2 IQ : Acquisition Time Objectives...Describe types of data acquisition modes....compute acquisition times for 2D and 3D scans.
3 2D Acquisitions The 2D mode acquires and reconstructs raw image data into two dimensional images whose brightness is proportional to the intensity of the MRI signal from the corresponding protons. This slice excitation method is followed by frequency and phase encoding to produce an image. Because the spatial encoding takes place in two dimensions the images are labeled 2D. Phase Encoding 2D slices Frequency Encoding
4 2D Multi-Planar Acquiring multiple images within a single acquisition or time frame means that the order of slice excitation is odd-numbered first, then even numbered images. All slices are acquired within the TR period Multi-Planar acquisition All 5 slices in the same TR
5 2D Sequential Acquiring an image sequentially means that all the excitation pulses ( # of phase steps x NEX x TR ) are delivered for one location before the process is repeated at another location. 1 Sequential acquisition Each slice is completed before moving on to the next slice
6 3D Acquisition In volume or 3D imaging, a wide RF pulse is delivered to excite an entire scan volume or slab. Spatial encoding must then be done in the phase, frequency and slice axes. Slice Encoding Phase Encoding 3D Volume Frequency Encoding
7 3D Acquisition 3D Slab
8 Progress Check 2D: TR x Phase Matrix x NEX = Acquisition Time 3D: TR x Phase Matrix x NEX x # of Slices = Acq. Time TR- TR is measured in milliseconds and is the time between RF pulses that repeat the pattern. Phase Matrix - The system collects data from the slice once for each phase encoding step. NEX- The number of times each set of phase encoding steps is repeated is called a NEX or Number of EXcitations. Calculate the scan times below using the appropriate formula
9 15. I.Q.: Acquisition Time 16. There are multiple acquisition modes: 2D: sequential and multi-planar 3D / volume MRS/ spectroscopy Cine 2D scan time formula: TR x phase x NEX 60,000 3D scan time formula: TR x phase x NEX x slices 60,000 2D & 3D are standard Go to QLB Pomp and ZIP
10 Module 9: IQ: Spatial Resolution
11 Spatial Resolution Objectives...Recall and explain the scan parameters that affect spatial resolution....compute spatial resolution given specific scan parameters....evaluate images acquired with varying spatial resolution.
12 Spatial Resolution How close two objects can be before they can be distinguished as separate objects.
13 Spatial Resolution Three parameters affect spatial resolution: FOV Matrix Slice Thickness Small FOV s, large matrices and small slice thickness result in high spatial resolution. #Phase Row FOV (cm) #Frequency Column Voxel Slice Thickness (mm)
14 FOV FOV determines the amount of anatomy displayed on the image. The larger the FOV, the larger the pixel size, and therefore the resolution is decreased. FOV can be changed in 1 cm intervals. Pixel size Pixel size 24 cm FOV 16 cm FOV
15 Progress Check Calculating Pixel Size, Area and Volume FOV / # phase steps = phase dimension FOV / # frequency steps = frequency dimension Phase x frequency = pixel area Pixel area x slice thickness = voxel volume Calculate the voxel volume for the image parameters listed below
16 Progress Check Slice Thickness change Image A: Thickness = 3 mm Image B: thickness = 5 mm Image C: Thickness = 10 mm SNR: Spatial resolution: Contrast: Time:
17 Progress Check FOV change Image A: FOV = 18 cm Image B: FOV = 22 cm Image C: FOV= 25 cm SNR: Spatial resolution: Contrast: Time:
18 Module 10: IQ: SNR
19 IQ: SNR Objectives View images for SNR and spatial resolution trade-offs....quantify (calculate) the change in SNR when a parameter has been changed.
20 Signal to Noise Ratio SNR is the ratio of the amplitude of the MR signal to the amplitude of the noise. Noise is the undesirable signal that is generated from the patient, the environment and the system electronics. signal noise
21 Signal to Noise Ratio SNR TIME TR NEX Resolution FOV Matrix Slice Thick Exception to rule : Receive Bandwidth
22 Receive Bandwidth Readout Window + 16 khz X gradient TR Readout Window + 32 khz X gradient TR
24 Receive Bandwidth Tradeoffs RBW SNR Chemical TE Motion Shift Artifact H 2 0 Chemical shift The frequency difference between protons bound in fat versus water Fat (+) (-) approx. 220 Hz ( 1.5T ) approx. 143 Hz ( 1.0 T ) approx. 74 Hz (.5T ) approx. 35 Hz (.2T )
25 I.Q.: Spatial Resolution/SNR 18. Voxel volume = I.Q. Tradeoffs FOV x FOV x slice phase freq SNR Isotropic voxels occur when all sides (height, width, depth) are equal. Time RBW is an exception Resolution
29 I.Q. : Image Contrast Objectives Identify scan timing parameters that determine image contrast....identify changes in image contrast and SNR when scan parameters change.
30 Contrast Weighting Images are labeled according to the internal factors predominately responsible for the variations in signal intensities The goal in MR is to select a pulse sequence and timing parameters that will cause one of the three contrast mechanisms to predominate over the other two.
31 Contrast Weighting T1-weighted images result when the variation in longitudinal regrowth creates transverse magnetization differences which predominate over other contrast mechanisms. T2-weighted images result when the rate of nuclear dephasing predominates over the other contrast mechanisms. Proton density-weighted images result when the number of nuclei in a tissue is predominately responsible for the image contrast.
37 T2* The sum of T2 and T2 effects that influence transverse decay. 1/T2 + 1/T2 = 1/T2* T2 is the transverse decay due to magnetic field inhomogeneities, chemical shift of the second kind and patient induced magnetic susceptibilities. T2 decay echo T2* decay
38 180 degree RF pulse Initial 90 0 pulse. Vectors decay and pulse applied. Fast catch up with slow components. Vectors are rephased.
39 Dephasing TE controls dephasing which controls T2 effects 37% 16% 5% 2% 0% S i g n a l csf fat brain-g brain-w 1 T2 2 T2 3 T2 4 T2 5 T2 Time (TE)
40 Spin Density Hydrogen content Mobility of hydrogen Field strength affects the number of hydrogen protons that are parallel with B O.
41 Tissue Contrast Relaxation recovery ( T1 ) The time it takes the net magnetization to return to B 0. Relaxation dephasing ( T2 ) The time it takes the protons to dephase and thus for the net vector to decay. Nuclear density ( PD ) The number of nuclei that comprise the vectors from various tissues.
42 Scan timing parameter chart TR (T1) TE (T2) Tissue Intensities for the brain T > white/gray = grayer CSF = brighter as TE > fat = dark/bright T : 1.5T : 1.0T : 0.5T : 0.2T 10-15ms minimum (fractional echo) white = light gray gray = gray CSF = dark fat = bright PD 2000 > minimum (fractional echo) gray = light gray white = gray CSF = dark fat = bright
49 TR Scan Parameter Trade-offs Class Activity SNR Sp. res. Time Cont. Cont. Cont. T1 PD T2 TE T1 PD T2 NEX Slice Thickness FOV Receive Bandwidth Contrast is directly affected by TR, TE, TI, and flip angle. SNR changes can enhance or obscure contrast but cannot change the image weighting from one type of contrast to another. Frequency Phase
50 I.Q.: Contrast Time (TR) T ms PD T1 T1 Saturation Min. Sat. Long TR Max Sat. Short TR T2 Dephase Min dephase Short TE Min. dephase Short TE T2 Time (TE) 300 ms T2 T2* Min sat. Long TR Min sat. Small flip Short TR ( ) Max dephase Long TE Short TE
51 Progress Check Slice Thickness change Image A: Thickness = 3 mm Image B: thickness = 5 mm Image C: Thickness = 10 mm SNR: Spatial resolution: Contrast: Time:
52 Progress Check FOV change Image A: FOV = 18 cm Image B: FOV = 22 cm Image C: FOV= 25 cm SNR: Spatial resolution: Contrast: Time:
Module : Spin Echo Spin Echo Objectives Review the SE PSD. Review the concepts of T, T, and T*. Spin Echo PSD RF Gz Gy 90 80 Gx Spin Echo - SE Spin echo is a standard pulse sequence on Signa MRi/LX and
Musculoskeletal MRI Technical Considerations Garry E. Gold, M.D. Professor of Radiology, Bioengineering and Orthopaedic Surgery Stanford University Outline Joint Structure Image Contrast Protocols: 3.0T
5 Factors Affecting the Signal-to-Noise Ratio 29 5 Factors Affecting the Signal-to-Noise Ratio In the preceding chapters we have learned how an MR signal is generated and how the collected signal is processed
5 An MRI sequence is an ordered combination of RF and gradient pulses designed to acquire the data to form the image. In this chapter I will describe the basic gradient echo, spin echo and inversion recovery
Basic Principles of Magnetic Resonance Contents: Jorge Jovicich firstname.lastname@example.org I) Historical Background II) An MR experiment - Overview - Can we scan the subject? - The subject goes into the magnet -
4/5/ FYS-KJM 474 contrast SNR MR-teori og medisinsk diagnostikk Kap 8 Image qualit, signal, contrast and noise resolution vailable MRparameters speed Main source of noise in MRI: Noise generated within
Introduction Techniques of Fat Suppression Ian Cameron, PhD FCCPM MR Physicist, The Ottawa Hospital, 501 Smyth Rd., Ottawa, ON, Canada, K1H 8L6 The Magnetic Resonance (MR) signal used to create MR images
32-Channel Head Coil Imaging at 3T Thomas Benner Athinoula A. Martinos Center for Biomedical Imaging, Department of Radiology, Massachusetts General Hospital and Harvard Medical School, Boston, MA, USA
SITE IMAGING MANUAL ACRIN 6698 Diffusion Weighted MR Imaging Biomarkers for Assessment of Breast Cancer Response to Neoadjuvant Treatment: A sub-study of the I-SPY 2 TRIAL Version: 1.0 Date: May 28, 2012
Overview Optimizing MRI Protocols (SNR, CNR, Spatial Accuracy) Geoffrey D. Clarke, Ph.D. University of TX Health Science Center San Antonio, TX Parameters Affecting Image Quality Manipulating Image Contrast
G16.4426/EL5823/BE6203 Medical Imaging MRI Instrumentation, Data Acquisition, Image Reconstruction Riccardo Lattanzi, Ph.D. Assistant Professor Department of Radiology, NYU School of Medicine Department
ACR MRI Accreditation Program: The Technologist s Role Geoffrey D. Clarke, Ph.D. University of Texas Health Science Center at San Antonio Radiological Sciences Division Overview ACR MRI Accreditation Program
The following defines how different file formats are de-identified with LONI tools. Each metadata attribute of each file can have the following operations performed: Operation keep Remove Replace Description
Nuclear Magnetic Resonance Introduction Atomic magnetism Nuclear magnetic resonance refers to the behaviour of atomic nuclei in the presence of a magnetic field. The first principle required to understand
Overview Optimizing MRI Protocols Clinical Practice & Compromises Geoffrey D. Clarke, Radiology Department University of Texas Health Science Center at San Antonio Pulse timing parameters for adjusting
Magnetic Resonance Imaging What are the uses of MRI? To begin, not only are there a variety of scanning methodologies available, but there are also a variety of MRI methodologies available which provide
April 24, 2015 Exam #3: Solution Key online now! Graded exams by Monday! Final Exam Monday, May 4 th, 10:30 a.m. Room: Perkins 107 1 A Classical Perspective A classical view will help us understand the
Diagnostik Ultrasound Basic physics, image reconstruction and signal processing Per Åke Olofsson Dpt of Biomedical Engineering, Malmö University Hospital, Sweden Ultrasound Real-time modality 17-WEEK FETAL
MRI for Paediatric Surgeons Starship David Perry Paediatric Radiologist Starship Children s Hospital CHILDREN S HEALTH What determines the brightness of a pixel in MRI? i.e. What determines the strength
Lisa M. The fundamentals of nuclear magnetic resonance (NMR) spectroscopy were explored on samples containing large numbers of protons. Mineral oil and dilluted solutions of CuSO 4 were placed in a permanent
Pulse Sequences and Acquisition Techniques for Breast MRI ACR Breast MRI Accreditation Program Launched May 2010 Ron Price Vanderbilt University Medical Center Nashville, TN 37232 Information available:
NMR Techniques Applied to Mineral Oil, Water, and Ethanol L. Bianchini and L. Coffey Physics Department, Brandeis University, MA, 02453 (Dated: February 24, 2010) Using a TeachSpin PS1-A pulsed NMR device,
Basics of MRI Professor Sir Michael Brady FRS FREng Department of Engineering Science Oxford University Michaelmas 2004 Lecture 1: MRI image formation Basic nuclear magnetic resonance NMR NMR spectroscopy
Diffusione e perfusione in risonanza magnetica E. Pagani, M. Filippi DW-MRI DIFFUSION-WEIGHTED MRI Principles Diffusion results from a microspic random motion known as Brownian motion THE RANDOM WALK How
Pulsed nuclear magnetic resonance to explore relaxation times Hannah Saddler, Adam Egbert, and Will Weigand (Dated: 26 October 2015) The method of pulsed nuclear magnetic resonance was employed in this
A Comparison of Five Methods for Signal Intensity Standardization in MRI Jan-Philip Bergeest, Florian Jäger Lehrstuhl für Mustererkennung, Friedrich-Alexander-Universität Erlangen-Nürnberg email@example.com
Generation and Detection of NMR Signals Hanudatta S. Atreya NMR Research Centre Indian Institute of Science NMR Spectroscopy Spin (I)=1/2h B 0 Energy 0 = B 0 Classical picture (B 0 ) Quantum Mechanical
MRI DATA PROCESSING Compiled by: Nicolas F. Lori and Carlos Ferreira Introduction Magnetic Resonance Imaging (MRI) is a clinical exam that is safe to the patient. Nevertheless, it s very important to attend
COST AID ASL post- processing Workshop This workshop runs thought the post- processing of ASL data using tools from the FMRIB Software Library (www.fmrib.ox.ac.uk.uk/fsl), we will primarily focus on the
C1 Medical Imaging Modalities & Characteristics 4005-759 Linwei Wang Major Types of Medical Imaging Modalities X-ray Imaging Computed Tomography (CT) Magnetic Resonance Imaging (MRI) Nuclear Imaging Positron
Proton magnetic resonance spectroscopy in the brain: Report of AAPM MR Task Group #9 Dick J. Drost a) Nuclear Medicine and MRI Department, St. Joseph s Health Centre, London, Ontario N6A 4L6, Canada William
Advanced Physics Labs SEP006 Pulsed NMR Pulsed NMR is widely used for chemical analysis, in Magnetic Resonance Imaging (MRI), and a number of other applications of magnetic resonance. In this lab you will
In vivo stem cell tracking in the myocardium of small laboratory animals: Feasibility and acquisition strategies on clinical magnetic resonance imaging (MRI) systems Piotr A. Wielopolski Erasmus MC Department
MRI of the Spine Jennifer Jones Radiology Manager Bristol Nuffield COILS Used to improve the diagnostic quality of scan images. Used in conjunction with the main bore of the magnet All scanners have different
http://www.magnet.fsu.edu Disclosure Advanced Neuro MRI: Imaging Techniques and Protocol Optimization Research funding provided by Siemens Healthcare. Chen Lin, PhD DABR Indiana University School of Medicine
1 Principles of functional Magnetic Resonance Imaging Martin A. Lindquist Department of Biostatistics; Johns Hopkins University Tor D. Wager Department of Psychology & Neuroscience; University of Colorado
Data acquisition There are certain things that we have to take into account before and after we take an FID (or the spectrum, the FID is not that useful after all). Some deal with the detection system.
The MRI Study Guide for Technologists Kenneth S. Meacham The MRI Study Guide for Technologists With 51 Illustrations Springer-Verlag New York Berlin Heidelberg London Paris Tokyo Hong Kong Barcelona Budapest
An Overview of Magnetic Resonance Imaging (MRI) Academic Resource Center Table of Contents What is MRI? General MRI Machine Who is it for? How does it work? Magnetization vector Magnetization gradients
MDCT Technology Kalpana M. Kanal, Ph.D., DABR Assistant Professor Department of Radiology University of Washington Seattle, Washington ACMP Annual Meeting 2008 - Seattle, WA Educational Objectives Historical
Advances in scmos Camera Technology Benefit Bio Research scmos camera technology is gaining in popularity - Why? In recent years, cell biology has emphasized live cell dynamics, mechanisms and electrochemical
THEORY, SIMULATION, AND COMPENSATION OF PHYSIOLOGICAL MOTION ARTIFACTS IN FUNCTIONAL MRI Douglas C. Noll* and Walter Schneider Departments of *Radiology, *Electrical Engineering, and Psychology University
REVIEW Magnetic Resonance: An Introduction to Ultrashort TE (UTE) Imaging Matthew D. Robson, PhD, Peter D. Gatehouse, DPhil, Mark Bydder, PhD, and Graeme M. Bydder, MB, ChB Abstract: The background underpinning
Physics of Medical X-Ray Imaging (1) Chapter 3 CHAPTER 3: DIGITAL IMAGING IN DIAGNOSTIC RADIOLOGY 3.1 Basic Concepts of Digital Imaging Unlike conventional radiography that generates images on film through
ParaVision 6 The Next Generation of MR Acquisition and Processing for Preclinical and Material Research Innovation with Integrity Preclinical MRI A new standard in Preclinical Imaging ParaVision sets a
09 MRI SCANNERS: A BUYER S GUIDE MRI SCANNERS: A BUYER S GUIDE D. Price, I. Delakis, C. Renaud and R. Dickinson Correspondence Centre for Evidence-based Purchasing 152C Skipton House 80 London Road London
Physiological Basis of the BOLD Signal Kerstin Preuschoff Social and Neural systems Lab University of Zurich Source: Arthurs & Boniface, 2002 From Stimulus to Bold Overview Physics of BOLD signal - Magnetic
Introduction to Magnetic Resonance Imaging Techniques Lars G. Hanson, firstname.lastname@example.org Danish Research Centre for Magnetic Resonance (DRCMR), Copenhagen University Hospital Hvidovre Latest document version:
Investigation of Magnetic Resonance Imaging Effects when Using Bone Conduction Implants Master of Science Thesis in Biomedical Engineering, MPBME Karl-Johan Fredén Jansson Department of Signals and Systems
runl I IUI%I/\L Magnetic Resonance Imaging SECOND EDITION Scott A. HuetteS Brain Imaging and Analysis Center, Duke University Allen W. Song Brain Imaging and Analysis Center, Duke University Gregory McCarthy
Nuclear Magnetic Resonance Practical Course M I. Physikalisches Institut Universität zu Köln May 15, 2014 Abstract Nuclear magnetic resonance (NMR) techniques are widely used in physics, chemistry, and
Instrumental Lab Nuclear Magnetic Resonance Dr Alex J. Roche 1 Nuclear Magnetic Resonance (NMR) Spectroscopy NMR is the most powerful analytical tool currently available to an organic chemist. NMR allows
Chapter 2 NMR in Inhomogeneous Fields Federico Casanova and Juan Perlo 2.1 Introduction Besides the redesigning of the hardware to excite and detect NMR signals from sample volumes external to the sensor,
Breast MRI: Pulse Sequences, Acquisition Protocols, and Analysis Objectives Ron Price Vanderbilt University Medical Center Nashville, TN 37232 1. Review background of MRI breast cancer imaging 2. Present
Jorge Jovicich September 4, 2001 Contents 1 Introduction 2 2 MRI: a brief historical background 2 3 A MR Experiment 3 3.1 Overview............................................... 3 3.2 Can we scan the subject?......................................
Particle Surface Area TM Introducing the Acorn Area TM The Acorn Area TM is a revolutionary instrument designed to measure the surface area of nanoparticles dispersed in a liquid. This patented nuclear
Digital Transmission of Analog Data: PCM and Delta Modulation Required reading: Garcia 3.3.2 and 3.3.3 CSE 323, Fall 200 Instructor: N. Vlajic Digital Transmission of Analog Data 2 Digitization process
Phantom Test Guidance for the ACR MRI Accreditation Program The American College of Radiology 1891 Preston White Dr Reston, VA 20191-4397 www.acr.org Contents 0.0 INTRODUCTION............................................................................
NMR Pulse Spectrometer PS 15 experimental manual NMR Pulse Spectrometer PS 15 Experimental Manual for MS Windows For: MS Windows software Winner Format: MS Word 2002 File: PS15 Experimental Manual 1.5.1.doc
Nuclear Magnetic Resonance Author: James Dragan Lab Partner: Stefan Evans Physics Department, Stony Brook University, Stony Brook, NY 794. (Dated: December 5, 23) We study the principles behind Nuclear
Journal of Magnetic Resonance 164 (2003) 321 328 www.elsevier.com/locate/jmr Single-scan longitudinal relaxation measurements in high-resolution NMR spectroscopy Nikolaus M. Loening, a, * Michael J. Thrippleton,
Pulsed Fourier Transform NR The rotating frame of reference The NR Eperiment. The Rotating Frame of Reference. When we perform a NR eperiment we disturb the equilibrium state of the sstem and then monitor
The Open-Access Journal for the Basic Principles of Diffusion Theory, Experiment and Application Incorporating Internal Gradient and Restricted Diffusion Effects in Nuclear Magnetic Resonance Log Interpretation
Donna M. Reeve, MS, DABR, DABMP Department of Imaging Physics Educational Objectives Discuss the importance of breast MRI quality control (QC). Provide an overview of the new ACR Breast MRI Accreditation
Nyt indenfor MS ved MR Update: MRI in Multiple sclerosis Hartwig Roman Siebner Danish Research Centre for Magnetic Resonance (DRCMR) Copenhagen University Hospital Hvidovre Dansk Radiologisk Selskabs 10.
Functional neuroimaging Imaging brain function in real time (not just the structure of the brain). The brain is bloody & electric Blood increase in neuronal activity increase in metabolic demand for glucose
Performance testing for Precision 500D Classical R/F System John M. Boudry, Ph.D. Image Quality Systems Engineer GE Healthcare Technologies Outline System background Image Quality Signature Test (IQST)
MIAP1 Revised July 2015 MEDICAL IMAGING ADVISORY PANEL 1 Course Syllabus Magnetic Resonance Imaging Level 1 Page 1 of 13 MIAP1 MRI Course Syllabus Guide Under Revision 07 July 2015 Please refer to Medical
GE Healthcare MR Field Notes The sensitivity area relates to the diameter of the coil. By Increasing coil size, the area of sensitivity also increases. But as with most MR princicloser a coil s position
MRI in the wild Matt Rosen, Ph.D MGH/A.A. Martinos Center/Harvard University Novel Methods for Phytosequestration ARPA-E Workshop Chicago, 23 July 2015 Wild MRI? High-field MRI instruments are of limited
Protein Dynamics by NMR Why NMR is the best! Key Points NMR dynamics divided into 2 regimes: fast and slow. How protein mobons affect NMR parameters depend on whether they are faster or slower than the
14.1 An Introduction to NMR Spectroscopy A. The Basics of Nuclear Magnetic Resonance (NMR) Spectroscopy nuclei with odd atomic number have a S = ½ with two spin states (+1/2 and -1/2) 1 H NMR (proton NMR):
Lecture #7 (2D NMR) Basics of multidimensional NMR (2D NMR) 2D NOESY, COSY and TOCSY 2/23/15 Utility of Resonance Assignments Resonance Assignments: Assignment of frequency positions of resonances (peaks)
The American College of Radiology, with more than 30,000 members, is the principal organization of radiologists, radiation oncologists, and clinical medical physicists in the United States. The College
REVIEW Magnetic Resonance in Medicine 00:00 00 (2014) Recommended Implementation of Arterial Spin-Labeled Perfusion MRI for Clinical Applications: A Consensus of the ISMRM Perfusion Study Group and the
Your consent to our cookies if you continue to use this website.