Experiment # 3: Pipe Flow


 Annabel Carpenter
 2 years ago
 Views:
Transcription
1 ME 05 Mechanical Engineering Lab Page ME 05 Mechanical Engineering Laboratory Spring Quarter 00 Experiment # 3: Pipe Flow Objectives: a) Calibrate a pressure transducer and two different flowmeters (paddlewheel and orifice plate); b) Use the flowmeter and pressure transducer to measure the friction factor for pipes of different diameter, of different lengths, and for different flow rates. Check for Reynolds number scaling and compare with the Moody diagram; c) Measure minor losses in fittings and compare with empirical rules of thumb; d) Use a hydraulic analog of a Wheatstone bridge to test rules of thumb for minor losses. Introduction Volumetric flow rate, pressure, and head losses are key fundamental quantities in analyzing and designing piping systems. This experiment will introduce you to basic measurement techniques and to some principles of pipe flow. In this experiment three basic devices a pressure transducer, an orifice plate flowmeter and a paddlewheel flowmeter are calibrated and compared against standard practice, and then used to make fundamental measurements of losses in pipes, fittings, and piping networks. PreLab Reading Review relevant material from your undergraduate fluid mechanics courses, including (i) Reynolds number, (ii) losses in straight pipes and the Moody diagram, (iii) Bernoulli s equation and the mechanical energy balance, (iv) orifice meters, and (v) minor losses in fittings. Some of this material is presented below, but this lab handout is not a substitute for more extensive background reading. PreLab Work Prepare and submit an outline that includes: Calibrations to perform Data sets to collect Possible sources of experimental uncertainty and a plan for quantifying these errors Brief description of the work plan Any equations or physical parameters that may be needed during the laboratory session (See general lab guidelines & print out grading sheet from website).
2 ME 05 Mechanical Engineering Lab Page PreLab Exercises:. Hydrodynamic losses in pipe flow are characterized by measuring the pressure drop P over a length of pipe L. If you anticipate using flow rates of 0.5 gals/min through /4 i.d. smoothwall tubing, and want a pressure drop of 0 kpa, what length, L, of tubing should you use? Express your answer in meters. Note: You will note that this problem statement uses mixed units, which unfortunately are a fact of life in engineering calculations. You should know how to do unit conversions accurately and quickly. A good rule of thumb is to convert all units to SI before doing any numerical calculations. Hint: Assume that the working fluid is water at 0 C, and refer to a standard Moody diagram to complete this task.. The Validyne pressure transducer measures pressure differences between the two sides of a stainless steel plate (diaphragm). It will be calibrated by applying hydrostatic pressure to one side. If water at 0 C is the working fluid, what range of water heights should be used to calibrate the device over a range of differential pressures from 0 0 kpa? 3. The kit includes /8, /4, and 3/8 i.d. tubes. If water at 0 C is the working fluid and the transition Reynolds number is taken as,000, calculate the velocity and the volumetric flow rate for transition from laminar to turbulent flow for each sized tube. Record both velocities and volumetric flow rates in your notebook for future reference. 4. The paddlewheel flowmeter works on the principle that the oncoming flow rotates the paddlewheel at a frequency that is related to the flow rate. There will be some backflow as the vane of the paddlewheel sweeps forward. Consider the hypothetical situation where the flow rate vs. frequency relation is exactly linear. What would that tell you about the backflow? 5. In a standard fluids text we find the following rules of thumb for the ratio of equivalent length to pipe diameter L e /D for minor losses due to: L e /D Standard elbow: 30 Standard tee: flow through run 0 flow through branch 60 Consider the flow of water at Q = liters/min. through a ¼ diameter tube containing an elbow. Use the rule of thumb to estimate the pressure drop across the elbow. Express your answer in Pascals. 6. Referring to the pipe network shown in Figure, and with the aid of the development in the handout, manipulate the energy balance to obtain a working equation for the head losses as follows.
3 ME 05 Mechanical Engineering Lab Page 3 A) If p3 p p3 0, i.e. the bridge is balanced, and (as is true of our setup), all the tubing between point and points and 3 is the same diameter and length, the fittings are identical, and the elevation at points and 3 are the same, what is the left hand side of equation (3)? B) Now if in addition the diameter of the tubing at the outlets 4 and 5 is identical what is the relationship between u4 and u 5? C) With all this in mind, if the outlets 4 and 5 are held such that the water exits into the atmosphere, what is the working equation relating the elevations at 4 and 5 and the losses in legs A and B? LA Pump 3 LB Figure : A simple pipe network equivalent to a Wheatstone bridge. Equipment Omega paddlewheel flowmeter Validyne pressure transducer with bleeding screwdriver Water reservoir and sump pump Teflon tubing and fitting assortment Flow needle valve Orifice plate Bucket Balance Stopwatch Oscilloscope and power supply Thermometer
4 ME 05 Mechanical Engineering Lab Page 4 Technical Data Orifice plate Upstream pipe diameter Orifice diameter = 9.53 mm = 4.76 mm System Description You will need to set up a simple method to calibrate a pressure transducer by providing a known pressure difference between the two sides of the transducer. In addition you will have to construct a waterbench to perform measurements that allow you to calibrate the pressure transducer and the two kinds of flowmeters, and investigate head loss in pipe flow, fittings, and pipe networks. Although you will decide the specific arrangement, Fig. shows generically the layout of the flow loop. orifice plate pipe section pressure transducer return needle valve paddlewhee flow meter pump water supply Figure : Flowloop schematic. Theoretical orifice relations An orifice plate is one of the most common flow measurement devices. Using a control volume approach shown in Fig. 3, it is possible to obtain an expression for the flow coefficient in terms of the flow rate Q, the pressure difference P P across the orifice plate, and the geometrical parameters of the flowmeter. Applying conservation of mass for steady flow,
5 ME 05 Mechanical Engineering Lab Page 5 A, () V AV and Bernoulli s equation between position to position, V P gz V P gz, () Orifice plate Pipe D d Control volume Differential pressure transducer Figure 3: Flow in the vicinity of an orifice plate. we find that if Z =Z : V P P ( A / A ), (3) where V is the flow velocity, A is the area, is the density, g is the acceleration due to gravity and Z is the elevation. This can be rewritten in terms of the volumetric flow rate as a function of the pressure difference: Q V A A ( P ( A P ) / A ). (4) For the orifice plate meter shown in Fig. 3, the area A is not given by the orifice diameter d, but rather the diameter of the vena contracta, (where the flow has a minimum crosssectional area). This area is unknown and will change with the flow rate. Consequently, (4)
6 ME 05 Mechanical Engineering Lab Page 6 is often written with A d / 4, and a discharge coefficient C D is added to account for the combination of these geometric effects and viscous losses: ( P P) Q C D A, (5) 4 where d / D, where d is the orifice diameter and D is the diameter of the pipe. The discharge coefficient for an orifice plate meter is not constant and is found experimentally by measuring both Q and (P P ) and applying equation (5). Values for C have been measured for standardized tap locations, which allow flow rates to be measured from a pressure drop across the orifice plate. Figure 4 shows the typical dependence of C as a function of geometry and Reynolds number, Re. D D Figure 4: Discharge coefficient curves for a standard orifice plate flowmeter. Head Loss in Pipe Flows There is a pressure drop when a fluid flows in a pipe because energy is required to overcome the viscous or frictional forces exerted by the walls of the pipe on the moving fluid. In addition to the energy lost due to frictional forces, the flow also loses energy (or pressure) as it goes through fittings, such as valves, elbows, contractions and expansions. This loss in pressure is often due to the fact that flow separates locally as it moves through such fittings. The pressure loss in pipe flows is commonly referred to as head loss. The frictional losses are referred to as major losses (h l ) while losses through fittings, etc, are called minor losses (h lm ). Together they make up the total head losses (h lt ) for pipe flows.
7 ME 05 Mechanical Engineering Lab Page 7 Mechanical Energy Equation for Pipe Flows The mechanical energy equation between any two points and for steady incompressible flow is: P V P V gz gz h lt. (6) (It also be noted that for flow without losses, h lt = 0, and the energy equation reduces to Bernoulli s Equation.) The terms in parentheses represent the mechanical energy per unit mass at a particular crosssection in the pipe. Hence, the difference between the mechanical energy at two locations, i.e. the total head loss, results from the conversion of mechanical energy to thermal energy due to frictional effects. For an incompressible flow, conservation of mass determines V (since, V A V A ) and so the terms involving the fluid velocity are determined by geometry. If the elevation at position is known, the change in the gravitational potential is known. The net result is that if the pipe diameter is constant and the elevation does not change, the head loss is manifested simply as a pressure loss. Major Losses The major head loss in pipe flows is expressed in the following way: L V h l f, (7) D where L and D are the length and diameter of the pipe, respectively, and V is the average fluid velocity through the pipe. This may be taken as a definition of the friction factor, f. In general, the friction factor is a function of the Reynolds number Re and the nondimensional surface roughness / D, and is determined experimentally. The plot of f vs. Re is usually referred to as the Moody Diagram, after L. F. Moody who first published this data in this form. Minor Losses The head losses associated with fittings such as elbows, tees, couplings, etc. are referred to as minor losses. In some cases, such as short pipes with multiple fittings, these losses are actually a large percentage of the total head loss and hence are not really minor. Minor losses are expressed as either V h lm K, (8a) where K is the Loss Coefficient and must be determined experimentally for each situation, or as Le V hlm f, (8b) D wherein the loss is expressed in terms of the (known) friction factor and an equivalent Le / D. For example, an elbow creates a loss that is roughly equivalent to a pipe of length of 30 pipe diameters (see the table in Prelab Question 4). Loss coefficients, K and/or equivalent length
8 ME 05 Mechanical Engineering Lab Page 8 ratios Le / D can be found in a variety of handbooks: data for specific simple fittings are available in most undergraduate Fluid Mechanics texts. Pipe networks: the hydraulic analog of a Wheatstone bridge Consider the pipe system shown schematically in Figure. We are interested in describing the pressure loss through all the legs of this simple network. If both legs A and B exit into the atmosphere, then the pressure differentials downstream of junctions and 3 can be defined as: p p p A and p p p B 3 a a (9) where p a is atmospheric pressure. The energy equation for these two branches yields: and (0a,b) Assume that leg A of the network consists of only a straight tube uniform tube and therefore the head loss hla is LA VA hla fa. () D The head loss in leg B of the network, h LB, includes losses through the pipe itself but also any minor losses due to the insertion of elbows, etc. In general, A L V L V h f f, () B B eb B LB B B DB DB where we have chosen to express the minor losses in terms of equivalent pipe lengths, L eb. Subtracting (0a) from (0b) we obtain: 3
9 ME 05 Mechanical Engineering Lab Page 9 This particular pipe network is analogous to a Wheatstone bridge. The purpose of the electrical version of such a bridge is to be able to measure small changes in resistance accurately. In the hydraulic analog we measure small changes in head loss. When the bridge is balanced, i.e. there is no flow through the leg L 3, the pressures at points and 3 must be the same: (otherwise, the pressure gradient would drive a flow through the leg). In our laboratory setup, the leg L 3 viewed from the side is shaped in an arc as shown in Figure 5. A small tightly fitting sphere is placed in the tube. Any flow in the leg will exert a drag on the sphere and it will rise above the center. By contrast, a no flow condition will result in the sphere positioned at zero degrees from the vertical since p3 p p3 0. Thus, monitoring the sphere position allows a coarse measurement of bridge balance. o Figure 5: Schematic of the sphere in the arched tube comprising the center leg. Experimental Procedure: General You will be making a variety of measurements with water, the physical properties of which are temperature dependent. For this reason, it is very important that you know the temperature of the water for each measurement. Week One ) Calibration of the pressure transducer The first step is to calibrate the output voltage from the Validyne differential pressure transducer. Apply known pressure differences to the two sides of the transducer using hydrostatic pressure. Five to ten data points should be obtained, ranging from a zero pressure differential to a pressure differential of about 0 kpa. Perform a linear leastsquares analysis of the data before week two. Try both linear and a quadratic fits and compute the goodness of fit. If a linear fit is sufficiently accurate, record the slope and intercept of the resulting line for later use in data acquisition.
10 ME 05 Mechanical Engineering Lab Page 0 ) Calibration of the paddlewheel and orifice plate flowmeters Paddlewheel flowmeter: The paddlewheel flowmeter outputs a pulse train whose frequency is related to the flow rate. Calibrate the paddlewheel using varying flow rates by measuring the frequency as a function of flow rate. Five to ten data points should be obtained. Determine the rising and falling cutoff flow rates, i.e. the discharge below which the paddlewheel is motionless or erratic. You will notice that at low flow rates the frequency is erratic, and that the frequency fluctuates at all flow rates. Do your best to get an average reading from the oscilloscope. What might cause such fluctuations? Orifice plate flowmeter: Measure the pressure drop across the orifice plate as a function of flow rate. Five to ten points should be obtained. These data will be used to determine the discharge coefficient C D for the orifice plate as a function of the Reynolds number. 3) Investigation of major losses Prepare 6 8 lengths of the three different diameter tubing. Using the paddlewheel to measure flow rate and Tees as pressure taps, obtain data for pressure drop over a given length as a function of flow rate for the three different sized tubes. Since the Moody diagram is for long, straight tubes, try to make your tube runs as long and as straight as feasible. Obtain 50 data points for each tube over the maximum range of flow rates possible. These data will be used to determine the friction factor, f, as a function of the Reynolds number. These values will also be compared to the standard Moody diagram, so you should perform calculations on some of the data during the experiment to make sure the comparison is reasonable. Complete these calculations and the comparison with the Moody diagram before week two. Week Two 4) Major losses Depending on the quality of your data from week one, you may chose to check calibrations and/or repeat your measurements of major losses. 5) Investigation of minor losses Using the paddlewheel to measure flow rate and tees as pressure taps, measure the minor losses for an elbow, a tee, and a straight coupling as a function of flow rate. Use only one diameter tube and make sure your data are taken in the turbulent regime. 6) Investigation of a simple pipe network Set up the flow system shown in Figure of this handout incorporating the section of arced tubing between the points and 3. Use ¼ tubing for these experiments. It is best to place a needle valve before the branch so that the flow rate can be controlled. a) Prepare two 6 long lengths of ¼ tubing to serve as legs A and B. Set the flow rate with the needle valve in the midrange of the pump and make sure that the tube exits are at the same elevation. Curiously, although the two legs are identical tubing and identical lengths, the bridge may be slightly out of balance. This could be due to a number of factors, including different coiling of the two legs, burrs and rough edges where the tube was cut, slight differences in the
11 ME 05 Mechanical Engineering Lab Page losses in the two elbows at points and 3, etc. By raising or lowering the exit tubes, determine which of the branches has the larger loss, and shorten the appropriate tube in order to bring the bridge into balance. b) Experiment with the effect of raising or lowering one tube exit elevation on the bridge balance. In this way, you will obtain some feeling for the response time of the middle leg. Since the small sphere is tightly fitting, there is some time lag between a change in hydraulic resistance and the motion of the sphere. Experiment also with the effect of throttling the flow with your finger. Explain the reasons for what you observe. Change the flow rate and observe whether the bridge remains in balance or not. If so, why? If not, why not? c) Reestablish a balanced bridge by returning the flow rate to the original setting. From this point on, do not change the needle valve, as it is important for these next steps to be done at constant flow rate. Add an elbow to one of the legs and observe the resulting imbalance. Raise or lower one tube to reestablish the balance and record the elevation change necessary to accomplish this. This datum will be used to compute the minor loss using the mechanical energy balance. d) Using the empirical rule of thumb that an elbow creates a loss equivalent to 30 pipe diameters of smooth, straight pipe, shorten the leg containing the elbow by an appropriate length. Observe the bridge balance or imbalance. If imbalanced, measure the change in elevation of one of the tube exits required to reestablish balance. Experiment Report Pressure transducer The Validyne pressure transducer produces a voltage related to the pressure difference across a thin plate. If the deflection of the plate follows the laws of linear elasticity, the pressure will be linearly related to the voltage and the device is said to be a linear transducer. Perform a linear least squares analysis of the data. Try both linear and a quadratic fits and compute the goodness of fit. Discuss the degree to which this is a linear transducer. Paddlewheel flowmeter The paddlewheel flowmeter produces a pulse signal, the frequency of which is related to the fluid velocity in the pipe. Perform a least squares fit of your data, using different polynomial fits and a power law relation. Find a suitable fitting function and record your fit. To what degree is the paddlewheel a linear transducer? Are there reasons to expect either linear or non linearity in the calibration? Discuss.
12 ME 05 Mechanical Engineering Lab Page Orifice plate flowmeter The theoretical relation between Q and P is a nonlinear one, namely Q ( const.) Cd P. Using log log scales, plot your data points for Q as a function of P. Do the data appear to m fall along a straight line, indicating that a power law relation of the type Q K(P) might apply? If so, what is m? If not, why not? Use the measurements of Q vs. P, compute the discharge coefficient Cd for all the data. Plot Cd vs. the Reynolds number Re and compare against standard curves. Head loss in pipe flow Calculate the friction factors for each flow rate and tube size and plot all the data as a function of the Reynolds number. Use different plotting symbols for different tube diameters and check for Reynolds number scaling. Compare your data with the standard Moody diagram and discuss. Minor losses Express your results for minor losses through elbows, tees and couplings both as loss coefficients, K, and as equivalent lengths, Le / D. Compare your results with literature results for K, and with the common empirical rules of thumb for Le / D. Piping network Compute the loss coefficient and the equivlent length, Le / D, for an elbow as measured by the bridge technique. Compare it against your direct measurement and also against the standard rule of thumb.
Experiment 3 Pipe Friction
EML 316L Experiment 3 Pipe Friction Laboratory Manual Mechanical and Materials Engineering Department College of Engineering FLORIDA INTERNATIONAL UNIVERSITY Nomenclature Symbol Description Unit A crosssectional
More informationFLUID FLOW Introduction General Description
FLUID FLOW Introduction Fluid flow is an important part of many processes, including transporting materials from one point to another, mixing of materials, and chemical reactions. In this experiment, you
More informationHead Loss in Pipe Flow ME 123: Mechanical Engineering Laboratory II: Fluids
Head Loss in Pipe Flow ME 123: Mechanical Engineering Laboratory II: Fluids Dr. J. M. Meyers Dr. D. G. Fletcher Dr. Y. Dubief 1. Introduction Last lab you investigated flow loss in a pipe due to the roughness
More informationCHE 253M Experiment No. 3 LIQUID FLOW MEASUREMENT
Rev 8/15 AW/GW CHE 253M Experiment No. 3 LIQUID FLOW MEASUREMENT The objective of this experiment is to familiarize the student with several types of flowmeasuring devices commonly used in the laboratory
More informationChapter 8: Flow in Pipes
Objectives 1. Have a deeper understanding of laminar and turbulent flow in pipes and the analysis of fully developed flow 2. Calculate the major and minor losses associated with pipe flow in piping networks
More informationFigure 1. Head losses in a pipe
53:071 Principles of Hydraulics Laboratory Experiment #1 ENERGY AND HYDRAULIC GRADE LINES IN WATER PIPE SYSTEMS Principle The energy of a real fluid decreases as it moves through a pipe. The energy budget
More informationMichael Montgomery Marketing Product Manager Rosemount Inc. Russ Evans Manager of Engineering and Design Rosemount Inc.
ASGMT / Averaging Pitot Tube Flow Measurement Michael Montgomery Marketing Product Manager Rosemount Inc. Russ Evans Manager of Engineering and Design Rosemount Inc. Averaging Pitot Tube Meters Introduction
More informationExperiment (13): Flow channel
Introduction: An open channel is a duct in which the liquid flows with a free surface exposed to atmospheric pressure. Along the length of the duct, the pressure at the surface is therefore constant and
More informationME 305 Fluid Mechanics I. Part 8 Viscous Flow in Pipes and Ducts
ME 305 Fluid Mechanics I Part 8 Viscous Flow in Pipes and Ducts These presentations are prepared by Dr. Cüneyt Sert Mechanical Engineering Department Middle East Technical University Ankara, Turkey csert@metu.edu.tr
More informationSwissmetro travels at high speeds through a tunnel at low pressure. It will therefore undergo friction that can be due to:
I. OBJECTIVE OF THE EXPERIMENT. Swissmetro travels at high speeds through a tunnel at low pressure. It will therefore undergo friction that can be due to: 1) Viscosity of gas (cf. "Viscosity of gas" experiment)
More informationApplied Fluid Mechanics
Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and
More informationFluid Mechanics Prof. T. I. Eldho Department of Civil Engineering Indian Institute of Technology, Bombay. Lecture No. # 36 Pipe Flow Systems
Fluid Mechanics Prof. T. I. Eldho Department of Civil Engineering Indian Institute of Technology, Bombay Lecture No. # 36 Pipe Flow Systems Welcome back to the video course on Fluid Mechanics. In today
More informationChapter 10. Flow Rate. Flow Rate. Flow Measurements. The velocity of the flow is described at any
Chapter 10 Flow Measurements Material from Theory and Design for Mechanical Measurements; Figliola, Third Edition Flow Rate Flow rate can be expressed in terms of volume flow rate (volume/time) or mass
More informationFluid Mechanics Definitions
Definitions 91a1 Fluids Substances in either the liquid or gas phase Cannot support shear Density Mass per unit volume Specific Volume Specific Weight % " = lim g#m ( ' * = +g #V $0& #V ) Specific Gravity
More informationFIGURE P8 50E FIGURE P8 62. Minor Losses
8 48 Glycerin at 40 C with r 1252 kg/m 3 and m 0.27 kg/m s is flowing through a 4cmdiameter horizontal smooth pipe with an average velocity of 3.5 m/s. Determine the pressure drop per 10 m of the pipe.
More informationPipe Loss Experimental Apparatus
Pipe Loss Experimental Apparatus Kathleen Lifer, Ryan Oberst, Benjamin Wibberley Ohio Northern University Ada, OH 45810 Email: bwibberley@onu.edu Abstract The objective of this project was to develop
More informationdu u U 0 U dy y b 0 b
BASIC CONCEPTS/DEFINITIONS OF FLUID MECHANICS (by Marios M. Fyrillas) 1. Density (πυκνότητα) Symbol: 3 Units of measure: kg / m Equation: m ( m mass, V volume) V. Pressure (πίεση) Alternative definition:
More informationModeling Fluid Systems
Modeling Fluid Systems The prevalent use of fluid (hydraulic) circuitry in machines tool applications, aircraft control systems, and similar operations occurs because of such factors such as accuracy,
More informationA LAMINAR FLOW ELEMENT WITH A LINEAR PRESSURE DROP VERSUS VOLUMETRIC FLOW. 1998 ASME Fluids Engineering Division Summer Meeting
TELEDYNE HASTINGS TECHNICAL PAPERS INSTRUMENTS A LAMINAR FLOW ELEMENT WITH A LINEAR PRESSURE DROP VERSUS VOLUMETRIC FLOW Proceedings of FEDSM 98: June 5, 998, Washington, DC FEDSM98 49 ABSTRACT The pressure
More informationApplied Fluid Mechanics
Applied Fluid Mechanics Sixth Edition Robert L. Mott University of Dayton PEARSON Prentkv Pearson Education International CHAPTER 1 THE NATURE OF FLUIDS AND THE STUDY OF FLUID MECHANICS 1.1 The Big Picture
More informationOpen channel flow Basic principle
Open channel flow Basic principle INTRODUCTION Flow in rivers, irrigation canals, drainage ditches and aqueducts are some examples for open channel flow. These flows occur with a free surface and the pressure
More informationCHME 302 CHEMICAL ENGINEERING LABOATORYI EXPERIMENT 3021 FLOW MEASUREMENT TEST
CHME 302 CHEMICAL ENGINEERING LABOATORYI EXPERIMENT 3021 FLOW MEASUREMENT TEST OBJECTIVE The purpose of the experiment is to demonstrate the working principles of four most common devices used to measure
More informationPressure drop in pipes...
Pressure drop in pipes... PRESSURE DROP CALCULATIONS Pressure drop or head loss, occurs in all piping systems because of elevation changes, turbulence caused by abrupt changes in direction, and friction
More informationLaminar and Turbulent flow. Flow Sensors. Reynolds Number. Thermal flow Sensor. Flow and Flow rate. R = Mass Flow controllers
Flow and Flow rate. Laminar and Turbulent flow Laminar flow: smooth, orderly and regular Mechanical sensors have inertia, which can integrate out small variations due to turbulence Turbulent flow: chaotic
More informationTransient Mass Transfer
Lecture T1 Transient Mass Transfer Up to now, we have considered either processes applied to closed systems or processes involving steadystate flows. In this lecture we turn our attention to transient
More informationFlow Measurement Options for Pipeline and Open Channel Flow
Flow Measurement Options for Pipeline and Open Channel Flow October 2013 Presented by Molly Skorpik  2013 Montana Association of Dam and Canal Systems Conference Irrigation Training and Research Center
More informationPipe FlowFriction Factor Calculations with Excel
Pipe FlowFriction Factor Calculations with Excel Course No: C03022 Credit: 3 PDH Harlan H. Bengtson, PhD, P.E. Continuing Education and Development, Inc. 9 Greyridge Farm Court Stony Point, NY 10980
More informationApplied Fluid Mechanics
Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 7. General Energy Equation
More informationhosetohose coupling pumptohose coupling
pipe_02 A homeowner plans to pump water from a stream in their backyard to water their lawn. A schematic of the pipe system is shown in the figure. 3 m 1 m inlet pipetopump coupling stream reentrant
More informationME 315  Heat Transfer Laboratory. Experiment No. 7 ANALYSIS OF ENHANCED CONCENTRIC TUBE AND SHELL AND TUBE HEAT EXCHANGERS
ME 315  Heat Transfer Laboratory Nomenclature Experiment No. 7 ANALYSIS OF ENHANCED CONCENTRIC TUBE AND SHELL AND TUBE HEAT EXCHANGERS A heat exchange area, m 2 C max maximum specific heat rate, J/(s
More informationThe Viscosity of Fluids
Experiment #11 The Viscosity of Fluids References: 1. Your first year physics textbook. 2. D. Tabor, Gases, Liquids and Solids: and Other States of Matter (Cambridge Press, 1991). 3. J.R. Van Wazer et
More informationBasic Equations, Boundary Conditions and Dimensionless Parameters
Chapter 2 Basic Equations, Boundary Conditions and Dimensionless Parameters In the foregoing chapter, many basic concepts related to the present investigation and the associated literature survey were
More informationEffects of Beta Ratio and Reynold s Number on Coefficient of Discharge of Orifice Meter
J Agric Rural Dev 7(&2), 556, June 29 ISSN 886 K wl I cj x Dbœqb zj Available online at http://www.banglajol.info/index.php/jard JARD Journal of Agriculture & Rural Development Effects of Beta Ratio
More informationOUTCOME 3 TUTORIAL 5 DIMENSIONAL ANALYSIS
Unit 41: Fluid Mechanics Unit code: T/601/1445 QCF Level: 4 Credit value: 15 OUTCOME 3 TUTORIAL 5 DIMENSIONAL ANALYSIS 3 Be able to determine the behavioural characteristics and parameters of real fluid
More information4.What is the appropriate dimensionless parameter to use in comparing flow types? YOUR ANSWER: The Reynolds Number, Re.
CHAPTER 08 1. What is most likely to be the main driving force in pipe flow? A. Gravity B. A pressure gradient C. Vacuum 2.What is a general description of the flow rate in laminar flow? A. Small B. Large
More informationFLOW MEASUREMENT 2001 INTERNATIONAL CONFERENCE DERIVATION OF AN EXPANSIBILITY FACTOR FOR THE VCONE METER
FLOW MEASUREMENT 200 INTERNATIONAL CONFERENCE DERIVATION OF AN EXPANSIBILITY FACTOR FOR THE VCONE METER Dr D G Stewart, NEL Dr M ReaderHarris, NEL Dr R J W Peters, McCrometer Inc INTRODUCTION The VCone
More informationBasics and Concepts. 2.1 Introduction Force, Weight, and Mass Density Specific Gravity Pressure...
2 Fluids Basics and Concepts TOPIC PAGE 2.1 Introduction... 22 2.2 Force, Weight, and Mass... 22 2.3 Density... 23 2.4 Specific Gravity... 23 2.5 Pressure... 23 2.6 Temperature... 25 2.7 Viscosity... 26
More informationDimensional analysis is a method for reducing the number and complexity of experimental variables that affect a given physical phenomena.
Dimensional Analysis and Similarity Dimensional analysis is very useful for planning, presentation, and interpretation of experimental data. As discussed previously, most practical fluid mechanics problems
More informationForce. Net Force Mass. Acceleration = Section 1: Weight. Equipment Needed Qty Equipment Needed Qty Force Sensor 1 Mass and Hanger Set 1 Balance 1
Department of Physics and Geology Background orce Physical Science 1421 A force is a vector quantity capable of producing motion or a change in motion. In the SI unit system, the unit of force is the Newton
More informationDifferential Relations for Fluid Flow. Acceleration field of a fluid. The differential equation of mass conservation
Differential Relations for Fluid Flow In this approach, we apply our four basic conservation laws to an infinitesimally small control volume. The differential approach provides point by point details of
More informationTesting Protocol for Differential Pressure Measurement Devices API MPMS Chapter 22.2
Testing Protocol for Differential Pressure Measurement Devices API MPMS Chapter 22.2 Steve Baldwin Chevron Energy Technology Company Houston, Texas USA Casey Hodges CEESI Measurement Solutions. Nunn, Colorado
More informationGrant Agreement No. 228296 SFERA. Solar Facilities for the European Research Area SEVENTH FRAMEWORK PROGRAMME. Capacities Specific Programme
Grant Agreement No. 228296 SFERA Solar Facilities for the European Research Area SEVENTH FRAMEWORK PROGRAMME Capacities Specific Programme Research Infrastructures Integrating Activity  Combination of
More informationLecture 5 Hemodynamics. Description of fluid flow. The equation of continuity
1 Lecture 5 Hemodynamics Description of fluid flow Hydrodynamics is the part of physics, which studies the motion of fluids. It is based on the laws of mechanics. Hemodynamics studies the motion of blood
More informationOpen Channel Flow. M. Siavashi. School of Mechanical Engineering Iran University of Science and Technology
M. Siavashi School of Mechanical Engineering Iran University of Science and Technology W ebpage: webpages.iust.ac.ir/msiavashi Email: msiavashi@iust.ac.ir Landline: +98 21 77240391 Fall 2013 Introduction
More informationFluid Dynamics Basics
Fluid Dynamics Basics Bernoulli s Equation A very important equation in fluid dynamics is the Bernoulli equation. This equation has four variables: velocity ( ), elevation ( ), pressure ( ), and density
More informationPROCESS CONTROL INDUSTRIAL MAINTENANCE TECHNICIAN CERTIFICATE
PROCESS CONTROL INDUSTRIAL MAINTENANCE TECHNICIAN CERTIFICATE Process Control (T5552) Course 1 Introduction to Process Control Course 2 Instrument Tags Course 3 Piping and Instrumentation Diagrams Course
More informationChapter 13 OPENCHANNEL FLOW
Fluid Mechanics: Fundamentals and Applications, 2nd Edition Yunus A. Cengel, John M. Cimbala McGrawHill, 2010 Lecture slides by Mehmet Kanoglu Copyright The McGrawHill Companies, Inc. Permission required
More informationMeasure Pressure and Flow
Measure Pressure and Flow Install the EMANT300 USB DAQ and Stepper Motor Training Kit 1. Connect up the EMANT300 USB DAQ and Pressure and Air pump Adaptor as shown. EMANT300 USB DAQ Pressure and Air pump
More informationDEVELOPMENT OF HIGH SPEED RESPONSE LAMINAR FLOW METER FOR AIR CONDITIONING
DEVELOPMENT OF HIGH SPEED RESPONSE LAMINAR FLOW METER FOR AIR CONDITIONING Toshiharu Kagawa 1, Yukako Saisu 2, Riki Nishimura 3 and Chongho Youn 4 ABSTRACT In this paper, we developed a new laminar flow
More informationFor Water to Move a driving force is needed
RECALL FIRST CLASS: Q K Head Difference Area Distance between Heads Q 0.01 cm 0.19 m 6cm 0.75cm 1 liter 86400sec 1.17 liter ~ 1 liter sec 0.63 m 1000cm 3 day day day constant head 0.4 m 0.1 m FINE SAND
More informationHydraulic losses in pipes
Hydraulic losses in pipes Henryk Kudela Contents 1 Viscous flows in pipes 1 1.1 Moody Chart.................................... 2 1.2 Types of Fluid Flow Problems........................... 5 1.3 Minor
More informationWEEK 6: FORCE, MASS, AND ACCELERATION
Name Date Partners WEEK 6: FORCE, MASS, AND ACCELERATION OBJECTIVES To develop a definition of mass in terms of an object s acceleration under the influence of a force. To find a mathematical relationship
More informationThe Viscosity of Fluids
Experiment #11 The Viscosity of Fluids References: 1. Your first year physics textbook. 2. D. Tabor, Gases, Liquids and Solids: and Other States of Matter (Cambridge Press, 1991). 3. J.R. Van Wazer et
More informationFLUID MECHANICS IM0235 DIFFERENTIAL EQUATIONS  CB0235 2014_1
COURSE CODE INTENSITY PREREQUISITE COREQUISITE CREDITS ACTUALIZATION DATE FLUID MECHANICS IM0235 3 LECTURE HOURS PER WEEK 48 HOURS CLASSROOM ON 16 WEEKS, 32 HOURS LABORATORY, 112 HOURS OF INDEPENDENT
More informationExperimentation and Computational Fluid Dynamics Modelling of Roughness Effects in Flexible Pipelines
Experimentation and Computational Fluid Dynamics Modelling of Roughness Effects in Flexible Pipelines Sophie Yin Jeremy Leggoe School of Mechanical and Chemical Engineering Daniel Teng Paul Pickering CEED
More informationCEE 370 Fall 2015. Laboratory #3 Open Channel Flow
CEE 70 Fall 015 Laboratory # Open Channel Flow Objective: The objective of this experiment is to measure the flow of fluid through open channels using a Vnotch weir and a hydraulic jump. Introduction:
More informationThese slides contain some notes, thoughts about what to study, and some practice problems. The answers to the problems are given in the last slide.
Fluid Mechanics FE Review Carrie (CJ) McClelland, P.E. cmcclell@mines.edu Fluid Mechanics FE Review These slides contain some notes, thoughts about what to study, and some practice problems. The answers
More informationmeasurement, but almost any pipe elbow can be calibrated Elbow meters are not as potentially accurate as venturi,
Lecture 14 Flow Measurement in Pipes I. Elbow Meters An elbow in a pipe can be used as a flow measuring device much in the same way as a venturi or orifice plate The head differential across the elbow
More informationHeat Transfer Prof. Dr. Ale Kumar Ghosal Department of Chemical Engineering Indian Institute of Technology, Guwahati
Heat Transfer Prof. Dr. Ale Kumar Ghosal Department of Chemical Engineering Indian Institute of Technology, Guwahati Module No. # 04 Convective Heat Transfer Lecture No. # 03 Heat Transfer Correlation
More informationHEAVY OIL FLOW MEASUREMENT CHALLENGES
HEAVY OIL FLOW MEASUREMENT CHALLENGES 1 INTRODUCTION The vast majority of the world s remaining oil reserves are categorised as heavy / unconventional oils (high viscosity). Due to diminishing conventional
More informationAdvanced Differential Pressure Flowmeter Technology VCONE FLOW METER TECHNICAL BRIEF
Advanced Differential Pressure Flowmeter Technology VCONE FLOW METER TECHNICAL BRIEF Table of Contents Section 1  General Introduction 1.1 1 Principles Of Operation 1.2 1 Reshaping The Velocity Profile
More informationFluids and Solids: Fundamentals
Fluids and Solids: Fundamentals We normally recognize three states of matter: solid; liquid and gas. However, liquid and gas are both fluids: in contrast to solids they lack the ability to resist deformation.
More informationMinor losses include head losses through/past hydrants, couplers, valves,
Lecture 10 Minor Losses & Pressure Requirements I. Minor Losses Minor (or fitting, or local ) hydraulic losses along pipes can often be estimated as a function of the velocity head of the water within
More informationChapter 2. Derivation of the Equations of Open Channel Flow. 2.1 General Considerations
Chapter 2. Derivation of the Equations of Open Channel Flow 2.1 General Considerations Of interest is water flowing in a channel with a free surface, which is usually referred to as open channel flow.
More informationThe Most General Applications of Bernoulli s Equation
The Most General Applications of Bernoulli s Equation Bởi: OpenStaxCollege Torricelli s Theorem [link] shows water gushing from a large tube through a dam. What is its speed as it emerges? Interestingly,
More information2.0 BASIC CONCEPTS OF OPEN CHANNEL FLOW MEASUREMENT
2.0 BASIC CONCEPTS OF OPEN CHANNEL FLOW MEASUREMENT Open channel flow is defined as flow in any channel where the liquid flows with a free surface. Open channel flow is not under pressure; gravity is the
More informationBackwater Rise and Drag Characteristics of Bridge Piers under Subcritical
European Water 36: 735, 11. 11 E.W. Publications Backwater Rise and Drag Characteristics of Bridge Piers under Subcritical Flow Conditions C.R. Suribabu *, R.M. Sabarish, R. Narasimhan and A.R. Chandhru
More informationApplied Fluid Mechanics
Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and
More informationMechanical Pressure Measurement David Gardellin, P.E.
Mechanical Pressure Measurement David Gardellin, P.E. Bourdon tubetype pressure sensors are the most common for industrial use in the family of elastic pressure elements. They have been in use for over
More informationLab 5: Conservation of Energy
Lab 5: Conservation of Energy Equipment SWS, 1meter stick, 2meter stick, heavy duty bench clamp, 90cm rod, 40cm rod, 2 double clamps, brass spring, 100g mass, 500g mass with 5cm cardboard square
More informationHEAT TRANSFER AUGMENTATION THROUGH DIFFERENT PASSIVE INTENSIFIER METHODS
HEAT TRANSFER AUGMENTATION THROUGH DIFFERENT PASSIVE INTENSIFIER METHODS P.R.Hatwar 1, Bhojraj N. Kale 2 1, 2 Department of Mechanical Engineering Dr. Babasaheb Ambedkar College of Engineering & Research,
More informationGraphical Presentation of Data
Graphical Presentation of Data Guidelines for Making Graphs Titles should tell the reader exactly what is graphed Remove stray lines, legends, points, and any other unintended additions by the computer
More informationFILTRATION. Introduction. Batch Filtration (Small Scale) Controlling parameters: Pressure drop, time for filtration, concentration of slurry
FILTRATION Introduction Separation of solids from liquids is a common operation that requires empirical data to make predictions of performance. These data are usually obtained from experiments performed
More informationChapter 5 MASS, BERNOULLI AND ENERGY EQUATIONS
Fluid Mechanics: Fundamentals and Applications, 2nd Edition Yunus A. Cengel, John M. Cimbala McGrawHill, 2010 Chapter 5 MASS, BERNOULLI AND ENERGY EQUATIONS Lecture slides by Hasan Hacışevki Copyright
More informationMATLAB AS A PROTOTYPING TOOL FOR HYDRONIC NETWORKS BALANCING
MATLAB AS A PROTOTYPING TOOL FOR HYDRONIC NETWORKS BALANCING J. Pekař, P. Trnka, V. Havlena* Abstract The objective of this note is to describe the prototyping stage of development of a system that is
More informationEXPERIMENT NUMBER 7 PERFORMANCE TEST OF A CENTRIFUGAL PUMP
EXPERIMENT NUMBER 7 PERFORMANCE TEST OF A CENTRIFUGAL PUMP OBJECTIVE The primary objectives of this experiment is to measure the performance of a centrifugal pump and compare the results to the manufacturers
More informationFluid Flow Instrumentation
Fluid Flow Instrumentation In the physical world, mechanical engineers are frequently required to monitor or control the flow of various fluids through pipes, ducts and assorted vessels. This fluid can
More informationFILTRATION. Introduction
FILTRATION Introduction Separation of solids from liquids is a common operation that requires empirical data to make predictions of performance. These data are usually obtained from experiments performed
More informationManufacturing Equipment Modeling
QUESTION 1 For a linear axis actuated by an electric motor complete the following: a. Derive a differential equation for the linear axis velocity assuming viscous friction acts on the DC motor shaft, leadscrew,
More informationWhen the fluid velocity is zero, called the hydrostatic condition, the pressure variation is due only to the weight of the fluid.
Fluid Statics When the fluid velocity is zero, called the hydrostatic condition, the pressure variation is due only to the weight of the fluid. Consider a small wedge of fluid at rest of size Δx, Δz, Δs
More informationFluent Software Training TRN Boundary Conditions. Fluent Inc. 2/20/01
Boundary Conditions C1 Overview Inlet and Outlet Boundaries Velocity Outline Profiles Turbulence Parameters Pressure Boundaries and others... Wall, Symmetry, Periodic and Axis Boundaries Internal Cell
More informationChapter 11 SERVO VALVES. Fluid Power Circuits and Controls, John S.Cundiff, 2001
Chapter 11 SERVO VALVES Fluid Power Circuits and Controls, John S.Cundiff, 2001 Servo valves were developed to facilitate the adjustment of fluid flow based on the changes in the load motion. 1 Typical
More informationA MTR FUEL ELEMENT FLOW DISTRIBUTION MEASUREMENT PRELIMINARY RESULTS
A MTR FUEL ELEMENT FLOW DISTRIBUTION MEASUREMENT PRELIMINARY RESULTS W. M. Torres, P. E. Umbehaun, D. A. Andrade and J. A. B. Souza Centro de Engenharia Nuclear Instituto de Pesquisas Energéticas e Nucleares
More informationExperimental Evaluation of the Discharge Coefficient of a CentrePivot Roof Window
Experimental Evaluation of the Discharge Coefficient of a CentrePivot Roof Window Ahsan Iqbal #1, Alireza Afshari #2, Per Heiselberg *3, Anders Høj **4 # Energy and Environment, Danish Building Research
More informationDepartment of Chemical Engineering, National Institute of Technology, Tiruchirappalli 620 015, Tamil Nadu, India
Experimental Thermal and Fluid Science 32 (2007) 92 97 www.elsevier.com/locate/etfs Studies on heat transfer and friction factor characteristics of laminar flow through a circular tube fitted with right
More informationChapter 8 Fluid Flow
Chapter 8 Fluid Flow GOALS When you have mastered the contents of this chapter, you will be able to achieve the following goals: Definitions Define each of the following terms, and use it in an operational
More informationEnergy and Flow Measurement for Hydronic Systems
Energy and Flow Measurement for Hydronic Systems Presented By: George Szyszko Applications Consultant MCR for ONICON Incorporated Why Measure Thermal Energy? To control something, you must first measure
More informationL r = L m /L p. L r = L p /L m
NOTE: In the set of lectures 19/20 I defined the length ratio as L r = L m /L p The textbook by Finnermore & Franzini defines it as L r = L p /L m To avoid confusion let's keep the textbook definition,
More informationENERGYand WORK (PART I and II) 9MAC
ENERGYand WORK (PART I and II) 9MAC Purpose: To understand work, potential energy, & kinetic energy. To understand conservation of energy and how energy is converted from one form to the other. Apparatus:
More informationME332 FLUID MECHANICS LABORATORY
ME332 FLUID MECHANICS LABORATORY Mihir Sen Department of Aerospace and Mechanical Engineering University of Notre Dame Notre Dame, IN 46556 Contents January 8, 2003 Spring 2003 Before break: Experiments
More informationInternational Journal of Latest Research in Science and Technology Volume 4, Issue 2: Page No.161166, MarchApril 2015
International Journal of Latest Research in Science and Technology Volume 4, Issue 2: Page No.161166, MarchApril 2015 http://www.mnkjournals.com/ijlrst.htm ISSN (Online):22785299 EXPERIMENTAL STUDY
More information2. Parallel pump system Q(pump) = 300 gpm, h p = 270 ft for each of the two pumps
Pumping Systems: Parallel and Series Configurations For some piping system designs, it may be desirable to consider a multiple pump system to meet the design requirements. Two typical options include parallel
More informationPractice Problems on Pumps. Answer(s): Q 2 = 1850 gpm H 2 = 41.7 ft W = 24.1 hp. C. Wassgren, Purdue University Page 1 of 16 Last Updated: 2010 Oct 29
_02 A centrifugal with a 12 in. diameter impeller requires a power input of 60 hp when the flowrate is 3200 gpm against a 60 ft head. The impeller is changed to one with a 10 in. diameter. Determine the
More informationLecture 6  Boundary Conditions. Applied Computational Fluid Dynamics
Lecture 6  Boundary Conditions Applied Computational Fluid Dynamics Instructor: André Bakker http://www.bakker.org André Bakker (20022006) Fluent Inc. (2002) 1 Outline Overview. Inlet and outlet boundaries.
More informationAppendix 4C. Open Channel Theory
4C1 Appendix 4C Open Channel Theory 4C2 Appendix 4.C  Table of Contents 4.C.1 Open Channel Flow Theory 4C3 4.C.2 Concepts 4C3 4.C.2.1 Specific Energy 4C3 4.C.2.2 Velocity Distribution Coefficient
More informationWhat is the most obvious difference between pipe flow and open channel flow????????????? (in terms of flow conditions and energy situation)
OPEN CHANNEL FLOW 1 3 Question What is the most obvious difference between pipe flow and open channel flow????????????? (in terms of flow conditions and energy situation) Typical open channel shapes Figure
More informationPSS 27.2 The Electric Field of a Continuous Distribution of Charge
Chapter 27 Solutions PSS 27.2 The Electric Field of a Continuous Distribution of Charge Description: Knight ProblemSolving Strategy 27.2 The Electric Field of a Continuous Distribution of Charge is illustrated.
More informationAP1 Oscillations. 1. Which of the following statements about a springblock oscillator in simple harmonic motion about its equilibrium point is false?
1. Which of the following statements about a springblock oscillator in simple harmonic motion about its equilibrium point is false? (A) The displacement is directly related to the acceleration. (B) The
More informationPrelab Exercises: Hooke's Law and the Behavior of Springs
59 Prelab Exercises: Hooke's Law and the Behavior of Springs Study the description of the experiment that follows and answer the following questions.. (3 marks) Explain why a mass suspended vertically
More informationApplication of the Orifice Meter for Accurate Gas Flow Measurement page 1. Application of the Orifice Meter for Accurate Gas Flow Measurement.
Application of the Orifice Meter for Accurate Gas Flow Measurement page 1 DANIEL MEASUREMENT AND CONTROL WHITE PAPER Application of the Orifice Meter for Accurate Gas Flow Measurement www.daniel.com Summary
More information