# FLOW MEASUREMENT 2001 INTERNATIONAL CONFERENCE DERIVATION OF AN EXPANSIBILITY FACTOR FOR THE V-CONE METER

Save this PDF as:

Size: px
Start display at page:

Download "FLOW MEASUREMENT 2001 INTERNATIONAL CONFERENCE DERIVATION OF AN EXPANSIBILITY FACTOR FOR THE V-CONE METER"

## Transcription

1 FLOW MEASUREMENT 200 INTERNATIONAL CONFERENCE DERIVATION OF AN EXPANSIBILITY FACTOR FOR THE V-CONE METER Dr D G Stewart, NEL Dr M Reader-Harris, NEL Dr R J W Peters, McCrometer Inc INTRODUCTION The V-Cone meter is a differential pressure device that has been developed and tested since 985 and is now well understood and accepted as a viable flow measurement device. It is widely used in many industrial applications to measure a variety of fluids over a wide range of temperatures and pressures. The choice of the V-Cone can be based on both the short upstream lengths required and the ease of operation over a wide turndown ratio relative to an orifice plate. Over the past 0 years it has been increasingly accepted in the Gas Industry as a reliable metering device. Due to the fact that gas is a compressible fluid it is necessary to apply an expansibility correction factor, (or the Y factor in the USA), resulting in the well known mass flow equation: q m Cd π 2 = d 2 pρ β () Kinghorn [] in his 986 paper on this subject stated, The expansibility coefficient,, compensates for the fact that changes in the pressure of the gas as it flows through the meter result in changes in its density. For nozzles and Venturi meters this can be computed on the assumption that the flow is adiabatic since it is constrained by the walls of the meter. For orifice plates, however, the three-dimensional expansion which takes place requires an empirical determination of values of the expansibility coefficient. The expansibility factor equations for Venturi meters/nozzles and orifice plates are given by ISO 567-:99 as: κτ = κ 2 κ β τ ( κ ) 2 κ 2 κ β τ τ ( β ) p κp (Venturi tubes) (2) = (orifice plates) (3) There is currently a draft revision of ISO 567 containing a revision of Eq. (3) by Reader-Harris [2]. The flow through the V-Cone was initially assumed to be adiabatic, similar to the Venturi meter. However, in 99, Dahlstrom [3] presented the following empirical equation for the V-Cone expansibility factor based on only two V-Cone meters: = ( β ) p κp ()

2 Consequently, it was decided that a more detailed examination to determine the expansibility factor for the standard V-Cone should be undertaken. NEL were chosen to carry out this work and to derive the applicable equation. This paper details the experimental process, the results obtained, and the final derived expansibility equation for the standard V-Cone. 2 OBJECTIVES The key objectives of these tests were as follows:. To determine the expansibility factor equation for the standard V-Cone meter. 2. To investigate any dependence of the expansibility factor on pipe diameter. 3. To investigate any dependence of the expansibility factor on Reynolds number. 3 EXPERIMENTAL METHOD To determine the expansibility factor it is necessary to carry out tests at constant mass flowrate (i.e. constant Reynolds number) whilst varying the static pressure and differential pressure at the meter. In order to achieve this constant flowrate a sonic nozzle was used as the reference meter. The static pressure and differential pressure at the V-Cone were varied using a valve downstream of the V-Cone. To achieve the objectives stated above, the following tests were carried out: Three 6 V-Cones (β = 0.75, 0.55, and 0.5) were tested at kg/s, giving an approximate pipe Reynolds number, Re D 0.5 * 0 6. These are test numbers, 2, and 3. Next, one of these 6 V-Cones (β = 0.5) was tested at a second flowrate of 0.5 kg/s, giving a pipe Reynolds number, Re D 0.25 * 0 6. This is test number. One V-Cone (β = 0.55) was tested at flowrates of 0.33 kg/s and 0.66 kg/s, to give the same approximate pipe Reynolds number as for the 6 tests. These are test numbers 5 and 6. On completion of these tests it was felt that there was insufficient data from which to derive an equation for expansibility in a V-Cone, particularly at high beta values, with only one data set with β > Subsequently, it was decided to carry out the following additional tests: Another V-Cone (β = 0.65) was tested at flowrates of 0.33 kg/s and 0.66 kg/s, to give the same approximate pipe Reynolds numbers as for the 6 tests. These are test numbers 7 and 8. A 3 V-Cone (β = 0.75) was tested at a flowrate of 0.5 kg/s, again to give the same approximate pipe Reynolds number as for the 6 tests. This is test number 9. Details of the V-Cones and tests are summarised in Table. Table - Details of V-Cones and Tests Test No. Meter serial number Size (in) β (-) Flowrate (kg/s)

3 The V-Cones were installed one at a time in the NEL High Pressure Gravimetric Rig in the Gas Flow Lab with 3, or 6 pipework upstream and downstream of the V-Cone as appropriate to the meter size. The reference flowrate was measured using a sonic nozzle upstream of the V- Cone. The temperature and pressure were measured upstream of the sonic nozzle. The static pressure upstream of the V-Cone, and the differential pressure across it were measured, as were the temperature downstream of the V-Cone and the barometric pressure. A control valve was located some distance downstream of the V-Cone to enable the static/differential pressure at the V-Cone to be varied. DISCUSSION OF RESULTS In order to discuss the results in a satisfactory manner it is first necessary to describe the method used for determining the expansibility factor from the data collected.. Method of Determining Expansibility Factor From the flow equation for a differential pressure device, Eq. (), it can be seen that there are two unknowns from our test results, the discharge coefficient, C d, and the expansibility factor,. Eq. () can be rearranged to give: C = d qm π d 2 β 2 pρ (5) The functional form of the equation for V-Cone expansibility factor must be decided before proceeding further with the analysis. If Eq. (2) is expanded for small p/p and small β, then it, Eq. (3), and the revised orifice plate expansibility factor equation in ISO/DIS [2] are all of the form: = ( a + bβ ) p κp (6) Accordingly, Figures A. to A.9 (in Appendix A) show the experimental values of C d plotted against p κ p. The resultant plot has a reasonably linear form, as can be seen in all the graphs. It can be seen from Eq. (6) that as p κ p tends to zero, the expansibility factor tends to unity. Therefore, if a best fit linear curve is calculated through the data in the form y = mx + c, then the intercept c can be taken as the discharge coefficient C d as will be unity. From the slope m, the term in brackets in Eq. (6) can be determined. In order to evaluate this term in brackets, i.e. constant and dependence on β, there needs to be data over a wide enough range of β values. After the additional two meters were tested, data was available for β values of 0.5, 0.55, 0.65, and This is sufficient to be able to develop a reliable equation for the expansibility factor. 3

4 .2 Analysis of Results Using the method described above, the resulting values for m and c are given in Table 2 below. Table 2. Linear fits through data from tests to 9. Test No. Size (inch) β Flow (kg/s) m c The values of m and c give the linear fit through the C d against p/κp data for each test. The parameter c is the measured value of C d, obtained by extrapolating the linear fit to the y-axis, where is unity. Dividing the m values by this C d value gives us the slope of the equation for the expansibility factor,. These slopes for each test are given below in Table 3: Table 3. Slopes of derived expansibility equation for each test. Test No. Size (inch) β Flow (kg/s) slope The final expansibility equation must be derived from the results in Table 3 above. To achieve this, the slopes of the data from each test are shown below in Figure plotted against β.

5 y = x R 2 = β (-) Figure. Variation of expansibility slope with β. Altering the power of β yielded no significant improvement in the fit of the slopes from each test. By taking the best-fit line through the slopes from each test, the overall expansibility equation can be determined. This best-fit line is shown in Figure. Therefore the resulting equation for the expansibility factor in a V-Cone meter is: = ( β ) p κ p (7).3 Effect of Reynolds Number and Pipe Diameter As described in Section 2, two key objectives of the tests were to show that the expansibility factor was not significantly affected by pipe diameter or by Reynolds number. It can be seen from Figure that this is the case. The two values for β = 0.75 (β = 0.32) are in good agreement, and come from a 6 V-Cone at kg/s and a 3 V-Cone at 0.5 kg/s, both these tests giving a Reynolds number of approximately 0.5 x 0 6. Tests 2 and 5 were undertaken at the same Reynolds number for β = 0.55 (β = 0.09) but with 6 and V-Cone respectively, and the trend as to which pipe diameter gives the highest value for the slope is the opposite to that for β = Equally there are three cases of the same meter having been tested at two different flowrates, and hence Reynolds numbers. Tests 3/, 5/6, and 7/8 show reasonable agreement within their respective pairs and there is no trend as to which Reynolds number gives the higher value for the slope. This confirms that the expansibility factor is not affected by either the pipe diameter or the Reynolds number.. Comparison with Dahlstrom s Equation As stated in the introduction, Dahlstrom [3] presented Eq. () as the expansibility factor for the standard V-Cone meter. Dahlstrom s equation was derived from only five points from two meters. Figures A.0 to A.8 show the test results from this work along with the derived equation for expansibility, Eq. (6), compared with Dahlstrom s equation, Eq. (). The orifice 5

6 plate and Venturi tube equations are also shown on the graph as a guide to how the V-Cone expansibility compares with that from both these devices. It can be seen that Dahlstrom s equation is reasonably close to the new equation, Eq. (6), for many of the tests. However, the number of points and different meters used in the derivation of the new equation give rise to increased confidence in it. It is also evident that the V-Cone expansibility is somewhat different from that of the Venturi tube, and indeed lies between the orifice plate and Venturi tube equations, slightly closer to the Venturi. 5 CONCLUSIONS Six V-Cone flowmeters were tested in air at NEL s Flow Centre in order to determine the expansibility factor of a gas flowing through a V-Cone. By testing each meter at a nominally constant flowrate and by varying the static pressure and differential pressure at the meter, it was possible to obtain the necessary data to determine the recommended equation for the expansibility factor, Eq. (7). In addition, no significant effect on the expansibility factor could be attributed to a change in Reynolds number or pipe diameter. 6 LIST OF NOMENCLATURE a, b Coefficients in Eq. (5) β Effective diameter ratio [-] C d Discharge coefficient [-] Expansibility factor [-] d (Throat) diameter [m] κ Isentropic exponent [-] p Static pressure [Pa] ρ Density [kg/m 3 ] p p = p p Differential pressure [Pa] τ Pressure ratio, τ [-] q m Mass flowrate [kg/s] Re D Pipe reynolds number, Re D m π dµ = [-] 7 REFERENCES [] F.C. Kinghorn. The expansibility correction for orifice plates: EEC data. In Proc. Int. Conf. on Flow Measurment in the Mid 80 s (Paper 5.2). National Engineering Laboratory East Kilbride, Glasgow, June 986. [2] International Standards Organisation. Measurement of fluid flow by means of pressure differential devices inserted in circular cross-section conduits running full, Part 2: Orifice plates. Geneva: International Standards Organisation. ISO/DIS 567-2, May [3] M.J. Dahlstrom. V-Cone meter: Gas measurement for the real world. North Sea Flow Measurement Workshop, Peebles, Scotland, 99. 6

7 APPENDIX A - DIAGRAMS OF RESULTS Fig. A. - Results from test. Test No. 6 inch β = 0.75 kg/s y = x + 55 R 2 = p / κp Fig. A.2 - Results from test 2. Test No. 2 6 inch β = 0.55 kg/s y = x R 2 = p / κp 7

8 Fig. A.3 - Results from test 3. Test No. 3 6 inch β = 0.5 kg/s y = x R 2 = p / κp Fig. A. - Results from test. Test No. 6 inch β = kg/s y = x R 2 = p / κp 8

9 Fig. A.5 - Results from test 5. Test No. 5 inch β = kg/s y = x R 2 = p / κp Fig. A.6 - Results from test 6. Test No. 6 inch β = kg/s y = x R 2 = p / κp 9

10 Fig. A.7 - Results from test 7. Test No. 7 inch β = kg/s y = x R 2 = p / κp Fig. A.8 - Results from test 8. Test No. 8 inch β = kg/s y = x + 32 R 2 = p / κp 0

11 Fig. A.9 - Results from test 9. Test No. 9 3 inch β = kg/s y = x + 85 R 2 = p / κp Fig. A.0 - Derived expansibility factor values from test compared with Eq. (7) along with Test data p / κp (-)

12 Fig. A. - Derived expansibility factor values from test 2 compared with Eq. (7) along with Test 2 data p / κp (-) Fig. A.2 - Derived expansibility factor values from test 3 compared with Eq. (7) along with Test 3 data p / κp (-) 2

13 Fig. A.3 - Derived expansibility factor values from test compared with Eq. (7) along with Test data p / κp (-) Fig. A. - Derived expansibility factor values from test 5 compared with Eq. (7) along with Test 5 data p / κp (-) 3

14 Fig. A.5 - Derived expansibility factor values from test 6 compared with Eq. (7) along with Test 6 data p / κp (-) Fig. A.6 - Derived expansibility factor values from test 7 compared with Eq. (7) along with Test 7 data p / κp (-)

15 Fig. A.7 - Derived expansibility factor values from test 8 compared with Eq. (7) along with Test 8 data p / κp (-) Fig. A.8 - Derived expansibility factor values from test 9 compared with Eq. (7) along with Test 9 data p / κp (-) 5

### New Method for Accurate High Reynolds Metering Uses Water Calibration for Significant Cost Savings White Paper

New Method for Accurate High Reynolds Metering Uses Water Calibration for Significant Cost Savings White Paper Page 1 of 3 White Paper Jonathan Hollist, V-Cone Engineer Nick Voss, V-Cone Product Manager

### Custody Transfer Measurement. with the V-Cone Flowmeter

Custody Transfer Measurement with the V-Cone Flowmeter Stephen A. Ifft McCrometer Inc. Hemet, California, USA Abstract This paper will discuss the approval of the McCrometer V-Cone flowmeter for custody

### Michael Montgomery Marketing Product Manager Rosemount Inc. Russ Evans Manager of Engineering and Design Rosemount Inc.

ASGMT / Averaging Pitot Tube Flow Measurement Michael Montgomery Marketing Product Manager Rosemount Inc. Russ Evans Manager of Engineering and Design Rosemount Inc. Averaging Pitot Tube Meters Introduction

### Advanced Differential Pressure Flowmeter Technology V-CONE FLOW METER TECHNICAL BRIEF

Advanced Differential Pressure Flowmeter Technology V-CONE FLOW METER TECHNICAL BRIEF Table of Contents Section 1 - General Introduction 1.1 1 Principles Of Operation 1.2 1 Reshaping The Velocity Profile

### FLUID FLOW Introduction General Description

FLUID FLOW Introduction Fluid flow is an important part of many processes, including transporting materials from one point to another, mixing of materials, and chemical reactions. In this experiment, you

### GOOD PRACTICE GUIDE AN INTRODUCTION TO DIFFERENTIAL-PRESSURE FLOW METERS

GOOD PRACTICE GUIDE AN INTRODUCTION TO DIFFERENTIAL-PRESSURE FLOW METERS www.tuvnel.com Introductory guide to differential-pressure flow meters This introductory guide to differential-pressure (Δp) flow

### Experiment 3 Pipe Friction

EML 316L Experiment 3 Pipe Friction Laboratory Manual Mechanical and Materials Engineering Department College of Engineering FLORIDA INTERNATIONAL UNIVERSITY Nomenclature Symbol Description Unit A cross-sectional

### Testing Protocol for Differential Pressure Measurement Devices API MPMS Chapter 22.2

Testing Protocol for Differential Pressure Measurement Devices API MPMS Chapter 22.2 Steve Baldwin Chevron Energy Technology Company Houston, Texas USA Casey Hodges CEESI Measurement Solutions. Nunn, Colorado

### Chapter 10. Flow Rate. Flow Rate. Flow Measurements. The velocity of the flow is described at any

Chapter 10 Flow Measurements Material from Theory and Design for Mechanical Measurements; Figliola, Third Edition Flow Rate Flow rate can be expressed in terms of volume flow rate (volume/time) or mass

### UNIT Engineering: Measurement Technology Flow (SCQF level 6)

National Unit Specification: general information CODE H0W6 12 SUMMARY This Unit can be delivered as part of a National Qualification Group Award but can also be taken as a free-standing Unit. This Unit

### Head Loss in Pipe Flow ME 123: Mechanical Engineering Laboratory II: Fluids

Head Loss in Pipe Flow ME 123: Mechanical Engineering Laboratory II: Fluids Dr. J. M. Meyers Dr. D. G. Fletcher Dr. Y. Dubief 1. Introduction Last lab you investigated flow loss in a pipe due to the roughness

### RITTER Multiple Sonic Nozzle Calibration System

RITTER Multiple Sonic Nozzle Calibration System »Wouldn t it be great to eliminate doubtful measurement results by using proven measurement technology? Providing the most precise results ensures and increases

### Theory overview of flow measurement using differential pressure devices based on ISO-5167 standard.

Theory overview of flow measurement using differential pressure devices based on ISO-567 standard. rian FL40 flow computer description. Flow Cad software users manual. Technical note, Differential pressure

### measurement, but almost any pipe elbow can be calibrated Elbow meters are not as potentially accurate as venturi,

Lecture 14 Flow Measurement in Pipes I. Elbow Meters An elbow in a pipe can be used as a flow measuring device much in the same way as a venturi or orifice plate The head differential across the elbow

### Effects of Beta Ratio and Reynold s Number on Coefficient of Discharge of Orifice Meter

J Agric Rural Dev 7(&2), 5-56, June 29 ISSN 8-86 K wl I cj x Dbœqb zj Available online at http://www.banglajol.info/index.php/jard JARD Journal of Agriculture & Rural Development Effects of Beta Ratio

### Grant Agreement No. 228296 SFERA. Solar Facilities for the European Research Area SEVENTH FRAMEWORK PROGRAMME. Capacities Specific Programme

Grant Agreement No. 228296 SFERA Solar Facilities for the European Research Area SEVENTH FRAMEWORK PROGRAMME Capacities Specific Programme Research Infrastructures Integrating Activity - Combination of

### Gas Flow Measurement Software A Report for

Gas Flow Measurement Software A Report for National Measurement System Directorate Department of Trade & Industry 151 Buckingham Palace Road London, SW1W 9SS Report No: 332/99 Date: September 1999 The

### Equations of Lines Derivations

Equations of Lines Derivations If you know how slope is defined mathematically, then deriving equations of lines is relatively simple. We will start off with the equation for slope, normally designated

### THEORETICAL UNCERTAINTY OF ORIFICE FLOW MEASUREMENT page 1 THEORETICAL UNCERTAINTY OF ORIFICE FLOW MEASUREMENT INTRODUCTION COEFFICIENT OF DISCHARGE

THEORETICAL UNCERTAINTY OF ORIFICE FLOW MEASUREMENT page 1 DANIEL MEASUREMENT AND CONTROL WHITE PAPERS THEORETICAL UNCERTAINTY OF ORIFICE FLOW MEASUREMENT www.daniel.com INTRODUCTION Orifice meters are

### CHME 302 CHEMICAL ENGINEERING LABOATORY-I EXPERIMENT 302-1 FLOW MEASUREMENT TEST

CHME 302 CHEMICAL ENGINEERING LABOATORY-I EXPERIMENT 302-1 FLOW MEASUREMENT TEST OBJECTIVE The purpose of the experiment is to demonstrate the working principles of four most common devices used to measure

### Calculation of flow rate from differential pressure devices orifice plates

Calculation of flow rate from differential pressure devices orifice plates Peter Lau EMATEM - Sommerschule Kloster Seeon August -, 008 Contents Introduction - Why using dp-devices when other principles

### is the stagnation (or total) pressure, constant along a streamline.

70 Incompressible flow (page 60): Bernoulli s equation (steady, inviscid, incompressible): p 0 is the stagnation (or total) pressure, constant along a streamline. Pressure tapping in a wall parallel to

### Experiment # 3: Pipe Flow

ME 05 Mechanical Engineering Lab Page ME 05 Mechanical Engineering Laboratory Spring Quarter 00 Experiment # 3: Pipe Flow Objectives: a) Calibrate a pressure transducer and two different flowmeters (paddlewheel

### The Unique Accelabar Flow Meter

The Unique Accelabar Flow Meter The Accelabar is a new and unique flow meter that combines two differential pressure technologies to produce operating ranges never before attainable in a single flow meter.

### 191: Calibration and Standards in Flow Measurement. Richard Paton National Engineering Laboratory, Scotland, UK 1 GENERAL PRINCIPLES

191: Calibration and Standards in Measurement Richard Paton National Engineering Laboratory, Scotland, UK 1 General Principles 1 2 Gravimetric Calibration of Liquid meters 2 3 Volumetric Calibration of

### CHE 253M Experiment No. 3 LIQUID FLOW MEASUREMENT

Rev 8/15 AW/GW CHE 253M Experiment No. 3 LIQUID FLOW MEASUREMENT The objective of this experiment is to familiarize the student with several types of flow-measuring devices commonly used in the laboratory

### V-Cone, An Alternative to Orifice Meter in Wet Gas Applications

V-Cone, An Alternative to Orifice Meter in Wet Gas Applications Stephen A. Ifft McCrometer Inc. Hemet, California, USA Abstract This paper will discuss the use of the V-Cone differential pressure flowmeter

### Flow Measurement in Pipes and Ducts. Flow Measurement in Pipes and Ducts, Course #503. Presented by:

Flow Measurement in Pipes and Ducts, Course #503 Presented by: PDH Enterprises, LLC PO Box 942 Morrisville, NC 27560 www.pdhsite.com This course is about measurement of the flow rate of a fluid flowing

### Technical Publication

Technical Publication Flow Meter Selection for Improved Gas Flow Measurements: Comparison of Differential Pressure and Thermal Dispersion Technologies Art Womack, Sr. Applications Engineer Fluid Components

### AP Physics 1. Calculating the value of Pi Example 2015 2016 1 2

AP Physics 1 Kevin J. Kukla 201 2016 1 AP Physics 1 Lab Journal Guidelines Calculating the value of Pi Example 201 2016 1 2 Lab Journal Guidelines (I) Purpose of Lab Lab Question: The purpose of this lab

### The Use of Control Valve Sizing Equations with Simulation Based Process Data

The Use of Control Valve Sizing Equations with Simulation Based Process Data Marc L. Riveland Director, Advanced Technologies Fisher Valves, Emerson Process Management Keywords: Control Valve, Thermodynamics,

### BLOWERS EXHAUSTERS CONTINENTAL INDUSTRIE ASME PTC 10 PERFORMANCE TEST PROCEDURE. Blowers & Exhausters. Page 1 of 6 PURPOSE STANDARDS INSTRUMENTS

PURPOSE Page 1 of 6 Test purpose is the measure of the following compressor operating parameters: flow of gas handled pressure rice produced shaft power requirement efficiency surge limit STANDARDS Test

### Correcting a Coriolis Meter for Two Phase Oil & Gas Flow

Correcting a Coriolis Meter for Two Phase Oil & Gas Flow Chris Mills, NEL 1 INTRODUCTION In the oil & gas industry, the requirement for flow measurement of two phase oil and gas flow is prevalent. This

### Rosemount Orifice Plate in Wet Gas

00870-0100-4810 January 2006 Page 1 Rosemount Orifice Plate in Wet Gas This paper discusses the phenomenon of wet gas measurement, definitions, and the need to accurately measure this mixture. This paper

### Applied Fluid Mechanics

Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 7. General Energy Equation

### Experimental Evaluation of the Discharge Coefficient of a Centre-Pivot Roof Window

Experimental Evaluation of the Discharge Coefficient of a Centre-Pivot Roof Window Ahsan Iqbal #1, Alireza Afshari #2, Per Heiselberg *3, Anders Høj **4 # Energy and Environment, Danish Building Research

### ME 239: Rocket Propulsion. Nozzle Thermodynamics and Isentropic Flow Relations. J. M. Meyers, PhD

ME 39: Rocket Propulsion Nozzle Thermodynamics and Isentropic Flow Relations J. M. Meyers, PhD 1 Assumptions for this Analysis 1. Steady, one-dimensional flow No motor start/stopping issues to be concerned

### WHITE PAPER. A New Approach to Accurate Water Flow Measurement - The Study of the FPI Mag Accuracy

WHITE PAPER A New Approach to Accurate Water Flow Measurement - The Study of the FPI Mag Accuracy A New Approach to Accurate Water Flow Measurement - The Study of the FPI Mag Accuracy Author: Nicholas

### Pressure drop in pipe lines for compressed air: comparison between experimental and theoretical analysis M. Carello, A. Ivanov, L.

Pressure drop in pipe lines for compressed air: comparison between experimental and theoretical analysis M. Carello, A. Ivanov, L. Mazza /,70729 Tbrmo, Abstract In this paper a test methodology to determine

### Vortex Shedding Meters Class # Curtis Gulaga Business Development Manager CB Engineering Ltd. #20, Street S.E. Calgary, Alberta, Canada

Vortex Shedding Meters Class # 8150 Curtis Gulaga Business Development Manager CB Engineering Ltd. #20, 5920 11 Street S.E. Calgary, Alberta, Canada 1.0 Introduction Vortex meters have proven to be repeatable,

### Fluid Mechanics Prof. T. I. Eldho Department of Civil Engineering Indian Institute of Technology, Bombay. Lecture No. # 36 Pipe Flow Systems

Fluid Mechanics Prof. T. I. Eldho Department of Civil Engineering Indian Institute of Technology, Bombay Lecture No. # 36 Pipe Flow Systems Welcome back to the video course on Fluid Mechanics. In today

### Flow Disturbance Cone Meter Testing

1. Introduction North Sea Flow Measurement Workshop Flow Disturbance Cone Meter Testing Gordon Stobie, GS Flow Ltd., formerly ConocoPhillips Company Richard Steven, CEESI Kim Lewis, DP Diagnostics Ltd

### Swissmetro travels at high speeds through a tunnel at low pressure. It will therefore undergo friction that can be due to:

I. OBJECTIVE OF THE EXPERIMENT. Swissmetro travels at high speeds through a tunnel at low pressure. It will therefore undergo friction that can be due to: 1) Viscosity of gas (cf. "Viscosity of gas" experiment)

### Chapter 8: Flow in Pipes

Objectives 1. Have a deeper understanding of laminar and turbulent flow in pipes and the analysis of fully developed flow 2. Calculate the major and minor losses associated with pipe flow in piping networks

### Testing methods applicable to refrigeration components and systems

Testing methods applicable to refrigeration components and systems Sylvain Quoilin (1)*, Cristian Cuevas (2), Vladut Teodorese (1), Vincent Lemort (1), Jules Hannay (1) and Jean Lebrun (1) (1) University

### Applied Fluid Mechanics

Applied Fluid Mechanics Sixth Edition Robert L. Mott University of Dayton PEARSON Prentkv Pearson Education International CHAPTER 1 THE NATURE OF FLUIDS AND THE STUDY OF FLUID MECHANICS 1.1 The Big Picture

### Module 2 : Convection. Lecture 20a : Illustrative examples

Module 2 : Convection Lecture 20a : Illustrative examples Objectives In this class: Examples will be taken where the concepts discussed for heat transfer for tubular geometries in earlier classes will

### By Authority Of THE UNITED STATES OF AMERICA Legally Binding Document

By Authority Of THE UNITED STATES OF AMERICA Legally Binding Document By the Authority Vested By Part 5 of the United States Code 552(a) and Part 1 of the Code of Regulations 51 the attached document has

### A LAMINAR FLOW ELEMENT WITH A LINEAR PRESSURE DROP VERSUS VOLUMETRIC FLOW. 1998 ASME Fluids Engineering Division Summer Meeting

TELEDYNE HASTINGS TECHNICAL PAPERS INSTRUMENTS A LAMINAR FLOW ELEMENT WITH A LINEAR PRESSURE DROP VERSUS VOLUMETRIC FLOW Proceedings of FEDSM 98: June -5, 998, Washington, DC FEDSM98 49 ABSTRACT The pressure

### Establishment of the Gas Flow Calibration Facility at CSIR-NPL, India

Establishment of the Gas Flow Calibration Facility at CSIR-PL, India Speaker: Shiv Kumar Jaiswal Fluid Flow Measurement Standards CSIR-ational Physical Laboratory (PLI) Dr. K.S. Krishnan Road ew Delhi

### Applied Fluid Mechanics

Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and

### Transient Mass Transfer

Lecture T1 Transient Mass Transfer Up to now, we have considered either processes applied to closed systems or processes involving steady-state flows. In this lecture we turn our attention to transient

### CO 2 41.2 MPa (abs) 20 C

comp_02 A CO 2 cartridge is used to propel a small rocket cart. Compressed CO 2, stored at a pressure of 41.2 MPa (abs) and a temperature of 20 C, is expanded through a smoothly contoured converging nozzle

### Performance 4. Fluid Statics, Dynamics, and Airspeed Indicators

Performance 4. Fluid Statics, Dynamics, and Airspeed Indicators From our previous brief encounter with fluid mechanics we developed two equations: the one-dimensional continuity equation, and the differential

### Gas Flowmeter Calibrations with the Working Gas Flow Standard

Gas Flowmeter Calibrations with the Working Gas Flow Standard NIST Special Publication 50-80 John D. Wright, Jean-Philippe Kayl, Aaron N. Johnson, and Gina M. Kline November 3, 009 Fluid Metrology Group

### What is the "Best" Transfer Standard for Gas Flow?

What is the "Best" Transfer Standard for Gas Flow? Flow Systems 220 Bunyan Ave. Berthoud, CO 80513 USA Phone: 800-466-3569 Fax: 970-532-0748 www.flowsystemsinc.com WHAT IS THE BEST TRANSFER STANDARD FOR

### Fluid Mechanics: Static s Kinematics Dynamics Fluid

Fluid Mechanics: Fluid mechanics may be defined as that branch of engineering science that deals with the behavior of fluid under the condition of rest and motion Fluid mechanics may be divided into three

### Prof. U.S.P. Shet, Prof. T. Sundararajan and Prof. J.M. Mallikarjuna. 4.7 Comparison of Otto, Diesel and Dual Cycles:

.7 Comparison of Otto, Diesel and Dual Cycles: The important variable factors which are used as the basis for comparison of the cycles are compression ratio, peak pressure, heat addition, heat rejection

### Homework 9. Problems: 12.31, 12.32, 14.4, 14.21

Homework 9 Problems: 1.31, 1.3, 14.4, 14.1 Problem 1.31 Assume that if the shear stress exceeds about 4 10 N/m steel ruptures. Determine the shearing force necessary (a) to shear a steel bolt 1.00 cm in

### V-CONE METERING PROGRAM User Manual

Flow Computer Division V-CONE METERING PROGRAM User Manual Form A6104 March 2002 Revision Tracking Sheet March 2002 This manual may be revised from time to time to incorporate new or updated information.

### High Speed Aerodynamics Prof. K. P. Sinhamahapatra Department of Aerospace Engineering Indian Institute of Technology, Kharagpur

High Speed Aerodynamics Prof. K. P. Sinhamahapatra Department of Aerospace Engineering Indian Institute of Technology, Kharagpur Module No. # 01 Lecture No. # 06 One-dimensional Gas Dynamics (Contd.) We

### Chapter 28 Fluid Dynamics

Chapter 28 Fluid Dynamics 28.1 Ideal Fluids... 1 28.2 Velocity Vector Field... 1 28.3 Mass Continuity Equation... 3 28.4 Bernoulli s Principle... 4 28.5 Worked Examples: Bernoulli s Equation... 7 Example

### Flow Measurement. Reference - Flow Measurement FUNDAMENTALS. ρ = fluid density (lb/ft3 or kg/m3)

Flow Measurement FUDAMETALS Flowmetering terms can often seem cryptic Here are definitions of some of the most commonly used ACCURACY Accuracy is a quantity defining the limit that errs will not exceed

### FIGURE P8 50E FIGURE P8 62. Minor Losses

8 48 Glycerin at 40 C with r 1252 kg/m 3 and m 0.27 kg/m s is flowing through a 4-cm-diameter horizontal smooth pipe with an average velocity of 3.5 m/s. Determine the pressure drop per 10 m of the pipe.

### Air Flow Measurements

ME-EM 30 ENERGY LABORATORY Air Flow Measurements Pitot Static Tube A slender tube aligned with the flow can measure local velocity by means of pressure differences. It has sidewall holes to measure the

### CONTROL VALVE PRESSURE DROP AND SIZING

CONTENT Chapter Description Page I Purpose of Control Valve II Type and Main Components of Control Valve 3 III Power 5 IV. Pressure Drop Across Control Valve 7 V. Symbols and Units 10 VI. Unit Conversion

### Modeling Fluid Systems

Modeling Fluid Systems The prevalent use of fluid (hydraulic) circuitry in machines tool applications, aircraft control systems, and similar operations occurs because of such factors such as accuracy,

### Application of the Orifice Meter for Accurate Gas Flow Measurement page 1. Application of the Orifice Meter for Accurate Gas Flow Measurement.

Application of the Orifice Meter for Accurate Gas Flow Measurement page 1 DANIEL MEASUREMENT AND CONTROL WHITE PAPER Application of the Orifice Meter for Accurate Gas Flow Measurement www.daniel.com Summary

### Air Flow Optimization via a Venturi Type Air Restrictor

, July 3-5, 013, London, U.K. Air Flow Optimization via a Venturi Type Air Restrictor Anshul Singhal, Mallika Parveen, Member, IAENG Abstract The aim of this project is to create a flow restriction device

### pressure inside the valve just before the final release of air. in pipeline systems resulting from 2p 1 p a m C D A o

BY SRINIVASA LINGIREDDY, DON J. WOOD, AND NAFTALI ZLOZOWER It is a common practice to locate air valves at high elevations along water transmission mains. Improper sizing of an air valve could lead to

### Chapter 17. For the most part, we have limited our consideration so COMPRESSIBLE FLOW. Objectives

Chapter 17 COMPRESSIBLE FLOW For the most part, we have limited our consideration so far to flows for which density variations and thus compressibility effects are negligible. In this chapter we lift this

### API Standard Sizing, Selection, & Installation of Pressure Relieving Devices Part I, Sizing & Selection

3.3.3.1 520-I-01/03 Question 1: Is this a correct interpretation of Section 3.3.3.1.3 in that the backpressure was specified to not exceed the allowable overpressure? Question 2: If my interpretation of

### Fluid Mechanics Definitions

Definitions 9-1a1 Fluids Substances in either the liquid or gas phase Cannot support shear Density Mass per unit volume Specific Volume Specific Weight % " = lim g#m ( ' * = +g #V \$0& #V ) Specific Gravity

### Applied Fluid Mechanics

Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and

### What is the most obvious difference between pipe flow and open channel flow????????????? (in terms of flow conditions and energy situation)

OPEN CHANNEL FLOW 1 3 Question What is the most obvious difference between pipe flow and open channel flow????????????? (in terms of flow conditions and energy situation) Typical open channel shapes Figure

### The Cartesian Plane The Cartesian Plane. Performance Criteria 3. Pre-Test 5. Coordinates 7. Graphs of linear functions 9. The gradient of a line 13

6 The Cartesian Plane The Cartesian Plane Performance Criteria 3 Pre-Test 5 Coordinates 7 Graphs of linear functions 9 The gradient of a line 13 Linear equations 19 Empirical Data 24 Lines of best fit

### Technical Note PP 841-TN Pressure Capability of HDPE PE4710 and PVC Pipe

Technical Note PP 841-TN Pressure Capability of HDPE PE4710 and PVC Pipe Too frequently, the comparison between High Density Polyethylene (HDPE) and Polyvinyl Chloride (PVC) pipe is made primarily on pressure

### THE SPRING CONSTANT. Apparatus: A spiral spring, a set of weights, a weight hanger, a balance, a stop watch, and a twometer

THE SPRING CONSTANT Objective: To determine the spring constant of a spiral spring by Hooe s law and by its period of oscillatory motion in response to a weight. Apparatus: A spiral spring, a set of weights,

### Laminar and Turbulent flow. Flow Sensors. Reynolds Number. Thermal flow Sensor. Flow and Flow rate. R = Mass Flow controllers

Flow and Flow rate. Laminar and Turbulent flow Laminar flow: smooth, orderly and regular Mechanical sensors have inertia, which can integrate out small variations due to turbulence Turbulent flow: chaotic

### HEAVY OIL FLOW MEASUREMENT CHALLENGES

HEAVY OIL FLOW MEASUREMENT CHALLENGES 1 INTRODUCTION The vast majority of the world s remaining oil reserves are categorised as heavy / unconventional oils (high viscosity). Due to diminishing conventional

### Water hammering in fire fighting installation

Water hammering in fire fighting installation Forward One of major problems raised in the fire fighting network installed at Pioneer company for pharmaceutical industry /Sulaymania was the high water hammering

### 5.1 Writing Linear Equations in Slope-Intercept Form. 1. Use slope-intercept form to write an equation of a line.

5.1 Writing Linear Equations in Slope-Intercept Form Objectives 1. Use slope-intercept form to write an equation of a line. 2. Model a real-life situation with a linear function. Key Terms Slope-Intercept

### Flow Measurement Options for Pipeline and Open Channel Flow

Flow Measurement Options for Pipeline and Open Channel Flow October 2013 Presented by Molly Skorpik - 2013 Montana Association of Dam and Canal Systems Conference Irrigation Training and Research Center

### INTRODUCTION TO FLUID MECHANICS

INTRODUCTION TO FLUID MECHANICS SIXTH EDITION ROBERT W. FOX Purdue University ALAN T. MCDONALD Purdue University PHILIP J. PRITCHARD Manhattan College JOHN WILEY & SONS, INC. CONTENTS CHAPTER 1 INTRODUCTION

### Electrochemical Kinetics ( Ref. :Bard and Faulkner, Oldham and Myland, Liebhafsky and Cairns) R f = k f * C A (2) R b = k b * C B (3)

Electrochemical Kinetics ( Ref. :Bard and Faulkner, Oldham and Myland, Liebhafsky and Cairns) 1. Background Consider the reaction given below: A B (1) If k f and k b are the rate constants of the forward

### Experiment (13): Flow channel

Introduction: An open channel is a duct in which the liquid flows with a free surface exposed to atmospheric pressure. Along the length of the duct, the pressure at the surface is therefore constant and

### EFFECT OF REYNOLDS NUMBER IN CORIOLIS FLOW MEASUREMENT

EFFECT OF REYNOLDS NUMBER IN CORIOLIS FLOW MEASUREMENT Christof Huber, Michael Nuber and Martin Anklin Endress+Hauser Flowtec AG, Kägenstr. 7, 4153 Reinach, Switzerland Abstract In hydrocarbon measurement

### SIMPLIFIED METHOD FOR ESTIMATING THE FLIGHT PERFORMANCE OF A HOBBY ROCKET

SIMPLIFIED METHOD FOR ESTIMATING THE FLIGHT PERFORMANCE OF A HOBBY ROCKET WWW.NAKKA-ROCKETRY.NET February 007 Rev.1 March 007 1 Introduction As part of the design process for a hobby rocket, it is very

### (a) Calculate the voidage (volume fraction occupied by voids) of the bed.

SOLUTIONS TO CAPTER 6: FLOW TROUG A PACKED BED OF PARTICLES EXERCISE 6.1: A packed bed of solid particles of density 500 kg/m 3, occupies a depth of 1 m in a vessel of cross-sectional area 0.04 m. The

Desuperheater overview Steam used in process plants can be superheated, that is, heated to a temperature above saturation. The excess of temperature above its saturation is called 'superheat'. Desuperheated

### MATLAB AS A PROTOTYPING TOOL FOR HYDRONIC NETWORKS BALANCING

MATLAB AS A PROTOTYPING TOOL FOR HYDRONIC NETWORKS BALANCING J. Pekař, P. Trnka, V. Havlena* Abstract The objective of this note is to describe the prototyping stage of development of a system that is

### Appendix C: Graphs. Vern Lindberg

Vern Lindberg 1 Making Graphs A picture is worth a thousand words. Graphical presentation of data is a vital tool in the sciences and engineering. Good graphs convey a great deal of information and can

### GOOD PRACTICE GUIDE THE CALIBRATION OF FLOW METERS

GOOD PRACTICE GUIDE THE CALIBRATION OF FLOW METERS www.tuvnel.com The calibration of flow meters This good practice guide covers the general principles of calibration when applied to the flow meters or

### LECTURE 28 to 29 ACCUMULATORS FREQUENTLY ASKED QUESTIONS

LECTURE 28 to 29 ACCUMULATORS FREQUENTLY ASKED QUESTIONS 1. Define an accumulator and explain its function A hydraulic accumulator is a device that stores the potential energy of an incompressible fluid

### BASIC UNDERSTANDING OF FLOW CALCULATIONS AND ESTIMATES MAKES SIZING VALVES SIMPLER

BASIC UNDERSTANDING OF FLOW CALCULATIONS AND ESTIMATES MAKES SIZING VALVES SIMPLER Valve size often is described by the nominal size of the end connections but a more important measure is the flow that

### International Journal of Latest Research in Science and Technology Volume 4, Issue 2: Page No.161-166, March-April 2015

International Journal of Latest Research in Science and Technology Volume 4, Issue 2: Page No.161-166, March-April 2015 http://www.mnkjournals.com/ijlrst.htm ISSN (Online):2278-5299 EXPERIMENTAL STUDY

### Wet gas compressor capacity limits

Understanding Centrifugal Compressor Performance in a Connected Process System Scott Golden, Scott A. Fulton and Daryl W. Hanson Process Consulting Services Inc., Houston, Texas Wet gas compressor capacity

### Section 1.1 Linear Equations: Slope and Equations of Lines

Section. Linear Equations: Slope and Equations of Lines Slope The measure of the steepness of a line is called the slope of the line. It is the amount of change in y, the rise, divided by the amount of