MEDICAL IMAGING 2nd Part Computed Tomography


 Alexis Hall
 3 years ago
 Views:
Transcription
1 MEDICAL IMAGING 2nd Part Computed Tomography
2 Introduction 2 In the last 30 years Xray Computed Tomography development produced a great change in the role of diagnostic imaging in medicine. In convetional projection radiograpgy the image represents the 2D projection of a 3D structure. To obtain information on the real structure the number of projections have to be increased A tomogram is an image of a slice or a plane within the body The basics of CT is to take a series of conventional crosssectional xrays while the patient is rotated slightly around an axis between each exposure. A series of projection data is obtained, thes data are used to reconstruct crosssectional images.
3 Introduction : The austrian mathematician Radon develop a way to reconstruct the density distribution of an object if the line integrals (for every direction) are available : The physician Cormack develops a lot of theoretical work on x rays creating the basis of CT scanning (without knowing Radon work) 1971: The first CT system has been created by Godfrey N. Hounsfield, he received the Nobel prize in 1979 (together with Cromack) 1972: The first clinical use of a CT system in London, the acquisition time of a couple of tomogram was 5 minutes
4 Introduction : The first II Generation CT system totalbody scanner has been realised in the U.S., the acquisition time for 1 tomogram was 1 minute : III and IV Generations CT with acquisition time for 1 tomogram lower than 10 secs 1983: First electron beam CT, very expensive (scarce diffusion) 1989: First helical CT, very low acquisition time (less than 1 sec) and able to explore a large body volume 1992: Dualslice CT scanner, able to acquire a large number of layers in a very low time 2000: Multislice CT scanner, multiple arrays of detectors; continuous development.
5 Computed Tomography Generations Firstgeneration scanner Secondgeneration scanner 5 Single source collimated in a pencil beam and single detector Linear scan Arbitrary number of rays and angular projections can be measured, scattered radiation is mostly undetected Partial fanbeam and detectors array. Linear scan Able to complete a full scan in less time than I Generation scanners
6 Computed Tomography Generations Thirdgeneration scanner Fourthgeneration scanner 6 Fanbeam and detectos array. Source and detectors rotate in synchrony Dramatic decrease of scan time Fanbeam and large ring of stationary detectos array. Single rotating source Collimators cannot be used in detectors (they must receive energy from a mobile source) high detection efficiency and scattering can cause problems
7 Computed Tomography Generations 7 Main characteristics of the first four CT scanners generations I II III IV Scan Time (sec) Number of detectors (for each layer) Movement Rotation + Transl. Rotation + Transl. Rotation Rotation Xray tube rotation Data amount (each tomogram) 57.6 kb 57.6 kb 1 MB 242 MB Matrix 80x80 80x80 256x x512
8 Computed Tomography Generations Fifthgeneration scanner: electron beam computed tomography (EBCT); no moving parts, a electronically steered electron beam is used, this beam hit one of the four tugsten anode strip encircling the patient. This produces xrays that are collimated in a fanbeam and revealed by a stationary ring of detectors. A full set of data are acquired in 50 milliseconds. Very expensive Sixthgeneration scanner: helical CT addresses the need for rapid volumetric data acquisition. Conventional arrangement of source and detctors (III and IV generations) which can continuously rotate. During tube rotation patient table is slightly moved through the sourcedetectors plane. The image acquisition is not sequential (as it is for previous CT generations) but volumetric 8 A) sequential scanning B) helical scanning
9 Computed Tomography Generations Seventhgeneration scanner: multislice CT scanners, a thick fanbeam is used and multiple detectors array are used to collect the xrays. This technology is in continuous development, recently (and 256) multiple detectors array has been developed. 9 With the 256slice system the entire heart can be covered in a single rotation Helical and multislice CT has made the requirements for new development in data processing techniques even more critical Future development: multislice scanners able to interact with a conebeam Due to the large volume that can be explored it will be possible to stop using mobile patient table, this will decrease the complexity in image reconstruction
10 CT Instrumentation 10 Most of the commercial medical CT scanners use just one xray source. Xray tubes usually wear out in less than one year and need frequent status control and calibration. Xrays generated by the tube require collimation and filtration. Typically fanbeam geometry (having degress of fan angle) is obtained through two pieces of lead that form a slit between them. Slit width is adjustable by the operator in order to obtain different fanbeam thickness, typically between 0.5 and 10 mm for single slice systems and 20 to 30 mm for multislice CT systems. Fanangle Thickness mm Xrays fanbeam
11 CT Instrumentation 11 Different types of CT detectors: Solid state detectors contain a scintillation crystal able to interact with xrays and produce photoelectrons that are captured by a photodiode and converted to electric current Xenon gas detectors long thin tubes containing compressed xenon gas, when ionized the gas generates a current between anode and cathode. Less efficient than solid state detectors but highly directional Slip ring solves the mechanical problem of continuous electrical contact with rotating gantry; the gantry holds the xray tube (that needs high voltage to work properly) and the detectors that generate hundreds of signals to be transmitted out
12 Data Acquisition If a thin beam of monoenergetic xrays, having an intensity of, propagates for a distance d through an omogenic material, the intensity of the attenuated beam is: I 0 12 I = Ie µd 0 I0 µ I Being µ the linear attenuation coefficient µ is a function of material atomic number and electronic density d If the beam propagates through different omogenic materials the intensity of the attenuated beam is: d µ i ( µ 1+ µ µ n) d i= I = I e = I e n I0 x µ 1 µ 2 µ n d I
13 Data Acquisition 13 If the material has a continuous variable attenuation coefficient the intensity of the attenuated beam is: I = I e 0 0 D dx Given a measurement of and knowledge of, the projection measurement is defined as: p I 0 I0 I = ln( ) = I 0 D µ dx Therefore the basic measurement of a CT scanner is a line integral of the linear attenuation coefficient x I 0 x D I
14 Data Acquisition A CT scanner has to be calibrated in order to obtain the reference intensity I 0 Multiple projections (along parallel lines) create a socalled attenuation profile 14 CT detectors are able to measure the line integrals of bidimensional distribution of linear attenuation coefficients Through a suitable number of attenuation profiles it is possible to identify the bidimensional distribution of attenuation coefficients and it is possible to reconstruct the bidimensional image. Attenuation profile
15 CT Numbers 15 Different CT have different xrays tubes, so the same object will produce differenc values of µ on different scanners (the value of µ is a function of: xray energy, tissue density and atomic number) In order to compare data from different scanners is not possible to calculate absolute values of linear attenuation coefficents but at each pixel is associated a numerical value called CT number A CT number is calculated in Hounsfield units (UH) and is defined as: µ w µ t µ w h = 1000 µ Being the water attenuation coefficient and the tissue attenuation coefficient A CT number is 10 times the percentual difference between µ and t µ w divided by µ w w µ t
16 CT Numbers 16 As an example, a tissue having a attenuation coefficient 4% bigger than water has a CT number: h t (1.04 µ w µ w) = 1000 = 40 µ w Clearly h=0 for water, that is the reference value. Since µ = 0 for air, it is possible to find that h=1000 for air. The largest CT number found naturally in the body are for bones, 1000 HU for average bone. CT numbers can surpass 3000 HU for metal and contrast agents A typical HU range is: / 3071 (4096 levels, 12 bits each pixel)
17 CT Numbers Examples of typical HU values 17
18 CT Numbers Windowing 18
19 CT Numbers Conventional CT scan without contrast showing possible tumor in the liver Conventional CT scan of the same patient using contrast
20 Image Formation 20 CT images are reconstructed as a numerical bidimensional matrix. Each element of the matrix is called pixel The number of pixels is variable between 256x256 and 1024x1024 For each couple of x and y, the pixel represent a micro bidimensional area and f(x,y) is the functional value of this area f(x,y) is a grey level value, positive integer power of 2 NxN matrix As a matter of fact this is a bidimensional representation of a tridimensional tissue volume f(x,y)
21 Image Representation 21 Each pixel of the bidimensional matrix represents the properties of a tissue volume called voxel Voxel dimension is a function of the field of view (FOV) In general the voxel crosssectional plane dimensions are lower than the one of z axis (alongray), voxels are parallelepiped Spatial resolution along z axis is lower than the one along x and y axes Anisotropy pixel y voxel x z
22 Parallel Ray Reconstruction 22 The basic CT measurements is a line integral of the effective linear attenuation coefficient within a crosssection µ We desire a picture of or a CT number over the entire crosssection. Can we reconstruct a picture of given a collection of its line integrals? µ The answer is yes! A line in the (x, y) plane is given by: { } L(, l θ) = ( x, y) xcosθ + ysinθ = l The unit normal to the line is oriented at an angle relative to x axis and the line is at distance l from the origin in the direction of the unit normal θ
23 Parallel Ray Reconstruction 23 The line integral of function f(x,y) is defined as: gl (, θ) = f( xy, ) δ( xcosθ + ysin θ ldxdy ) Being the line impulse defined as: δl ( xy, ) = δ( xcosθ + ycos θ l) The inpulse shifting property causes the integrand to be zero everywhere except on the line For a fixed θ glθ (, ), is called a projection θ l glθ (, ) For all and, is called the 2D Radon Transform of f( x, y)
24 Parallel Ray Reconstruction 24 If we make the following identifications f( x, y) = µ ( x, y) Unknown function to be reconstructed by CT I gl (, θ ) = ln( ) I CT Measurements It is possible to see that this mathematical abstraction characterizes the CT measurement situation It has to be noticed that this definition of projection corresponds to a collection of line integrals of parallel lines. This correspond to the geometry of 1G scanners only. However this formalism and methods can be applied also to reconstruct fanbeam projections. 0
25 Sinogram 25 glθ (, ) An image of with l and θ as rectilinear coordinates is called a SINOGRAM A sinogram is a pictorial representation of the Radon transform of and represents the data that are needed to reconstruct it There is no need to calculate the sinogram for angles higher than π since these projections are redundant f( x, y)
26 Backprojection glθ (, ) f( x, y) Let be the 2D Radon transform of. Consider the projection at θ = θ 0. There are an infinite number of functions that could give rise to this projection; it is not possible to determine uniquely form a single projection f( x, y) glθ (, ) l = l f( x, y) 0 Ll (, θ ) If takes on a large value at, then must be large 0 over the line (or somewhere over the line) 0 0 One way to create an image exploiting this property is to simply assign every point on the value of Ll (, θ ) gl ( 0, θ0) 0 0 l If we repeat this for every, the resulting function is called backprjection image and is given by bθ ( x, y) = g( xcosθ + ysin θθ, ) The backprojection image at a angle is consistent with the projection at θ 0 but its values are assigned with no prior information about the distribution of the image intensities θ 0 26
27 Backprojection It is possible to addup backprojection images for every angle, obtaining the backprojection summation image, also called laminogram (, ) π f (, ) b x y = bθ x y dθ 0 27 Not the correct way to reconstruct f( x, y) Backprojection for θ = 30 Laminogram (blurriness)
28 d µ r 0 I Ie µ r = d 0 Backprojection I Ie µ r = d Backprojection assumption x x x x x I = I0e µ µ µ µ µ µ = µ /5 x ( d + d + d + d + d) r Object density is distributed along the whole xray path Starlike Artifact Backprojection (2 projections) Backprojection (4 projections)
29 ProjectionSlice Theorem 29 Lets take the 1D Fourier transform of a projection with respect to l πρl G( ρθ, ) F gl (, θ) gl (, θ) e dl = = 1D Being { } j2 ρ the spatial frequency glθ (, ) If we substitute the analitical expression of it is possible to obtain j2πρl G( ρθ, ) = f ( x, y) δ( x cosθ+ y sin θ l) e dxdydl j2 l f( x, y) πρ δ( xcosθ ysin θ l) e dldxdy = + = f ( x, y) e j2 πρ ( xcosθ + ysin θ ) dxdy
30 ProjectionSlice Theorem 30 The last expression is reminescent of the 2D Fourier transform of j2 ( xu yv) F( u, v) = π + f ( x, y) e dxdy Being u and v the frequency variables in the x and y directions u = ρ cosθ v = ρ sinθ G( ρ, θ) = F( ρcos θ, ρsin θ) f( x, y) Making the following identifications: and leads to a very important equivalence, called the projectionslice theorem The 1D Fourier transform of a projection is a slice of the 2D Fourier transform of the object. The 1D Fourier transform of the projection, calculated along a direction, equals the Fourier transform of the object calculated along the same direction passing through the origin
31 ProjectionSlice Theorem Graphical representation of the theorem 31 ρ θ and can be interpretated as the polar coordinates of the 2D Fourier transform
32 The Fourier Method 32 A very simple image reconstruction method follows immediately from the projectionslice theorem Take the 1D Fourier transform of each projection, insert it with the corresponding correct angular orientation into the correct slice of the 2D Fourier plane, and take the inverse 2D Fourier transform f( x, y) = F G( ρ, θ ) 2D 1 { } This method is not widely used in CT due to some practical problems: Interpolating polar data onto a Cartesian grid 2D inverse Fourier transform is timeconsuming
33 Filtered Backprojection 33 The inverse Fourier transform of F( uv, ) can be written in polar coordinates 2π 2 ( cos sin ) (, ) ( cos, sin ) j πρ f x y F e x θ + y θ = d d 0 0 ρ θ ρ θ ρ ρ θ Using the projectionslice theorem π 2 ( cos sin ) (, ) (, ) j πρ f x y G e x θ + y θ d d 0 = ρ θ ρ ρ θ The term xcosθ y being gl (, θ ) = g( l, θ + π) θ + sin is constant considering integration over ρ f x y ρ G ρθ e dρ dθ π 2 (, ) (, ) j πρ = l 0 l= xcosθ+ ysinθ
34 Filtered Backprojection 34 The inner integral of the previous formula is an inverse 1D Fourier transform ρ The term makes it a filtering equation. The Fourier transform of a projection is multiplied by a frequency filter and inversetransformed After the inversetransform, the filtered projection is backprojected (this is done replacing l = xcosθ + ysinθ ), and then is performed a summation of all filtered projection ρ This reconstruction approach is called filtered backprojection and is faster and more flexible with respect to the Fourier method The term ρ is called ramp filter
35 Convolution Backprojection 35 From the convolution theorem of Fourier transforms follows π 1 { } 0 1D l= xcosθ+ ysinθ f( x, y) = F ρ g( l, θ ) d θ Defining cl () = F 1 1D { ρ } π 0 [ θ ] = cosθ+ sin f ( x, y ) = c ( l ) g ( l, ) d l x y θ θ π = gl (, θ ) cx ( cosθ + ysin θ ldld ) θ 0 Substitution of the convolution integral to obtain the equation for convolution backprojection
36 Convolution Backprojection 36 Performing a convolution rather than a filtering operation is generally more efficient if the inpulse response is narrow (it is so in this case) cl () does not exists since ρ is not integrable In practice ρ is windowed with suitable windowing functions Convolution backprojection algorithms use an approximate impulse response cl () = F D W 1 1 { ρ ( ρ )}
37 Reconstruction 37 Filtered backprojection uses three basic steps to reconstruct an image from a sinogram: filtering, backprojection and summation Convolution backprojection uses the same steps except that filtering is performed using a convolution Filtering step: each row of the sinogram is filtered through a ramp filter to obtain the filtered sinogram Ramp filter is essentially a highpass filter and its value for f=0 is 0 Highfrequency details are accentuated
38 Reconstruction 38 Backprojection is applied to each filtered projection, rather than to the raw data Last setp is the summation of filtered backprojections This step is performed using an accumulator concept No need to store hundreds of backprojection images
39 Reconstruction 39
40 CT Image Quality 40 CT image quality can be expressed as a function of: Spatial resolution in the transversal plane xy Longitudinal resolution and slice sensitivity profile Noise Some indicators of CT image quality will be introduced Contrast Artifacts
41 CT Image Quality 41 Spatial resolution is a function of the smalles details that can be identified in the image; it is stricly dependant on the detectors (and, in general, on the intruments used for data acquisition and reconstruction). Spatial resolution is measure of how closely lines can be resolved in an image, it is measured in mm or LP/cm (line pairs/cm). The clarity of the image is not the number of pixels in an image, in fact spatial resolution refers to the number of independent pixel values per unit length. Detectable lines Undetectable lines
42 CT Image Quality 42 Pixel dimensions can be considered the upper limit of the spatial resolution It is not possible to identify details having dimensions lower than pixel size Details having dimensions comparable to the pixel ones provides contributions to adjacent pixels
43 CT Image Quality 43 Slice sensitivity profile (SSP) is the impulse response of the CT system along the z axis (patient long axis). Thick slice Thin slice
44 CT Image Quality Image width affects contrast and noise of object Optimised slice width: imaged slice object size 44 Object 5mm 5mm Low contrast Low noise 1mm Better contrast More noise
45 CT Image Quality Noise is the random variation of color (grayscale), some regions of the image that should be uniform show coarseness 45 Noise is produced by the sensors and circuitry of the scanner Noise intensity can be evaluated considering a region of the image; the mean pixels intensity of the region is calcualted as well as standard deviation standard deviation Noise = % mean intensity Noise can be reduced through reconstruction filters Noise is in inverse proportion to layer thickness and radiation dose root square
46 CT Image Quality 46 Different kind of filters can be used in backprojection to optimize spatial resolution against noise: smooth, standard, detail, bone, etc. Smooth Filtering Low contrast Low noise Sharp Filtering Better contrast More noise
47 Artifacts 47 Artifacts can appear in CT images, due to different reasons Aliasing: if projections are undersampled higher frequency information will appear as lower frequency artifacts. An high number of detectors is needed, it is needed to sample at numerous angles Beam hardening: when travelling through te body the xrays beam spectrum component with low energy is almost completely absorbed and the mean energy increases (this artifacs affect the bones) Other: electornic or system drift (miscalibration), xray scatter (can be prevented using collimators at expense of lower efficiency), motion (typically a scan require 1 to 10 secs) Beam herdening artifact
CT scanning. By Mikael Jensen & Jens E. Wilhjelm Risø National laboratory Ørsted DTU. (Ver. 1.2 4/9/07) 20022007 by M. Jensen and J. E.
1 Overview CT scanning By Mikael Jensen & Jens E. Wilhjelm Risø National laboratory Ørsted DTU (Ver. 1.2 4/9/07) 20022007 by M. Jensen and J. E. Wilhjelm) As it can be imagined, planar Xray imaging has
More informationComputed Tomography Scientific/Technological fact sheet
Students material Medical Imaging The Glass Patient Computed Tomography Scientific/Technological fact sheet Introduction In conventional Xray imaging, the entire thickness of the body is projected on
More informationCone Beam Reconstruction Jiang Hsieh, Ph.D.
Cone Beam Reconstruction Jiang Hsieh, Ph.D. Applied Science Laboratory, GE Healthcare Technologies 1 Image Generation Reconstruction of images from projections. textbook reconstruction advanced acquisition
More informationBasic principles of computed tomography. MUDr. Lukáš Mikšík, KZM FN Motol
Basic principles of computed tomography MUDr. Lukáš Mikšík, KZM FN Motol Tomography tomos = slice; graphein = to write definition  imaging of an object by analyzing its slices Damien Hirst Autopsy with
More informationPrinciples of XRay Imaging
Principles of XRay Imaging 1 Already a few weeks after the discovery of Xrays in 1895 by Wilhelm Conrad R ontgen the first medical images with photographic plates and fluorescent screens were made. This
More informationTomographic Image Reconstruction
Tomographic Image Reconstruction 1 Introduction ffl Tomography is a noninvasive imaging technique allowing for the visualization of the internal structures of an object without the superposition of over
More informationXRays were discovered accidentally in 1895 by Wilhelm Conrad Röntgen
XRays were discovered accidentally in 1895 by Wilhelm Conrad Röntgen Due to their short wavelength, on the order of magnitude of cells, and their high energy, they can penetrate skin and other soft tissue.
More informationChapter 5 Xray imaging 5.1 The physics of diagnostic Xrays
Chapter 5 Xray imaging 5.1 The physics of diagnostic Xrays Learning objectives: How are Xrays produced in an Xray tube? How is the energy of the photons from an Xray tube controlled? Why are photons
More informationCT Image Reconstruction. Terry Peters Robarts Research Institute London Canada
CT Image Reconstruction Terry Peters Robarts Research Institute London Canada 1 Standard Xray Views Standard Radiograph acquires projections of the body, but since structures are overlaid on each other,
More information5.4 CT Number Accuracy and Noise
186 Chapter 5 5.4 CT Number Accuracy and Noise In many clinical practices, radiologists rely on the value of measured CT numbers to differentiate healthy tissue from disease pathology. Although this practice
More informationMRI: Magnetic resonance imaging:
MRI: Magnetic resonance imaging: Like CT, MRI produces detailed cross sectional images of body regions. However the physical principles of image acquisition are different and very complicated. The following
More informationIntroduction to Medical Imaging. Lecture 11: ConeBeam CT Theory. Introduction. Available conebeam reconstruction methods: Our discussion:
Introduction Introduction to Medical Imaging Lecture 11: ConeBeam CT Theory Klaus Mueller Available conebeam reconstruction methods: exact approximate algebraic Our discussion: exact (now) approximate
More informationChapter 16 Physics of Diagnostic XRays
1895, W. C. Roentgen Discovery of xray The first xray image: Fig.16.1 Radiology Diagnostic radiology (radiologist) Radiation therapy (therapeutic radiologist) Nuclear medicine 1. Production of XRay
More informationThe application of Fourier analysis in solving the Computed Tomography (CT) inverse problem.
The application of Fourier analysis in solving the Computed Tomography (CT) inverse problem. by Nasser M. Abbasi July 3, 2010 Abstract Digital signal processing plays an increasingly important rule in
More informationCSE168 Computer Graphics II, Rendering. Spring 2006 Matthias Zwicker
CSE168 Computer Graphics II, Rendering Spring 2006 Matthias Zwicker Last time Sampling and aliasing Aliasing Moire patterns Aliasing Sufficiently sampled Insufficiently sampled [R. Cook ] Fourier analysis
More informationMODELING AND IMPLEMENTATION OF THE MECHANICAL SYSTEM AND CONTROL OF A CT WITH LOW ENERGY PROTON BEAM
MODELING AND IMPLEMENTATION OF THE MECHANICAL SYSTEM AND CONTROL OF A CT WITH LOW ENERGY PROTON BEAM João Antônio Palma Setti, j.setti@pucpr.br Pontifícia Universidade Católica do Paraná / Rua Imaculada
More informationChapter 3 SYSTEM SCANNING HARDWARE OVERVIEW
Qiang Lu Chapter 3. System Scanning Hardware Overview 79 Chapter 3 SYSTEM SCANNING HARDWARE OVERVIEW Since all the image data need in this research were collected from the highly modified AS&E 101ZZ system,
More informationThinking ahead. Focused on life. REALIZED: GROUNDBREAKING RESOLUTION OF 80 µm VOXEL
Thinking ahead. Focused on life. REALIZED: GROUNDBREAKING RESOLUTION OF 80 µm VOXEL Xray ZOOM RECONSTRUCTION Flat Panel Detector (FPD) Automatic Positioning Function For ø 40 x H 40 mm, ø 60 x H 60 mm,
More informationEECS 556 Image Processing W 09. Interpolation. Interpolation techniques B splines
EECS 556 Image Processing W 09 Interpolation Interpolation techniques B splines What is image processing? Image processing is the application of 2D signal processing methods to images Image representation
More informationCHAPTER 3: DIGITAL IMAGING IN DIAGNOSTIC RADIOLOGY. 3.1 Basic Concepts of Digital Imaging
Physics of Medical XRay Imaging (1) Chapter 3 CHAPTER 3: DIGITAL IMAGING IN DIAGNOSTIC RADIOLOGY 3.1 Basic Concepts of Digital Imaging Unlike conventional radiography that generates images on film through
More informationCOMMISSIONING AND USE OF CONVENTIONAL CT
COMMISSIONING AND USE OF CONVENTIONAL CT Sasa Mutic Department of Radiation Oncology Siteman Cancer Center Mallinckrodt Institute of Radiology Washington University School of Medicine St. Louis, Missouri
More informationContents. Xray and Computed Tomography. Characterization of Xrays. Production of Xrays
J. E. Wilhjelm Ørsted TU Technical University of enmark, Bldg. 348, K2800 Kongens Lyngby, enmark. Xray and Computed Tomography Contents History and characterization of Xrays Conventional (projection)
More informationPurchasing a cardiac CT scanner: What the radiologist needs to know
Purchasing a cardiac CT scanner: What the radiologist needs to know Maria Lewis ImPACT St George s Hospital, London maria.lewis@stgeorges.nhs.uk CT scanner development Slice wars 1998 Increased zcoverage
More informationComputed Tomography Resolution Enhancement by Integrating HighResolution 2D XRay Images into the CT reconstruction
Digital Industrial Radiology and Computed Tomography (DIR 2015) 2225 June 2015, Belgium, Ghent  www.ndt.net/app.dir2015 More Info at Open Access Database www.ndt.net/?id=18046 Computed Tomography Resolution
More informationPart 3 tracking detectors material effects track models fitting with momentum
Part 3 tracking detectors material effects track models fitting with momentum passage of particles through matter particles traversing through a medium interact with that medium they loose energy ionization:
More informationSpiral CT: Single and Multiple Detector Systems. AAPM Refresher Course Nashville, TN July 28,1999
Spiral CT: Single and Multiple Detector Systems AAPM Refresher Course Nashville, TN July 28,1999 Mike McNittGray, PhD, DABR Assistant Professor UCLA Radiological Sciences mmcnittgray@mednet.ucla.edu XRay
More information5 Factors Affecting the SignaltoNoise Ratio
5 Factors Affecting the SignaltoNoise Ratio 29 5 Factors Affecting the SignaltoNoise Ratio In the preceding chapters we have learned how an MR signal is generated and how the collected signal is processed
More informationScan Time Reduction and Xray Scatter Rejection in Dual Modality Breast Tomosynthesis. Tushita Patel 4/2/13
Scan Time Reduction and Xray Scatter Rejection in Dual Modality Breast Tomosynthesis Tushita Patel 4/2/13 Breast Cancer Statistics Second most common cancer after skin cancer Second leading cause of cancer
More informationMultislice helical CT: Scan and reconstruction
Multislice helical CT: Scan and reconstruction Hui Hu a),b) General Electric Company, Applied Science Laboratory, Milwaukee, Wisconsin 532010414 Received 21 April 1998; accepted for publication 27 October
More informationBIOMEDICAL ULTRASOUND
BIOMEDICAL ULTRASOUND Goals: To become familiar with: Ultrasound wave Wave propagation and Scattering Mechanisms of Tissue Damage Biomedical Ultrasound Transducers Biomedical Ultrasound Imaging Ultrasonic
More informationGE Medical Systems Training in Partnership. Module 8: IQ: Acquisition Time
Module 8: IQ: Acquisition Time IQ : Acquisition Time Objectives...Describe types of data acquisition modes....compute acquisition times for 2D and 3D scans. 2D Acquisitions The 2D mode acquires and reconstructs
More informationState of Stress at Point
State of Stress at Point Einstein Notation The basic idea of Einstein notation is that a covector and a vector can form a scalar: This is typically written as an explicit sum: According to this convention,
More informationFundamentals of ConeBeam CT Imaging
Fundamentals of ConeBeam CT Imaging Marc Kachelrieß German Cancer Research Center (DKFZ) Heidelberg, Germany www.dkfz.de Learning Objectives To understand the principles of volumetric image formation
More informationComputerized Tomography, Xrays, and the Radon Transform
1 Computerized Tomography, Xrays, and the Radon Transform 1.1 Introduction The purpose of this chapter is to give an informal introduction to the subject of tomography. There are very few mathematical
More informationMDCT Technology. Kalpana M. Kanal, Ph.D., DABR Assistant Professor Department of Radiology University of Washington Seattle, Washington
MDCT Technology Kalpana M. Kanal, Ph.D., DABR Assistant Professor Department of Radiology University of Washington Seattle, Washington ACMP Annual Meeting 2008  Seattle, WA Educational Objectives Historical
More informationChapter 24. Wave Optics
Chapter 24 Wave Optics Wave Optics The wave nature of light is needed to explain various phenomena. Interference Diffraction Polarization The particle nature of light was the basis for ray (geometric)
More informationBone mineral density (BMD) calibration in Skyscan CTanalyser
Method note Bone mineral density (BMD) calibration in Skyscan CTanalyser Bone mineral density (BMD) is defined as the volumetric density of calcium hydroxyapatite (CaHA) in terms of g.cm 3. It is calibrated
More informationPolarization of Light
Polarization of Light References Halliday/Resnick/Walker Fundamentals of Physics, Chapter 33, 7 th ed. Wiley 005 PASCO EX997A and EX999 guide sheets (written by Ann Hanks) weight Exercises and weights
More informationExposure SURE. Low Dose Diagnostic Image Quality. Erin Angel, PhD. Manager, Clinical Sciences, CT Toshiba America Medical Systems, Inc.
Exposure Low Dose Diagnostic Image Quality Erin Angel, PhD Manager, Clinical Sciences, CT Toshiba America Medical Systems, Inc. Exposure: Low Dose Diagnostic Image Quality 1 Since its inception, Computed
More informationComputed Tomography. 62.1 Instrumentation. Ian A. Cunningham. DataAcquisition Geometries. Ian A. Cunningham
62 Computed Tomography Ian A. Cunningham Victoria Hospital, the John P. Robarts Research Institute, and the University of Western Ontario Philip F. Judy Brigham and Women s Hospital and Harvard Medical
More informationCorrecting the Lateral Response Artifact in Radiochromic Film Images from Flatbed Scanners
Correcting the Lateral Response Artifact in Radiochromic Film Images from Flatbed Scanners Background The lateral response artifact (LRA) in radiochromic film images from flatbed scanners was first pointed
More informationPET/CT QC/QA. Quality Control in PET. Magnus Dahlbom, Ph.D. Verify the operational integrity of the system. PET Detectors
Quality Control in PET PET/CT QC/QA Magnus Dahlbom, Ph.D. Division of Nuclear Medicine Ahmanson Biochemical Imaging Clinic David Geffen School of Medicine at UCLA Los Angeles Verify the operational integrity
More informationDICOM Correction Item
Correction Number DICOM Correction Item CP626 Log Summary: Type of Modification Clarification Rationale for Correction Name of Standard PS 3.3 2004 + Sup 83 The description of pixel spacing related attributes
More informationSynthetic Sensing: Proximity / Distance Sensors
Synthetic Sensing: Proximity / Distance Sensors MediaRobotics Lab, February 2010 Proximity detection is dependent on the object of interest. One size does not fit all For noncontact distance measurement,
More informationPHYSICAL METHODS, INSTRUMENTS AND MEASUREMENTS Vol. III  Medical and Industrial Tomography  W.B.Gilboy
MEDICAL AND INDUSTRIAL TOMOGRAPHY Department of Physics, University of Surrey, Guildford, Surrey, U.K. Keywords: Radiography, transmission tomography, emission tomography, microtomography, SPECT (single
More informationManual for simulation of EB processing. Software ModeRTL
1 Manual for simulation of EB processing Software ModeRTL How to get results. Software ModeRTL. Software ModeRTL consists of five thematic modules and service blocks. (See Fig.1). Analytic module is intended
More informationComputer Graphics. Course SS 2007 Antialiasing. computer graphics & visualization
Computer Graphics Course SS 2007 Antialiasing How to avoid spatial aliasing caused by an undersampling of the signal, i.e. the sampling frequency is not high enough to cover all details Supersampling 
More informationMethods of Computerized Images for Medical Diagnosis Seminar: Medical Images Summer Semester 2005
+ Methods of Computerized Images for Medical Diagnosis Seminar: Medical Images Summer Semester 2005 Author: TuBinh Dang Matr.: 6147230 For: Professor Domik University of Paderborn Tutor: Gitta Domik domik@upb.de
More informationDetailed simulation of mass spectra for quadrupole mass spectrometer systems
Detailed simulation of mass spectra for quadrupole mass spectrometer systems J. R. Gibson, a) S. Taylor, and J. H. Leck Department of Electrical Engineering and Electronics, The University of Liverpool,
More informationMedical Image Processing on the GPU. Past, Present and Future. Anders Eklund, PhD Virginia Tech Carilion Research Institute andek@vtc.vt.
Medical Image Processing on the GPU Past, Present and Future Anders Eklund, PhD Virginia Tech Carilion Research Institute andek@vtc.vt.edu Outline Motivation why do we need GPUs? Past  how was GPU programming
More informationLecture PowerPoints. Chapter 24 Physics: Principles with Applications, 7th edition Giancoli
Lecture PowerPoints Chapter 24 Physics: Principles with Applications, 7th edition Giancoli This work is protected by United States copyright laws and is provided solely for the use of instructors in teaching
More informationImage Analysis for Volumetric Industrial Inspection and Interaction
Image Analysis for Volumetric Industrial Inspection and Interaction Rasmus Larsen Mads F. Hansen, Hildur Olafsdottir, Thomas H. Mosbech, Lasse F. Laursen DTU Informatics (CT) reconstruction from few projections
More informationPhysics of Imaging Systems Basic Principles of Computer Tomography (CT) I
1 10/30/2015 Page 1 Master s Program in Medical Physics Physics of Imaging Systems Basic Principles of Computer Tomography (CT) I Chair in Faculty of Medicine Mannheim University of Heidelberg TheodorKutzerUfer
More informationCT simulator. Professor and Director Department of Radiation Oncology. University of Pittsburgh Cancer Institute Pittsburgh, Pennsylvania, USA
CT simulator M. Saiful Huq,, Ph.D., FAAPM, FInstP Professor and Director Department of Radiation Oncology University of Pittsburgh Cancer Institute Pittsburgh, Pennsylvania, USA Outline What is a CT simulator?
More informationWhat does DXA mean and why is it important? Principles of operation of DXA systems. How can we measure bone mineral density? What is radiation?
What does DXA mean and why is it important? Principles of operation of DXA systems Dr Wil Evans Head of Medical Physics University Hospital of Wales, Cardiff DXA stands for Dual energy Xray Absorptiometry
More informationCommon Core Unit Summary Grades 6 to 8
Common Core Unit Summary Grades 6 to 8 Grade 8: Unit 1: Congruence and Similarity 8G18G5 rotations reflections and translations,( RRT=congruence) understand congruence of 2 d figures after RRT Dilations
More informationC1 Medical Imaging Modalities & Characteristics. 4005759 Linwei Wang
C1 Medical Imaging Modalities & Characteristics 4005759 Linwei Wang Major Types of Medical Imaging Modalities Xray Imaging Computed Tomography (CT) Magnetic Resonance Imaging (MRI) Nuclear Imaging Positron
More informationIntroduction to acoustic imaging
Introduction to acoustic imaging Contents 1 Propagation of acoustic waves 3 1.1 Wave types.......................................... 3 1.2 Mathematical formulation.................................. 4 1.3
More informationOutline. Exponential Attenuation. Introduction. Simple exponential attenuation. Simple exponential attenuation. Simple exponential attenuation
Outline Exponential Attenuation Chapter 3 F.A. Attix, Introduction to Radiological Physics and Radiation Dosimetry Simple exponential attenuation and plural modes of absorption arrowbeam vs. broadbeam
More informationR/F. Efforts to Reduce Exposure Dose in Chest Tomosynthesis Targeting Lung Cancer Screening. 3. Utility of Chest Tomosynthesis. 1.
R/F Efforts to Reduce Exposure Dose in Chest Tomosynthesis Targeting Lung Cancer Screening Department of Radiology, National Cancer Center Hospital East Kaoru Shimizu Ms. Kaoru Shimizu 1. Introduction
More informationLaboratory 6: Diffraction and Interference
Laboratory 6: Diffraction and Interference Renjie Li ID: 804291044 Physics 4BL Lab 8 May 20th 2015 Partner: Christine Truong TA: Eddie S. 1 Introduction In this lab, we will be performing experiments that
More informationAPPLICATION OF XRAY COMPUTED TOMOGRAPHY IN SILICON SOLAR CELLS
APPLICATION OF XRAY COMPUTED TOMOGRAPHY IN SILICON SOLAR CELLS V.A. Popovich 1, W. Verwaal 2, M. Janssen 1, I. J. Bennett 3, I.M.Richardson 1, 1. Delft University of Technology, Department of Materials
More informationPage: 1 of 6 Page: 1 of 6
Page: 1 of 6 Page: 1 of 6 CR Basics and FAQ Overview Computed Radiography is a term used to describe a system that electronically records a radiographic image. Computed Radiographic systems use unique
More informationMatrix, Gradients & Signals Image formation & kpace
Outline 6.1 MR Image Formation, s & Signals Image formation & kpace Imaging s & Signals, and Amplitude selection encoding encoding Carolyn Kaut Roth, RT (R)(MR)(CT)(M)(CV) FSMRT CEO Imaging Education
More informationPSS 27.2 The Electric Field of a Continuous Distribution of Charge
Chapter 27 Solutions PSS 27.2 The Electric Field of a Continuous Distribution of Charge Description: Knight ProblemSolving Strategy 27.2 The Electric Field of a Continuous Distribution of Charge is illustrated.
More informationTWODIMENSIONAL XRAY DIFFRACTION
TWODIMENSIONAL XRAY DIFFRACTION BOB B. HE GQ WILEY ,. "'! :~! CONTENTS~\..~..). Preface 1. Introduction 1.1 XRay Technology and Its Brief History, 1.2 Geometry of Crystals, 2 1.2.1 Crystal Lattice
More informationThermal Diffusivity, Specific Heat, and Thermal Conductivity of Aluminum Oxide and Pyroceram 9606
Report on the Thermal Diffusivity, Specific Heat, and Thermal Conductivity of Aluminum Oxide and Pyroceram 9606 This report presents the results of phenol diffusivity, specific heat and calculated thermal
More informationMODERN VOXEL BASED DATA AND GEOMETRY ANALYSIS SOFTWARE TOOLS FOR INDUSTRIAL CT
MODERN VOXEL BASED DATA AND GEOMETRY ANALYSIS SOFTWARE TOOLS FOR INDUSTRIAL CT C. Reinhart, C. Poliwoda, T. Guenther, W. Roemer, S. Maass, C. Gosch all Volume Graphics GmbH, Heidelberg, Germany Abstract:
More informationScientific Data Visualization Foundation
Scientific Data Visualization Foundation Data Sources Scientific Visualization Pipelines Data Acquisition Methods GeoVisualization 1 Scientific Data Sources Common data sources: Scanning devices Computation
More informationTranslating Protocols Between Scanner Manufacturer and Model
Translating Protocols Between Scanner Manufacturer and Model Cynthia H. McCollough, PhD, FACR, FAAPM Professor of Radiologic Physics Director, CT Clinical Innovation Center Department of Radiology Mayo
More informationInstruction Sheet 554 828
08/13W07Hund Instruction Sheet 554 828 Xray image sensor (554 828) Precision slide Xray image sensor (554 829) Software Computed Tomography Pro (554 820) Sensor head on precision slide with USB module
More informationIMAGE FORMATION. Antonino Furnari
IPLab  Image Processing Laboratory Dipartimento di Matematica e Informatica Università degli Studi di Catania http://iplab.dmi.unict.it IMAGE FORMATION Antonino Furnari furnari@dmi.unict.it http://dmi.unict.it/~furnari
More informationMicrowave Scanning Technology for Material
ECNDT 2006  Th.4.2.2 Microwave Scanning Technology for Material Testing Arndt GÖLLER, hf sensor GmbH, Leipzig, Germany Abstract. Microwave scanners are introduced as imaging systems for nondestructive
More informationData. microcat +SPECT
Data microcat +SPECT microcat at a Glance Designed to meet the throughput, resolution and image quality requirements of academic and pharmaceutical research, the Siemens microcat sets the standard for
More informationFourier Transform and Image Filtering. CS/BIOEN 6640 Lecture Marcel Prastawa Fall 2010
Fourier Transform and Image Filtering CS/BIOEN 6640 Lecture Marcel Prastawa Fall 2010 The Fourier Transform Fourier Transform Forward, mapping to frequency domain: Backward, inverse mapping to time domain:
More informationThe accurate calibration of all detectors is crucial for the subsequent data
Chapter 4 Calibration The accurate calibration of all detectors is crucial for the subsequent data analysis. The stability of the gain and offset for energy and time calibration of all detectors involved
More informationNuclear Medicine Imaging Systems: The Scintillation Camera
Radiation Detection and Measurement Range of charged particles (e.g.,!: µm; ": mm) Range of high energy photons (cm) Two main types of interactions of high energy photons Compton scatter Photoelectric
More informationXray Production. Target Interactions. Principles of Imaging Science I (RAD119) Xray Production & Emission
Principles of Imaging Science I (RAD119) Xray Production & Emission Xray Production Xrays are produced inside the xray tube when high energy projectile electrons from the filament interact with the
More informationThere are certain things that we have to take into account before and after we take an FID (or the spectrum, the FID is not that useful after all).
Data acquisition There are certain things that we have to take into account before and after we take an FID (or the spectrum, the FID is not that useful after all). Some deal with the detection system.
More informationModern methods for solving inverse problems
Modern methods for solving inverse problems Jürgen Hesser Experimental Radiation Oncology University Medical Center Mannheim University of Heidelberg 1 Motivation Image Guided Radiation Therapy (IGRT)
More informationCT RADIATION DOSE REPORT FROM DICOM. Frank Dong, PhD, DABR Diagnostic Physicist Imaging Institute Cleveland Clinic Foundation Cleveland, OH
CT RADIATION DOSE REPORT FROM DICOM Frank Dong, PhD, DABR Diagnostic Physicist Imaging Institute Cleveland Clinic Foundation Cleveland, OH CT Patient comes out... Patient goes in... Big Black Box Radiology
More informationLecture 5: Xray measurement techniques
Lecture 5: Xray measurement techniques Contents 1 Introduction 1 2 Powder photographs 2 2.1 DebyeScherrer method...................... 2 2.2 Focussing method......................... 4 2.3 Pinhole photographs.......................
More informationCT Traceability  Prof. Wim Dewulf, Group T  KU Leuven
CT Traceability  Calibration and Accuracy Prof. Wim Dewulf, Group T  KU Leuven Outline Introduction: terminology and procedures Voxel size calibration Edge offset calibration Conclusions Outline Introduction:
More informationPowder diffraction and synchrotron radiation
Powder diffraction and synchrotron radiation Gilberto Artioli Dip. Geoscienze UNIPD CIRCe Center for Cement Materials single xl diffraction powder diffraction Ideal powder Powder averaging Textured sample
More informationUltrasound tomography
Peter Pazmany Catholic University Faculty of Information Technology www.itk.ppke.hu Medical diagnostic systems (Orvosbiológiai képalkotó rendszerek) Ultrasound tomography (Ultrahang tomográfia) Miklós
More informationUltrasonic Wave Propagation Review
Ultrasonic Wave Propagation Review Presented by: Sami ElAli 1 1. Introduction Ultrasonic refers to any study or application of sound waves that are higher frequency than the human audible range. Ultrasonic
More informationExperiment 5. Lasers and laser mode structure
Northeastern University, PHYS5318 Spring 2014, 1 1. Introduction Experiment 5. Lasers and laser mode structure The laser is a very important optical tool that has found widespread use in science and industry,
More informationFourier Transform and Its Medical Application 서울의대의공학교실 김희찬
Fourier Transform and Its Medical Application 서울의대의공학교실 김희찬 강의내용 Fourier Transform 의수학적이해 Fourier Transform 과신호처리 Fourier Transform 과의학영상응용 Integral transform a particular kind of mathematical operator
More informationMicroCT for SEM Nondestructive Measurement and Volume Visualization of Specimens Internal Microstructure in SEM MicroCT Innovation with Integrity
MicroCT for SEM Nondestructive Measurement and Volume Visualization of Specimens Internal Microstructure in SEM Innovation with Integrity MicroCT 3D Microscopy Using MicroCT for SEM MicroCT for SEM
More informationIMAGE PROCESSING AND DATA ANALYSIS IN COMPUTED TOMOGRAPHY
IMAGE PROCESSING AND DATA ANALYSIS IN COMPUTED TOMOGRAPHY E. D. SELEÞCHI 1, O. G. DULIU 2 1 University of Bucharest, Faculty of Physics, Romania 2 University of Bucharest, Department of Atomic and Nuclear
More informationFundamentals of modern UVvisible spectroscopy. Presentation Materials
Fundamentals of modern UVvisible spectroscopy Presentation Materials The Electromagnetic Spectrum E = hν ν = c / λ 1 Electronic Transitions in Formaldehyde 2 Electronic Transitions and Spectra of Atoms
More informationA System for Capturing High Resolution Images
A System for Capturing High Resolution Images G.Voyatzis, G.Angelopoulos, A.Bors and I.Pitas Department of Informatics University of Thessaloniki BOX 451, 54006 Thessaloniki GREECE email: pitas@zeus.csd.auth.gr
More informationPersonal Identity Verification (PIV) IMAGE QUALITY SPECIFICATIONS FOR SINGLE FINGER CAPTURE DEVICES
Personal Identity Verification (PIV) IMAGE QUALITY SPECIFICATIONS FOR SINGLE FINGER CAPTURE DEVICES 1.0 SCOPE AND PURPOSE These specifications apply to fingerprint capture devices which scan and capture
More informationPurpose of Time Series Analysis. Autocovariance Function. Autocorrelation Function. Part 3: Time Series I
Part 3: Time Series I Purpose of Time Series Analysis (Figure from Panofsky and Brier 1968) Autocorrelation Function Harmonic Analysis Spectrum Analysis Data Window Significance Tests Some major purposes
More informationQuality Assurance Testing of Gamma Camera and SPECT Systems. Sharon L. White University of Alabama at Birmingham July 28, 2009
Quality Assurance Testing of Gamma Camera and SPECT Systems Sharon L. White University of Alabama at Birmingham July 28, 2009 Disclaimers Gamma camera images and photographs of equipment are not intended
More informationDiagnostic Xrays and CT Scans  Xray vision
 Xray vision http://www.uab.edu/surgonc/cases/gi/case2/ctscanof.htm http://www.museumboerhaave.nl/aacollection/aajpegs/m22/9955.jpg  motivation and outline Xrays have been known for over 110 years.
More informationINTRODUCTION. A. Purpose
New York State Department of Health Bureau of Environmental Radiation Protection Guide for Radiation Safety/Quality Assurance Programs Computed Radiography INTRODUCTION A. Purpose This guide describes
More informationCOMP175: Computer Graphics. Lecture 1 Introduction and Display Technologies
COMP175: Computer Graphics Lecture 1 Introduction and Display Technologies Course mechanics Number: COMP 17501, Fall 2009 Meetings: TR 1:302:45pm Instructor: Sara Su (sarasu@cs.tufts.edu) TA: Matt Menke
More informationChapter 5: Diffraction and Beam Formation Using Arrays
Chapter 5: Diffraction and Beam Formation Using Arrays Bmode Imaging www.acuson.com rf signal! envelop Linear Scanning Beam Formation Using Arrays Focusing: Curved Linear Scanning Sector Steering Focusing
More informationBME464: Medical Imaging
BME464: Medical Imaging John DiCecco, PhD Department of Electrical, Computer, and Biomedical Engineering University of Rhode Island Kingston, RI USA diceccoj@ele.uri.edu September 1, 2013 John DiCecco,
More information