# The Calculation of G rms

Size: px
Start display at page:

Transcription

1 The Calculation of G rms QualMark Corp. Neill Doertenbach The metric of G rms is typically used to specify and compare the energy in repetitive shock vibration systems. However, the method of arriving at the G rms measurement (input signal filtering, cutoff frequency of the measurement) can have a dramatic effect on the value. It is important to understand how the measurement is made, and to understand its limitations, in order to use it effectively. This paper will describe the metric of G rms, how it is calculated in both the frequency and time domains and what factors can cause variations in G rms calculations. What is G rms? Repetitive shock (RS) vibration systems produce a continuously varying pseudorandom broad spectrum vibration. A typical real time signal from an accelerometer mounted on an RS table is shown in figure 1. The root mean square (rms) value of this signal can be calculated by squaring the magnitude of the signal at every point, finding the average (mean) value of the squared magnitude, then taking the square root of the average value. The resulting number is the G rms metric. (Note: Since this paper addresses G rms calculations specifically, all of the discussion here assumes a signal source that is representative of g s (acceleration). However, the discussion would apply equally well to any measured signal.) Figure 1. Typical acceleration signal from an RS vibration system

2 Consider for a moment what the case would be if the signal from the accelerometer on the table were a sine wave rather than the complex signal you see in figure 1. The rms value of a sine wave is easy to calculate from the real time signal it is simply the peak value of the sine wave times the square root of two. The resultant number would be the G rms value for that vibration level. Similarly, the rms value of the signal shown in figure 1 can be calculated. However, since this irregular signal is not described by a straight-forward equation, it is not possible to directly calculate the rms value like you can for a sine wave. Fortunately, there are other methods for calculating the rms value of the signal. Calculating G rms from a real time signal If you need to continuously measure the rms value of a signal in a system for example, to control the vibration level of a repetitive shock vibration system one straightforward method is to employ analog rms converter circuitry 1. rms to DC converter chips are available commercially that produce an output voltage proportional to the rms value of the input signal (figure 2). Adding an input filter to the circuit will limit the frequency range of the signal that is being measured. A microprocessor based control circuit could easily monitor the output rms signal, display a G rms value and control the vibration system to maintain a G rms setpoint. Figure 2. Block diagram of rms converter chip (Source: Analog Devices data sheet, AD736) Another method is to perform the rms calculation digitally. This is done by first sampling the analog input signal, yielding a sequence of numbers corresponding to the magnitude of the input signal at each sampled point. The rms calculation now is fairly straightforward. First, each value is squared. Then, all the values are averaged together. A final square root calculation yields the rms number. This method requires the use of the appropriate anti-aliasing filters and sampling circuitry to insure an accurate measurement of the signal across the frequency range of interest. Calculating G rms from a frequency domain signal Even though the G rms signal can be easily described as a time domain measurement, it is typically thought of as a frequency domain measurement taken from the Power

3 Spectrum, or Power Spectral Density, curve. A brief review of the basics of Fourier theory will make this method of determining G rms clearer. When G rms is calculated using Power Spectrum information it is often thought of as the area under the curve of the Power Spectrum display. More accurately, it is the square root of the integral of the Power Spectrum 2. This calculation results in the same G rms value as obtained through the time domain measurements thanks to Parseval s Theorem. Parseval s Theorem (figure 3) states that the energy of a signal is the same whether calculated in the time domain or the frequency domain 3. Since the Power Spectrum display is in units of G 2, the integral of the Power Spectrum, or the area under the curve, satisfies the right side of Parseval s Theorem, while the summation of the squared values of the digitally sampled time domain signal satisfy the left side of the equation. Taking the square root of each side results in equivalent G rms calculations. h 2 ( t) dt = H ( f ) 2 df Figure 3. Parseval s Theorem When you look at the Power Spectrum of a typical vibration signal (figure 4) one thing that can be confusing is the units of the Y axis. For a Power Spectrum, the units are shown as g 2/ Hz, or often G rms 2 /Hz. In this case, the rms is not referencing the rms calculation in the time domain described above. Rather, it is an indication of the measurement used for the sinusoidal components represented in the Fourier Transform. The Fourier Transform of a signal shows the frequency and amplitude of the sine waves that, when summed, would form the time domain signal. If the amplitude of these sine waves is measured as an rms value, then the resultant Y axis units for the Power Spectrum in the frequency domain is G 2 rms /Hz. Indeed, the definition of the Power Spectrum requires that the units be in this form. 2 While some spectrum analyzers will allow choices of Y axis units that include G 2 rms /Hz, g 2 peak /Hz, etc., the only units that result in a Power Spectrum (and hence that can be used to directly calculate G rms as described above) are G 2 rms /Hz.

4 Figure 4. Power Spectrum of an RS vibration system Variations that affect G rms calculations If you use the G rms metric to specify or compare the performance of RS machines it is very important that you understand how the measurement method, and specifically the cutoff frequency and input filtering, can affect the resultant calculation. Any use of the G rms metric should include a clear description of these parameters to allow it to be interpreted correctly. If you examine the Power Spectrum display and consider that G rms is proportional to the area under that curve, you can immediately see that the cutoff frequency used for the G rms calculation can greatly affect the value calculated. With a broad band signal such as is generated from a repetitive shock vibration system, the difference between a calculation based on a 2.5 khz cutoff frequency and a 5 khz cutoff frequency can be very dramatic (figure 5j). Comparing two systems based only on the G rms calculated to 2.5 khz can result in erroneous conclusions.

5 Figure 5. rms calculation using two different cutoff frequencies. Similarly, a specification of G rms in specific frequency bands can provide a method to more accurately specify a desired spectral content in a system. Some spectrum analyzers allow this calculation only over certain ranges. However, it is not difficult to calculate G rms over any desired range using stored data from the spectrum analyzer and a spreadsheet program. Another source of variation in G rms calculations can arise when rms converter circuitry is used. The analog input filter on the circuit affects the value in a way similar to the effects of cutoff frequency in the frequency domain measurement. By filtering the input you are limiting the frequency range of the signal that is used in the rms measurement. Just like in the frequency domain example above, a measurement done with rms converter circuitry with a 2 khz input filter will be very different from one done with a 5 khz input filter. Even more variables are introduced when you try to compare the G rms values from an analog rms converter circuit to those obtained from a spectrum analyzer. A typical rms converter circuit might have an input filter set at 5kHz. However, this filter may be a single or two pole filter with the cutoff being the 3dB point of the filter. This means that, while the energy beyond 5 khz has been attenuated as described by the roll-off of the filter, it has not been immediately reduced to zero past 5 khz, as is the case with the digital cutoff of the spectrum analyzer. Also, the analog filter in the rms circuitry attenuates the signal some before the 5kHz 3 db point as well. Consequently, the G rms value provided by the rms converter can be very different from the value provided by a spectrum analyzer even though they specify the same cutoff frequency.

6 These differences can vary based on the signal content. In the example being considered here, the presence or absence of a peak in the power spectrum at 5500 Hz would have no effect on the spectrum analyzer G rms value due to the sharp digital cutoff at 5000 Hz. However, if rms converter circuitry is used, some of the energy at 5500 Hz would still be included in the G rms measurement because it would not be completely attenuated by the filter. Conclusion It has been said that G rms is one of marketing s favorite specifications you can make it whatever you want, just by choosing the cutoff frequencies correctly! This isn t too far from the truth. The calculation can be made from both time domain and frequency domain data. Each method has it s own set of variables that can affect the calculation. There is no industry standard or preferred method for doing the calculation and different methods are used in different vibration systems. Before comparing G rms values from different machines it is important to know how the measurement was made. If you are using G rms as a test specification, be sure to specify the frequency range over which the measurement is taken, and filter specifications as necessary. This will help insure the accurate reproduction of your desired tests. References 1 Analog Devices, Application Note AN-256, RMS to DC converters ease measurement tasks 2 Steinberg, Dave S., Vibration Analysis for Electronic Equipment, John Wiley & Sons, Brighan, E. Orna, The Fast Fourier Transform, Prentice Hall, Inc., Englewood Cliffs, NJ, 1974

### ANALYZER BASICS WHAT IS AN FFT SPECTRUM ANALYZER? 2-1

WHAT IS AN FFT SPECTRUM ANALYZER? ANALYZER BASICS The SR760 FFT Spectrum Analyzer takes a time varying input signal, like you would see on an oscilloscope trace, and computes its frequency spectrum. Fourier's

### Electrical Resonance

Electrical Resonance (R-L-C series circuit) APPARATUS 1. R-L-C Circuit board 2. Signal generator 3. Oscilloscope Tektronix TDS1002 with two sets of leads (see Introduction to the Oscilloscope ) INTRODUCTION

### Lock - in Amplifier and Applications

Lock - in Amplifier and Applications What is a Lock in Amplifier? In a nut shell, what a lock-in amplifier does is measure the amplitude V o of a sinusoidal voltage, V in (t) = V o cos(ω o t) where ω o

### RANDOM VIBRATION AN OVERVIEW by Barry Controls, Hopkinton, MA

RANDOM VIBRATION AN OVERVIEW by Barry Controls, Hopkinton, MA ABSTRACT Random vibration is becoming increasingly recognized as the most realistic method of simulating the dynamic environment of military

### Introduction to Digital Audio

Introduction to Digital Audio Before the development of high-speed, low-cost digital computers and analog-to-digital conversion circuits, all recording and manipulation of sound was done using analog techniques.

### Tamura Closed Loop Hall Effect Current Sensors

Tamura Closed Loop Hall Effect Current Sensors AC, DC, & Complex Currents Galvanic Isolation Fast Response Wide Frequency Bandwidth Quality & Reliability RoHs Compliance Closed Loop Hall Effect Sensors

### Making Accurate Voltage Noise and Current Noise Measurements on Operational Amplifiers Down to 0.1Hz

Author: Don LaFontaine Making Accurate Voltage Noise and Current Noise Measurements on Operational Amplifiers Down to 0.1Hz Abstract Making accurate voltage and current noise measurements on op amps in

### RF Measurements Using a Modular Digitizer

RF Measurements Using a Modular Digitizer Modern modular digitizers, like the Spectrum M4i series PCIe digitizers, offer greater bandwidth and higher resolution at any given bandwidth than ever before.

### The front end of the receiver performs the frequency translation, channel selection and amplification of the signal.

Many receivers must be capable of handling a very wide range of signal powers at the input while still producing the correct output. This must be done in the presence of noise and interference which occasionally

### Computer Networks and Internets, 5e Chapter 6 Information Sources and Signals. Introduction

Computer Networks and Internets, 5e Chapter 6 Information Sources and Signals Modified from the lecture slides of Lami Kaya (LKaya@ieee.org) for use CECS 474, Fall 2008. 2009 Pearson Education Inc., Upper

### VCO Phase noise. Characterizing Phase Noise

VCO Phase noise Characterizing Phase Noise The term phase noise is widely used for describing short term random frequency fluctuations of a signal. Frequency stability is a measure of the degree to which

### Improving A D Converter Performance Using Dither

Improving A D Converter Performance Using Dither 1 0 INTRODUCTION Many analog-to-digital converter applications require low distortion for a very wide dynamic range of signals Unfortunately the distortion

### SOFTWARE FOR GENERATION OF SPECTRUM COMPATIBLE TIME HISTORY

3 th World Conference on Earthquake Engineering Vancouver, B.C., Canada August -6, 24 Paper No. 296 SOFTWARE FOR GENERATION OF SPECTRUM COMPATIBLE TIME HISTORY ASHOK KUMAR SUMMARY One of the important

### Direct and Reflected: Understanding the Truth with Y-S 3

Direct and Reflected: Understanding the Truth with Y-S 3 -Speaker System Design Guide- December 2008 2008 Yamaha Corporation 1 Introduction Y-S 3 is a speaker system design software application. It is

### T = 1 f. Phase. Measure of relative position in time within a single period of a signal For a periodic signal f(t), phase is fractional part t p

Data Transmission Concepts and terminology Transmission terminology Transmission from transmitter to receiver goes over some transmission medium using electromagnetic waves Guided media. Waves are guided

### SIGNAL PROCESSING FOR EFFECTIVE VIBRATION ANALYSIS

SIGNAL PROCESSING FOR EFFECTIVE VIBRATION ANALYSIS Dennis H. Shreve IRD Mechanalysis, Inc Columbus, Ohio November 1995 ABSTRACT Effective vibration analysis first begins with acquiring an accurate time-varying

### How to avoid typical pitfalls when measuring Variable Frequency Drives

How to avoid typical pitfalls when measuring Variable Frequency Drives Author: Dipl.-Ing. Thomas Jäckle, Head of R&D Increased political and public focus on optimizing resource utilization has been one

### PeakVue Analysis for Antifriction Bearing Fault Detection

August 2011 PeakVue Analysis for Antifriction Bearing Fault Detection Peak values (PeakVue) are observed over sequential discrete time intervals, captured, and analyzed. The analyses are the (a) peak values

### PHYS 331: Junior Physics Laboratory I Notes on Noise Reduction

PHYS 331: Junior Physics Laboratory I Notes on Noise Reduction When setting out to make a measurement one often finds that the signal, the quantity we want to see, is masked by noise, which is anything

### SIGNAL GENERATORS and OSCILLOSCOPE CALIBRATION

1 SIGNAL GENERATORS and OSCILLOSCOPE CALIBRATION By Lannes S. Purnell FLUKE CORPORATION 2 This paper shows how standard signal generators can be used as leveled sine wave sources for calibrating oscilloscopes.

### LAB 7 MOSFET CHARACTERISTICS AND APPLICATIONS

LAB 7 MOSFET CHARACTERISTICS AND APPLICATIONS Objective In this experiment you will study the i-v characteristics of an MOS transistor. You will use the MOSFET as a variable resistor and as a switch. BACKGROUND

### How to compute Random acceleration, velocity, and displacement values from a breakpoint table.

How to compute Random acceleration, velocity, and displacement values from a breakpoint table. A random spectrum is defined as a set of frequency and amplitude breakpoints, like these: 0.050 Acceleration

### The Effective Number of Bits (ENOB) of my R&S Digital Oscilloscope Technical Paper

The Effective Number of Bits (ENOB) of my R&S Digital Oscilloscope Technical Paper Products: R&S RTO1012 R&S RTO1014 R&S RTO1022 R&S RTO1024 This technical paper provides an introduction to the signal

### Lab #9: AC Steady State Analysis

Theory & Introduction Lab #9: AC Steady State Analysis Goals for Lab #9 The main goal for lab 9 is to make the students familar with AC steady state analysis, db scale and the NI ELVIS frequency analyzer.

### LOW COST MOTOR PROTECTION FILTERS FOR PWM DRIVE APPLICATIONS STOPS MOTOR DAMAGE

LOW COST MOTOR PROTECTION FILTERS FOR PWM DRIVE APPLICATIONS STOPS MOTOR DAMAGE Karl M. Hink, Executive Vice President Originally presented at the Power Quality 99 Conference ABSTRACT Motor protection

### Frequency Response of Filters

School of Engineering Department of Electrical and Computer Engineering 332:224 Principles of Electrical Engineering II Laboratory Experiment 2 Frequency Response of Filters 1 Introduction Objectives To

### Q1. The graph below shows how a sinusoidal alternating voltage varies with time when connected across a resistor, R.

Q1. The graph below shows how a sinusoidal alternating voltage varies with time when connected across a resistor, R. (a) (i) State the peak-to-peak voltage. peak-to-peak voltage...v (1) (ii) State the

### Acceleration levels of dropped objects

Acceleration levels of dropped objects cmyk Acceleration levels of dropped objects Introduction his paper is intended to provide an overview of drop shock testing, which is defined as the acceleration

### Department of Electrical and Computer Engineering Ben-Gurion University of the Negev. LAB 1 - Introduction to USRP

Department of Electrical and Computer Engineering Ben-Gurion University of the Negev LAB 1 - Introduction to USRP - 1-1 Introduction In this lab you will use software reconfigurable RF hardware from National

### FAST Fourier Transform (FFT) and Digital Filtering Using LabVIEW

FAST Fourier Transform (FFT) and Digital Filtering Using LabVIEW Wei Lin Department of Biomedical Engineering Stony Brook University Instructor s Portion Summary This experiment requires the student to

### PROCEDURE: 1. Measure and record the actual values of the four resistors listed in Table 10-1.

The answer to two questions will help you identify a series or parallel connection: (1) Will the identical current go through both components? f the answer is yes, the components are in series. (2) Are

### Principles of Adjustable Frequency Drives

What is an Adjustable Frequency Drive? An adjustable frequency drive is a system for controlling the speed of an AC motor by controlling the frequency of the power supplied to the motor. A basic adjustable

### Transistor Characteristics and Single Transistor Amplifier Sept. 8, 1997

Physics 623 Transistor Characteristics and Single Transistor Amplifier Sept. 8, 1997 1 Purpose To measure and understand the common emitter transistor characteristic curves. To use the base current gain

### FREQUENCY CONTROLLED AC MOTOR DRIVE

FREQUENCY CONTROLLED AC MOTOR DRIVE 1.0 Features of Standard AC Motors The squirrel cage induction motor is the electrical motor motor type most widely used in industry. This leading position results mainly

### CHAPTER 6 Frequency Response, Bode Plots, and Resonance

ELECTRICAL CHAPTER 6 Frequency Response, Bode Plots, and Resonance 1. State the fundamental concepts of Fourier analysis. 2. Determine the output of a filter for a given input consisting of sinusoidal

### MICROPHONE SPECIFICATIONS EXPLAINED

Application Note AN-1112 MICROPHONE SPECIFICATIONS EXPLAINED INTRODUCTION A MEMS microphone IC is unique among InvenSense, Inc., products in that its input is an acoustic pressure wave. For this reason,

### LAB 12: ACTIVE FILTERS

A. INTRODUCTION LAB 12: ACTIVE FILTERS After last week s encounter with op- amps we will use them to build active filters. B. ABOUT FILTERS An electric filter is a frequency-selecting circuit designed

### Dithering in Analog-to-digital Conversion

Application Note 1. Introduction 2. What is Dither High-speed ADCs today offer higher dynamic performances and every effort is made to push these state-of-the art performances through design improvements

### 14.5GHZ 2.2KW CW GENERATOR. GKP 22KP 14.5GHz WR62 3x400V

14.5GHZ 2.2KW CW GENERATOR GKP 22KP 14.5GHz WR62 3x400V UTILIZATION OF GKP 22KP GENERATOR With its characteristics of power stability whatever the load, very fast response time at a pulse, low ripple,

### electronics fundamentals

electronics fundamentals circuits, devices, and applications THOMAS L. FLOYD DAVID M. BUCHLA Lesson 1: Diodes and Applications Center-Tapped Full-wave Rectifier The center-tapped (CT) full-wave rectifier

### Basics of Digital Recording

Basics of Digital Recording CONVERTING SOUND INTO NUMBERS In a digital recording system, sound is stored and manipulated as a stream of discrete numbers, each number representing the air pressure at a

### Analysis/resynthesis with the short time Fourier transform

Analysis/resynthesis with the short time Fourier transform summer 2006 lecture on analysis, modeling and transformation of audio signals Axel Röbel Institute of communication science TU-Berlin IRCAM Analysis/Synthesis

### The Fourier Analysis Tool in Microsoft Excel

The Fourier Analysis Tool in Microsoft Excel Douglas A. Kerr Issue March 4, 2009 ABSTRACT AD ITRODUCTIO The spreadsheet application Microsoft Excel includes a tool that will calculate the discrete Fourier

### Switch Mode Power Supply Topologies

Switch Mode Power Supply Topologies The Buck Converter 2008 Microchip Technology Incorporated. All Rights Reserved. WebSeminar Title Slide 1 Welcome to this Web seminar on Switch Mode Power Supply Topologies.

### Physics 623 Transistor Characteristics and Single Transistor Amplifier Sept. 13, 2006

Physics 623 Transistor Characteristics and Single Transistor Amplifier Sept. 13, 2006 1 Purpose To measure and understand the common emitter transistor characteristic curves. To use the base current gain

### Loop Bandwidth and Clock Data Recovery (CDR) in Oscilloscope Measurements. Application Note 1304-6

Loop Bandwidth and Clock Data Recovery (CDR) in Oscilloscope Measurements Application Note 1304-6 Abstract Time domain measurements are only as accurate as the trigger signal used to acquire them. Often

### Lecture 1-6: Noise and Filters

Lecture 1-6: Noise and Filters Overview 1. Periodic and Aperiodic Signals Review: by periodic signals, we mean signals that have a waveform shape that repeats. The time taken for the waveform to repeat

### 1. Oscilloscope is basically a graph-displaying device-it draws a graph of an electrical signal.

CHAPTER 3: OSCILLOSCOPE AND SIGNAL GENERATOR 3.1 Introduction to oscilloscope 1. Oscilloscope is basically a graph-displaying device-it draws a graph of an electrical signal. 2. The graph show signal change

### CIRCUITS LABORATORY EXPERIMENT 3. AC Circuit Analysis

CIRCUITS LABORATORY EXPERIMENT 3 AC Circuit Analysis 3.1 Introduction The steady-state behavior of circuits energized by sinusoidal sources is an important area of study for several reasons. First, the

### Experiment 3: Double Sideband Modulation (DSB)

Experiment 3: Double Sideband Modulation (DSB) This experiment examines the characteristics of the double-sideband (DSB) linear modulation process. The demodulation is performed coherently and its strict

### Selecting IHLP Composite Inductors for Non-Isolated Converters Utilizing Vishay s Application Sheet

VISHAY DALE www.vishay.com Magnetics Selecting IHLP Composite Inductors for Non-Isolated Converters INTRODUCTION This application note will provide information to assist in the specification of IHLP composite

### LABORATORY 10 TIME AVERAGES, RMS VALUES AND THE BRIDGE RECTIFIER. Bridge Rectifier

LABORATORY 10 TIME AVERAGES, RMS VALUES AND THE BRIDGE RECTIFIER Full-wave Rectification: Bridge Rectifier For many electronic circuits, DC supply voltages are required but only AC voltages are available.

### Auto-Tuning Using Fourier Coefficients

Auto-Tuning Using Fourier Coefficients Math 56 Tom Whalen May 20, 2013 The Fourier transform is an integral part of signal processing of any kind. To be able to analyze an input signal as a superposition

### EXPERIMENT NUMBER 5 BASIC OSCILLOSCOPE OPERATIONS

1 EXPERIMENT NUMBER 5 BASIC OSCILLOSCOPE OPERATIONS The oscilloscope is the most versatile and most important tool in this lab and is probably the best tool an electrical engineer uses. This outline guides

### DIGITAL-TO-ANALOGUE AND ANALOGUE-TO-DIGITAL CONVERSION

DIGITAL-TO-ANALOGUE AND ANALOGUE-TO-DIGITAL CONVERSION Introduction The outputs from sensors and communications receivers are analogue signals that have continuously varying amplitudes. In many systems

### Analog Filters. A common instrumentation filter application is the attenuation of high frequencies to avoid frequency aliasing in the sampled data.

Analog Filters Filters can be used to attenuate unwanted signals such as interference or noise or to isolate desired signals from unwanted. They use the frequency response of a measuring system to alter

### Accurate Measurement of the Mains Electricity Frequency

Accurate Measurement of the Mains Electricity Frequency Dogan Ibrahim Near East University, Faculty of Engineering, Lefkosa, TRNC dogan@neu.edu.tr Abstract The frequency of the mains electricity supply

### Trigonometric functions and sound

Trigonometric functions and sound The sounds we hear are caused by vibrations that send pressure waves through the air. Our ears respond to these pressure waves and signal the brain about their amplitude

### SIGNAL PROCESSING & SIMULATION NEWSLETTER

1 of 10 1/25/2008 3:38 AM SIGNAL PROCESSING & SIMULATION NEWSLETTER Note: This is not a particularly interesting topic for anyone other than those who ar e involved in simulation. So if you have difficulty

### Decoding an Accelerometer Specification. What Sensor Manufacturer s Don t Tell You! David Lally VP Engineering PCB Piezotronics, Inc.

Decoding an Accelerometer Specification Sheet What Sensor Manufacturer s Don t Tell You! David Lally VP Engineering PCB Piezotronics, Inc. IMAC-XXVII February 2008 1 Overview Specification Sheet Provides

### PRACTICAL GUIDE TO DATA SMOOTHING AND FILTERING

PRACTICAL GUIDE TO DATA SMOOTHING AND FILTERING Ton van den Bogert October 3, 996 Summary: This guide presents an overview of filtering methods and the software which is available in the HPL.. What is

### SR2000 FREQUENCY MONITOR

SR2000 FREQUENCY MONITOR THE FFT SEARCH FUNCTION IN DETAILS FFT Search is a signal search using FFT (Fast Fourier Transform) technology. The FFT search function first appeared with the SR2000 Frequency

### Similar benefits are also derived through modal testing of other space structures.

PAGE 1 OF 5 PREFERRED RELIABILITY PRACTICES MODAL TESTING: MEASURING DYNAMIC STRUCTURAL CHARACTERISTICS Practice: Modal testing is a structural testing practice that provides low levels of mechanical excitation

### Introduction to Digital Filters

CHAPTER 14 Introduction to Digital Filters Digital filters are used for two general purposes: (1) separation of signals that have been combined, and (2) restoration of signals that have been distorted

### Basic Acoustics and Acoustic Filters

Basic CHAPTER Acoustics and Acoustic Filters 1 3 Basic Acoustics and Acoustic Filters 1.1 The sensation of sound Several types of events in the world produce the sensation of sound. Examples include doors

### QUICK START GUIDE FOR DEMONSTRATION CIRCUIT 956 24-BIT DIFFERENTIAL ADC WITH I2C LTC2485 DESCRIPTION

LTC2485 DESCRIPTION Demonstration circuit 956 features the LTC2485, a 24-Bit high performance Σ analog-to-digital converter (ADC). The LTC2485 features 2ppm linearity, 0.5µV offset, and 600nV RMS noise.

### DIODE CIRCUITS LABORATORY. Fig. 8.1a Fig 8.1b

DIODE CIRCUITS LABORATORY A solid state diode consists of a junction of either dissimilar semiconductors (pn junction diode) or a metal and a semiconductor (Schottky barrier diode). Regardless of the type,

### Chapter 8 - Power Density Spectrum

EE385 Class Notes 8/8/03 John Stensby Chapter 8 - Power Density Spectrum Let X(t) be a WSS random process. X(t) has an average power, given in watts, of E[X(t) ], a constant. his total average power is

### Programmable-Gain Transimpedance Amplifiers Maximize Dynamic Range in Spectroscopy Systems

Programmable-Gain Transimpedance Amplifiers Maximize Dynamic Range in Spectroscopy Systems PHOTODIODE VOLTAGE SHORT-CIRCUIT PHOTODIODE SHORT- CIRCUIT VOLTAGE 0mV DARK ark By Luis Orozco Introduction Precision

### TESTS OF 1 MHZ SIGNAL SOURCE FOR SPECTRUM ANALYZER CALIBRATION 7/8/08 Sam Wetterlin

TESTS OF 1 MHZ SIGNAL SOURCE FOR SPECTRUM ANALYZER CALIBRATION 7/8/08 Sam Wetterlin (Updated 7/19/08 to delete sine wave output) I constructed the 1 MHz square wave generator shown in the Appendix. This

### Current and Temperature Ratings

Document 361-1 Current and Temperature Ratings Introduction This application note describes: How to interpret Coilcraft inductor current and temperature ratings Our current ratings measurement method and

### Lab 1: The Digital Oscilloscope

PHYSICS 220 Physical Electronics Lab 1: The Digital Oscilloscope Object: To become familiar with the oscilloscope, a ubiquitous instrument for observing and measuring electronic signals. Apparatus: Tektronix

### Measuring Impedance and Frequency Response of Guitar Pickups

Measuring Impedance and Frequency Response of Guitar Pickups Peter D. Hiscocks Syscomp Electronic Design Limited phiscock@ee.ryerson.ca www.syscompdesign.com April 30, 2011 Introduction The CircuitGear

### Filter Comparison. Match #1: Analog vs. Digital Filters

CHAPTER 21 Filter Comparison Decisions, decisions, decisions! With all these filters to choose from, how do you know which to use? This chapter is a head-to-head competition between filters; we'll select

### Spectrum Level and Band Level

Spectrum Level and Band Level ntensity, ntensity Level, and ntensity Spectrum Level As a review, earlier we talked about the intensity of a sound wave. We related the intensity of a sound wave to the acoustic

### ANIMA: Non-Conventional Interfaces in Robot Control Through Electroencephalography and Electrooculography: Motor Module

Ninth LACCEI Latin American and Caribbean Conference (LACCEI 2011), Engineering for a Smart Planet, Innovation, Information Technology and Computational Tools for Sustainable Development, August 3-5, 2011,

### Lab 1. The Fourier Transform

Lab 1. The Fourier Transform Introduction In the Communication Labs you will be given the opportunity to apply the theory learned in Communication Systems. Since this is your first time to work in the

### Designing Gain and Offset in Thirty Seconds

Application Report SLOA097 February 2002 Designing Gain and Offset in Thirty Seconds Bruce Carter High Performance Linear ABSTRACT This document discusses how to design an operational amplifier (op amp)

### Network Theory Question Bank

Network Theory Question Bank Unit-I JNTU SYLLABUS: Three Phase Circuits Three phase circuits: Phase sequence Star and delta connection Relation between line and phase voltages and currents in balanced

### AVX EMI SOLUTIONS Ron Demcko, Fellow of AVX Corporation Chris Mello, Principal Engineer, AVX Corporation Brian Ward, Business Manager, AVX Corporation

AVX EMI SOLUTIONS Ron Demcko, Fellow of AVX Corporation Chris Mello, Principal Engineer, AVX Corporation Brian Ward, Business Manager, AVX Corporation Abstract EMC compatibility is becoming a key design

Introduction to Receivers Purpose: translate RF signals to baseband Shift frequency Amplify Filter Demodulate Why is this a challenge? Interference (selectivity, images and distortion) Large dynamic range

### Section 3. Sensor to ADC Design Example

Section 3 Sensor to ADC Design Example 3-1 This section describes the design of a sensor to ADC system. The sensor measures temperature, and the measurement is interfaced into an ADC selected by the systems

### A Sound Analysis and Synthesis System for Generating an Instrumental Piri Song

, pp.347-354 http://dx.doi.org/10.14257/ijmue.2014.9.8.32 A Sound Analysis and Synthesis System for Generating an Instrumental Piri Song Myeongsu Kang and Jong-Myon Kim School of Electrical Engineering,

### MATRIX TECHNICAL NOTES

200 WOOD AVENUE, MIDDLESEX, NJ 08846 PHONE (732) 469-9510 FAX (732) 469-0418 MATRIX TECHNICAL NOTES MTN-107 TEST SETUP FOR THE MEASUREMENT OF X-MOD, CTB, AND CSO USING A MEAN SQUARE CIRCUIT AS A DETECTOR

### The accelerometer designed and realized so far is intended for an. aerospace application. Detailed testing and analysis needs to be

86 Chapter 4 Accelerometer Testing 4.1 Introduction The accelerometer designed and realized so far is intended for an aerospace application. Detailed testing and analysis needs to be conducted to qualify

### SUMMARY. Additional Digital/Software filters are included in Chart and filter the data after it has been sampled and recorded by the PowerLab.

This technique note was compiled by ADInstruments Pty Ltd. It includes figures and tables from S.S. Young (2001): Computerized data acquisition and analysis for the life sciences. For further information

### AC 2012-3923: MEASUREMENT OF OP-AMP PARAMETERS USING VEC- TOR SIGNAL ANALYZERS IN UNDERGRADUATE LINEAR CIRCUITS LABORATORY

AC 212-3923: MEASUREMENT OF OP-AMP PARAMETERS USING VEC- TOR SIGNAL ANALYZERS IN UNDERGRADUATE LINEAR CIRCUITS LABORATORY Dr. Tooran Emami, U.S. Coast Guard Academy Tooran Emami received her M.S. and Ph.D.

### Active Vibration Isolation of an Unbalanced Machine Spindle

UCRL-CONF-206108 Active Vibration Isolation of an Unbalanced Machine Spindle D. J. Hopkins, P. Geraghty August 18, 2004 American Society of Precision Engineering Annual Conference Orlando, FL, United States

### Exploring the Necessity of the Hot Hipot Test

Exploring the Necessity of the Hot Hipot Test Introduction In an industry comprised of workers with varying electronics knowledge, a Hipot test can seem a daunting task for some. Indeed, many test operators

### Rotating Machinery Diagnostics & Instrumentation Solutions for Maintenance That Matters www.mbesi.com

13 Aberdeen Way Elgin, SC 29045 Cell (803) 427-0791 VFD Fundamentals & Troubleshooting 19-Feb-2010 By: Timothy S. Irwin, P.E. Sr. Engineer tsi@mbesi.com Rotating Machinery Diagnostics & Instrumentation

### What you will do. Build a 3-band equalizer. Connect to a music source (mp3 player) Low pass filter High pass filter Band pass filter

Audio Filters What you will do Build a 3-band equalizer Low pass filter High pass filter Band pass filter Connect to a music source (mp3 player) Adjust the strength of low, high, and middle frequencies

### H/L = Power supply. FN = Nominal Frequency. RN = Rated value of output burden. UN = Nominal input voltage. IN = Nominal input current.

Application : The transducer RISH CON - CA/CV (Fig.1) converts a sinusoidal AC Current or AC Voltage into a load independent DC Current or a load independent DC Voltage proportional to the measured value.

### ε: Voltage output of Signal Generator (also called the Source voltage or Applied

Experiment #10: LR & RC Circuits Frequency Response EQUIPMENT NEEDED Science Workshop Interface Power Amplifier (2) Voltage Sensor graph paper (optional) (3) Patch Cords Decade resistor, capacitor, and

### This document is downloaded from DR-NTU, Nanyang Technological University Library, Singapore.

This document is downloaded from DR-NTU, Nanyang Technological University Library, Singapore. Title Transcription of polyphonic signals using fast filter bank( Accepted version ) Author(s) Foo, Say Wei;

### EDEXCEL NATIONAL CERTIFICATE/DIPLOMA UNIT 5 - ELECTRICAL AND ELECTRONIC PRINCIPLES NQF LEVEL 3 OUTCOME 4 - ALTERNATING CURRENT

EDEXCEL NATIONAL CERTIFICATE/DIPLOMA UNIT 5 - ELECTRICAL AND ELECTRONIC PRINCIPLES NQF LEVEL 3 OUTCOME 4 - ALTERNATING CURRENT 4 Understand single-phase alternating current (ac) theory Single phase AC

### Cell Phone Vibration Experiment

Objective Cell Phone Vibration Experiment Most cell phones are designed to vibrate. But at what frequency do they vibrate? With an accelerometer, data acquisition and signal analysis the vibration frequency

### Basics of Structural Vibration Testing and Analysis

APPLICATION NOTE AN011 Basics of Structural Vibration Testing and Analysis Introduction Structural vibration testing and analysis contributes to progress in many industries, including aerospace, auto-making,