# Chapter 10. Flow Rate. Flow Rate. Flow Measurements. The velocity of the flow is described at any

Size: px
Start display at page:

Download "Chapter 10. Flow Rate. Flow Rate. Flow Measurements. The velocity of the flow is described at any"

Transcription

1 Chapter 10 Flow Measurements Material from Theory and Design for Mechanical Measurements; Figliola, Third Edition Flow Rate Flow rate can be expressed in terms of volume flow rate (volume/time) or mass flow rate (mass/time). Flow rate through pipe or duct can be defined using concept of control volume, where the amount of flow through volume in a given period of time is the flow rate. Figliola, 000 Flow Rate The velocity of the flow is described at any ^ ^ ^ point by: U( x, r, ϑ) = ue x + v e r + w e ϑ u, v, and w are scalar velocity magnitudes and ^ ^ ^ e x, e r, e ϑ are unit vectors. By conservation of mass, the amount of fluid density ρ that passes through control volume (CV) and accumulates within CV must be equal to 0. 1

2 Flow Rate o o For steady flows, m in = m out o where = ρ ^ m u* nda o A m is mass flow rate through area A For isometric flows, consider liquids as incompressible. ^ Q in =Q out where Q = u * nda A Where Q is the volumetric flow rate Flow Rate In pipe flow, if we know u(r,ϑ) at position x, in steady, incompressible flow, we can estimate the average velocity and, thus, average flow. r = 1 Q π u( r, ϑ) rdϑdr 0 0 r 1 is the pipe radius. Flow Rate If the average velocity is known, Q = UA. A method that will measure steady flow rates must be sensitive to either average mass flux ρu or U average velocity U.

3 Flow Rate The flow in a pipe is either laminar, turbulent, or transitional. 4Q = Ud1/ v = Using Reynolds number, Re πd 1v Turbulent flow: R e > 4000 v = kinematic viscosity Laminar flow: R e < 000 d 1 = diameter for circular pipe or hydraulic diameter (4r h ) for non-circular cross- sections. r h = wetted area/ wetted parameter Note: Stay away from transitions in flow Volume Flow Rate The direct implementation of equations for estimating the volume flow rate through a duct requires measurement of the velocity at points along several cross-sections of a flow control surface. This procedure is most often used for the one-time verification or calibration of system flow rates. Volume Flow Rate In using this technique in circular pipes, a number of discrete measuring positions are chosen along m flow cross-sections (radii) spaces at 360 o /m apart. A velocity probe is traversed along each flow cross-section with readings taken at each measurement position. 3

4 Figliola, 000 Obstruction Flow Meters Three types of meters that use obstruction: 1. Orifice plate. Venturi 3. Flow Nozzle Obstruction Flow Meters Obstruction flow meters rely on the Bernoulli effect. When a restriction in flow occurs, there is an increase in velocity (conservation of mass). When velocity goes up, pressure goes down. Therefore, there is a measurable pressure drop across the orifice/obstruction. p = p 1 p Q (p 1 p ) n n = 1 (laminar) n = ½ (turbulent) 4

5 Obstruction Meters Placed in line with pipe Due to vena contracta effect Due to inability of flow to instantaneously Figliola, 000 adjust to change in cross-section. Assume: 1. No external heat energy added. No shaft work done 3. Steady flow 4. One-dim flow Energy Equation p / γ + U / g = p / γ + U / g + h 1 L From conservation of mass, ū 1 = ū A /A 1 For incompressible flow, 1 1 Q ] 1 / I = UA = ( A /[ 1 ( A / A1) ] ) [ ( p1 p) / ρ + ghl1 This equation can be simplified using coefficients Obstruction Meters When the flow area changes abruptly, the effective flow area immediately downstream of the alteration will not necessarily be the same as the pipe flow area. This effect is brought about by an inability of a fluid to expand immediately upon encountering an expansion as a result of the inertia of each fluid particle. This forms a central core flow bounded by regions of slower moving recirculating eddies. 5

6 Obstruction Meters As a consequence, the pressure sensed with pipe wall taps located within the vena contracta region will correspond to the higher moving velocity within the vena contracta of unknown flow area, A. The unknown vena contracta area will be accounted for by introducing a contraction coefficient, C c. Energy Equation C c = A /A 0 contraction coefficient Q 1 / I = CcA0 /[ 1 ( CcA0 / A1) ] ( P) / ρ + gh1 C f = frictional head loss 1 / QI = CfCcA 0 /[ 1 ( CcA0 / A1) ] ( P) / ρ C = discharge coefficient combines C c and C f Energy Equation QI = CEA p / ρ where E is the velocity of the approach factor. 4 1 / E = 1/[1 ( Ao / A1 ) ] 1 / = 1/(1 β ) where β = d 0 /d 1 CE is often represented as flow coefficient K o The value of C varies depending on the flow, Reynolds number and the β ratio. 6

7 Energy Equation Because the magnitude of vena contracta and head loss vary along the length of meter the location of the pressure taps is critical. The flow behavior of orifice plate, venturi, and flow nozzle meters have been studied extensively and are therefore used without calibration through use of standard flow factor tables and hand books. Orifice Meter An orifice meter consists of a circular plate, containing a hole (orifice), which is inserted into a pipe such that the orifice is concentric with the pipe inside diameter. Several variations in the orifice design exist, but the square-edged orifice is common. Installation is simplified by housing the orifice plate between two pipe flanges. Orifice Meter With this technique, an orifice plate is interchangeable with others of different β values The simplicity of the installation and orifice design allows for a range of β values to be maintained on hand at modest expense. 7

8 Figliola, 000 Venturi Meter A venturi meter consists of a smooth converging contraction to a narrow throat followed by a shallow diverging section. The standard venturi can utilize either a 15 o or 7 o divergent section. The meter is installed between two flanges intended for this purpose. Pressure is sensed between a location upstream of the throat and a location at the throat such that equations are used with values for both A and β based on the throat diameter. Figliola, 000 8

9 Flow Nozzles A flow nozzle consists of a gradual contraction to a narrow throat. It needs less installation space than a venturi meter and has ~80% of the initial cost. A common form for the nozzle is the ASME long radius nozzle, in which the nozzle contraction is that of the quadrant of an ellipse with major axis aligned with the flow axis. Flow Nozzles The nozzle is typically installed in line, but can also be used at the inlet to and the outlet from a plenum or reservoir or at the outlet of a pipe. Pressure taps are usually located at one pipe diameter upstream of the nozzle inlet and at the nozzle throat by using either wall or throat taps. The flow rate is determined from equations with values for A and β based on the throat diameter. Figliola, 000 9

10 Sonic Nozzles Sonic nozzles are used to meter and to control the flow rate of compressible gases. They may take the form of any of the previously described obstruction meters. If the gas flow rate through and obstruction meter becomes sufficiently high, the sonic condition will be achieved at the meter throat. At the sonic condition, the gas velocity will equal the acoustic wave speed (speed of sound) of the gas. At that point, the throat is considered to be choked and the mass flow rate through the throat will be at a maximum for the given inlet conditions regardless of any further increase in pressure drop across the meter. Laminar Flow Elements Laminar flow elements force the fluid flow to be laminar and then use expressions for laminar flow to interpret the Pressure drop. Figliola, 000 Laminar Flow p 1 / γ + U1 / g = p / γ + U / g + hl1 Darcy-Weisback for common flow: hl1 = f( L / d1)( U1 / g) when f = 64 /(Re d1 ) f is the friction factor Q U1( πd1 / 4) = U( πd / 4) for d 1 =d U = Thusp1 p = γh L 1 using U = 4Q / πd1 = 1 U 4 Q = ( πd 1 /18µ)( p1 p / L) L = length of tube section 10

11 Advantages/Disadvantages Advantages: 1. High sensitivity at low flow rates. Can measure flow in either direction 3. Wide flow range 4. Indicated average flow in pulsing flow condition Disadvantages: 1. Susceptible to plugging, only use in clean fluids. All measured pressure drop, remains a system drop Insertion Volume Flow Meters The largest group of meters based on a phenomenon that is actually sensitive to the average velocity across a control surface of known area. Several of these designs are included in the discussion that follows. Electromagnetic Flow Meters The operating principle of an electromagnetic flow meter is based on the fundamental principle that an emf of electric potential, E, is induced in a conductor of length, L, which moves with a velocity, U, through a magnetic field of magnetic flux, B. 11

12 Figliola, 000 Electromagnetic Flow Meters The electromagnetic flow meter comes commercially as a packaged flow device, which is installed directly in line and connected to an external electronic output unit. Units are available using either permanent magnets, called dc units, or variable flux strength electromagnets, called ac units. Electromagnetic Flow Meters The electromagnetic flow meter has a low pressure loss associated due to its open tube, no obstruction design, and it is suitable for installations that can tolerate only a small pressure drop. This absence of internal parts is very attractive for metering corrosion and dirty fluids. The operating principle is independent of fluid density and viscosity, responding only to average velocity and there is no difficulty with measurements in either laminar or turbulent flows, provided that the velocity profile is reasonably symmetrical. 1

13 Vortex Shedding Meters Vortex shedding is a natural phenomenon where alternating vortices are shed in the wake of a body at a frequency that depends on the flow velocity past the body. The vortices formed on opposite sides of the body are carried downstream in the body s wake forming a vortex street, each vortex having an opposite sign of rotation. The vortex shedding phenomenon is used to sense average velocity in pipe flows in a vortex flow meter. Vortex Shedding Meters Figliola, 000 Vortex Shedding Meters The lower flow rate limit on vortex meters appears to be at Reynolds numbers near 10,000. This can be a problem in metering high-viscosity pipe flows. Density variations affect the strength of the shed vortex and this places a lower limit on fluid density which is based on the sensitivity of the vortex shedding detection equipment. The meter has no moving parts and relatively low pressure compared to obstruction meters. 13

14 Turbine Meters The exchange of momentum between the flow and a rotor turns the rotor at a rotational speed that is proportional to the flow rate. A rotor is encased in a bored housing that is inserted in the flow stream to be metered. Turbine meters have a low-pressure drop and are very accurate. Turbine Meters They are exceptionally repeatable, but are restricted to clean fluids because of possible fouling of their rotating parts. They are sensitive to temperature changes which affect fluid viscosity. Figliola,

### FLUID FLOW Introduction General Description

FLUID FLOW Introduction Fluid flow is an important part of many processes, including transporting materials from one point to another, mixing of materials, and chemical reactions. In this experiment, you

### A LAMINAR FLOW ELEMENT WITH A LINEAR PRESSURE DROP VERSUS VOLUMETRIC FLOW. 1998 ASME Fluids Engineering Division Summer Meeting

TELEDYNE HASTINGS TECHNICAL PAPERS INSTRUMENTS A LAMINAR FLOW ELEMENT WITH A LINEAR PRESSURE DROP VERSUS VOLUMETRIC FLOW Proceedings of FEDSM 98: June -5, 998, Washington, DC FEDSM98 49 ABSTRACT The pressure

### Laminar and Turbulent flow. Flow Sensors. Reynolds Number. Thermal flow Sensor. Flow and Flow rate. R = Mass Flow controllers

Flow and Flow rate. Laminar and Turbulent flow Laminar flow: smooth, orderly and regular Mechanical sensors have inertia, which can integrate out small variations due to turbulence Turbulent flow: chaotic

### Grant Agreement No. 228296 SFERA. Solar Facilities for the European Research Area SEVENTH FRAMEWORK PROGRAMME. Capacities Specific Programme

Grant Agreement No. 228296 SFERA Solar Facilities for the European Research Area SEVENTH FRAMEWORK PROGRAMME Capacities Specific Programme Research Infrastructures Integrating Activity - Combination of

### Head Loss in Pipe Flow ME 123: Mechanical Engineering Laboratory II: Fluids

Head Loss in Pipe Flow ME 123: Mechanical Engineering Laboratory II: Fluids Dr. J. M. Meyers Dr. D. G. Fletcher Dr. Y. Dubief 1. Introduction Last lab you investigated flow loss in a pipe due to the roughness

### 4.What is the appropriate dimensionless parameter to use in comparing flow types? YOUR ANSWER: The Reynolds Number, Re.

CHAPTER 08 1. What is most likely to be the main driving force in pipe flow? A. Gravity B. A pressure gradient C. Vacuum 2.What is a general description of the flow rate in laminar flow? A. Small B. Large

### Flow Measurement in Pipes and Ducts. Flow Measurement in Pipes and Ducts, Course #503. Presented by:

Flow Measurement in Pipes and Ducts, Course #503 Presented by: PDH Enterprises, LLC PO Box 942 Morrisville, NC 27560 www.pdhsite.com This course is about measurement of the flow rate of a fluid flowing

### Fluid Flow Instrumentation

Fluid Flow Instrumentation In the physical world, mechanical engineers are frequently required to monitor or control the flow of various fluids through pipes, ducts and assorted vessels. This fluid can

### Michael Montgomery Marketing Product Manager Rosemount Inc. Russ Evans Manager of Engineering and Design Rosemount Inc.

ASGMT / Averaging Pitot Tube Flow Measurement Michael Montgomery Marketing Product Manager Rosemount Inc. Russ Evans Manager of Engineering and Design Rosemount Inc. Averaging Pitot Tube Meters Introduction

### Basic Equations, Boundary Conditions and Dimensionless Parameters

Chapter 2 Basic Equations, Boundary Conditions and Dimensionless Parameters In the foregoing chapter, many basic concepts related to the present investigation and the associated literature survey were

### Chapter 8: Flow in Pipes

Objectives 1. Have a deeper understanding of laminar and turbulent flow in pipes and the analysis of fully developed flow 2. Calculate the major and minor losses associated with pipe flow in piping networks

### Air Flow Measurements

ME-EM 30 ENERGY LABORATORY Air Flow Measurements Pitot Static Tube A slender tube aligned with the flow can measure local velocity by means of pressure differences. It has sidewall holes to measure the

### INTRODUCTION TO FLUID MECHANICS

INTRODUCTION TO FLUID MECHANICS SIXTH EDITION ROBERT W. FOX Purdue University ALAN T. MCDONALD Purdue University PHILIP J. PRITCHARD Manhattan College JOHN WILEY & SONS, INC. CONTENTS CHAPTER 1 INTRODUCTION

### High Speed Aerodynamics Prof. K. P. Sinhamahapatra Department of Aerospace Engineering Indian Institute of Technology, Kharagpur

High Speed Aerodynamics Prof. K. P. Sinhamahapatra Department of Aerospace Engineering Indian Institute of Technology, Kharagpur Module No. # 01 Lecture No. # 06 One-dimensional Gas Dynamics (Contd.) We

### du u U 0 U dy y b 0 b

BASIC CONCEPTS/DEFINITIONS OF FLUID MECHANICS (by Marios M. Fyrillas) 1. Density (πυκνότητα) Symbol: 3 Units of measure: kg / m Equation: m ( m mass, V volume) V. Pressure (πίεση) Alternative definition:

### Flow Measurement Options for Pipeline and Open Channel Flow

Flow Measurement Options for Pipeline and Open Channel Flow October 2013 Presented by Molly Skorpik - 2013 Montana Association of Dam and Canal Systems Conference Irrigation Training and Research Center

### Lecture 6 - Boundary Conditions. Applied Computational Fluid Dynamics

Lecture 6 - Boundary Conditions Applied Computational Fluid Dynamics Instructor: André Bakker http://www.bakker.org André Bakker (2002-2006) Fluent Inc. (2002) 1 Outline Overview. Inlet and outlet boundaries.

### Energy and Flow Measurement for Hydronic Systems

Energy and Flow Measurement for Hydronic Systems Presented By: George Szyszko Applications Consultant MCR for ONICON Incorporated Why Measure Thermal Energy? To control something, you must first measure

### ABB Flowmeters Review of industrial processing flowmeters

White paper ABB Flowmeters Review of industrial processing flowmeters Understanding flowmeters and their pros and cons By Ron DiGiacomo, ABB Measurement Products One of the most critical measurements in

### Experiment 3 Pipe Friction

EML 316L Experiment 3 Pipe Friction Laboratory Manual Mechanical and Materials Engineering Department College of Engineering FLORIDA INTERNATIONAL UNIVERSITY Nomenclature Symbol Description Unit A cross-sectional

### Differential Relations for Fluid Flow. Acceleration field of a fluid. The differential equation of mass conservation

Differential Relations for Fluid Flow In this approach, we apply our four basic conservation laws to an infinitesimally small control volume. The differential approach provides point by point details of

### Heat Transfer Prof. Dr. Ale Kumar Ghosal Department of Chemical Engineering Indian Institute of Technology, Guwahati

Heat Transfer Prof. Dr. Ale Kumar Ghosal Department of Chemical Engineering Indian Institute of Technology, Guwahati Module No. # 04 Convective Heat Transfer Lecture No. # 03 Heat Transfer Correlation

### Applied Fluid Mechanics

Applied Fluid Mechanics Sixth Edition Robert L. Mott University of Dayton PEARSON Prentkv Pearson Education International CHAPTER 1 THE NATURE OF FLUIDS AND THE STUDY OF FLUID MECHANICS 1.1 The Big Picture

### Fluids and Solids: Fundamentals

Fluids and Solids: Fundamentals We normally recognize three states of matter: solid; liquid and gas. However, liquid and gas are both fluids: in contrast to solids they lack the ability to resist deformation.

### Liquid Flowmeters A Guide for Selecting a Flowmeter for Pressurized Systems

Liquid Flowmeters A Guide for Selecting a Flowmeter for Pressurized Systems Selecting a meter requires a first-rate technical assessment of the pressurized system and a thorough evaluation of the capabilities

### Using meters to measure steam flow

Page 1 of 6 April 1998 Plant Engineering SELECTING NEW TECHNOLOGY: PROBLEM SOLVING Using meters to measure steam flow Jesse L. Yoder, Ph.D., Senior Analyst, Automation Research Corp., Dedham, MA Key Concepts

### Air Flow Optimization via a Venturi Type Air Restrictor

, July 3-5, 013, London, U.K. Air Flow Optimization via a Venturi Type Air Restrictor Anshul Singhal, Mallika Parveen, Member, IAENG Abstract The aim of this project is to create a flow restriction device

### Advanced Differential Pressure Flowmeter Technology V-CONE FLOW METER TECHNICAL BRIEF

Advanced Differential Pressure Flowmeter Technology V-CONE FLOW METER TECHNICAL BRIEF Table of Contents Section 1 - General Introduction 1.1 1 Principles Of Operation 1.2 1 Reshaping The Velocity Profile

### Lecture 5 Hemodynamics. Description of fluid flow. The equation of continuity

1 Lecture 5 Hemodynamics Description of fluid flow Hydrodynamics is the part of physics, which studies the motion of fluids. It is based on the laws of mechanics. Hemodynamics studies the motion of blood

### Energy Systems Engineering Technology

College of Technology Instrumentation and Control Module # 4 Flow Measurement Document Intent: The intent of this document is to provide an example of how a subject matter expert might teach Flow Measurement.

### Chapter 13 OPEN-CHANNEL FLOW

Fluid Mechanics: Fundamentals and Applications, 2nd Edition Yunus A. Cengel, John M. Cimbala McGraw-Hill, 2010 Lecture slides by Mehmet Kanoglu Copyright The McGraw-Hill Companies, Inc. Permission required

### CO 2 41.2 MPa (abs) 20 C

comp_02 A CO 2 cartridge is used to propel a small rocket cart. Compressed CO 2, stored at a pressure of 41.2 MPa (abs) and a temperature of 20 C, is expanded through a smoothly contoured converging nozzle

### Chapter 5 MASS, BERNOULLI AND ENERGY EQUATIONS

Fluid Mechanics: Fundamentals and Applications, 2nd Edition Yunus A. Cengel, John M. Cimbala McGraw-Hill, 2010 Chapter 5 MASS, BERNOULLI AND ENERGY EQUATIONS Lecture slides by Hasan Hacışevki Copyright

### Theory overview of flow measurement using differential pressure devices based on ISO-5167 standard.

Theory overview of flow measurement using differential pressure devices based on ISO-567 standard. rian FL40 flow computer description. Flow Cad software users manual. Technical note, Differential pressure

### Open channel flow Basic principle

Open channel flow Basic principle INTRODUCTION Flow in rivers, irrigation canals, drainage ditches and aqueducts are some examples for open channel flow. These flows occur with a free surface and the pressure

### Basic Principles in Microfluidics

Basic Principles in Microfluidics 1 Newton s Second Law for Fluidics Newton s 2 nd Law (F= ma) : Time rate of change of momentum of a system equal to net force acting on system!f = dp dt Sum of forces

### Flow Sensors. - mass flow rate - volume flow rate - velocity. - stream line parabolic velocity profile - turbulent vortices. Methods of measurement

Flow Sensors Flow - mass flow rate - volume flow rate - velocity Types of flow - stream line parabolic velocity profile - turbulent vortices Methods of measurement - direct: positive displacement (batch

### FLOW MEASUREMENT 2001 INTERNATIONAL CONFERENCE DERIVATION OF AN EXPANSIBILITY FACTOR FOR THE V-CONE METER

FLOW MEASUREMENT 200 INTERNATIONAL CONFERENCE DERIVATION OF AN EXPANSIBILITY FACTOR FOR THE V-CONE METER Dr D G Stewart, NEL Dr M Reader-Harris, NEL Dr R J W Peters, McCrometer Inc INTRODUCTION The V-Cone

### Applied Fluid Mechanics

Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and

### Experiment (13): Flow channel

Introduction: An open channel is a duct in which the liquid flows with a free surface exposed to atmospheric pressure. Along the length of the duct, the pressure at the surface is therefore constant and

### Experiment # 3: Pipe Flow

ME 05 Mechanical Engineering Lab Page ME 05 Mechanical Engineering Laboratory Spring Quarter 00 Experiment # 3: Pipe Flow Objectives: a) Calibrate a pressure transducer and two different flowmeters (paddlewheel

### measurement, but almost any pipe elbow can be calibrated Elbow meters are not as potentially accurate as venturi,

Lecture 14 Flow Measurement in Pipes I. Elbow Meters An elbow in a pipe can be used as a flow measuring device much in the same way as a venturi or orifice plate The head differential across the elbow

### Fluid Mechanics Prof. S. K. Som Department of Mechanical Engineering Indian Institute of Technology, Kharagpur

Fluid Mechanics Prof. S. K. Som Department of Mechanical Engineering Indian Institute of Technology, Kharagpur Lecture - 20 Conservation Equations in Fluid Flow Part VIII Good morning. I welcome you all

### NUMERICAL ANALYSIS OF THE EFFECTS OF WIND ON BUILDING STRUCTURES

Vol. XX 2012 No. 4 28 34 J. ŠIMIČEK O. HUBOVÁ NUMERICAL ANALYSIS OF THE EFFECTS OF WIND ON BUILDING STRUCTURES Jozef ŠIMIČEK email: jozef.simicek@stuba.sk Research field: Statics and Dynamics Fluids mechanics

### THEORETICAL UNCERTAINTY OF ORIFICE FLOW MEASUREMENT page 1 THEORETICAL UNCERTAINTY OF ORIFICE FLOW MEASUREMENT INTRODUCTION COEFFICIENT OF DISCHARGE

THEORETICAL UNCERTAINTY OF ORIFICE FLOW MEASUREMENT page 1 DANIEL MEASUREMENT AND CONTROL WHITE PAPERS THEORETICAL UNCERTAINTY OF ORIFICE FLOW MEASUREMENT www.daniel.com INTRODUCTION Orifice meters are

### FLUID FLOW STREAMLINE LAMINAR FLOW TURBULENT FLOW REYNOLDS NUMBER

VISUAL PHYSICS School of Physics University of Sydney Australia FLUID FLOW STREAMLINE LAMINAR FLOW TURBULENT FLOW REYNOLDS NUMBER? What type of fluid flow is observed? The above pictures show how the effect

### Open Channel Flow. M. Siavashi. School of Mechanical Engineering Iran University of Science and Technology

M. Siavashi School of Mechanical Engineering Iran University of Science and Technology W ebpage: webpages.iust.ac.ir/msiavashi Email: msiavashi@iust.ac.ir Landline: +98 21 77240391 Fall 2013 Introduction

### Application of the Orifice Meter for Accurate Gas Flow Measurement page 1. Application of the Orifice Meter for Accurate Gas Flow Measurement.

Application of the Orifice Meter for Accurate Gas Flow Measurement page 1 DANIEL MEASUREMENT AND CONTROL WHITE PAPER Application of the Orifice Meter for Accurate Gas Flow Measurement www.daniel.com Summary

### 191: Calibration and Standards in Flow Measurement. Richard Paton National Engineering Laboratory, Scotland, UK 1 GENERAL PRINCIPLES

191: Calibration and Standards in Measurement Richard Paton National Engineering Laboratory, Scotland, UK 1 General Principles 1 2 Gravimetric Calibration of Liquid meters 2 3 Volumetric Calibration of

### Chapter 28 Fluid Dynamics

Chapter 28 Fluid Dynamics 28.1 Ideal Fluids... 1 28.2 Velocity Vector Field... 1 28.3 Mass Continuity Equation... 3 28.4 Bernoulli s Principle... 4 28.5 Worked Examples: Bernoulli s Equation... 7 Example

### RITTER Multiple Sonic Nozzle Calibration System

RITTER Multiple Sonic Nozzle Calibration System »Wouldn t it be great to eliminate doubtful measurement results by using proven measurement technology? Providing the most precise results ensures and increases

### Dynamic Process Modeling. Process Dynamics and Control

Dynamic Process Modeling Process Dynamics and Control 1 Description of process dynamics Classes of models What do we need for control? Modeling for control Mechanical Systems Modeling Electrical circuits

### Dimensional Analysis

Dimensional Analysis An Important Example from Fluid Mechanics: Viscous Shear Forces V d t / / / / / / / / / / / / / / / / / / / / / / / / / / / / / / / / / Ƭ = F/A = μ V/d More generally, the viscous

### Urban Hydraulics. 2.1 Basic Fluid Mechanics

Urban Hydraulics Learning objectives: After completing this section, the student should understand basic concepts of fluid flow and how to analyze conduit flows and free surface flows. They should be able

### Civil Engineering Hydraulics Open Channel Flow. Adult: Where s your costume? What are you supposed to be?

Civil Engineering Hydraulics Calvin: Trick or treat! Adult: Where s your costume? What are you supposed to be? Calvin: I m yet another resource-consuming kid in an overpopulated planet, raised to an alarming

### Aeration Air & Digester Gas Flow Metering Using Thermal Mass Technology. HWEA 2011 Conference Craig S. Johnson

Aeration Air & Digester Gas Flow Metering Using Thermal Mass Technology HWEA 2011 Conference Craig S. Johnson Presentation Overview Introduction Aeration Air & Digester gas challenges Gas flow metering

### Distinguished Professor George Washington University. Graw Hill

Mechanics of Fluids Fourth Edition Irving H. Shames Distinguished Professor George Washington University Graw Hill Boston Burr Ridge, IL Dubuque, IA Madison, Wl New York San Francisco St. Louis Bangkok

### Control ball valves for severe services. Author: Michele Ferrante, PARCOL S.p.A., Italy

Control ball valves for severe services Author: Michele Ferrante, PARCOL S.p.A., Italy Control valves are primarily classified according to the type of their obturator motion which can be linear or rotary.

### A MTR FUEL ELEMENT FLOW DISTRIBUTION MEASUREMENT PRELIMINARY RESULTS

A MTR FUEL ELEMENT FLOW DISTRIBUTION MEASUREMENT PRELIMINARY RESULTS W. M. Torres, P. E. Umbehaun, D. A. Andrade and J. A. B. Souza Centro de Engenharia Nuclear Instituto de Pesquisas Energéticas e Nucleares

### Pressure drop in pipes...

Pressure drop in pipes... PRESSURE DROP CALCULATIONS Pressure drop or head loss, occurs in all piping systems because of elevation changes, turbulence caused by abrupt changes in direction, and friction

### CE 6303 MECHANICS OF FLUIDS L T P C QUESTION BANK PART - A

CE 6303 MECHANICS OF FLUIDS L T P C QUESTION BANK 3 0 0 3 UNIT I FLUID PROPERTIES AND FLUID STATICS PART - A 1. Define fluid and fluid mechanics. 2. Define real and ideal fluids. 3. Define mass density

### ME 305 Fluid Mechanics I. Part 8 Viscous Flow in Pipes and Ducts

ME 305 Fluid Mechanics I Part 8 Viscous Flow in Pipes and Ducts These presentations are prepared by Dr. Cüneyt Sert Mechanical Engineering Department Middle East Technical University Ankara, Turkey csert@metu.edu.tr

### Chapter 2. Derivation of the Equations of Open Channel Flow. 2.1 General Considerations

Chapter 2. Derivation of the Equations of Open Channel Flow 2.1 General Considerations Of interest is water flowing in a channel with a free surface, which is usually referred to as open channel flow.

### Diffusion and Fluid Flow

Diffusion and Fluid Flow What determines the diffusion coefficient? What determines fluid flow? 1. Diffusion: Diffusion refers to the transport of substance against a concentration gradient. ΔS>0 Mass

### Fundamentals of Fluid Mechanics

Sixth Edition. Fundamentals of Fluid Mechanics International Student Version BRUCE R. MUNSON DONALD F. YOUNG Department of Aerospace Engineering and Engineering Mechanics THEODORE H. OKIISHI Department

### Flow. 1.1 Introduction. 1.2 Units of Flow. 1.3 Measurement of Flow

1 Flow 1.1 Introduction Industrial flow measurements include measuring of flow rate of solids, liquids and gases. There are two basic ways of measuring flow ; one on volumetric basis and the other on weight

### Contents. Microfluidics - Jens Ducrée Physics: Fluid Dynamics 1

Contents 1. Introduction 2. Fluids 3. Physics of Microfluidic Systems 4. Microfabrication Technologies 5. Flow Control 6. Micropumps 7. Sensors 8. Ink-Jet Technology 9. Liquid Handling 10.Microarrays 11.Microreactors

### Pipe Flow-Friction Factor Calculations with Excel

Pipe Flow-Friction Factor Calculations with Excel Course No: C03-022 Credit: 3 PDH Harlan H. Bengtson, PhD, P.E. Continuing Education and Development, Inc. 9 Greyridge Farm Court Stony Point, NY 10980

### Thermal Dispersion Mass Flow

Thermatel Thermal Dispersion Mass Flow Measurement Handbook Table of Contents Introduction...1 What is Mass Flow Measurement?...1 Types of Flow Transmitters...4 Differential Pressure...4 Orifice...4 Venturi...6

### CEE 370 Fall 2015. Laboratory #3 Open Channel Flow

CEE 70 Fall 015 Laboratory # Open Channel Flow Objective: The objective of this experiment is to measure the flow of fluid through open channels using a V-notch weir and a hydraulic jump. Introduction:

### Design and Analysis of Various Bluff Objects for Vortex flow meters

Design and Analysis of Various Bluff Objects for Vortex flow meters T. Milinda Purna 1, L. Aarya 2, K. Lakshmi Nageswari 3 1 Assistant Professor, Department Of Electronics & Instrumentation, Sree Vidyanikethan

### Flow Measuring Techniques in Steady and Pulsating Compressible Flows. Fredrik Laurantzon

Flow Measuring Techniques in Steady and Pulsating Compressible Flows by Fredrik Laurantzon November 21 Technical Reports from Royal Institute of Technology KTH Mechanics SE-1 44 Stockholm, Sweden Akademisk

### EXAMPLE: Water Flow in a Pipe

EXAMPLE: Water Flow in a Pipe P 1 > P 2 Velocity profile is parabolic (we will learn why it is parabolic later, but since friction comes from walls the shape is intuitive) The pressure drops linearly along

### Chokes. Types Reasons Basics of Operations Application

Chokes Types Reasons Basics of Operations Application Most Common Chokes Positive: Fixed orifice Disassemble to change bean Adjustable Provides variable orifice size through external adjustment Schematic

### Lecture 8 - Turbulence. Applied Computational Fluid Dynamics

Lecture 8 - Turbulence Applied Computational Fluid Dynamics Instructor: André Bakker http://www.bakker.org André Bakker (2002-2006) Fluent Inc. (2002) 1 Turbulence What is turbulence? Effect of turbulence

### A Comparison of Analytical and Finite Element Solutions for Laminar Flow Conditions Near Gaussian Constrictions

A Comparison of Analytical and Finite Element Solutions for Laminar Flow Conditions Near Gaussian Constrictions by Laura Noelle Race An Engineering Project Submitted to the Graduate Faculty of Rensselaer

### Custody Transfer Measurement. with the V-Cone Flowmeter

Custody Transfer Measurement with the V-Cone Flowmeter Stephen A. Ifft McCrometer Inc. Hemet, California, USA Abstract This paper will discuss the approval of the McCrometer V-Cone flowmeter for custody

### Lecture 4 Classification of Flows. Applied Computational Fluid Dynamics

Lecture 4 Classification of Flows Applied Computational Fluid Dynamics Instructor: André Bakker http://www.bakker.org André Bakker (00-006) Fluent Inc. (00) 1 Classification: fluid flow vs. granular flow

### FLOW METER SPECIFICATION AND APPLICATION ANALYSIS Class #41

FLOW METER SPECIFICATION AND APPLICATION ANALYSIS Class #41 Curtis Gulaga Business Development Manager CB Engineering Ltd. #20, 5920 11 th Street SE Calgary, AB T2H 2M4 1.0 Introduction Today s working

### HEAT TRANSFER ANALYSIS IN A 3D SQUARE CHANNEL LAMINAR FLOW WITH USING BAFFLES 1 Vikram Bishnoi

HEAT TRANSFER ANALYSIS IN A 3D SQUARE CHANNEL LAMINAR FLOW WITH USING BAFFLES 1 Vikram Bishnoi 2 Rajesh Dudi 1 Scholar and 2 Assistant Professor,Department of Mechanical Engineering, OITM, Hisar (Haryana)

### L r = L m /L p. L r = L p /L m

NOTE: In the set of lectures 19/20 I defined the length ratio as L r = L m /L p The textbook by Finnermore & Franzini defines it as L r = L p /L m To avoid confusion let's keep the textbook definition,

### 2.0 BASIC CONCEPTS OF OPEN CHANNEL FLOW MEASUREMENT

2.0 BASIC CONCEPTS OF OPEN CHANNEL FLOW MEASUREMENT Open channel flow is defined as flow in any channel where the liquid flows with a free surface. Open channel flow is not under pressure; gravity is the

### Steady Flow: Laminar and Turbulent in an S-Bend

STAR-CCM+ User Guide 6663 Steady Flow: Laminar and Turbulent in an S-Bend This tutorial demonstrates the flow of an incompressible gas through an s-bend of constant diameter (2 cm), for both laminar and

### Turbulence Modeling in CFD Simulation of Intake Manifold for a 4 Cylinder Engine

HEFAT2012 9 th International Conference on Heat Transfer, Fluid Mechanics and Thermodynamics 16 18 July 2012 Malta Turbulence Modeling in CFD Simulation of Intake Manifold for a 4 Cylinder Engine Dr MK

### Water hammering in fire fighting installation

Water hammering in fire fighting installation Forward One of major problems raised in the fire fighting network installed at Pioneer company for pharmaceutical industry /Sulaymania was the high water hammering

### ME332 FLUID MECHANICS LABORATORY

ME332 FLUID MECHANICS LABORATORY Mihir Sen Department of Aerospace and Mechanical Engineering University of Notre Dame Notre Dame, IN 46556 Contents January 8, 2003 Spring 2003 Before break: Experiments

### Calculation of flow rate from differential pressure devices orifice plates

Calculation of flow rate from differential pressure devices orifice plates Peter Lau EMATEM - Sommerschule Kloster Seeon August -, 008 Contents Introduction - Why using dp-devices when other principles

### THE THERMAL FLOW METER, A GAS METER FOR ENERGY MEASUREMENT

THE THERMAL FLOW METER, A GAS METER FOR ENERGY MEASUREMENT Kazuto Otakane, Tokyo Gas Co., Ltd Katsuhito Sakai, Tokyo Gas Co., Ltd Minoru Seto, Tokyo Gas Co., Ltd 1. INTRODUCTION Tokyo Gas s new gas meter,

### Testing Protocol for Differential Pressure Measurement Devices API MPMS Chapter 22.2

Testing Protocol for Differential Pressure Measurement Devices API MPMS Chapter 22.2 Steve Baldwin Chevron Energy Technology Company Houston, Texas USA Casey Hodges CEESI Measurement Solutions. Nunn, Colorado

### Adaptation of General Purpose CFD Code for Fusion MHD Applications*

Adaptation of General Purpose CFD Code for Fusion MHD Applications* Andrei Khodak Princeton Plasma Physics Laboratory P.O. Box 451 Princeton, NJ, 08540 USA akhodak@pppl.gov Abstract Analysis of many fusion

### Testing methods applicable to refrigeration components and systems

Testing methods applicable to refrigeration components and systems Sylvain Quoilin (1)*, Cristian Cuevas (2), Vladut Teodorese (1), Vincent Lemort (1), Jules Hannay (1) and Jean Lebrun (1) (1) University

### XI / PHYSICS FLUIDS IN MOTION 11/PA

Viscosity It is the property of a liquid due to which it flows in the form of layers and each layer opposes the motion of its adjacent layer. Cause of viscosity Consider two neighboring liquid layers A

### AN EXPERIMENTAL STUDY OF EXERGY IN A CORRUGATED PLATE HEAT EXCHANGER

International Journal of Mechanical Engineering and Technology (IJMET) Volume 6, Issue 11, Nov 2015, pp. 16-22, Article ID: IJMET_06_11_002 Available online at http://www.iaeme.com/ijmet/issues.asp?jtype=ijmet&vtype=6&itype=11

### Piping Hydraulic Line Design and Sizing Software KLM Technology Group

Piping Hydraulic Line Design and Sizing Software KLM Technology Group Practical Engineering Guidelines for Processing Plant Solutions #03-12 Block Aronia, Jalan Sri Perkasa 2 Taman Tampoi Utama 81200 Johor

### CE 204 FLUID MECHANICS

CE 204 FLUID MECHANICS Onur AKAY Assistant Professor Okan University Department of Civil Engineering Akfırat Campus 34959 Tuzla-Istanbul/TURKEY Phone: +90-216-677-1630 ext.1974 Fax: +90-216-677-1486 E-mail:

### Heat transfer in Flow Through Conduits

Heat transfer in Flow Through Conduits R. Shankar Suramanian Department of Chemical and Biomolecular Engineering Clarkson University A common situation encountered y the chemical engineer is heat transfer

### - momentum conservation equation ρ = ρf. These are equivalent to four scalar equations with four unknowns: - pressure p - velocity components

J. Szantyr Lecture No. 14 The closed system of equations of the fluid mechanics The above presented equations form the closed system of the fluid mechanics equations, which may be employed for description

### Hydraulic losses in pipes

Hydraulic losses in pipes Henryk Kudela Contents 1 Viscous flows in pipes 1 1.1 Moody Chart.................................... 2 1.2 Types of Fluid Flow Problems........................... 5 1.3 Minor