# PIEZOMETER AIRFLOW MEASURING RING

Save this PDF as:

Size: px
Start display at page:

## Transcription

1 PIEZOMETER AIRFLOW MEASURING RING M.K. Plastics is pleased to introduce The Piezometer Airflow Measuring Ring, now available as an accessory for our steel Axijet fans. The system consists of a Piezometer ring mounted at the throat and a static pressure tap mounted on the face of the inlet cone. A differential pressure transducer and digital display can also be provided. Based on laboratory tests, the system was determined to be accurate within +/-5%. HOW IT WORKS The System is based on the principle of a flow nozzle. The inlet cone of the fan is used as the flow nozzle, and the flow can be calculated by measuring the static pressure drop through the inlet cone. The pressure drop is measured from the tap located on the face of the inlet cone to the piezometer ring in thethroat. The inlet tap is connected to the high-pressure side of the transducer and the piezometer ring is connected to the low-pressure side (see diagram below). 1

2 TAP LOCATIONS FOR IN-LINE VERTICAL AXIJET FANS Measurement of Airflow The equations below are accurate for flow estimation for flows from 40% to 100% of wide-open volume. According to testing done previously at M.K. Plastics, several factors affect the accuracy of this method of determining flow. The equations below assume the following: There are no vanes or other obstructions in or near the inlet Even flow entering the funnel (no pre-swirl) Standard wheel to inlet cone overlap Accurate determination of air density at the inlet Free inlet (consult M.K. Plastics for ducted inlet factors) Non-Standard Density Method One of the following equations is used to measure the flow: ACFM = C1 * A * A = Actual inlet funnel throat area (Sq. ft) available from M.K. Plastics ΔP = The differential in static pressure from the piezometer ring and the front pressure tap (inch-w.g.) P = Air density (pounds mass/cubic ft) C1 = Factor (standard centrifugal Axijets: 735.8, mixed flow Vertical Axijets: , centrifugal Vertical Axijets: ). 2

3 Standard Density Method The equations can be simplified by assuming standard density and assuming funnel dimensions match drawing dimensions. For the F factor, consult M.K. Plastics. The equation then becomes the following For standard air (p = lb/ft3): ACFM = F F = Factor (consult M.K. Plastics) ΔP = The differential in static pressure from the piezometer ring and the front pressure tap (inch-w.g.) 3

### Differential Pressure Sensors for air and non-corrosive gases, with calibration certificate

1 919 1919P01 1919P03 QBM65/C, QBM75-1U/C Differential Pressure Sensors for air and non-corrosive gases, with calibration certificate QBM751-1/C QBM65/C QBM75/C Highly accurate measurements with calibration

### DREXELBROOK. USonic-R Series. Ultrasonic Level Measurement System. User Friendly

DREXELBROOK A Leader In Level Measurement Solutions USonic-R Series Ultrasonic Level Measurement System User Friendly Set the measurement range directly in inches, feet, millimeters, centimeters, or meters

### INLINE flowmeter for continuous flow measurement

INLINE flowmeter for continuous flow measurement Economic integration in pipe systems without any additional piping 3-wire frequency pulse version to directly interface with PLC s (both PNP and NPN) Connection

### key- precision Tecum cuil Ludere ipsae. Possem Precision level detection When space and price matter Pointek CLS 100

Pointek CLS 100 Compact design and short insertion length. Ideal for constricted spaces and for small vessels or pipes up to 100 mm (4 ). Tecum cuil Ludere ipsae. Possem Precision level detection When

### MTI10 Insertion and MTL10 In-line Thermal Mass Flowmeter and Temperature Transmitter

Thermal Mass meter and Temperature Transmitter Description The MTI10 insertion thermal mass flowmeter and the MTL10 in-line thermal mass flowmeter provide accurate mass flow measurement of clean, dry gases

### TAC I/NETTM MR-VAV-AX. Application Specific MicroRegulator TM

TAC I/NETTM TM TAC s TM has an extensive range ofpre-engineered VAV control sequences, which combined with a fully integrated actuator and airflow transducer, makes it a low Total Installed Cost solution

### PolyGard Carbon Monoxide CO Transmitter ADTX3 1110

PolyGard Carbon Monoxide CO Transmitter ADTX3 1110 DESCRIPTION CO transmitter including digital measurement value processing and temperature compensation for the continuous monitoring of the ambient air

### Automation System TROVIS 6400 TROVIS 6493 Compact Controller

Automation System TROVIS 6400 TROVIS 6493 Compact Controller For panel mounting (front frame 48 x 96 mm/1.89 x 3.78 inch) Application Digital controller to automate industrial and process plants for general

### Online Infrared Flue Gas Analyzer Gasboard 3000

Online Infrared Flue Gas Analyzer Gasboard 3000 O 2 CO CO2 SO 2 NO NO x Use of measurement methods Proprietary Infrared NDIR Detectors - Micro-flow in ppm range: CO, NO, SO2, CO2 - Dual beam in % volume

### Digital inductive conductivity transmitter

Digital inductive conductivity transmitter Optimal solution for conductivity measurements in difficult fluids (polluted, dirty,...) PEEK/PPA version for CIP applications Large range of process connections

### Contents. Document information

User Manual Contents Document information... 2 Introduction... 3 Warnings... 3 Manufacturer... 3 Description... Installation... Configuration... Troubleshooting...11 Technical data...12 Device Scope: PCB

### CAMS. Combustion Airflow Management System. Proven solutions for a tough industry

CAMS Combustion Airflow Management System Proven solutions for a tough industry CAMS TM Combustion Airflow Management System Product Description The Air Monitor Power CAMS TM Combustion Airflow Management

### DSL-340 MkIII Double Pass Dust Monitor Measures 0-1000mg/m 3 using DDP

Ideal for monitoring dust levels in the exhaust gas of industrial combustion or air filtration processes. Innovative Dynamic Detection Principle (DDP) measurement technique Immune to gradual reductions

### Signet 9900 Transmitter

Signet 9900 Transmitter Member of the SmartPro Family of Instruments Features Panel Mount Field Mount Multiple sensor types supported with one instrument Dial-type digital bar graph NEW! View ph/orp sensor

### THERMAL ANEMOMETRY ELECTRONICS, SOFTWARE AND ACCESSORIES

TSI and TSI logo are registered trademarks of TSI Incorporated. SmartTune is a trademark of TSI Incorporated. THERMAL ANEMOMETRY ELECTRONICS, SOFTWARE AND ACCESSORIES IFA 300 Constant Temperature Anemometry

### Series DFMT Digital Paddlewheel Flowmeter

Series DFMT Digital Paddlewheel Flowmeter Bulletin F-DFMT Specifications - Installation Operating Instructions -/8 [00] -/ [8] -/8 [00] A The Series DFMT Digital Paddlewheel Flow Transmitter provides instantaneous

### Plastic tube flowmeters Series PT/PS

Plastic tube flowmeters Variable area flowmeter for liquids and gases Low cost, excellent readability and light weight Simple installation (flanged, threaded or socket ends for solvent or fusion welding

### 2.24. Pump Controller 704. Data Sheet

Pump Controller 704 2.24 There are several reasons to choose an MJK pump controller. Pump controller 704, as described in this data sheet, was introduced in 1989. It has been updated several times with

### COMBIMASS. Technical Data COMBIMASS eco-bio +

COMBIMASS Technical Data THE SYSTEM COMBIMASS The field transmitters of the COMBIMASS eco series are suitable for gas flow measurement and cover a wide range of different applications. The instruments

### Thermal Mass Flow Meter (Series MST)

Overview MaxiFlo TM MST series Thermal Mass Flow Meter is the instrument of choice for reliable and accurate measurement of mass flows for various gases. It measures the mass flow of gas based on constant

### 4 Posiflow Proportional Solenoid Valves

4 Posiflow Solenoid Brass or Stainless Steel Bodies 1/4" to 1/2" NPT 2/2 Features Flow rates adjustable between 0% and 100% of rating Flow rate can also be regulated by a range of electrical inputs (sensors,

### Programmable Terminal. NPSeries. Easy Setup with the Low-cost, Compact NP Series

Programmable Terminal NPSeries Easy Setup with the Low-cost, Compact NP Series Simpler and More Versatile.OMRON s Compact NP Series Connect! Easier connection to OMRON PLCs and Temperature Controllers,

### Ventostat Wall Mount. Telaire Wall Mount CO 2. , Humidity and Temperature Transmitters. GE Measurement & Control. Features:

GE Measurement & Control Ventostat Wall Mount Telaire Wall Mount, Humidity and Temperature Transmitters Features: Patented, Absorption Infrared Gas sensing engine provides high accuracy in a compact low

### Pressure transmitter (2 wire) / switch for continuous or On/Off Control

transmitter (2 wire) / switch for continuous or On/Off Control Indication, monitoring, transmitting and continuous or On/Off control in one device Output signal 4...20 ma, 2-wire for continuous control

### SENSORS. Fiber Optic Sensors - S70. Advanced fiber optic amplifiers for high speed and low contrast applications

Fiber Optic Sensors - S7 Advanced fiber optic amplifiers for high speed and low contrast applications DIN rail mountable models with dual digital displays speed models: 2 μs...5 ms Super high speed models:

### Input signal Maximum Range Accuracy. Thermocouple E -50 to 700 C (-58 to 1832 F) ±1 C

F4 Process Controller Installation and Operation Guide SAFETY ALERTS The symbols below are used on the equipment and throughout this document to draw the user s attention to important operational and safety

### Open channel & partially filled pipe Ultrasonic flow meters

Open channel & partially filled pipe Ultrasonic flow meters SUITABLE FOR PARTIALLY FILLED PIPES AND OPEN CHANNEL FAMOUS PROFESIONAL ULTRASONIC FLOWMETERS MANUFACTURER, ADVANCED TECHNOLOGY, SERIES PARTIALLY

### Rackmount Intelligent Pressure Scanners

Rackmount Intelligent Pressure Scanners NetScanner System 9816/98RK-1 FEATURES ±0.05% Accuracy 100 Hz EU Throughput Autonegotiating 10/100/1G Base T Ethernet - TCP and UDP Protocol Pressure Ranges up to

### VIBRATION MONITORING OF FANS INSTALLED IN ROAD TUNNELS

VIBRATION MONITORING OF FANS INSTALLED IN ROAD TUNNELS CEMB SPA Via Risorgimento, 9 Mandello del Lario (Lc), Italy stm@cemb.com www.cemb.com INSTRUMENT FOR MONITORING VIBRATIONS OF FANS INSTALLED IN ROAD

### Valve Proving System VPS 504 Series S06 (120 VAC) Series S05 (24 VDC)

Valve Proving System VPS 504 Series S06 (120 VAC) Series S05 (24 VDC) VPS 504 S06 (120 VAC) is: CSA Certified File # 1637485 CSA Requirement No. 4-01 (USA) Technical Information Letter R-15 (Canada) UL

### 4-Wire Air Duct Smoke Detector. Protection Systems. A UTC Fire & Security Company. Models DH-60 FEATURES APPLICATION DESCRIPTION SAMPLING TUBES

4-Wire Air Duct Smoke Detector Models DH-60 F-70.05 FEATURES Approvals/Listing UL 268A Listed File S1064 Factory Mutual Approved CSFM Listed #3240-1076:121 NYC MEA Listed #134-95-E City of Los Angeles

### Sanitary Type Electronic Flow Switches

Industries Pharmaceutical Sanitary Type Electronic Flow Switches Food and Beverage Biotechnology Semiconductor Water Treatment S Series Ameritrol, Inc. Instruments and Controls Features No Moving Parts

### MLC 9000+ TM Maximum Productivity and Versatility for Control without Compromise!

Data Sheet Maximum Productivity and Versatility for Control without Compromise! Basic Bus Communication Module DESCRIPTION The Basic Bus Communication Module (BCM) is part of the MLC 9000+ DIN rail mount

### Wireless Data Logger Monitoring Systems

Wireless Data Logger Monitoring Systems Table of Contents About MadgeTech, Inc. Wireless Systems & Components Wireless Data Loggers 2000A Wireless Data Logger Models MadgeTech Cloud Services MadgeTech

### Vroom Hardware manual ver. 1.00 Code 114VROOHWE00. Vroom CANBUS USER INTERFACE WITH LCD GRAPHIC DISPLAY AND WITH TEMPERATURE AND HUMIDITY SENSOR

Vroom CANBUS USER INTERFACE WITH LCD GRAPHIC DISPLAY AND WITH TEMPERATURE AND HUMIDITY SENSOR ENGLISH HARDWARE MANUAL ver. 1.00 CODE 114VROOHWE00 page 1 of 22 Important Important Read these instructions

### FLOW CALCULATOR INSTRUCTION MANUAL MESURES BAMOPHOX 759 26-06-2007 759 M1 02 E MES FLOW CALCULATOR 759-02/1

BAMOPHOX 759 E - M FLOW CALCULATOR INSTRUCTION MANUAL MESURES 22, Rue de la Voie des Bans - Z.I. de la Gare - 95100 ARGENTEUIL Tél : (33) 01 30 25 83 20 - Web : www.bamo.fr Fax : (33) 01 34 10 16 05 -

### TEMON 8-C. Doc. N MO-0370-ING TEMPERATURE MONITOR DEVICE TYPE TEMON 8-C OPERATION MANUAL. Microener - Copyright 2010 FW 2.2 Date 01.12.2008 Rev.

TEMPERATURE MONITOR DEVICE TYPE TEMON 8-C OPERATION MANUAL Microener - Copyright 2010 FW 2.2 Date 01.12.2008 Rev. 0 1. Generality 3 2. Introduction 3 3. Accessories and Options 3 4. Installation 3 5. Connection

### Repeaters and Mimics Data Sheet

Repeaters and Mimics Data Sheet IDR series repeaters and mimics allow vital information from a Notifier fire system to be communicated to points around a building remote from the fire alarm control panel.

### 2/8.10.01. These vacuum switches are used in measuring ranges between -1 and 0 bar.

CATALOGUE > Release 8.5 > Electronic vacuum switches Series SWM Electronic miniature vacuum switches Series SWM These vacuum switches are used in measuring ranges between -1 and 0 bar. Small dimension

### WEATHER STATION FOR SOLAR FARM MONITORING

WEATHER STATION FOR SOLAR FARM MONITORING SOLAR FARM MONITORING SYSTEM: Measures global, horizontal, & background irradiance. Measures wind speed, wind direction, ambient temperature, and relative humidity.

### ISO-9001:2000 certified. EE300 Series Electromechanical Inventory Level Measuring System. Aplus Finetek Sensor, Inc.

ISO-9001:2000 certified EE300 Series Electromechanical Inventory Level Measuring System Aplus Finetek Sensor, Inc. PRODUCT INTRODUCTION OPERATING PRINCIPLE The EE300 series level measuring system is a

### Regards from the Team at Rheology Solutions.

Rheology Solutions is the sole Australian distributor of this product range and we welcome the opportunity of discussing your application requirements. We hope the information you are seeking is contained

### 713 Open Channel Flowmeter and Transmitter. All-in-one package for open channel flow measurement.

and Transmitter All-in-one package for open channel flow measurement. 3.1 NEW Class 1, Division 1 Approved Sensor As our products are continuously improved, we reserve the right to make any change in the

### Preliminary. Dual-line FLOW rate/totalizers. Pr ovu Model PD6200 Analog Input. Pr ovu Model PD6300 Pulse Input. Analog Inputs.

Dual-line FLOW rate/totalizers Pr ovu PD6200 Analog Input ProVu Se r i e s Pr ovu PD6300 Pulse Input Analog Inputs 0-20 ma, 4-20 ma, 0-5 V, 1-5 V, and ±10 V Inputs Isolated 24 VDC @ 200 ma Transmitter

### Interface Level And Point Level Analyzers & Sensors

Interface Level And Point Level Analyzers & Sensors Model 2511 Analyzer Model 2505 System Model 2120 System Royce now offers a complete line of Interface Level and Point Level Analyzers that make it possible

### Unidirectional Transmitter/ Receiver Units Introduction

Unidirectional Transmitter/ Receiver Units Introduction Wireless Input/Output (I/O) connects directly to analog, discrete and pulse transducer signals. The signals are transmitted by radio and either re-created

### WD-AMX Water Detection Controllers

Page 1 of 5 WD-AMX Water Detection Controllers Features: Benefit: LED Status of leak status VFC output Audible alarm Auto or manual reset alarm output Uses an isolated AC signal which prevents oxidation

### API Marine. BMS Presentation. Ballast Measurement Systems

API Marine BMS Presentation Ballast Measurement Systems BMS4 Typical Configuration BMS4 Typical Configuration BMS4 Extended Configuration BMS4 System (24 tanks) Air regulator Terminal block Drain/Air Filter

### Temperature & Humidity SMS Alert Controller

Temperature & Humidity Alert Controller METERS 3 simple steps starting the unit: Insert the SIM card Plug in the sensors connectors Connect the AC power cord. Specifications: AC 90~260V Auto Select Internal

### 40.3. Control Relays and Timers

.3 Contents EASY500/700/800 Intelligent Relays........ EASY/MFD Expansion Modules............ MFD Intelligent Relays................... EASY/MFD Communication Modules....... EASY/MFD Power Supplies, Accessories

### Non-invasive CLAMP-ON ultrasonic flowmeter for liquids

Non-invasive CLAMP-ON ultrasonic flowmeter for liquids Model CU100: non-invasive installation flowmeter, with transducers installed outside the pipe Installation in horizontal or vertical pipe Easy and

### SHARKY 775 COMPACT ENERGY METER ULTRASONIC

APPLICATION The ultrasonic compact energy meter can be used for measuring the energy consumption in heating / cooling application for billing purposes. FEATURES 4 Approval for ultrasonic meter with dynamic

### JNIOR. Overview. Get Connected. Get Results. JNIOR Model 310. JNIOR Model 312. JNIOR Model 314. JNIOR Model 410

The INTEG is an Ethernet I/O (digital, analog) device that monitors and controls a small set of process signals. functions as both basic I/O for integration with another application or system AND as a

### RISH CON - Hz. Salient Features : Application : Product Features: FREQUENCY TRANSDUCER

Application : The RISH CON - Hz transducer is used for frequency measurement. The output signal is proportional to measured frequency and is either load independent DC Current or load independent DC Voltage.

### Quick Connect. quick - simple - efficient. www.g-mw.de

Quick Connect quick - simple - efficient www.g-mw.de Phone: +49 9103 7129-0 Fax: +49 9103 7129-207 Innovative connection technology to plug three single-phase current transformers to multifunctional power

### Zlinx Wireless I/O. Peer-to-Peer and Modbus I/O B&B ELECTRONICS PRODUCT INFORMATION

Modular, Customizable Wire Replacement 128 / 256 Bit AES Encryption Software Selectable RF Transmit Power Software Selectable Over-the-air Data Rate Modbus ASCII /RTU Compatible Wide Operating Temperature

### Chassis, Power Supply, and Element Management

Chassis, Power Supply, and Element Management C-COR CHP Max5000 Converged Headend Platform Accelerate deployment of new, revenue generating advanced services Innovative technology creating converged high-

### O C M III. Open Channel Flow Monitor

O C M III Open Channel Flow Monitor Part 1. General 1.1 Scope A. This section describes the requirements for an ultrasonic flow transmitter with sensor and temperature compensation probe. B. Under this

### Power Supply and Indicator Unit Type 5024-1. Fig. 1 Type 5024-1. Mounting and Operating Instructions EB 9539 EN

Power Supply and Indicator Unit Type 5024-1 Fig. 1 Type 5024-1 Mounting and Operating Instructions EB 9539 EN Edition October 2003 Application 1 Application The Type 5024-1 Power Supply and Indicator Unit

### REGTRONIC OPERATING INSTRUCTIONS

REGTRONIC OPERATING INSTRUCTIONS 1.FEATURES Electrical connection: M12 8-pin connector Preset pressure range 0.05-10 bar with possible full scale regulation. 10 100 mbar adjustable deadband Supply pressure:

### THYRISTORS T1-1PH from 850A to 2700A Specifications Universal 1 Phase Thyristor Unit 55-77-03-62, March 2012

Technical Information THYRISTORS T1-1PH from 850A to 2700A Specifications Universal 1 Phase Thyristor Unit 55-77-03-62, March 2012 GENERAL DESCRIPTION THYRISTORS T1-1PH is a full digital and universal

### The Unique Accelabar Flow Meter

The Unique Accelabar Flow Meter The Accelabar is a new and unique flow meter that combines two differential pressure technologies to produce operating ranges never before attainable in a single flow meter.

### Why a torque measuring system?

T-Sense P R O D U C T B U L L E T I N 660 Optical Torque Measuring Systems Introduction The use of a T-Sense torque measuring system means efficiency improvement, overload protection and prevention of

### Transmitter Interface Program

Transmitter Interface Program Operational Manual Version 3.0.4 1 Overview The transmitter interface software allows you to adjust configuration settings of your Max solid state transmitters. The following

### Thermal flow sensors TA10-ZG8c and TA10-ZG9c for exact and stable, long-term measuring of lower flow velocities (Laminar Flow)

Thermal flow sensors TA10-ZG8c and TA10-ZG9c for exact and stable, long-term measuring of lower flow velocities (Laminar Flow) Drawing 8c Drawing 9c U391_TA10C_DoP_e_150421 www.hoentzsch.com 1 of 5 Measured

### WEA-Base. User manual for load cell transmitters. UK WEA-Base User manual for load cell transmitters Version 3.2 UK

WEA-Base User manual for load cell transmitters 1 Contents 1. Technical data... 3 2. Assembly... 4 2.1 Power supply... 4 2.2 Load cells... 4 2.3 RS-485... 4 2.4 Relays... 5 2.5 Digital input... 5 2.6 Analogue

T-SERIES INDUSTRIAL INCLINOMETER ANALOG INTERFACE T-Series industrial inclinometers are compact high performance sensors used to determine inclination in roll and pitch axes with excellent precision and

### M2M The measure of efficiency

M2M The measure of efficiency The importance of control Measurement and advanced analysis of electrical parameters To make equipment as efficient as possible, it is essential to know the details of its

### SLIM LINE FUME HOOD MONITOR

SLIM LINE FUME HOOD MONITOR Phoenix Controls FHM420 Fume Hood Monitor is sized to fit any fume hood and includes icons for simple and universal operation. The FHM420 slim line monitor works in conjunction

### Flow Monitoring System

FMS-3 Product Data Sheet PS-00028 April 1997 Flow Monitoring System FMS-3 Flow Monitoring System The Micro Motion FMS-3 Flow Monitoring System is a microprocessor-based batch controller with rate, total,

### Introduction. Refrigerant Leak Detecting System. Feature and Benefits

PSC European Refrigeration Controls Catalogue Catalog Section 9 Product Bulletin PD-GAS-E Refrigerant Leak Detecting System Introduction This range of refrigerant leak detecting systems is designed for

### Everything you need for protection scheme testing

Power System Simulator for Testing Protection Relays and Schemes Everything you need for protection scheme testing The is the only instrument with the high power, flexibility and software to perform full

### T7560A,B,C Digital Wall Module

T7560A,B,C Digital Wall Module HONEYWELL EXCEL 5000 OPEN SYSTEM BEFORE INSTALLATION All wiring must comply with local electrical codes and ordinances or as specified on installation wiring diagrams. Digital

### Typical Aluminum GFM Mass Flow Meter

MASS FLOW METERS Design Features n Rigid metallic construction. n Maximum pressure of 1000 psig (70 bars). n Leak integrity 1 x 10-9 of helium. n NIST traceable certification. n Built-in tiltable LCD readout.

### 1.44 kw Programmable DC Power Supplies XLN Series

Data sheet 1.44 kw Programmable DC Power Supplies XLN Series New Family of High Density System Power Supplies The B&K Precision XLN series are compact, programmable, single-output DC power supplies, suitable

### red-y smart series product information Thermal Mass Flow Meters and Controllers for Gases

red-y smart series product information Thermal Mass Flow Meters and Controllers for Gases Reliable and accurate: Thermal Mass Flow Meters and Controllers Reliable technology and industry standard interfaces

### Measurement. SITRANS F US Inline. Transmitter FUS080/FUE080 4/219

Siemens AG 2010 Flow Measurement Overview The transmitter is available in an IP67/NEMA X/6 enclosure and is designed for use in the flowmeters series: SONOKIT (1- or 2-track) FUS380 (2-track) FUE380 (2-track)

### Universal Voltage Terminal Photoelectric E3JM

Universal Voltage Terminal Photoelectric Universal AC/DC Photoelectric Sensor with Built-in Timer Easy-to-wire terminal block Built-in timer offers selectable ON-delay, OFF-delay and one-shot Relay or

### User s Manual Laminar Flow Elements

501:440-13 User s Manual Laminar Flow Elements 10920 Madison Avenue Cleveland, Ohio 44102 Tel: 216.281.1100 email:meriam@meriam.com Web:www.meriam.com 1 TABLE OF CONTENTS Subject Page Introduction... 3

### OXIGENO DISUELTO. Dissolved Oxygen. Install & forget. Sistema autolimpiable. Autoclean system

Sistema autolimpiable Dissolved Oxygen Autoclean system Applications: - water treatment - activated sludge - de-nitrification - fish pond Install & forget For special applications, the D.O monitor can

### Technical data. General specifications. Indicators/operating means. 30 Hz Multiplex operation 30 Hz / n, n = number of sensors, n 5

Model Number Single head system Features Parameterization interface for the application-specific adjustment of the sensor setting via the service program ULTRA 000 programmable switch outputs Hysteresis

### Rosemount 333 HART Tri-Loop

Product Data Sheet September 2014 00813-0100-4754, Rev GA Rosemount 333 HART Tri-Loop HART-to-Analog Signal Converter Convert a digital HART signal into three additional analog signals Easy to configure

### Problem / Solution. The problem:

Problem / Solution Solar Photovoltaic (PV) technology has shown significant development in recent years and this has led to many UK homeowners investing in Solar PV installations. Solar electricity is

### Control Devices Catalogue. Control Devices. Preferred and Standard Products

Control Devices Catalogue Control Devices Preferred and Standard Products KOP.J Timer Electronic 22.5 mm Characteristics Mounting KOP.J 22.5 mm mono or multifunction high immunity to interference surface

### RM17TE 183...528 V AC. Main. Product or component type. Product specific application. Relay monitored parameters 250 V DC 5 A DC

Characteristics multifunction control relay RM17-TE - range 183..528 V AC Complementary Reset time Maximum switching voltage Minimum switching current Maximum switching current [Us] rated supply voltage

### TECHNICAL DATASHEET #TD1404AX PWM CONTROLLED SOLENOID DRIVER

TECHNICAL DATASHEET #TD1404AX PWM CONTROLLED SOLENOID DRIVER (PWM Input, 1.2A or 2A Output, Metal Box or PCB) PCB Board - P/N: PWMC-PCB-2A, PWMC-PCB-1.2A Packaged Driver (metal box with 1.5 m (5 ft.) cable)

### In-Line Electronic Flow Switches

In-Line Electronic Flow Switches Industries Petrochemical Refining Oil Production Water Treatment Pharmaceutical Food and Beverage Pulp and Paper Power Production Gas Processing Mining Biotechnology Semiconductor

### CruzPro VAF110. AC Volts, Amps, Frequency, kw Monitor

CruzPro VAF110 AC Volts, Amps, Frequency, kw Monitor Table of Contents Introduction............................ 3 Specifications........................... 4 Installation..............................5

### Rittal TopTherm fan-and-filter unit

Rittal TopTherm fan-and-filter unit System Climate Control More pressure. Higher air flow rates. 2 Assembly without tools Straightforward assembly, exchange and maintenance with no need for tools of any

### VSO MAX HP High Flow Proportional Valve

VSO MAX HP High Flow Proportional Valve Typical Applications Ventilators Insufflators Anesthesia Delivery & Monitors Pressure & Flow Control Mass Flow Control Product Specifications Physical Properties

### MILLENIUM INSTALLATION MANUAL NTR 735 A. Simple Automation Control Module (MAS)

MILLENIUM Table of contents 1. INTRODUCTION 1 2. HARDWARE DESCRIPTION 2 3. INSTALLATION 5 4. CONNECTION 6 5. USER SAFETY AND PROTECTION OF EQUIPMENT 8 1. Introduction The MILLENIUM series has been designed

### SUMMARY of PRODUCTS. HVAC Systems Control Airflow, Space, & Building Pressurization. Accurate airflow measurement for demanding applications

SUMMARY of PRODUCTS HVAC Systems Control Airflow, Space, & Building Pressurization Accurate airflow measurement for demanding applications Airflow Measuring Stations FAN-E AIRFLOW MEASURING STATION. The

### USER S GUIDE. EE210 - Humidity and Temperature Transmitter for demanding Climate Control Applications GENERAL CAUTION CONNECTION DIAGRAM

USER S GUIDE EE210 - Humidity and Temperature Transmitter for demanding Climate Control Applications GENERAL The EE210 transmitter, available for wall or duct mounting, is designed for the highly accurate

### Druck DPI 104. GE Industrial Sensing. Digital pressure indicator User manual - K394

GE Industrial Sensing 99 Washington Street Melrose, MA 02176 Fax 781-665-0780 TestEquipmentDepot.com Druck DPI 104 Digital pressure indicator User manual - K394 A1.1 A1.2 B1 1 A2 B2 2 3 4 5 6 A1.3 8 7

### Product Information Sheet UH36 Hi Pot Tester

Product Information Sheet UH36 ETL Prüftechnik GmbH Telefon: +49 711 83 99 39-0 E-Mail: info@etl-prueftechnik.de Carl-Peters-Straße 23 D-70825 Korntal-Münchingen Telefax: +49 711 83 99 39-9 Internet: www.etl-prueftechnik.de

### DUCT SMOKE DETECTOR D4120

Purpose/Application National and local safety standards and codes recognize the ability of air duct systems to transfer smoke, toxic gases, and flame from area to area. Sometimes smoke can be of such quantity

### UDSA - UDSBD unit heater

APPLICATIONS Heating >> FACTORIES >> WAREHOUSES >> HALLS >> SHOWROOMS >> GREENHOUSES >> RETAIL >>... UDSA - UDSBD unit heater www.reznor.eu V3 gas fired unit heater Reznor V3 unit heaters are one of the

### Fiber Optic Selector Guide for Analog & Digital Data Links, Contact Closures & Multiplexers

Fiber Optic Selector Guide for Analog & Digital Data Links, Contact Closures & Multiplexers Analog Data Links Multi-Channel Contact Closures Multiplexers Digital Data Links - Contact Closures Power Supply