HOW TO MEASURE PRESSURE HOW TO MEASURE FLOW

Save this PDF as:

Size: px
Start display at page:

Transcription

1 HOW TO MEASURE PRESSURE HOW TO MEASURE FLOW by G.J.Matthews Airflow Developments Limited

2 HOW TO MEASURE PRESSURE. FUNDAMENTAL PRINCIPLES. Due to the depth of the earth s atmosphere the air around us exerts a pressure, known as atmospheric pressure. This can be measured with a barometer and varies depending upon the weather conditions. It also varies according to altitude above or below sea level at which it is made. Pressure measurements in heating systems are usually made relative to atmospheric pressure, and may be corrected to Standard Conditions to make possible a comparison between readings. The internationally agreed Standard Conditions are for an air density of 1.2 kg/m³. ( Atmospheric air at a temperature of 16 C, a pressure of 100,000 Pa. and a relative humidity of 65% has a density of 1.2 kg/m³, but these conditions are not part of the definition.) The only pressure acting at any point in a stationary fluid is Hydrostatic pressure which acts equally in all directions. If the fluid is set in motion the pressure still persists, but may change in magnitude. If we consider a very thin disc placed in the fluid, and moving with it, the static pressure as it is called will act on the two faces of the disc. P S INFINITELY THIN DISC. P S 2

3 If the disc is stationary the pressure acting on the two surfaces will be the same providing the presence of the disc does not disturb the flow. P S P S NO DISTURBANCE However if the stationary disc is turned through 90 its front face will be subjected to not only static pressure, but also an additional dynamic pressure equal to the velocity pressure in incompressible flow, due to the impact. P S V P D = ½ x x V² P D = DENSITY OF THE FLUID The velocity pressure is due entirely to the motion of the flow and depends upon the velocity and the density of the fluid. 3

4 The algebraic sum of the static pressure and the velocity pressure is called the total pressure. MEASURING TOTAL AND STATIC PRESSURE. A tube placed in a duct facing into the direction of the flow will measure the total pressure in the duct. If frictional losses are neglected, the mean total pressure at any cross section throughout the duct system is constant. Static pressure can only be determined accurately by measuring it in a manner such that the velocity pressure has no influence on the measurement at all. This is carried out by measuring it through a small hole at the wall of the duct; or a series of holes positioned at right angles to the flow in a surface lying parallel to the lines of flow. The pitot static tube is an example of this. Figure 1 shows the principle of the pitot static tube. Figure 1. Principle of the pitot static tube. 4

5 It will be seen that connecting tubes to a manometer makes the measurement of pressure. INSTRUMENTS FOR MEASURING PRESSURE. [1]. U TUBE MANOMETER. Although probably the oldest method of measuring low pressures, the simple U tube has much to commend it. If a U shaped glass tube is half filled with a liquid, e.g. water, and a pressure is applied to one end of the limb, the other being open to atmosphere, the liquid will move to balance the pressure. The weight of liquid so displaced will be proportional to the pressure applied. As the difference in height of the two columns of liquid and the density are known the pressure can be calculated. Each millimetre height difference of water column represents approximately 10 Pascals. Figure 2. Principle of the U-tube manometer. A disadvantage of the U tube is that the scale has to be constantly moved to line up with the moving zero 5

6 Alternatively with zero taken at the centre point the scale length is halved with subsequent loss of resolution. [2]. LIQUID - IN GLASS MANOMETERS. The disadvantages of the simple U tube manometer are overcome and other advantages incorporated in single limb industrial manometers in which it is only necessary to read one liquid level. In one such design, one of the limbs of the U tube is replaced by a reservoir, thus substantially increasing the surface area. A pressure applied to this reservoir causes the level of fluid to move a small, calculable amount. The same volume of fluid displaced in the glass limb produces a considerable change in the level. This nearly doubles the resolution compared with the U tube manometer for vertical instruments and gives much greater magnification when the limb is inclined. The manometer fluid may be plain water, but problems can arise from algae growing in the tube causing the density of the fluid to alter. Special blends of paraffin are often used and these have several advantages: a free moving meniscus, no staining of the tube, and expanded scales due to low relative densities. Where higher pressures are required, denser fluids are used, of which mercury is often used. For very low pressures the manometer limb is inclined to improve the resolution further. More precision can be achieved with adjustable - range limbs and it is possible to achieve from Pa. to Pa. with only two limbs. Figure 4: Examples of Single Limb Manometers 6

7 Figure 3. Precision portable manometer set. [3]. DIAL GAUGES. Dial pressure gauges are primarily employed for reading high pressures. At very low pressures they tend to exhibit unacceptable hysteresis errors unless they are very high quality instruments. Due to short scale length, the resolution is not usually very good. [4]. PRESSURE TRANSDUCERS. As an alternative to the fundamental liquid - filled portable manometer, electronic pressure transducer based instruments are available for laboratory or site use. They are generally compact as to be hand held, and eliminate the use of fluids, giving an acceptable level of accuracy for normal ventilating and related air movement measurement. The capacitance transducer employs a precision diaphragm moving between fixed electrodes. This causes capacitance changes proportional to a differential pressure. 7

8 The piezo resistive pressure sensor contains a silicon chip with an integral sensing diaphragm and four piezo-resistors; pressure applied on the diaphragm causes it to flex changing the resistance; this causes a low level output voltage proportional to pressure. A pressure transducer - based instrument allows continuous monitoring using a recorder, or input to electronic storage or control equipment. Figure 6. Pocket Manometer DB2. Microprocessor technology allows for the pressure readings to be converted into velocity readings when using a pitot static tube. In some instances the probe calibration factor may be inputted separately allowing other pressure measuring probes to be used with direct velocity readout. Several readings can be stored, with MIN, MAX and AVE functions. In some instances, duct area can be inputted to allow direct volume flow measurement 8

9 Figure 7. Airflow MEDM 5k Micromanometer. 9

10 Figure 8. PVM100 Micromanometer. 10

11 HOW TO MEASURE VOLUME FLOW RATE. FORMULAE. or Volume flow rate = Mass flow rate / Density. Volume flow rate = Velocity x duct cross-section area. The volume flow rate in a system can be measured at the entrance to the system, at the exit from the system, or somewhere within the system itself. This could involve measuring the total flow rate or the flow rate in a portion of the system. Wherever it is measured, it should be a prerequisite that the flow should be swirl - free. There are several methods with which the flow rate can be measured. [1]. IN - LINE FLOWMETERS (STANDARD PRIMARY DEVICES) BS1042 Part 1 : 1990 describes such devices as the venturi nozzle, the orifice plate, and the conical inlet. The venturi nozzle and the orifice plate may be used at the inlet to or the outlet from a system as well as between two sections of an airway. The conical inlet draws air from a free space at the entrance to a system. With regard to measuring the volume flow rate of general purpose fans, the requirements of BS1042 with regard to the lengths of straight duct upstream of the flow meter is reduced and BS848 Part 1 : 1980 details these changes together with the associated uncertainty of measurement. The general equation for these differential flowmeters is: where q m = [ d ² / 4 ] 2 u P qm = mass flow rate ( kg / s ) = flow coefficient = expansibility factor d = throat diameter u = upstream density ( kg / m³ ) P = pressure difference ( Pa.) It is not proposed to go into too much detail with these devices as both BS1042 and BS848 are complete documents and should be referred to. 11

12 [2]. PITOT - STATIC TUBE TRAVERSE. A commonly used method of accurately establishing the air flow rate in a duct is by traversing the duct with a pitot - static tube connected to a precision manometer. A conveniently accessible part of the duct should be selected; preferably where there is a straight parallel section of duct of at least 5 diameters downstream of any bend, obstruction or abrupt change of section. Velocity readings are required at the prescribed points as shown in figures 6 and 7. The air velocity at the point of measurement can easily be calculated from the velocity pressure reading according to the following formulae: The standard formula for calculating velocity from velocity pressure is: V = p v (m/s) This is only correct for standard air of 1.2 kg/m³ For non - standard air conditions the formula becomes: V = {100000/p a } {T/289} {100000/( p s )} {p v } (m/s) where V = air velocity ( m / s ) pv = velocity pressure ( Pa.) pa = atmospheric pressure ( Pa.) ps = static pressure ( Pa.) T = absolute temperature ( K ) (= t + 273) t = airstream temperature ( C ) The expressions {100000/pa}, {T/289} and {100000/( ps)} are corrections for atmospheric pressure, air temperature, and duct pressure to bring the measured value of pv to the equivalent of standard air. If the duct static pressure is less than 2500 Pa. it can normally be ignored. Similarly, the other two expressions can be ignored during site testing if they are not likely to affect the equivalent pv by 2 or 3%. When averaging the readings taken on a duct traverse it is strictly correct to average the air velocities (which is equivalent to averaging the square roots 12

13 of the velocity pressures). In practice, however, no great error will be introduced by taking a simple average of the velocity pressures when calculating the mean velocity providing the majority of the readings do not vary by more than about 25% from the mean value. 13

14 Figure 9. Measuring points for circular ducts. Log linear rule for traverse points on 3 diameters. Figure 10. Measuring points in rectangular ducts. Log Tchebycheff rule. 14

15 [3]. ANEMOMETER DUCT TRAVERSE. Anemometers are instruments, which measure air velocity. Two popular types are the Rotating vane anemometer and the Thermal anemometer. The rotating vane type is basically a mechanical device where a shrouded rotating vane rotates like a windmill, with delicate clockwork gearing recording the number of revolutions of the vane on a multi - point dial. These anemometers record the linear movement of air past the instrument in metres or feet for as long as it is held in the airstream. By noting the time with a stop watch the velocity in m/s or ft/min can be determined. In more recent anemometers as each blade of the rotating vane passes a pick up, it is counted electronically, the signal fed into electronic circuitry with its own time base and displays the measured velocity directly and instantaneously without any need for external timing. The anemometer head is normally 100 mm diameter in size, but heads as small as 16 mm are available. It can be used in larger ducts to traverse the duct to obtain mean duct velocity as with the pitot - static tube. Obviously the larger size head ( when compared to the pitot - static tube) interferes with the airflow in smaller ducts with the result that it causes the local duct velocity to increase to a significant effect. Figure 11 shows the typical increase in velocity reading when placed in different sized ducts. It is possible to have one anemometer head centrally located in its own test duct and be independently calibrated for volume flow rate. The Airflow 100mm rotating vane anemometers can be fitted with an Aircone flow hood which allows the volume flow rate through small grilles to be measured easily and accurately. The volume flow rate can be measured using an ATP600 Airflow Measuring Hood for larger sized grilles. Alternatively, it is recommended that a cardboard skirt of at least 2 x the length of the shortest side is taped around the grille and an anemometer traverse carried out across the entrance of the cardboard skirt to obtain the mean velocity. 15

16 FIGURE 11:-100 mm ANEMOMETER HEAD BLOCKAGE EFFECT mm DIA. DUCT PERCENTAGE INCREASE IN VELOCITY DUE TO HEAD BLOCKAGE mm DIA. DUCT 400mm DIA. DUCT 500mm DIA. DUCT DUCT AREA (sq.metres) 16

17 Figure 12. LCA30iS intrinsically safe vane anemometer. 17

18 Figure 13: Various vane Anemometers from AIRFLOW 18

19 Figure 15: ProHood Airflow Measuring Hood. [4]. THERMAL ANEMOMETERS. A wet finger in the air will detect the direction of the wind because a drop in temperature is felt on the surface facing the wind. Thermal anemometers act in a similar way, in that the passage of air takes heat away from a heated element at a rate dependant upon the velocity. This element is mounted at the end of a probe that can be inserted into the airstream. Velocity readout is direct and instantaneous. The method of heating the element ( which can be a small length of thin wire, a simple thermocouple or a thermister ) and the read - out technique determine the quality of the instrument. Because it is a thermal device, it is 19

20 important to compensate for variations in ambient temperature, pressure, and the composition of the gas / air being measured. The probe is normally small similar to that of the pitot - static tube, and is particularly appropriate for use at low velocities where a vane anemometer ( due to mechanical friction effects ) or pitot - static tube would not be sensitive enough. Its principal disadvantage is that they take only a point velocity reading and a traverse is needed to obtain an average velocity reading. Figure 16: TA35 thermal anemometer. 20

21 [5]. WILSON FLOW GRID. The Wilson flow grid consists of a row of parallel tubes with closed ends and forms what can be described as an open fence across the duct at right angles to the axis of flow. Some of the tubes are perforated with small holes facing upstream ( sensing total pressure ), whilst other tubes have holes on the downstream side to sense throat sub - static pressure. Both sets of tubes are connected to separate manifolds which provide two average pressure signals. The difference between the two signal constitutes an enhanced output signal. Figure 17. The Wilson flow grid. The presence of the grid in the duct creates a partial blockage in the cross - sectional area of the duct resulting in an increase in local velocity between the tubes. The positive pressure is derived from the total pressure sampled from the distribution of the forward facing holes. The negative pressure is derived from a combination of the depressed static pressure between the tube and the suction zone immediately behind the tubes. 21

22 The holes in the tubes are positioned such that an acceptable sampling system is utilised. The output signal can be compared to the mean duct velocity pressure that would be present in the duct without the grid being present. This is commonly termed the magnification factor. The output signal can also be plotted against the volume flow rate in the duct. This is possible because each Wilson flow grid is manufactured to suit a given duct size. It is possible to use a single double tube in a duct section, but because there is no effective manifold the reading can be slightly erratic. It should also be noted that the single tube type only senses the flow rate in one plane which is not ideal. 22

23 TABLE OF CONVERSION FACTORS FOR PRESSURE AND VOLUME FLOW RATE. In the table below, I have tried to include all of the most popular combinations, with the preferred S.I. unit shown in blue VOLUME FLOW RATE. 1 m³/s = l/s = 3600 m³/h = cfm = 60 m³/min 1 l/s = 3.60 m³/h = cfm = 0.06 m³/min = m³/s 1 m³/h = cfm = m³/min= m³/s = l/s 1 cfm = m³/min = m³/s = l/s = m³/h 1 m³/min = m³/s = l/s = 60 m³/h = cfm VELOCITY. 1 m/s = ft/min 1 ft/min = m/s PRESSURE. 1 Pa = 0.01 mbar = in.wg = mm.wg = psi. 1 mbar = in.wg = mm.wg = psi. = 100 Pa. 1 in.wg. = 25.4 mm.wg = psi. = Pa. = mbar. 1 mm.wg = psi. = Pa. = mbar = in.wg. 1 psi. = Pa. = mbar = in.wg. = mm.wg. It is recommended that the following multiples and sub-multiples of the S.I. units are used in fan technology: Pressure: kilopascal = kpa = 1000Pa. Barometric pressure: millibar = mbar = 100 Pa. Volume flow rate: litre per second = l/s = 10-3 m 3 /s 23

6. ENERGY PERFORMANCE ASSESSMENT OF FANS AND BLOWERS

6. ENERGY PERFORMANCE ASSESSMENT OF FANS AND BLOWERS 6.1 Introduction This section describes the method of testing a fan installed on site in order to determine the performance of the fan in conjunction

INDUSTRIAL PROCESS AIR HANDLING & MEASUREMENT. Air Velocity/Volume Measurement

3-06 INTRODUCTION INDUSTRIAL PROCESS AIR HANDLING & MEASUREMENT Air Velocity/Volume Measurement The movement of air is a factor in almost all modern industrial facilities. Industrial air movement is usually

Michael Montgomery Marketing Product Manager Rosemount Inc. Russ Evans Manager of Engineering and Design Rosemount Inc.

ASGMT / Averaging Pitot Tube Flow Measurement Michael Montgomery Marketing Product Manager Rosemount Inc. Russ Evans Manager of Engineering and Design Rosemount Inc. Averaging Pitot Tube Meters Introduction

Chapter 9. Pressure. Pressure. Pressure and Velocity Measurements

Chapter 9 Pressure and Velocity Measurements Material from Theory and Design for Mechanical Measurements; Figliola, Third Edition Pressure Pressure is force per unit area. It acts inward or outward, normal

Head Loss in Pipe Flow ME 123: Mechanical Engineering Laboratory II: Fluids

Head Loss in Pipe Flow ME 123: Mechanical Engineering Laboratory II: Fluids Dr. J. M. Meyers Dr. D. G. Fletcher Dr. Y. Dubief 1. Introduction Last lab you investigated flow loss in a pipe due to the roughness

Fluid Mechanics Definitions

Definitions 9-1a1 Fluids Substances in either the liquid or gas phase Cannot support shear Density Mass per unit volume Specific Volume Specific Weight % " = lim g#m ( ' * = +g #V \$0& #V ) Specific Gravity

Performance 4. Fluid Statics, Dynamics, and Airspeed Indicators

Performance 4. Fluid Statics, Dynamics, and Airspeed Indicators From our previous brief encounter with fluid mechanics we developed two equations: the one-dimensional continuity equation, and the differential

Flow Measurement Options for Pipeline and Open Channel Flow

Flow Measurement Options for Pipeline and Open Channel Flow October 2013 Presented by Molly Skorpik - 2013 Montana Association of Dam and Canal Systems Conference Irrigation Training and Research Center

Accurate Air Flow Measurement in Electronics Cooling

Accurate Air Flow Measurement in Electronics Cooling Joachim Preiss, Raouf Ismail Cambridge AccuSense, Inc. E-mail: info@degreec.com Air is the most commonly used medium to remove heat from electronics

Outdated Publication, for historical use. CAUTION: Recommendations in this publication may be obsolete.

IRRIGATION MANAGEMENT S E R I E S Irrigation Water Measurement Danny H. Rogers Extension Agricultural Engineer Richard D. Black Extension Agricultural Engineers Cooperative Extension Service Kansas State

Chapter 3.5: Fans and Blowers

Part I: Objective type questions and answers Chapter 3.5: Fans and Blowers 1. The parameter used by ASME to define fans, blowers and compressors is a) Fan ration b) Specific ratio c) Blade ratio d) Twist

Gauge Pressure, Absolute Pressure, and Pressure Measurement

Gauge Pressure, Absolute Pressure, and Pressure Measurement Bởi: OpenStaxCollege If you limp into a gas station with a nearly flat tire, you will notice the tire gauge on the airline reads nearly zero

δy θ Pressure is used to indicate the normal force per unit area at a given point acting on a given plane.

2 FLUID PRESSURES By definition, a fluid must deform continuously when a shear stress of any magnitude is applied. Therefore when a fluid is either at rest or moving in such a manner that there is no relative

FIGURE P8 50E FIGURE P8 62. Minor Losses

8 48 Glycerin at 40 C with r 1252 kg/m 3 and m 0.27 kg/m s is flowing through a 4-cm-diameter horizontal smooth pipe with an average velocity of 3.5 m/s. Determine the pressure drop per 10 m of the pipe.

Measure Pressure and Flow

Measure Pressure and Flow Install the EMANT300 USB DAQ and Stepper Motor Training Kit 1. Connect up the EMANT300 USB DAQ and Pressure and Air pump Adaptor as shown. EMANT300 USB DAQ Pressure and Air pump

2013 RESNET Conference Orlando, FL

2013 RESNET Conference Orlando, FL Why do we measure AirFlow? Heat/energy transfer medium We need The right amount (mass) In the right place At the right time With the correct temperature and humidity

Metrology Society of Australia MSA Test Method Calibration of Pressure Calibrators, Indicators and Transducers

Metrology Society of Australia MSA Test Method 1-2008 Calibration of Pressure Calibrators, Indicators and Transducers Revision No. 5 June 2008 Contents: 1. Introduction 2. Scope 3. Desired outcomes of

Understanding Pressure and Pressure Measurement

Freescale Semiconductor Application Note Rev 1, 05/2005 Understanding Pressure and Pressure Measurement by: David Heeley Sensor Products Division, Phoenix, Arizona INTRODUCTION Fluid systems, pressure

h b b h Q u da u 1 1 dy dz u 1 dy 1 dz u b h b h

P3.18 An incompressible fluid flows steadily through the rectangular duct in the figure. The exit velocity profile is given by u umax(1 y /b )(1 z /h ). (a) Does this profile satisfy the correct boundary

Fluid Mechanics. Fluid Statics [3-1] Dr. Mohammad N. Almasri. [3] Fall 2010 Fluid Mechanics Dr. Mohammad N. Almasri [3-1] Fluid Statics

1 Fluid Mechanics Fluid Statics [3-1] Dr. Mohammad N. Almasri Fluid Pressure Fluid pressure is the normal force exerted by the fluid per unit area at some location within the fluid Fluid pressure has the

Experiment # 3: Pipe Flow

ME 05 Mechanical Engineering Lab Page ME 05 Mechanical Engineering Laboratory Spring Quarter 00 Experiment # 3: Pipe Flow Objectives: a) Calibrate a pressure transducer and two different flowmeters (paddlewheel

Basics and Concepts. 2.1 Introduction Force, Weight, and Mass Density Specific Gravity Pressure...

2 Fluids Basics and Concepts TOPIC PAGE 2.1 Introduction... 22 2.2 Force, Weight, and Mass... 22 2.3 Density... 23 2.4 Specific Gravity... 23 2.5 Pressure... 23 2.6 Temperature... 25 2.7 Viscosity... 26

CHE 253M Experiment No. 3 LIQUID FLOW MEASUREMENT

Rev 8/15 AW/GW CHE 253M Experiment No. 3 LIQUID FLOW MEASUREMENT The objective of this experiment is to familiarize the student with several types of flow-measuring devices commonly used in the laboratory

AP2 Fluids. Kinetic Energy (A) stays the same stays the same (B) increases increases (C) stays the same increases (D) increases stays the same

A cart full of water travels horizontally on a frictionless track with initial velocity v. As shown in the diagram, in the back wall of the cart there is a small opening near the bottom of the wall that

Grant Agreement No. 228296 SFERA. Solar Facilities for the European Research Area SEVENTH FRAMEWORK PROGRAMME. Capacities Specific Programme

Grant Agreement No. 228296 SFERA Solar Facilities for the European Research Area SEVENTH FRAMEWORK PROGRAMME Capacities Specific Programme Research Infrastructures Integrating Activity - Combination of

Albaha University Faculty of Engineering Mechanical Engineering Department Lecture 9: Pressure measurement Ossama Abouelatta o_abouelatta@yahoo.com Mechanical Engineering Department Faculty of Engineering

When the fluid velocity is zero, called the hydrostatic condition, the pressure variation is due only to the weight of the fluid.

Fluid Statics When the fluid velocity is zero, called the hydrostatic condition, the pressure variation is due only to the weight of the fluid. Consider a small wedge of fluid at rest of size Δx, Δz, Δs

NEBB STANDARDS SECTION-8 AIR SYSTEM TAB PROCEDURES

NEBB STANDARDS SECTION-8 AIR SYSTEM TAB PROCEDURES 8.1 INTRODUCTION Testing, adjusting, and balancing of HVAC systems can best be accomplished by following a series of systematic procedures. The NEBB TAB

Chapter 10. Flow Rate. Flow Rate. Flow Measurements. The velocity of the flow is described at any

Chapter 10 Flow Measurements Material from Theory and Design for Mechanical Measurements; Figliola, Third Edition Flow Rate Flow rate can be expressed in terms of volume flow rate (volume/time) or mass

Name: Class: Date: Exam 4--PHYS 101--F14 Multiple Choice Identify the choice that best completes the statement or answers the question. 1. A wheel, initially at rest, rotates with a constant acceleration

ENGINEERING DATA. Fan Performance Troubleshooting Guide

Information and Recommendations for the Engineer ENGINEERING DATA Aerovent TC Ventco Fiber-Aire Twin City Fan & Blower TC Axial Clarage Fan Performance Troubleshooting Guide ED- Introduction Trying to

Velocity and Motion sensors (1)

Velocity and Motion sensors (1) Many sensors is now under operation to monitor linear and angular velocity and detect motion however the most used are Encoders, Tacho-generators Pyroelectic sensors Both

Laminar and Turbulent flow. Flow Sensors. Reynolds Number. Thermal flow Sensor. Flow and Flow rate. R = Mass Flow controllers

Flow and Flow rate. Laminar and Turbulent flow Laminar flow: smooth, orderly and regular Mechanical sensors have inertia, which can integrate out small variations due to turbulence Turbulent flow: chaotic

Practice Problems on Boundary Layers. Answer(s): D = 107 N D = 152 N. C. Wassgren, Purdue University Page 1 of 17 Last Updated: 2010 Nov 22

BL_01 A thin flat plate 55 by 110 cm is immersed in a 6 m/s stream of SAE 10 oil at 20 C. Compute the total skin friction drag if the stream is parallel to (a) the long side and (b) the short side. D =

CO 2 41.2 MPa (abs) 20 C

comp_02 A CO 2 cartridge is used to propel a small rocket cart. Compressed CO 2, stored at a pressure of 41.2 MPa (abs) and a temperature of 20 C, is expanded through a smoothly contoured converging nozzle

Electric vacuum pump for brake servo unit

Service. Self-Study Programme 257 Electric vacuum pump for brake servo unit Design and Function Cars fitted with a petrol engine in combination with an automatic gearbox which comply with the EU 4 Standard

FM Flow Measurement by Venturi and Orifice meter

FM 308 - Flow Measurement by Venturi and Orifice meter Objectives: 1. To find the coefficient of discharge for venturi meter. 2. To find the coefficient of discharge for orifice meter. Theory: Venturi

Experiment 3 Pipe Friction

EML 316L Experiment 3 Pipe Friction Laboratory Manual Mechanical and Materials Engineering Department College of Engineering FLORIDA INTERNATIONAL UNIVERSITY Nomenclature Symbol Description Unit A cross-sectional

Averaging Pitot Tubes; Fact and Fiction

Averaging Pitot Tubes; Fact and Fiction Abstract An experimental investigation has been undertaken to elucidate effects of averaging stagnation pressures on estimated velocities for pressure averaging

Exhaust Calculation Booklet

Exhaust Calculation Booklet American Dryer Corporation 88 Currant Road Fall River MA 02720-4781 Telephone: (508) 678-9000 / Fax: (508) 678-9447 e-mail: techsupport@amdry.com ADC Part No. 450450 Exhaust

PARTIALLY CLOSED VALVE EFFECTS ON THE V-CONE FLOWMETER

PARTIALLY CLOSED VALVE EFFECTS ON THE V-CONE FLOWMETER Stephen A. Ifft (McCrometer Inc., Hemet, CA 9545, USA) ABSTRACT Research conducted from 199 through 1996 indicates that the V-Cone flowmeter is less

FLOW MEASUREMENT 2001 INTERNATIONAL CONFERENCE DERIVATION OF AN EXPANSIBILITY FACTOR FOR THE V-CONE METER

FLOW MEASUREMENT 200 INTERNATIONAL CONFERENCE DERIVATION OF AN EXPANSIBILITY FACTOR FOR THE V-CONE METER Dr D G Stewart, NEL Dr M Reader-Harris, NEL Dr R J W Peters, McCrometer Inc INTRODUCTION The V-Cone

E 490 Fundamentals of Engineering Review. Fluid Mechanics. M. A. Boles, PhD. Department of Mechanical & Aerospace Engineering

E 490 Fundamentals of Engineering Review Fluid Mechanics By M. A. Boles, PhD Department of Mechanical & Aerospace Engineering North Carolina State University Archimedes Principle and Buoyancy 1. A block

Chapter 13 Fluids. Copyright 2009 Pearson Education, Inc.

Chapter 13 Fluids 13-1 Phases of Matter The three common phases of matter are solid, liquid, and gas. A solid has a definite shape and size. A liquid has a fixed volume but can be any shape. A gas can

Figure 1.6 shows a column of mercury held in place by a sealing plate. Recall from physics that at the bottom of the column of mercury

Lecture no. 7 1.6 PRESSURE Pressures, like temperatures, can be expressed by either absolute or relative scales. Pressure is defined as "normal force per unit area." Figure 1.5 shows a cylinder of water.

Aeration Air & Digester Gas Flow Metering Using Thermal Mass Technology. HWEA 2011 Conference Craig S. Johnson

Aeration Air & Digester Gas Flow Metering Using Thermal Mass Technology HWEA 2011 Conference Craig S. Johnson Presentation Overview Introduction Aeration Air & Digester gas challenges Gas flow metering

Air Conditioning Clinic

Air Conditioning Clinic Air Conditioning Fans One of the Equipment Series February 2012 TRG-TRC013-EN Air Conditioning Fans One of the Equipment Series A publication of Trane Preface Air Conditioning

Air Flow Measurements

ME-EM 30 ENERGY LABORATORY Air Flow Measurements Pitot Static Tube A slender tube aligned with the flow can measure local velocity by means of pressure differences. It has sidewall holes to measure the

Chapter 12 Industrial Ventilation

Chapter 1 Industrial Ventilation Introduction Industrial Ventilation is an important method for reducing employee exposures to airborne contaminants. Ventilation is used to reduce or remove contaminates

Applied Fluid Mechanics

Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and

oil liquid water water liquid Answer, Key Homework 2 David McIntyre 1

Answer, Key Homework 2 David McIntyre 1 This print-out should have 14 questions, check that it is complete. Multiple-choice questions may continue on the next column or page: find all choices before making

Experimental Evaluation of the Discharge Coefficient of a Centre-Pivot Roof Window

Experimental Evaluation of the Discharge Coefficient of a Centre-Pivot Roof Window Ahsan Iqbal #1, Alireza Afshari #2, Per Heiselberg *3, Anders Høj **4 # Energy and Environment, Danish Building Research

Gas Thermometer and Absolute Zero

Chapter 1 Gas Thermometer and Absolute Zero Name: Lab Partner: Section: 1.1 Purpose Construct a temperature scale and determine absolute zero temperature (the temperature at which molecular motion ceases).

These slides contain some notes, thoughts about what to study, and some practice problems. The answers to the problems are given in the last slide.

Fluid Mechanics FE Review Carrie (CJ) McClelland, P.E. cmcclell@mines.edu Fluid Mechanics FE Review These slides contain some notes, thoughts about what to study, and some practice problems. The answers

Applied Fluid Mechanics

Applied Fluid Mechanics Sixth Edition Robert L. Mott University of Dayton PEARSON Prentkv Pearson Education International CHAPTER 1 THE NATURE OF FLUIDS AND THE STUDY OF FLUID MECHANICS 1.1 The Big Picture

PIEZOMETER AIRFLOW MEASURING RING

PIEZOMETER AIRFLOW MEASURING RING M.K. Plastics is pleased to introduce The Piezometer Airflow Measuring Ring, now available as an accessory for our steel Axijet fans. The system consists of a Piezometer

Fluid Flow Measurement Selection and Sizing (ENGINEERING DESIGN GUIDELINE)

KLM Technology Group Sdn. Bhd. Guidelines for Processing Plant Page : 1 of 80 (ENGINEERING DESIGN GUIDELINE) Author: A L Ling Checked by: Karl Kolmetz TABLE OF CONTENT INTRODUCTION 5 Scope 5 Flow Measurement

TERMINAL UNIT AIRFLOW MEASUREMENTS

Page: 1 of 6 TERMINAL UNIT AIRFLOW MEASUREMENTS Dan Int-Hout Chief Engineer Page: 2 of 6 OVERVIEW The control of airflow rate in VAV systems is important for several reasons, including acoustics, ventilation,

PITOT-STATIC TUBE OR PITOT TUBE FOR MEASURING FLOW RATES?

XVII IMEKO World Congress Metrology in the 3rd Millennium June 22 27, 2003, Dubrovnik, Croatia PITOT-STATIC TUBE OR PITOT TUBE FOR MEASURING FLOW RATES? Eddy Jacques, Pierre Wauters Department of Mechanical

FLUID FLOW Introduction General Description

FLUID FLOW Introduction Fluid flow is an important part of many processes, including transporting materials from one point to another, mixing of materials, and chemical reactions. In this experiment, you

Airflow through Mine Openings and Ducts Chapter 5

Airflow through Mine Openings and Ducts Chapter 5 Fundamentals of Airflow Ventilation the application of the principles of fluid mechanics & thermodynamics to the flow of air in underground openings Fluid

Effects of Atmospheric Pressure on Gas Measurement

Effects of Atmospheric Pressure on Gas Measurement Class # 1390.1 March 2012 / White paper by Denis Rutherford Regional Sales Manager Central US Schneider Electric Telemetry & Remote SCADA Solutions 6650

Fluid Flow Instrumentation

Fluid Flow Instrumentation In the physical world, mechanical engineers are frequently required to monitor or control the flow of various fluids through pipes, ducts and assorted vessels. This fluid can

Why Study Fluids? Solids and How They Respond to Forces. Solids and How They Respond to Forces. Crystal lattice structure:

States of Matter Gas In a gas, the molecules are far apart and the forces between them are very small Solid In a solid, the molecules are very close together, and the form of the solid depends on the details

FLUID MECHANICS. Problem 2: Consider a water at 20 0 C flows between two parallel fixed plates.

FLUID MECHANICS Problem 1: Pressures are sometimes determined by measuring the height of a column of liquid in a vertical tube. What diameter of clean glass tubing is required so that the rise of water

14-1. Fluids in Motion There are two types of fluid motion called laminar flow and turbulent flow.

Fluid Dynamics Sections Covered in the Text: Chapter 15, except 15.6 To complete our study of fluids we now examine fluids in motion. For the most part the study of fluids in motion was put into an organized

LAB #2: Forces in Fluids

Princeton University Physics 103/105 Lab Physics Department LAB #2: Forces in Fluids Please do NOT attempt to wash the graduated cylinders once they have oil in them. Don t pour the oil in the graduated

EGN 3353C Fluid Mechanics

Static equilibrium of above element leads to Δ P= P P1 = ρg Δ z (*) γ s specific weight ΔP ρ (and is usually negligible in gases for small changes in depth) If point 1 is taken at the free surface open

INTERNATIONAL FIRE TRAINING CENTRE

INTERNATIONAL FIRE TRAINING CENTRE FIREFIGHTER INITIAL HYDRAULICS Throughout this note he means he/she and his means his/hers. Areas In bold type are considered to be of prime importance. INTRODUCTION

UNIVERSITY OF MINNESOTA DULUTH DEPARTMENT OF CHEMICAL ENGINEERING ChE MEASUREMENTS WITH FLOW METERS

UNIVERSITY OF MINNESOTA DULUTH DEPARTMENT OF CHEMICAL ENGINEERING ChE 3211-4211 MEASUREMENTS WITH FLOW METERS OBJECTIVE The purpose of this experiment is to calculate the coefficient of discharge from

A LAMINAR FLOW ELEMENT WITH A LINEAR PRESSURE DROP VERSUS VOLUMETRIC FLOW. 1998 ASME Fluids Engineering Division Summer Meeting

TELEDYNE HASTINGS TECHNICAL PAPERS INSTRUMENTS A LAMINAR FLOW ELEMENT WITH A LINEAR PRESSURE DROP VERSUS VOLUMETRIC FLOW Proceedings of FEDSM 98: June -5, 998, Washington, DC FEDSM98 49 ABSTRACT The pressure

Industrial Instrumentation Prof. A. Barua Department of Electrical Engineering Indian Institute of Technology Kharagpur

Industrial Instrumentation Prof. A. Barua Department of Electrical Engineering Indian Institute of Technology Kharagpur Lecture - 11 Capacitance Transducers This is lesson 11 of Industrial Instrumentation.

Centrifugal Fans and Pumps are sized to meet the maximum

Fans and Pumps are sized to meet the maximum flow rate required by the system. System conditions frequently require reducing the flow rate. Throttling and bypass devices dampers and valves are installed

Experiment (13): Flow channel

Introduction: An open channel is a duct in which the liquid flows with a free surface exposed to atmospheric pressure. Along the length of the duct, the pressure at the surface is therefore constant and

Laboratory Experience with a Model Jet Turbine

Session 2166 Laboratory Experience with a Model Jet Turbine John E. Matsson Oral Roberts University Abstract This paper describes the experience gained from the operation of a JetCat model turbojet engine

Stack Sampling Stack sampling or source sampling

Stack Monitoring Stack Sampling Stack sampling or source sampling may be defined as a method of collecting representative samples of pollutant laden air/gases at the place of origin of pollutants to determine

FLUID MECHANICS. TUTORIAL No.7 FLUID FORCES. When you have completed this tutorial you should be able to. Solve forces due to pressure difference.

FLUID MECHANICS TUTORIAL No.7 FLUID FORCES When you have completed this tutorial you should be able to Solve forces due to pressure difference. Solve problems due to momentum changes. Solve problems involving

AMD-23-TD. Thermal Dispersion Air Measuring Station with VCD-23 Control Damper. Application and Design. Ratings. Features and Control Options.

Application and esign Thermal ispersion Air Measuring Station with VC-23 Control amper The combines the functionality of a highly accurate thermal dispersion airflow measuring station and a low leakage

Practice Problems on Bernoulli s Equation. V car. Answer(s): p p p V. C. Wassgren, Purdue University Page 1 of 17 Last Updated: 2010 Sep 15

bernoulli_0 A person holds their hand out of a car window while driving through still air at a speed of V car. What is the maximum pressure on the person s hand? V car 0 max car p p p V C. Wassgren, Purdue

VOLU-probe/SS. Stainless Steel Pitot Airflow Traverse Probes. Accurate airflow measurement for demanding applications

VOLU-probe/SS Stainless Steel Pitot Airflow Traverse Probes Accurate airflow measurement for demanding applications VOLU-probe/SS The VOLU-probe/SS Stainless Steel Pitot Airflow Traverse Probe is ideally

EVALUATION OF EXHAUST VENTILATION SYSTEMS

HSCI 348 Industrical Hygiene Instrumentation Techniques Laboratory No. 5 EVALUATION OF EXHAUST VENTILATION SYSTEMS INTRODUCTION: Of major concern to the Industrial Hygienist is the evaluation and control

Gases and Pressure SECTION 1. Main Ideas

SECTION 1 Gases and Pressure Key Terms pressure millimeters of mercury partial pressure newton atmosphere of pressure Dalton s law of partial pressures barometer pascal In the chapter States of Matter,

Min-218 Fundamentals of Fluid Flow

Excerpt from "Chap 3: Principles of Airflow," Practical Mine Ventilation Engineerg to be Pubished by Intertec Micromedia Publishing Company, Chicago, IL in March 1999. 1. Definition of A Fluid A fluid

USER GUIDE. 5 in 1 Environmental Meter with CFM/CMM. Model 45170CM

USER GUIDE 5 in 1 Environmental Meter with CFM/CMM Model 45170CM Introduction Thank you for choosing the Extech Model 45170CM, 5 in 1 Environmental Meter. The 45170CM measures Air Flow, Air Velocity, Relative

Minor losses include head losses through/past hydrants, couplers, valves,

Lecture 10 Minor Losses & Pressure Requirements I. Minor Losses Minor (or fitting, or local ) hydraulic losses along pipes can often be estimated as a function of the velocity head of the water within

Technical Publication

Technical Publication Flow Meter Selection for Improved Gas Flow Measurements: Comparison of Differential Pressure and Thermal Dispersion Technologies Art Womack, Sr. Applications Engineer Fluid Components

AKRON EDUCTORS TROUBLESHOOTING GUIDE OPERATION & THEORY OF EDUCTORS GENERAL OPERATING AND MAINTENANCE INSTRUCTIONS

AKRON EDUCTORS TROUBLESHOOTING GUIDE OPERATION & THEORY OF EDUCTORS GENERAL OPERATING AND MAINTENANCE INSTRUCTIONS Products Include: 60 gpm eductors Style 3060, 3061, 3062 & 3070 95 gpm eductors Style

THE THERMAL FLOW METER, A GAS METER FOR ENERGY MEASUREMENT

THE THERMAL FLOW METER, A GAS METER FOR ENERGY MEASUREMENT Kazuto Otakane, Tokyo Gas Co., Ltd Katsuhito Sakai, Tokyo Gas Co., Ltd Minoru Seto, Tokyo Gas Co., Ltd 1. INTRODUCTION Tokyo Gas s new gas meter,

ME 305 Fluid Mechanics I. Part 4 Integral Formulation of Fluid Flow

ME 305 Fluid Mechanics I Part 4 Integral Formulation of Fluid Flow These presentations are prepared by Dr. Cüneyt Sert Mechanical Engineering Department Middle East Technical University Ankara, Turkey

AOE 3104 Aircraft Performance Problem Sheet 2 (ans) Find the Pressure ratio in a constant temperature atmosphere:

AOE 3104 Aircraft Performance Problem Sheet 2 (ans) 6. The atmosphere of Jupiter is essentially made up of hydrogen, H 2. For Hydrogen, the specific gas constant is 4157 Joules/(kg)(K). The acceleration

UNIK 5600/5700 Marine Certified Pressure Sensing Platform

GE Measurement & Control UNIK 5600/5700 Marine Certified Pressure Sensing Platform The new UNIK 5600/5700 carries marine certification for most zones on-board ship, as well as Intrinsically Safe certifications.

OPERATING INSTRUCTIONS

OPERATING INSTRUCTIONS Use Vane anemometers are used for the velocity measurement of directed air flows being free from eddies and turbulences. Because of the small, sturdy construction the instrument

> Introduction < upper side of table: g) Volume flow rate (V) in m 3 /h and in m 3 /s

1.1 TESTING METHODS AND STANDARDS The characteristic curves of Mistral fans published in catalogues of each fan series, have been established by testing methods and equipment approved by the UNI 7179-73P

du u U 0 U dy y b 0 b

BASIC CONCEPTS/DEFINITIONS OF FLUID MECHANICS (by Marios M. Fyrillas) 1. Density (πυκνότητα) Symbol: 3 Units of measure: kg / m Equation: m ( m mass, V volume) V. Pressure (πίεση) Alternative definition:

Laboratory 11. Fluid Statics, Pressure, and Buoyant Forces

Laboratory 11 Fluid Statics, Pressure, and Buoyant Forces I. Introduction In this exercise we will investigate some properties of non-moving fluids and the forces they exert on objects immersed in them.

Pressure. Pressure is one of those words we frequently use, perhaps knowing intuitively what it means. In science, we define pressure as follows:

Weather reports in the media provide information on variables such as temperature, precipitation and wind speed. In this chapter, we discuss three physical quantities that help determine weather: (1) Temperature,

Fluent Software Training TRN Boundary Conditions. Fluent Inc. 2/20/01

Boundary Conditions C1 Overview Inlet and Outlet Boundaries Velocity Outline Profiles Turbulence Parameters Pressure Boundaries and others... Wall, Symmetry, Periodic and Axis Boundaries Internal Cell

12. Pressure sensors

12. Pressure s Questions Task a) What is a piezo-resistive effect? b) What is the difference between absolute, gauge and differential pressure? c) What is the relation between output voltage (e.g. U X