general idea in this measurement is to provide the circuit input with a periodic chirp waveform, from the source output of the DSA, over the frequency range of interest. Connecting Channel#1 of the DSA to V - and Channel#2 of the DSA to V out of the circuit, coupled with selecting a frequency response measurement, yields a measurement of open-loop magnitude response for the student.
4 C1 15nF R kΩ Input 5 R3 1kΩ V VCC VCC VS- BAL1 BAL2 5 VS+ 1 U2 6 Output VDD 741 Figure 1. Op-amp circuit used to measure open loop frequency response of µa741 op-amp. One might ask why it would matter what circuit our students use if we are interested in an openloop frequency response measurement (i.e. from V - to V out ). The short answer is that an op-amp circuit is considerably more stable with negative feedback, so it is often much easier to measure open-loop characteristics operating within a closed loop system. The longer answer regarding why this particular circuit is used is perhaps more subtle. One might initially be tempted to have students simply use an inverting amplifier circuit such as one shown in Figure 2. In this circuit, assuming a periodic chirp input (which is spectrally flat), and taking a closer look at the voltage V - (the op-amp inverting input) as a function of frequency, we realize that the amplitudes of the low frequency components are extremely small compared to the higher frequency components. As a result, the A/D range for Channel #1 on the DSA would be dominated by high frequency components, and low frequency (low amplitude) measurements of V- (which form the denominator of the open-loop frequency response) would render open-loop frequency response measurements extremely challenging for students to make. 15V VDD R4 1kΩ -15V VCC Input 5 R3 1kΩ VCC VS- BAL2 BAL1 5 VS+ 1 U2 6 6 Output 741 VDD Figure 2. Typical inverting amplifier configuration (showing a linear gain of -1). 15V VDD
5 In having our students use the low pass amplifier configuration from Figure 1, the node voltage at V- is considerably more level as a function of frequency, which contributes to higher quality measurements of open-loop frequency response. The initial DSA settings for the open-loop circuit are shown in Table 1. Note, we use digital averaging (available as a choice on the Agilent 3567) to filter measurement noise. For this configuration, the circuit output and op-amp input V - have been connected to channel 2 and channel 1 of the DSA, respectively. Table 1 Initial DSA Setting for Open-loop Frequency Response Measurement Parameter Source Input Measure Freq Trace coordinate Average Auto Scale Setting On/Periodic Chirp 6 mvrms Freq Response (Ch2/Ch1) Start/&Span/4 Hz db/mag & x-axis/log On/1 On/off There are two measurements for the open loop frequency response, spanning the frequency ranges of 1 to 4 Hz and 128 to 512 Hz. The open-loop measurement magnitude response of the 741 op-amp in the range of 1 to 4 Hz is shown in Figure 3. As measured, the open-loop frequency response has a DC gain of approximately 19 db (28), and a break frequency of about 4 Hz. From the measurement data on the linear portion of the magnitude response curve from Figure 3, the gain bandwidth product can be calculated. Here the gain-bandwidth is approximately 1.1 MHz. GBP 8.87/2 1(1 ) 1.1MHz These results can be compared with the theoretical specifications of a real µa741 op-amp which predict an open loop DC gain of about 16 db (2), a break frequency of about 5 Hz, and a gain bandwidth product of approximately 1 MHz (shown in Figure 4).
6 Gain in db Magnitude (db) X: 1 Y: 19.1 Open-Loop Frequency Response of 741 Op-Amp X: 4 Y: X: 1 Y: Figure 3. Open-loop magnitude response (measured) for a µa741 op-amp. 12 Open Loop Magnitude Response Frequency in Hz Figure 4. Open-loop magnitude response (theoretical) for a µa741 op-amp.
7 Magnitude (db) In the second step of the lab, students measure the closed-loop magnitude response for an inverting amplifier (Figure 2) for two different gains (-1 and -1), over a frequency range of 128Hz to 51.2 khz. Settings from Table 1 are used; however the DSA periodic chirp amplitude routed to the circuit input is decreased to 1 mv rms to reduce high frequency slew rate nonlinearities. The open-loop and closed-loop magnitude response measurements for two different gains (-1 and -1) are shown in Figure 5. Theoretical open-loop and closed-loop magnitude response predictions are shown in Figure 6. After seeing both theoretical and measured results in the frequency domain, students appreciate that inverting amplifiers using real-world op-amps cannot provide constant gain over an infinite range of frequencies. Furthermore, in overlaying the open-loop characteristic with the closed-loop characteristic, students understand that real-world op-amps circuits seem to behave like ideal op-amp circuits if we restrict the input signal bandwidth. Deviation from ideal op-amp behavior in an inverting amplifier circuit occurs when we attempt to exceed the gain-bandwidth limitation of the op-amp itself. At that point, the closed-loop magnitude response seems to be dominated by the open-loop characteristic of the chosen op-amp, and students readily see this as they make their DSA measurements, perform MultiSim TM AC analysis simulations, and plot predicted magnitude responses from Matlab TM. 8 7 Open-Loop and Closed-Loop Frequency Responses of 741 Op-Amp Open-Loop Frequency Response Closed-Loop Frequency Response with Gain 1 Closed-Loop Frequency Response with Gain Figure 5. Open-loop and closed-loop magnitude responses (measured).
8 7 4 Gain in db 12 Closed Loop and Open Loop Magnitude Responses Open Loop Frequency Response 1 Closed Loop Frequency Responses (6, 4, and 2 db gain) , 1, 1, 1,, Figure 6. Open-loop and closed-loop magnitude responses (Predicted from MatlabTM, where a 1 st order transfer function has been assumed for the op-amp). In the third step of the lab, a 2 Ω load resistance is connected at the output of the amplifier (to ground) with a gain of -1 (Figure 7). The same settings from Table 1 are used in this step, however the periodic chirp amplitude from DSA routed to the circuit input is decreased to 1 mv rms (to minimize high frequency slew rate nonlinearities, and to keep output currents below the rated currents for a A741 op-amp). At this point, students immediately see (in real-time) the effect of loading and unloading the amplifier circuit (Figure 8). They see that applying a load resistor causes the magnitude response curve to shift downward at higher frequencies, and they see that the curve does not shift downward very much at lower frequencies. R4 1kΩ -15V VCC Input 5 R3 1kΩ VCC VS- BAL2 BAL1 5 VS+ 1 U2 6 Output R5 2Ω VDD VDD 15V Figure 7. Inverting amplifier configuration (gain of -1) showing load resistor.
9 Magnitude (db) 42 Closed-Loop Frequency Response of 741 Op-Amp Unloaded Closed-Loop Frequency Response Loaded Closed-Loop Frequency Response Figure 8. Measured magnitude response of the loaded and un-loaded inverting amplifier circuit with µa741 op-amp. The output impedance of the amplifier can be found by calculating the difference between the magnitude response of the loaded and unloaded frequency response. More specifically, examining the output side of Figure 7, we could consider this real world op-amp as an ideal opamp (with zero output impedance) in series with R out (Figure 9). By adding the load resistor, students use voltage division techniques to calculate Rout as a function of frequency. Vout* Rout Vout Rload = 2 ohms Figure 9. Simplified diagram for circuit from Figure 7, where we treat the real world op-amp as an ideal op-amp in series with R out. The measured amplifier output impedance is shown in Figure 1. Note that the amplifier impedance is very close to zero at low frequency (meaning that the amplifier behaves in an ideal way), and it goes up to about 43Ω above 4 khz (or as we approach the portion of the gain-
10 Output Impedance (Ohm) bandwidth curve that is dominated by the open-loop op-amp characteristic). From this investigation, students realize again that real-world op-amps circuits seem to behave like ideal op-amp circuits if we restrict the input signal bandwidth. 5 Closed-Loop Output Impedance of 741 Op-Amp Figure 1. Output impedance as a function of frequency for inverting amplifier configuration. In the fourth step of the lab, students measure the total harmonic distortion of a resistive loaded inverting amplifier, constructed with a 741 op-amp. The amplifier is designed for a gain of -3. The measurements shown here are the closed-loop circuit output and differential input of total harmonic distortions for resistive loaded and unloaded conditions. One important feature of op-amp circuits with negative feedback is that the closed loop system will often be highly linear even if the op-amp itself displays some nonlinearities. The normal method for measuring the linearity of an amplifier is to use a pure sinusoid as an input and measure the THD of the output. The effect of any non-linearity is to generate harmonics of the input frequency. Total Harmonic Distortion relates the root-squared sum (RSS) of the harmonic components to the amplitude of the fundamental, and is usually expressed either in db or as a percent. The DSA settings for this measurement are given in Table 2. Figure 11 shows the spectra of both the output (top trace) and the differential input (V-) (bottom trace) for the unloaded inverting amplifier with gain of -3. Note however, that the output is almost a perfect sinusoid,
11 Log Magnitude Log Magnitude indicating the closed loop system is highly linear, while the differential input voltage is slightly more distorted. Table 2 DSA Setting for Total Harmonic Distortion Key Bottom Selected Source On/Fixed Sine Input 3.5 Vot (-Peak) Measure Pwr Spec/Ch 2 Auto Scale On/off Trace coordinator Log/Mag & x-axis/lin Marker Function Harmonic /THD Display Format Up/Down Active Trace A/Output Active Trace B/ Differential Input Freq Start/124 Hz- Span/12.8 KHz 2 Distortion in Unloaded 741 Op-Amp at Output Measurement Magnitude Distortion in Unloaded 741 Op-Amp at Differential Input Measurement Magnitude Figure 11. The output distortion and the differential input distortion of unloaded amplifier (for pure sinusoidal input). More dramatic is to measure distortion at circuit output and differential input of the loaded amplifier (again, with gain of -3). A load resistor of 1kΩ is connected at the output where the DSA settings are given in Table 2. The output distortion and the differential input distortion of
12 Log magnitude Log magnitude loaded amplifier are shown in Figure 12. Note that the op-amp itself becomes much more nonlinear when connected to a resistive load. The closed loop system is still quite linear even though the open loop system is now much more non-linear than before, as evidence by the THD (approximately 6%) at the differential input. Note the significant distortion in the differential input, while the output is relatively undistorted. From this exercise, students learn that a realworld op-amp can in fact exhibit nonlinearities (especially as it drives more of a load), and that negative feedback makes a circuit with an imperfect op-amp behave in a much more ideal (linear) way. Said another way, a non-linear open loop system becomes a linear closed loop system. 2 Distortion in Loaded 741 Op-Amp at Output Measurement Magnitude Distortion in Loaded 741 Op-Amp at Differential Input Measurement Magnitude Figure 12. Output distortion and differential input distortion of loaded amplifier. Conclusion A successful methodology was provided to students in order to measure µa741op-amp circuit parameters in the frequency domain. As part of this lab, students used the Agilent 3567A Dynamic Signal Analyzer (DSA) to measure the open-loop and closed-loop frequency responses, and output resistance, of an inverting amplifier circuit. They also compared the Total Harmonic Distortion (THD) at the output and inverting input, for unloaded and resistive load conditions. All measurements were relatively easy to accomplish using the Agilent 3567A, and provided students a tremendous visual tool as they completed their lab exercises. Student learning was reinforced by using Matlab TM predictions, MultiSim TM simulations, and measurements, all in the frequency domain. Informal student feedback and results from one hourly examination suggest
13 to us that students learned exactly how real world op-amps differ from ideal op-amps. The most importantly, they learned that a real-world op-amp in a circuit with negative feedback can appear as if it is ideal, if one does not try to push the gain-bandwidth or slew rate capabilities of the device. References: 1. Peterson, B., Hartnett, R., and Gross, K., Analog and Digital Filter Design, unpublished textbook, Schaumann, R., and Van Valkenburg, M., Design of Analog Filters, Oxford University Press, Inc., New York, Van Valkenburg, M., Analog Filter Design, Holt, Rinehart and Winston, Inc., New York, 1982.
isim ACTIVE FILTER DESIGNER NEW, VERY CAPABLE, MULTI-STAGE ACTIVE FILTER DESIGN TOOL Michael Steffes Sr. Applications Manager 12/15/2010 SIMPLY SMARTER Introduction to the New Active Filter Designer Scope
Author: Don LaFontaine Making Accurate Voltage Noise and Current Noise Measurements on Operational Amplifiers Down to 0.1Hz Abstract Making accurate voltage and current noise measurements on op amps in
WHAT IS AN FFT SPECTRUM ANALYZER? ANALYZER BASICS The SR760 FFT Spectrum Analyzer takes a time varying input signal, like you would see on an oscilloscope trace, and computes its frequency spectrum. Fourier's
Electronics Operational Amplifier Internal design of an operational amplifier LD Physics Leaflets Discrete assembly of an operational amplifier as a transistor circuit P220.127.116.11 Objects of the experiment
Fundamentals of Signature Analysis An In-depth Overview of Power-off Testing Using Analog Signature Analysis www.huntron.com 1 www.huntron.com 2 Table of Contents SECTION 1. INTRODUCTION... 7 PURPOSE...
ROCHESTER INSTITUTE OF TECHNOLOGY MICROELECTRONIC ENGINEERING Selected Filter Circuits Dr. Lynn Fuller Webpage: http://people.rit.edu/lffeee 82 Lomb Memorial Drive Rochester, NY 146235604 Tel (585) 4752035
Loop Bandwidth and Clock Data Recovery (CDR) in Oscilloscope Measurements Application Note 1304-6 Abstract Time domain measurements are only as accurate as the trigger signal used to acquire them. Often
Designing Stable Compensation Networks for Single Phase Voltage Mode Buck Regulators Technical Brief December 3 TB47. Author: Doug Mattingly Assumptions This Technical Brief makes the following assumptions:.
Lecture 24 Inductance and Switching Power Supplies (how your solar charger voltage converter works) Copyright 2014 by Mark Horowitz 1 Roadmap: How Does This Work? 2 Processor Board 3 More Detailed Roadmap
Network Analyzer Fundamentals and Two Tone Linearity Name: Name: Name: Objective: To become familiar with the basic operation of a network analyzer To use the network analyzer to characterize the in-band
FREQUENCY RESPONSE ANALYZERS Dynamic Response Analyzers Servo analyzers When you need to stabilize feedback loops to measure hardware characteristics to measure system response BAFCO, INC. 717 Mearns Road
RF Network Analyzer Basics A tutorial, information and overview about the basics of the RF Network Analyzer. What is a Network Analyzer and how to use them, to include the Scalar Network Analyzer (SNA),
EE 242 EXPERIMENT 5: COMPUTER SIMULATION OF THREE-PHASE CIRCUITS USING PSPICE SCHEMATICS 1 Objective: To build, simulate, and analyze three-phase circuits using OrCAD Capture Pspice Schematics under balanced
Distributed by: www.jameco.com 1-800-831-4242 The content and copyrights of the attached material are the property of its owner. LF411 Low Offset, Low Drift JFET Input Operational Amplifier General Description
Network Analyzer Operation 2004 ITTC Summer Lecture Series John Paden Purposes of a Network Analyzer Network analyzers are not about computer networks! Purposes of a Network Analyzer Measures S-parameters
LM1036 Dual DC Operated Tone/Volume/Balance Circuit General Description The LM1036 is a DC controlled tone (bass/treble), volume and balance circuit for stereo applications in car radio, TV and audio systems.
Laboratory Manual and Supplementary Notes CoE 494: Communication Laboratory Version 1.2 Dr. Joseph Frank Dr. Sol Rosenstark Department of Electrical and Computer Engineering New Jersey Institute of Technology
SOP-8 DIP-8 Pin assignment: 1. CD 8. VO2 2. FC2 7. Gnd 3. FC1 6. Vcc 4. Vin 5. VO1 General Description The TS34119 is a low power audio amplifier, it integrated circuit intended (primarily) for telephone
MEASUREMENT UNCERTAINTY IN VECTOR NETWORK ANALYZER W. Li, J. Vandewege Department of Information Technology (INTEC) University of Gent, St.Pietersnieuwstaat 41, B-9000, Gent, Belgium Abstract: Precision
A B LOW NOISE JFET QUAD OPERATIONAL AMPLIFIERS WIDE COMMONMODE (UP TO V + CC ) AND DIFFERENTIAL VOLTAGE RANGE LOW INPUT BIAS AND OFFSET CURRENT LOW NOISE e n = 15nV/ Hz (typ) OUTPUT SHORTCIRCUIT PROTECTION
The Effect of Network Cabling on Bit Error Rate Performance By Paul Kish NORDX/CDT Table of Contents Introduction... 2 Probability of Causing Errors... 3 Noise Sources Contributing to Errors... 4 Bit Error
ITRODUCTIO Taking the Mystery out of the Infamous Formula, "SR = 6.0 + 1.76dB," and Why You Should Care by Walt Kester MT-001 TUTORIAL You don't have to deal with ADCs or DACs for long before running across
2006-1171: INCREASING PRODUCTIVITY AND AVOIDING CIRCUIT SIMULATION ERRORS IN MULTISIM John Hackworth, Old Dominion University John R. Hackworth is Program Director for the Electrical Engineering Technology
Comments on Audio Power Amplifier Design Handbook by Douglas Self Samuel Groner February 13, 2011 1 Introduction Douglas Self s writing on audio power amplifiers provides the reader with invaluable information.
VECTOR NETWORK ANALYZER PLANAR TR1300/1 DATA SHEET Frequency range: 300 khz to 1.3 GHz Measured parameters: S11, S21 Dynamic range of transmission measurement magnitude: 130 db Measurement time per point:
1. Introduction Oscilloscope, Function Generator, and Voltage Division In this lab the student will learn to use the oscilloscope and function generator. The student will also verify the concept of voltage
Simulating Envelope Tracking with Keysight Advanced Design System Application Note 1.0 Introduction Modern modulated signals often have high peak-to-average power ratios (PAPR). Use of such high PAPR signals
Ordering number: EN1321E Monolithic Linear IC LA4261 3.5 W 2-Channel AF Power Amplifier for Home Stereos and Music Centers Features. Minimum number of external parts required (No input capacitor, bootstrap
PAG 1/8 PREAMPLIFIERS - Class A FET Preamplifier 07X FET Phono Preamplifier 06X The Preamplifier 07X is Coda s flagship preamplifier, offering distortion-free preamplification and unsurpassed sonic performance
Blackmer Pre-Trimmed IC Voltage Controlled Amplifiers THAT 2180A, 2180B, 2180C FEATURES Wide Dynamic Range: >120 db Wide Gain Range: >130 db Exponential (db) Gain Control Low Distortion: < 0.01 % (2180A)
APSYN420A/B Specification 1.24 0.65-20.0 GHz Low Phase Noise Synthesizer 1 Introduction The APSYN420 is a wideband low phase-noise synthesizer operating from 0.65 to 20 GHz. The nominal output power is
Current Probes, More Useful Than You Think Training and design help in most areas of Electrical Engineering Copyright 1998 Institute of Electrical and Electronics Engineers. Reprinted from the IEEE 1998
Evaluating AC Current Sensor Options for Power Delivery Systems State-of-the-art isolated ac current sensors based on CMOS technology can increase efficiency, performance and reliability compared to legacy
Understanding the Fundamental Principles of Vector Network Analysis Application Note 1287-1 Table of Contents Page Introduction 2 Measurements in Communications Systems 2 Importance of Vector Measurements
INTEGRATED CIRCUITS DATA SHEET power amplifier with diagnostic facility Supersedes data of March 1994 File under Integrated Circuits, IC01 1996 Jan 08 FEATURES Requires very few external components High
AVX EMI SOLUTIONS Ron Demcko, Fellow of AVX Corporation Chris Mello, Principal Engineer, AVX Corporation Brian Ward, Business Manager, AVX Corporation Abstract EMC compatibility is becoming a key design
TDA7294 100V - 100W DMOS AUDIO AMPLIFIER WITH MUTE/ST-BY VERY HIGH OPERATING VOLTAGE RANGE (±40V) DMOS POWER STAGE HIGH OUTPUT POWER (UP TO 100W MU- SIC POWER) MUTING/STAND-BY FUNCTIONS NO SWITCH ON/OFF
Programmable Single-/Dual-/Triple- Tone Gong Preliminary Data SAE 800 Bipolar IC Features Supply voltage range 2.8 V to 18 V Few external components (no electrolytic capacitor) 1 tone, 2 tones, 3 tones
F O R T H E L O V E O F M U S I C F O R T H E L O V E O F M U S I C MODEL 2202IQ (1991-MSRP $549.00) OWNER'S MANUAL AND INSTALLATION GUIDE INTRODUCTION Congratulations on your decision to purchase a LINEAR
Data Transmission Concepts and terminology Transmission terminology Transmission from transmitter to receiver goes over some transmission medium using electromagnetic waves Guided media. Waves are guided
POWER FTOR CORRECTION INTRODUCTION Modern electronic equipment can create noise that will cause problems with other equipment on the same supply system. To reduce system disturbances it is therefore essential
Ver 3537 E. Analysis of Circuits () Key: [A]= easy... [E]=hard E. Circuit Analysis Problem Sheet (Lectures & ). [A] One of the following circuits is a series circuit and the other is a parallel circuit.
Helping Customers Innovate, Improve & Grow Application Note Signal Types and Terminations Introduction., H, LV, Sinewave, Clipped Sinewave, TTL, PECL,,, CML Oscillators and frequency control devices come
VCO Phase noise Characterizing Phase Noise The term phase noise is widely used for describing short term random frequency fluctuations of a signal. Frequency stability is a measure of the degree to which
INSTITUTE OF PHYSICS PUBLISHING Eur. J. Phys. 25 (2004) 581 591 EUROPEAN JOURNAL OF PHYSICS PII: S0143-0807(04)76273-X A wave lab inside a coaxial cable JoãoMSerra,MiguelCBrito,JMaiaAlves and A M Vallera
Servo Drive Current Loop Tuning Procedure (intended for Analog input PWM output servo drives) The standard tuning values used in ADVANCED Motion Controls drives are conservative and work well in over 90%
University of Pennsylvania Moore School of Electrical Engineering ESE319 Electronic Circuits - Modeling and Measurement Techniques 1. Introduction. Students are often frustrated in their attempts to execute
EE ELECTRICAL ENGINEERING See beginning of Section H for abbreviations, course numbers and coding. The * denotes labs which are held on alternate weeks. A minimum grade of C is required for all prerequisite
Introduction to Receivers Purpose: translate RF signals to baseband Shift frequency Amplify Filter Demodulate Why is this a challenge? Interference (selectivity, images and distortion) Large dynamic range
Application Report SLVA79 - October 1999 Understanding the Terms and Definitions of ltage Regulators Bang S. Lee Mixed Signal Products ABSTRACT This report provides an understanding of the terms and definitions
Voltage/current converter opamp circuits This worksheet and all related files are licensed under the Creative Commons Attribution License, version 1.0. To view a copy of this license, visit http://creativecommons.org/licenses/by/1.0/,
Progress In Electromagnetics esearch C, Vol. 42, 191 203, 2013 CONDUCTED EMISSION MEASUEMENT OF A CELL PHONE POCESSO MODULE Fayu Wan *, Junxiang Ge, and Mengxiang Qu Nanjing University of Information Science
Application Note #49 RF Amplifier Output Voltage, Current, Power, and Impedance Relationship By: Jason Smith; Manager Applications Engineer and Pat Malloy; Sr. Applications Engineer How much output voltage,
Pacific Gas and Electric Company Power System Harmonics What are power system harmonics? Ideally, voltage and current waveforms are perfect sinusoids. However, because of the increased popularity of electronic
Section 2150 Undergraduate Research Projects for Engineering Technology Students Chih-Ping Yeh, Jeannie Drew, Chris Rockwell, Hai-Chun Chien Division of Engineering Technology Wayne State University Abstract
Designing the NEWCARD Connector Interface to Extend PCI Express Serial Architecture to the PC Card Modular Form Factor Abstract This paper provides information about the NEWCARD connector and board design
Code Blue IP Wireless IP WIRELESS NETWORK Code Blue offers an IP Wireless solution for those areas where phone lines are not available or trenching costs are prohibitive. As more users desire ubiquitous
Chapter Frequency response. Introduction The frequency response of a system is a frequency dependent function which expresses how a sinusoidal signal of a given frequency on the system input is transferred
What Does Rail-to-Rail Operation Really Mean? 2004 Microchip Technology Incorporated. All Rights Reserved. What does Rail-to-Rail Operation really mean? 1 Agenda What does Rail-to-Rail output operation
Session 1448 Notes on Modifying an EET Associate Degree Curriculum to Improve Graduate Placement James Stewart, William Lin DeVry College of Technology North Brunswick, New Jersey / Purdue School of Engineering
Experiment 7: Familiarization with the Network Analyzer Measurements to characterize networks at high frequencies (RF and microwave frequencies) are usually done in terms of scattering parameters (S parameters).
Simple Signal Injector Aids Control-Loop Analysis A signal-injection circuit for control-loop analysis is flat from dc to 200 khz, isolated from chassis ground and easily constructed with a readily available
J Indian Inst Sci, July Aug 2006, 86, 315 325 Indian Institute of Science Application of network analyzer in measuring the performance functions of power supply B SWAMINATHAN* AND V RAMANARAYANAN Power
Environmental Monitoring with Sensors: Hands-on Exercise Now that you ve seen a few types of sensors, along with some circuits that can be developed to condition their responses, let s spend a bit of time
Direct Digital Amplification (DDX ) The Evolution of Digital Amplification Tempo Semiconductor 2013 Table of Contents Table of Contents... 2 Table of Figures... 2 1. DDX Technology Overview... 3 2. Comparison
ECE 584 Laboratory Experiments 2. The Vector Network Analyzer Introduction: In this experiment we will learn to use a Vector Network Analyzer to measure the magnitude and phase of reflection and transmission
Website: www.ijetae.com (SSN 50-459, SO 900:008 Certified Journal, Volume 3, ssue, February 03) Analysis of AC-DC Converter Based on Power Factor and THD Shiney.S.Varghese, Sincy George Department of Electrical
Technical Datasheet Scalar Network Analyzer Model 8003-10 MHz to 40 GHz The Giga-tronics Model 8003 Precision Scalar Network Analyzer combines a 90 db wide dynamic range with the accuracy and linearity
Yamaha Power Amplifier White Paper August 2008 Table of Contents 1. About EEEngine...2 1.1. Introduction...2 1.2. Explanation of different amplifier topologies...2 2. Yamaha technology...5 2.1. Dual mono-amplifier
INTEGRATED CIRCUITS DATA SHEET 24 W BTL or 2 x 12 W stereo car radio File under Integrated Circuits, IC01 January 1992 GENERAL DESCRIPTION The is a class-b integrated output amplifier encapsulated in a
C esonant Circuits Andrew McHutchon April 20, 203 Capacitors and Inductors There is a lot of inconsistency when it comes to dealing with reactances of complex components. The format followed in this document
Application note Signal conditioning for pyroelectric passive infrared (PIR) sensors Sylvain Colliard-Piraud Introduction Pyroelectric passive infrared (PIR) sensors are widely used in daily life. They
PHYSCS 111 LABORATORY Experiment #3 Current, Voltage and Resistance in Series and Parallel Circuits This experiment is designed to investigate the relationship between current and potential in simple series
Topic Bandwidth-dependent transformation of noise data from frequency into time domain and vice versa Authors Peter Bormann (formerly GeoForschungsZentrum Potsdam, Telegrafenberg, D-14473 Potsdam, Germany),
Agilent PN 89400-13 Extending Vector Signal Analysis to 26.5 GHz with 20 MHz Information Bandwidth Product Note The Agilent Technologies 89400 series vector signal analyzers provide unmatched signal analysis
Session #3133 A Low Cost Power Quality and Energy Savings Laboratory for Undergraduate Education and Research Abdullatif Bagegni, Vance Poteat Merrimack College Abstract This paper describes the design
EDM SOFTWARE ENGINEERING DATA MANAGEMENT SOFTWARE MODERN, UPATED INTERFACE WITH INTUITIVE LAYOUT DRAG & DROP SCREENS, GENERATE REPORTS WITH ONE CLICK, AND UPDATE SOFTWARE ONLINE ipad APP VERSION AVAILABLE
BELASIGNA 250 and 300 for Low-Bandwidth Applications APPLICATION NOTE Introduction This application note describes the use of BELASIGNA 250 and BELASIGNA 300 in low bandwidth applications. The intended
Product: Vector Network Analyzer R&S ZVB Performing Amplifier Measurements with the Vector Network Analyzer ZVB Application Note This document describes typical measurements that are required to be made
a FEATURES Small Size: 4 Channels/lnch Low Power: 35 mw () High Accuracy:.25% Max Nonlinearity (K Grade) High CMR: 13 db (Gain = 1 V/V) Wide Bandwidth: 5 khz Full-Power () High CMV Isolation: 2 V pk Continuous