1 Centers for Quantitative Imaging Excellence (CQIE) LEARNING MODULE QUANTITATIVE IMAGING IN MULTICENTER CLINICAL TRIALS: PET American College of Radiology Clinical Research Center v.1
2 Centers for Quantitative Imaging Excellence (CQIE) LEARNING MODULE This module is intended to provide a brief overview of general and PET-specific issues relevant to quantitative imaging for clinical trials. For additional information related to quantitative imaging for clinical trials, please see USEFUL LINKS on the CQIE web page (address provided below). For additional CQIE program information and qualification materials (MOP, forms, etc.), please refer to the CQIE web page. CQIE MOP and program information: After you have reviewed this module please let us know by submitting the learning module attestation. The link is provided at the end of this module.
3 What is Quantitative Imaging? As defined by the Toward Quantitative Imaging (TQI) task force of the Radiological Society of North America (RSNA): Quantitative imaging is the extraction of quantifiable features from medical images for the assessment of normal or the severity, degree of change, or status of a disease, injury, or chronic condition relative to normal. Quantitative imaging includes the development, standardization, and optimization of anatomical, functional, and molecular imaging acquisition protocols, data analysis, display methods, and reporting structures. These features permit the validation of accurately and precisely obtained image-derived metrics with anatomically and physiologically relevant parameters, including treatment response and outcome, and the use of such metrics in research and patient care. Buckler, et al., A Collaborative Enterprise for Multi-Stakeholder Participation in the Advancement of Quantitative Imaging, Radiology; Volume 258: Number 3, March 2011
4 What is Quantitative Imaging? Which imaging parameters are quantitative? Morphology Volume, 3D techniques (vct, vmr) Cellularity, density, composition of tissue Function Perfusion (DCE-MRI and DSC-MRI) Metabolic activity (PET) Molecule movement (DWI) MR Spectroscopy (MRS)
5 Quantitative Imaging as Biomarker What is an imaging biomarker? any anatomic, physiologic, biochemical, or molecular parameter detectable with one or more imaging methods used to help establish the presence and/or severity of disease. Ideally, a biomarker is an objectively measurable characteristic (versus a qualitative observation). Why use biomarkers? To speed the development of safe and effective medical therapies and procedures. Smith, et al., Biomarkers in Imaging: Realizing Radiology s Future, Radiology; Volume 227, Number 3, June 2003
6 Imaging Biomarkers in Clinical Trials Uses: Buckler, et al., A Collaborative Enterprise for Multi-Stakeholder Participation in the Advancement of Quantitative Imaging, Radiology; Volume 258: Number 3, March 2011
7 Imaging Biomarkers in Clinical Trials Needs: Standardization - the consistent performance of imaging, and adherence to protocols, for every research study performed at a given clinical site. Harmonization - the identification and implementation of mechanisms to control for inconsistencies of data between the different sites, particularly to ensure that imaging data generated with different systems are comparable. Facilitating the Use of Imaging Biomarkers in Therapeutic Clinical Trials, M.Graham, RSNA/SNM/FDA Two Topic Imaging Workshop, 2010
8 Imaging Standardization/Harmonization Why? Good Science Reliable decision making based on medical imaging requires comprehensive standards and tools to maintain integrity and ensure quality of results. For results to be of benefit to researchers and patients, the results must be accurate, comparable and reproducible (across patients, time-points, and institutions). Standardize imaging equipment (when possible) Standardize image/data acquisition Standardize image/data reconstruction Standardize image/data processing Standardize image interpretation
9 CQIE PET Qualification - Rationale Qualify NCI cancer centers in quantitative imaging methodologies: Static and dynamic PET, PET/CT (body and brain) Standardized qualification process and assessment, across 58 NCI-designation Cancer Centers Promote imaging standardization and harmonization within multi-center clinical trials.
10 CQIE PET - Qualification Imaging Clinical Test Cases 2 Brain, 2 Body- AC, NAC and CTAC files Anonymized and submitted via sftp Phantoms Uniform Cylinder- AC, NAC and CTAC files 2 bed position body FOV scan, using routine body protocol 1 bed position brain FOV scan, using routine brain protocol 1 bed position body FOV dynamic scan- 25 minute acquisition reconstructed in pre-defined time bins and summed ACR Phantom- AC NAC and CTAC files-don t forget to remove the 6 spheres and 6 mounting posts attached to the spheres 2 bed position body FOV scan, using routine body protocol 1 bed position brain FOV scan, using routine brain protocol Submitted via sftp
11 CQIE PET - Standardized QC Standardized QC helps to ensure the quantitative data generated within a clinical trial is comparable within an institution and across institutions. Sites are expected to incorporate the CQIE standardized QC measures into their existing QC activities. Sites are required to adhere to the CQIE standardized QC measures to maintain qualification status.
12 CQIE PET - Standardized QC QC is an important function of image quality and patient safety. Benefits of routine QC include Verification of operational integrity of imaging systems Early identification of technical issues Consistent, high image quality Consistent quantitative accuracy - SUV and Tracer Kinetics Quantitative imaging increases importance of QC. Participation in multicenter clinical research trials increases the importance of QC.
13 Standardized QC PET, PET/CT Test Purpose Frequency Physical Inspection To check gantry covers in tunnel and patient handling system. Daily Daily QC Check To test proper operation of scanner (per manufacturer s recommendations) Daily Normalization Uniformity Cross-Calibration Image quality To adjust system response to activity inside the field of view (FOV). To assess Transverse and Axial Uniformity across image planes by imaging a uniformly filled object. To monitor and identify discrepancies between the PET scanner and the dose calibrator. To check hot and cold spot image quality of standardized image quality phantom. As recommended by manufacturer (at least every 3 months); after software upgrades and after hardware service At lease every 3 months, after new setups and after software upgrades, At least every 3 months, after scanner upgrades, after new setups, and after modifications to the dose calibrator At least annually Documentation of the tests are subject to audit on annual renewal for qualification
14 Standardized QC - CT Test Standardized QC Tests for CT (as part of PET/CT) Water CT Number & Standard Deviation Artifacts Frequency Daily- Technologist Daily- Technologist Scout Prescription & Alignment Light Accuracy Imaged Slice Thickness-Slice Sensitivity Profile(SSP) Table Travel/Slice Positioning Accuracy Radiation Beam Width High-Contrast (Spatial) Resolution Low-Contrast Sensitivity and Resolution Image Uniformity and Noise CT Number Accuracy Artifact Evaluation Dosimetry/CTDI Annually Annually Annually Annually Annually Annually Annually Annually Annually Annually Documentation of the tests are subject to audit on annual renewal for qualification
15 Standardized QC Dose Calibrator Test Constancy (precision) Clock Accuracy Linearity Accuracy Geometry (position and volume) Frequency Each day of use and after equipment repair Daily Quarterly Annually and after equipment repair Upon installation and after replacement or repair of the chamber Accuracy of F18 Measurements Upon installation; at least once since June 2009 Documentation of the tests are subject to audit on annual renewal for qualification
16 Refresher - What is SUV? Standardized Uptake Value Semi-quantitative measure of the tracer uptake in a given region SUV Weight ConcTissue InjAct Weight Conc Tissue = Activity Concentration in Tissue (or region of a phantom) as measured in an ROI InjAct = Injected Activity at the start of the PET Emission acquisition Weight = Weight of patient (or active volume of phantom) Weight is the most common method of SUV normalization but factors like Lean Body Mass and Body Surface Area can also be employed
17 Technical Factors Affecting SUV Strict adherence to research protocol Accurate recording of time, activity and weight Dose calibrator accuracy, constancy, low background Scanner normalization, calibration Image reconstruction algorithm PET corrections: attenuation, scatter, randoms, decay ROI definition/methods for SUV SUV Normalization factor (body weight, body surface, lean body mass)
18 Quantitative Accuracy With so many factors contributing to the accuracy of SUV calculation it is of the utmost importance to ensure all instruments are functioning within acceptable limits at all times. All times must be recorded to the minute and all clocks must agree with the PET scanner clock to the minute. Assay Time, Injection Time, Residual Assay Time, Scan Start Time.
19 Strict Adherence to Protocol Deviation from protocol invalidates quantitative data. Increase or decrease in circulation time alters SUV Blood Glucose and Insulin levels alter SUV when using some tracers Patient behavior before and during uptake period can affect tracer distribution and SUV Moving the patient between scans when instructed not to may interfere with or invalidate core lab analysis Failure to perform additional scans with IV contrast may interfere with or invalidate core lab analysis
20 Accurate Data Entry If weight, net injected dose, time of dose assay or injection are recorded or entered incorrectly, any SUV calculated for the resulting images will be incorrect Good Research Practice Weigh every person on the scale in the PET Facility; do not take the patient s word Ensure all clocks in the PET Facility are synchronized to the PET scanner clock and to the minute
21 Dose Calibrator The dose calibrator must be maintained and checked in accordance with NRC guidelines The dose calibrator clock must be in agreement with the PET scanner clock to the minute All doses must be assayed with the dose calibrator background at 0 Remove all radioactive sources from the area Check to ensure background is 0 before and after the assay of every dose Keep a spare dose calibrator dip stick handy at all times for use in the event of contamination Clean even small amounts of radioactive material from the chamber if detected If background above 0 is unavoidable record the background reading and the time measured before and after dose assay
22 Image Reconstruction Algorithm Changing the reconstruction algorithm can affect the activity distribution visualized in the final image In all research trials, it is important to use the same reconstruction algorithm with the same parameters for all patients This will help ensure data across patients and time points can be compared both qualitatively and quantitatively
23 PET Corrections Normalization Corrects for non-uniformities in detector response Attenuation Corrects for decrease in detected events due to absorption and scatter Scatter Corrects for detected events that underwent a scattering event Randoms Corrects for detected events that originated from two different annihilation events DeadTime Corrects for a reduction is detected counts caused by elevated count rate Decay Corrects for the decreased activity during the scan due to radioactive decay. Cross-Calibration/SUV Converts the reconstructed counts into concentration (or SUV) in the final image
24 Scanner Normalization Corrects for nonuniformities in event detection over the full scanner Sources of nonuniformities: Variation in amount of scintillation light collected due to crystal nonuniformities and detector design (detector effect) Difference in detection sensitivity due to angle of incidence Most manufacturers use a uniform cylinder acquisition to generate the normalization correction Uniform cylinder
25 Routine Scanner QC Purpose of scanner QC is to ensure that the system performing as expected and expose any issue before they have a clinical impact QC program should involve daily, weekly, monthly, quarterly and annual tests QC programs should be reviewed regularly and updated should any deficiencies be found Scanner QC helps ensure that resulting images are quantitatively reliable
26 ROI Definition How the boundaries of an ROI are defined affects quantitative results A short Example: Max Max Pixel in ROI Peak Avg. in 1 cm ROI CT Avg. in ROI using boundaries defined on CT To the right: Actual Diameter is 16 mm Max = Peak = CT = 1.538
27 Normalization Factor-Body Weight It is essential to accurately record patient weight Ensure your scale is accurate Never record weight as reported by the patient or test subject.
28 Important Points Quantitative, multi-center clinical trials require reliable, quantitatively accurate image data This is obtained by properly maintaining all imaging related equipment, strictly following the research protocol, and accurately recording all required patient related information A small number of non-compliant cases can invalidate an entire trial
30 Please follow the link below to report your review of this module https://www.surveymonkey.com/s/cqie_quantitativeimaging_pet CQIE PET questions should be directed to American College of Radiology Clinical Research Center 1818 Market Street - Suite 1600 Philadelphia, PA Thank you!
Quantitative Imaging In Clinical Trials Using PET/CT: Update Paul Kinahan, Robert Doot Imaging Research Laboratory Department of Radiology University of Washington, Seattle, WA Supported by RSNA Quantitative
SITE IMAGING MANUAL ACRIN 6698 Diffusion Weighted MR Imaging Biomarkers for Assessment of Breast Cancer Response to Neoadjuvant Treatment: A sub-study of the I-SPY 2 TRIAL Version: 1.0 Date: May 28, 2012
Quality Control in PET PET/CT QC/QA Magnus Dahlbom, Ph.D. Division of Nuclear Medicine Ahmanson Biochemical Imaging Clinic David Geffen School of Medicine at UCLA Los Angeles Verify the operational integrity
Physics testing of image detectors Parameters to test Spatial resolution Contrast resolution Uniformity/geometric distortion Features and Weaknesses of Phantoms for CR/DR System Testing Dose response/signal
Quality Assurance Testing of Gamma Camera and SPECT Systems Sharon L. White University of Alabama at Birmingham July 28, 2009 Disclaimers Gamma camera images and photographs of equipment are not intended
(201) 825-9500 / (800) 631-3826 THRYOID UPTAKE SYSTEMS 51 Captus 3000 Thyroid Uptake And Well System The Ultimate... User Friendly... Intuitive Performing Thyroid Uptake measurements has never been easier.
Today s agenda Computed Tomography Presented by: Dina Hernandez, BSRT, RT (R), CT, QM Krista Bush, RT, MBA Leonard Lucey, JD ACR Quality & Safety MIPPA overview and CMS requirements CT accreditation How
The American College of Radiology, with more than 30,000 members, is the principal organization of radiologists, radiation oncologists, and clinical medical physicists in the United States. The College
Quality Control and Maintenance Programs Cari Borrás, D.Sc., FACR, FAAPM Visiting Professor DOIN-DEN / UFPE Recife, Pernambuco, Brazil Co-Chair, IUPESM Health Technology Task Group 1 Medical Imaging Equipment
Scan for mobile link. Positron Emission Tomography - Computed Tomography (PET/CT) Positron emission tomography (PET) uses small amounts of radioactive materials called radiotracers, a special camera and
The Importance of Following the PROTOCOL in Clinical Trials Presentation Objectives: Upon completion of this presentation, participants will be able to: Describe the following terms: Protocol, Protocol
JOURNAL OF APPLIED CLINICAL MEDICAL PHYSICS, VOLUME 16, NUMBER 4, 2015 Quality control of CT systems by automated monitoring of key performance indicators: a two-year study Patrik Nowik, a Robert Bujila,
Whitepapers on Imaging Infrastructure for Research Paper 1. General Workflow Considerations Bradley J Erickson, Tony Pan, Daniel J Marcus, CTSA Imaging Informatics Working Group Introduction The use of
QUALITY Tissue Benefits Mimicking of s RMI recognized RMI as the ultrasound standard phantoms for quality are performance the ultrasound labs ofand hospitals, manufacturers. clinics Sophisticated and ultrasound
American College of Radiology CT Accreditation Program Testing Instructions (Revised July 24, 2015) This guide provides all of the instructions necessary for clinical tests, phantom tests and general submission
PET and PET/CT in Clinical Trials Nathan C. Hall, M.D., Ph.D. The Ohio State University Medical Center CALGB Imaging Core Lab CALGB Imaging Committee Outline Introduction to PET Positron Emitter Physics
MODELING AND IMPLEMENTATION OF THE MECHANICAL SYSTEM AND CONTROL OF A CT WITH LOW ENERGY PROTON BEAM João Antônio Palma Setti, email@example.com Pontifícia Universidade Católica do Paraná / Rua Imaculada
AN INVESTIGATION INTO THE USEFULNESS OF THE ISOCS MATHEMATICAL EFFICIENCY CALIBRATION FOR LARGE RECTANGULAR 3 x5 x16 NAI DETECTORS Frazier L. Bronson CHP Canberra Industries, Inc. 800 Research Parkway,
Module 8: IQ: Acquisition Time IQ : Acquisition Time Objectives...Describe types of data acquisition modes....compute acquisition times for 2D and 3D scans. 2D Acquisitions The 2D mode acquires and reconstructs
Advanced Molecular Imaging Vereos PET/CT Truly digital PET imaging Philips proprietary Digital Photon Counting technology Vereos PET/CT is the first commercially available scanner to offer truly digital
Nuclear Medicine Accreditation Program Requirements OVERVIEW... 2 MEDICARE IMPROVEMENT F PATIENTS AND PROVIDERS ACT OF 2008 (MIPPA)... 2 MANDATY ACCREDITATION TIME REQUIREMENTS... 3 WITHDRAWN, ADDED, REPLACEMENT
Usefulness and limits of the SUV for tumor characterization and patient monitoring in FDG PET/CT Irène Buvat Imaging and Modelling in Neurobiology and Cancerology lab U8165 CNRS - Paris 7 & Paris 11 Universities
Rb 82 Cardiac PET Scanning Protocols and Dosimetry Deborah Tout Nuclear Medicine Department Central Manchester University Hospitals Overview Rb 82 myocardial perfusion imaging protocols Acquisition Reconstruction
Translating Protocols Between Scanner Manufacturer and Model Cynthia H. McCollough, PhD, FACR, FAAPM Professor of Radiologic Physics Director, CT Clinical Innovation Center Department of Radiology Mayo
DWI Case Study - QIBA PDF- MRI Technical Committee QIBA Annual Meeting May 21, 2014 What is IMI? The Innovative Medicines Initiative (IMI) is the world s largest biomedical /healthcare public-private initiative
Automotive Applications of 3D Laser Scanning Kyle Johnston, Ph.D., Metron Systems, Inc. 34935 SE Douglas Street, Suite 110, Snoqualmie, WA 98065 425-396-5577, www.metronsys.com 2002 Metron Systems, Inc
The American College of Radiology Accreditation Overview Dina Hernandez RT(R), QM, CT Program Specialist CT/MR Accreditation American College of Radiology May 5, 2008 American College of Radiology Quality
What is COMPASS? Answer: COMPASS is IBA Dosimetry s innovative 3D dosimetry analysis package. It is comprised of the MatriXX or MatriXX Evolution ion chamber array for measurement procurement and the COMPASS
X-Rays were discovered accidentally in 1895 by Wilhelm Conrad Röntgen Due to their short wavelength, on the order of magnitude of cells, and their high energy, they can penetrate skin and other soft tissue.
APPENDIX N Data Validation Using Data Descriptors Data validation is often defined by six data descriptors: 1) reports to decision maker 2) documentation 3) data sources 4) analytical method and detection
Current Industry Neuroimaging Experience in Clinical Trials Jerome Barakos, M.D. Melbourne Australia March 28, 2012 Synarc Experience and Expertise Largest imaging service provider dedicated to clinical
ACR MRI Accreditation Program: The Technologist s Role Geoffrey D. Clarke, Ph.D. University of Texas Health Science Center at San Antonio Radiological Sciences Division Overview ACR MRI Accreditation Program
(201) 825-9500 / (800) 631-3826 THYROID UPTAKE SYSTEMS 49 Captus 3000 Thyroid Uptake And Well System The Ultimate... User Friendly... Intuitive Performing Thyroid Uptake measurements has never been easier.
Multi-slice Helical CT Scanning of the Chest Comparison of different low-dose acquisitions Lung cancer is the main cause of deaths due to cancer in human males and the incidence is constantly increasing.
Positron Emission Tomography - For Patients A physician s written order is required for any PET-CT tests. How should I prepare for my PET-CT? PET-CT is more complicated than most other tests you may be
www.neurologica.com Critical Care Clinic Operating Room Radiology ED Trauma OR Neuro/Spine Pediatric Plastics Thoracic General Interventional Radiology ENT ED Trauma Trauma Stroke Critical Care Neuro Pediatric
Accreditation Is Coming Accreditation Is Coming Accreditation Is Coming Yes, It Is! Currently, Accreditation is Voluntary However, Some of the Accrediting Organizations Are Lobbying the Congress to Adopt
Scan Time Reduction and X-ray Scatter Rejection in Dual Modality Breast Tomosynthesis Tushita Patel 4/2/13 Breast Cancer Statistics Second most common cancer after skin cancer Second leading cause of cancer
Quality Control of Full Field Digital Mammography Units Melissa C. Martin, M.S., FACMP, FACR, FAAPM Melissa@TherapyPhysics.com 310-612-8127 ACMP Annual Meeting Virginia Beach, VA May 2, 2009 History of
Summary. 111 Detection and staging of recurrent prostate cancer is still one of the important clinical problems in prostate cancer. A rise in PSA or biochemical recurrence (BCR) is the first sign of recurrent
An Innovative Method for Dead Time Correction in Nuclear Spectroscopy Upp, Daniel L.; Keyser, Ronald M.; Gedcke, Dale A.; Twomey, Timothy R.; and Bingham, Russell D. PerkinElmer Instruments, Inc. ORTEC,
CT RADIATION DOSE REPORT FROM DICOM Frank Dong, PhD, DABR Diagnostic Physicist Imaging Institute Cleveland Clinic Foundation Cleveland, OH CT Patient comes out... Patient goes in... Big Black Box Radiology
RADIATION THERAPY Dosimetry Pioneers since 1922 NEW New Product Releases 2015 One phantom multiple options 4D Patient and Machine QA Modular Phantom for Stay flexible. Go modular. } Revolutionary new modular
Donna M. Reeve, MS, DABR, DABMP Department of Imaging Physics Educational Objectives Discuss the importance of breast MRI quality control (QC). Provide an overview of the new ACR Breast MRI Accreditation
The Ultimate... User Friendly... Intuitive Captus 3000 Thyroid Uptake And Well System Performing Thyroid Uptake measurements has never been easier. The Captus 3000 combines a totally new software program
COMMISSIONING AND USE OF CONVENTIONAL CT Sasa Mutic Department of Radiation Oncology Siteman Cancer Center Mallinckrodt Institute of Radiology Washington University School of Medicine St. Louis, Missouri
Moving Forward What does this mean for the Medical Physicist and the Imaging Community? John M. Boone, Ph.D., FAAPM, FACR Professor and Vice Chairman of Radiology University of California Davis Medical
Technical Note: 9 Use of the VALIDATOR Dosimetry System for Quality Assurance and Quality Control of Blood Irradiators 1- Introduction The VALIDATOR, model TN-ID-60, is a compact, and stand-alone dosimetry
WHERE IN THE WORLD IS JILL LIPOTI? HELLO FROM NEW JERSEY CRCPD - National Symposium on Fusion Imaging and Multimodalities February 18-20, 2004 Kansas City, Missouri New Jersey s Requirements As They Pertain
Truly digital PET imaging Philips proprietary Digital Photon Counting technology Vereos PET/CT is the first commercially available scanner to offer truly digital PET, resulting in significantly improved
CT for the production floor: Recent developments in Hard- and Software for industrial CT Systems GE s Industrial X-ray and CT Forum, Cologne July 7 th 2015 Oliver Brunke GE Sensing & Inspection Technologies
Quality Reports With PlanIQ * *Export only. 510(k) pending. Patent pending. Your Most Valuable QA and Dosimetry Tools Objective Plan Assessment? How good are my treatment plans? Quality assurance of radiation
Low energy ion scattering study of 4 on Cu(1) Chapter 8. Low energy ion scattering study of 4 on Cu(1) 8.1. Introduction For a better understanding of the reconstructed 4 surfaces one would like to know
A software tool for Quality Assurance of Computed / Digital Radiography (CR/DR) systems Nikunj Desai a and Daniel J Valentino a,b a icr Company Inc, 2580 West 237th Street, Torrance, CA 90505, USA b Department
Embedded Systems in Healthcare Pierre America Healthcare Systems Architecture Philips Research, Eindhoven, the Netherlands November 12, 2008 About the Speaker Working for Philips Research since 1982 Projects
Quality Control and Non-Imaging Instrumentation What can go wrong? Kathy Thomas MHA, CNMT, PET SNM Annual Meeting - Toronto Objectives Explain the purpose of quality control procedures on non-imaging instrumentation
Ultrasound - Dosimetry Gail ter Haar Joint Physics Department, Royal Marsden Hospital: Institute of Cancer Research, Sutton, Surrey UK Measurable in water Exposure Reduced by: tissue acoustic properties
Purchasing a cardiac CT scanner: What the radiologist needs to know Maria Lewis ImPACT St George s Hospital, London firstname.lastname@example.org CT scanner development Slice wars 1998 Increased z-coverage
IAC Ch 41, p.1 641 41.7 (136C) X-ray machines used for stereotactically guided breast biopsy. 41.7(1) Definitions. In addition to the definitions provided in rules 641 38.2(136C), 641 40.2(136C), and 641
Imaging g Inside the Human Body with Light Arjun G. Yodh University of Pennsylvania Department of Physics & Astronomy Goals Dreams Red/Near-infrared Transmission Window Why is this hard? How does light
The Challenge of CT Dose Records Kimberly Applegate, MD, MS, FACR Professor of Radiology and Pediatrics Emory University *financial disclosures: -Springer Textbook contracts -AIM advisory board for patient
The phantom portion of the American College of Radiology ACR Computed Tomography CT accreditation program: Practical tips, artifact examples, and pitfalls to avoid a Cynthia H. McCollough b) and Michael
Personal Identity Verification (PIV) IMAGE QUALITY SPECIFICATIONS FOR SINGLE FINGER CAPTURE DEVICES 1.0 SCOPE AND PURPOSE These specifications apply to fingerprint capture devices which scan and capture
PET Accreditation Program Requirements OVERVIEW... 2 MEDICARE IMPROVEMENT F PATIENTS AND PROVIDERS ACT OF 2008 (MIPPA)... 2 MANDATY ACCREDITATION TIME REQUIREMENTS... 3 WITHDRAWN, ADDED, REPLACEMENT UNITS...
American Association of Physicists in Medicine Radiation Therapy Committee Task Group 53: Quality assurance for clinical radiotherapy treatment planning Benedick Fraass a) University of Michigan Medical
GE Healthcare explore Locus Ultra pre-clinical CT dynamic computed tomography High-performance, dynamic, volume CT With acquisition speeds as fast as one second, the explore Locus Ultra CT scanner from
A Quantapoint White Paper ABSTRACT Specifying Laser Scanning Services to Help Ensure Trusted and Accurate Results As-built documentation created using laser scanning is rapidly being adopted as the tool
Imaging Informatics Ann Scherzinger, PhD, CIIP C-TRIC Lecture Series, May 18, 2011 Science of Information and Information Processing! Breadth, depth of current activities in human medical imaging informatics!
Molecular Imaging in Early Phase Oncology Trials Lessons Learned Potential Role for SNM Network Susan Galbraith MB BChir PhD Vice-President Oncology Discovery Medicine & Clinical Biomarkers Bristol-Myers
Thinking ahead. Focused on life. REALIZED: GROUNDBREAKING RESOLUTION OF 80 µm VOXEL X-ray ZOOM RECONSTRUCTION Flat Panel Detector (FPD) Automatic Positioning Function For ø 40 x H 40 mm, ø 60 x H 60 mm,
MODERN VOXEL BASED DATA AND GEOMETRY ANALYSIS SOFTWARE TOOLS FOR INDUSTRIAL CT C. Reinhart, C. Poliwoda, T. Guenther, W. Roemer, S. Maass, C. Gosch all Volume Graphics GmbH, Heidelberg, Germany Abstract:
Note: This is a courtesy copy and is not the official version of this rule. The official, legally effective version of this rule is available through www.lexisnexic.com/bookstore (Phone: (800) 223-1940).
Performance testing for Precision 500D Classical R/F System John M. Boudry, Ph.D. Image Quality Systems Engineer GE Healthcare Technologies Outline System background Image Quality Signature Test (IQST)
Radiation Therapy in Oncology Colin Winfield Elekta Limited Introduction... Needs and demands of Radiotherapy Uses and benefits of DICOM in RT Basic RT Process Practice 10 years ago; today; look to the
Role of the Medical Physicist in Clinical Implementation of Breast Tomosynthesis Bob Liu, Ph.D. Department of Radiology Massachusetts General Hospital And Harvard Medical School Digital Breast Tomosynthesis
Micro-CT for SEM Non-destructive Measurement and Volume Visualization of Specimens Internal Microstructure in SEM Innovation with Integrity Micro-CT 3D Microscopy Using Micro-CT for SEM Micro-CT for SEM
Synthetic Sensing: Proximity / Distance Sensors MediaRobotics Lab, February 2010 Proximity detection is dependent on the object of interest. One size does not fit all For non-contact distance measurement,
HIGH-PERFORMANCE INSPECTION VEHICLE FOR RAILWAYS AND TUNNEL LININGS. HIGH-PERFORMANCE INSPECTION VEHICLE FOR RAILWAY AND ROAD TUNNEL LININGS. The vehicle developed by Euroconsult and Pavemetrics and described
German Mobile Telecommunication Research Programme, July 25th, 2006 SAR Distribution in Test Animals Exposed to RF Radiation Verónica Berdiñas Torres, Andreas Christ, Niels Kuster IT IS Foundation, ETH
THE DOSIMETRIC EFFECTS OF OPTIMIZATION TECHNIQUES IN IMRT/IGRT PLANNING 1 Multiple PTV Head and Neck Treatment Planning, Contouring, Data, Tips, Optimization Techniques, and algorithms AAMD 2013, San Antonio,
CT QC Under the ACR QC Manual Douglas Pfeiffer, MS, DABR Boulder Community Hospital What? There s a Manual??! YES!!! Almost! Doesn t have a pretty cover yet! Should be out by RSNA! Really seriously I mean
Proper Implementation of Industrial CT Scanning to Reduce Inspection Costs & Get to Production Faster Jesse Garant, JG&A Metrology Center Traditional Metrology and Inspection Tactile Devices (Touch Probe)
G4 PET/X-ray P R O D U C T N O T E Pre-clinical in vivo Imaging Key Benefits: High performance, ultra-sensitive PET technology designed for pre-clinical imaging 3D PET and X-ray whole body images Purpose
1 Overview CT scanning By Mikael Jensen & Jens E. Wilhjelm Risø National laboratory Ørsted DTU (Ver. 1.2 4/9/07) 2002-2007 by M. Jensen and J. E. Wilhjelm) As it can be imagined, planar X-ray imaging has
Positron Emission Tomography (PET) Technologist Scope of Practice and Performance Standards Prepared by: Society of Nuclear Medicine and Molecular Imaging Technologist Section Approved: January 26, 2013