FAST Fourier Transform (FFT) and Digital Filtering Using LabVIEW


 Joseph Carson
 2 years ago
 Views:
Transcription
1 FAST Fourier Transform (FFT) and Digital Filtering Using LabVIEW Wei Lin Department of Biomedical Engineering Stony Brook University Instructor s Portion Summary This experiment requires the student to use LabVIEW to perform signal analysis on the acquired analog signals. Students should be familiar with the Fast Fourier Transform (FFT) and digital filtering using LabVIEW Uses This lecture applies to all courses of virtual instrumentation. Equipment List Computers LabVIEW 8.5 Express NIELVIS benchtop workstation References Lecture Slides of Data Analysis Using LabVIEW VIs from the project Data Acquisition Using NIDAQmx Student s Portion Introduction The students should learn the basic LabVIEW programming techniques for the FFT and digital filtering. They will modify two VIs developed in the 1
2 project Data Acquisition Using NIDAQmx and add frequency spectrum analysis and digital filter in the two VIs. Objectives Theory Perform FFT analysis using LabVIEW Frequency Spectrum Express VI Design and test digital filter using LabVIEW Filter Express VI The Fast Fourier Transform (FFT) and the power spectrum are powerful tools for analyzing and measuring signals from plugin data acquisition (DAQ) devices. It can measure the frequency components within the signal. FFTbased measurement requires digitization of a continuous signal. According to the Nyquist criterion, the sampling frequency, F s, must be at least twice the maximum frequency component in the signal. If this criterion is violated, a phenomenon known as aliasing occurs. When the Nyquist criterion is violated, frequency components above half the sampling frequency appear as frequency components below half the sampling frequency, resulting in an erroneous representation of the signal. This is called frequency aliasing. Therefore, before a signal is digitized, antialiasing filters are used to attenuate the frequency components at and above half the sampling frequency to a level below the dynamic range of the analogtodigital converter (ADC). Spectral leakage is another issue in FFT analysis. It is the result of an assumption in the FFT algorithm that the time record is exactly repeated throughout all time and that signals contained in a time record are thus periodic at intervals that correspond to the length of the time record. If the time record has a nonintegral number of cycles, this assumption is violated and spectral leakage occurs. To alleviate the spectrum leakage, a predefine window function is applied to the signals to be analyzed. Filters alter or remove unwanted frequencies from your signal. Depending on the frequency range that they either pass or attenuate, they can be classified into the following types: A lowpass filter passes low frequencies, but attenuates high frequencies. A highpass filter passes high frequencies, but attenuates low frequencies. A bandpass filter passes a certain band of frequencies. A bandstop filter attenuates a certain band of frequencies. Digital filters have the following advantages over their analog counterparts: 2
3 They are programmable in terms of filter order, cutoff frequencies, and amount of ripple. They are stable and predictable. They do not drift with temperature or humidity and do not require precision components. They have a superior performancetocost ratio. Therefore, it is recommended to use digital filters in instrumentation except for the antialiasing filter. LabVIEW has two express VIs for FFT analysis and digital filtering. They are frequency spectrum express VI and filter express VI. Lab Procedure Experiment 1: Create a LabVIEW application of frequency spectrum analysis: 1. Launch LabVIEW. 2. Load finite data acquisition VI. 3. Add the frequency spectrum VI (Express>Signal Analysis >Spectral). Configure the output of the express VI as linear in result section and set window as None. 4. Connect the waveform data from DAQmx read VI to the signal terminal of the frequency spectrum express VI. 5. Add a graph indicator on the front panel and connect the output (FFT(RMS)) of the frequency spectrum VI (FFT) to the graph. 6. Keep the ELVIS unit off. Connect the output of function generator FUNC OUT to ACH0 using connection wire on the prototype board. 7. Turn ELVIS unit on including the prototype board. 8. Launch ELVIS application and choose function generator. 9. Enter the parameters for the controls of the LabVIEW. Recommended sampling frequency is 1000 Hz and number of samples is Set the frequency, and function type of the signal generator (recommended function: sine wave, frequency 20 Hz. and run the VI and observe the frequency spectrum. In this case, you should collected integer number of cycles of the signal. Therefore, no spectrum leakage is present. 11. Change the signal frequency or the number of samples so that there is a disconnection at the end of the signal (non integer number of cycles collected) and observe the spectrum leakage. 12. You may need to add write Meas file VI to same the signal and the FFT result for your lab report. 13. Save the VI 3
4 Experiment 2, Create a LabVIEW application of digital filter: 1. Use the VI from previous experiment. 2. Add two more graph indictors on the front panel for the filtered signal and the spectrum of the filtered signal. 3. Add a filter express VI (Express>Signal Analysis>Filter) and connect the output from the DAQmx VI to the signal terminal of the filter VI 4. Connect the output of filter to the graph indicator of filtered signal. 5. Add another frequency spectrum express VI (Express>Signal Analysis>Spectral) on the block diagram and connect the filtered signal to its input. 6. Connect the output of the above frequency spectrum express VI to the graph indicator of the filtered signal spectrum. 7. Combine the two dynamic data, the spectrum of the original signal and the spectrum of the filtered signal using signal manipulation VI (Express>Signal Manipulate>merge signals) and connect it to the Write LabVIEW Measurement Data express VI (Express >Output>write meas data). In the wizard, select a. Ask user to choose file b. Ask each iteration c. If a file already exists, overwrite file d. X value columns: one column per channel 8. Run the application and record the results in files. The following are the suggestions for the test of digital filters. a. Use square wave at low frequency, far below the sampling frequency. (20 Hz) b. Change the cutoff frequency of the filter. c. Change the type of filter, e.g. lowpass or highpass. d. Change the order of the filter. (optional) e. Change the topology of the filter. (optional) Experiment 3. (Extra credits): Using FFT to measure the cutoff frequencies of the band pass filter you made in the previous lab. The expected solution will use one or two measurements to estimate the cutoff frequencies instead of using multiple sinusoidal signals. Lab Report 1. Objective. 2. Theory (What it FFT? Filter?) 3. Data: Demonstrate frequency spectrum of the tested signal and filter effects. 4. Due on 11/14/2008 4
5 Reference VIs FFT VI Filter VI 5
Data Acquisition Using NIDAQmx
Instructor s Portion Data Acquisition Using NIDAQmx Wei Lin Department of Biomedical Engineering Stony Brook University Summary This experiment requires the student to use NIDAQmx to acquire voltage
More informationADC Familiarization. Dmitry Teytelman and Dan Van Winkle. Student name:
Dmitry Teytelman and Dan Van Winkle Student name: RF and Digital Signal Processing US Particle Accelerator School 15 19 June, 2009 June 16, 2009 Contents 1 Introduction 2 2 Exercises 2 2.1 Measuring 48
More informationLab 3: Introduction to Data Acquisition Cards
Lab 3: Introduction to Data Acquisition Cards INTRODUCTION: In this lab, you will be building a VI to display the input measured on a channel. However, within your own VI you will use LabVIEW supplied
More informationCell Phone Vibration Experiment
Objective Cell Phone Vibration Experiment Most cell phones are designed to vibrate. But at what frequency do they vibrate? With an accelerometer, data acquisition and signal analysis the vibration frequency
More informationAnalog Filters. A common instrumentation filter application is the attenuation of high frequencies to avoid frequency aliasing in the sampled data.
Analog Filters Filters can be used to attenuate unwanted signals such as interference or noise or to isolate desired signals from unwanted. They use the frequency response of a measuring system to alter
More informationRightMark Audio Analyzer
RightMark Audio Analyzer Version 2.5 2001 http://audio.rightmark.org Tests description 1 Contents FREQUENCY RESPONSE TEST... 2 NOISE LEVEL TEST... 3 DYNAMIC RANGE TEST... 5 TOTAL HARMONIC DISTORTION TEST...
More informationCONVERTERS. Filters Introduction to Digitization DigitaltoAnalog Converters AnalogtoDigital Converters
CONVERTERS Filters Introduction to Digitization DigitaltoAnalog Converters AnalogtoDigital Converters Filters Filters are used to remove unwanted bandwidths from a signal Filter classification according
More informationAnalog Representations of Sound
Analog Representations of Sound Magnified phonograph grooves, viewed from above: The shape of the grooves encodes the continuously varying audio signal. Analog to Digital Recording Chain ADC Microphone
More informationFast Fourier Transforms and Power Spectra in LabVIEW
Application Note 4 Introduction Fast Fourier Transforms and Power Spectra in LabVIEW K. Fahy, E. Pérez Ph.D. The Fourier transform is one of the most powerful signal analysis tools, applicable to a wide
More informationSAMPLE: EXPERIMENT 10 Bandpass Filter / Fourier Analysis
SAMPLE: EXPERIMENT 10 Bandpass Filter / Fourier Analysis  This experiment is an excerpt from: Electric
More informationLAB 4 DIGITAL FILTERING
LAB 4 DIGITAL FILTERING 1. LAB OBJECTIVE The purpose of this lab is to design and implement digital filters. You will design and implement a first order IIR (Infinite Impulse Response) filter and a second
More informationLab 4 Op Amp Filters
Lab 4 Op Amp Filters Figure 4.0. Frequency Characteristics of a BandPass Filter Adding a few capacitors and resistors to the basic operational amplifier (op amp) circuit can yield many interesting analog
More informationDigital to Analog Conversion Using Pulse Width Modulation
Digital to Analog Conversion Using Pulse Width Modulation Samer ElHajMahmoud Electronics Engineering Technology Program Texas A&M University Instructor s Portion Summary The purpose of this lab is to
More informationLab 4  Data Acquisition
Spring 11 Lab 4  Data Acquisition Lab 41 Lab 4  Data Acquisition Format This lab will be conducted during your regularly scheduled lab time in a group format. Each student is responsible for learning
More informationChapter 3 DiscreteTime Fourier Series. by the French mathematician Jean Baptiste Joseph Fourier in the early 1800 s. The
Chapter 3 DiscreteTime Fourier Series 3.1 Introduction The Fourier series and Fourier transforms are mathematical correlations between the time and frequency domains. They are the result of the heattransfer
More informationFIR Filter Design. FIR Filters and the zdomain. The zdomain model of a general FIR filter is shown in Figure 1. Figure 1
FIR Filters and the Domain FIR Filter Design The domain model of a general FIR filter is shown in Figure. Figure Each  box indicates a further delay of one sampling period. For example, the input to
More informationIn modern electronics, it is important to be able to separate a signal into different
Introduction In modern electronics, it is important to be able to separate a signal into different frequency regions. In analog electronics, four classes of filters exist to process an input signal: lowpass,
More informationBasics on Digital Signal Processing
Basics on Digital Signal Processing Introduction Vassilis Anastassopoulos Electronics Laboratory, Physics Department, University of Patras Outline of the Course 1. Introduction (sampling quantization)
More informationchapter Introduction to Digital Signal Processing and Digital Filtering 1.1 Introduction 1.2 Historical Perspective
Introduction to Digital Signal Processing and Digital Filtering chapter 1 Introduction to Digital Signal Processing and Digital Filtering 1.1 Introduction Digital signal processing (DSP) refers to anything
More informationAnalog signals are those which are naturally occurring. Any analog signal can be converted to a digital signal.
3.3 Analog to Digital Conversion (ADC) Analog signals are those which are naturally occurring. Any analog signal can be converted to a digital signal. 1 3.3 Analog to Digital Conversion (ADC) WCB/McGrawHill
More informationFUNDAMENTALS OF MODERN SPECTRAL ANALYSIS. Matthew T. Hunter, Ph.D.
FUNDAMENTALS OF MODERN SPECTRAL ANALYSIS Matthew T. Hunter, Ph.D. AGENDA Introduction Spectrum Analyzer Architecture Dynamic Range Instantaneous Bandwidth The Importance of Image Rejection and AntiAliasing
More informationChapter 5. Basic Filters
Chapter 5 Basic Filters 39 CHAPTER 5. BASIC FILTERS 5.1 PreLab The answers to the following questions are due at the beginning of the lab. If they are not done at the beginning of the lab, no points will
More informationDepartment of Electrical and Computer Engineering BenGurion University of the Negev. LAB 1  Introduction to USRP
Department of Electrical and Computer Engineering BenGurion University of the Negev LAB 1  Introduction to USRP  11 Introduction In this lab you will use software reconfigurable RF hardware from National
More informationAircraft cabin noise synthesis for noise subjective analysis
Aircraft cabin noise synthesis for noise subjective analysis Bruno Arantes Caldeira da Silva Instituto Tecnológico de Aeronáutica São José dos Campos  SP brunoacs@gmail.com Cristiane Aparecida Martins
More information2 of 6 09/27/ :42 PM Figure 2: A system that has a sampling frequency at f s (a) will digitize signals with frequencies below f s /2 as well as
1 of 6 09/27/2006 05:42 PM Tutorial: Basics of choosing and designing the best filter for an effective dataacquisition system By Bonnie C. Baker, Senior Applications Engineer, Texas Instruments, Inc.
More informationPerformance of an IF sampling ADC in receiver applications
Performance of an IF sampling ADC in receiver applications David Buchanan Staff Applications Engineer Analog Devices, Inc. Introduction The concept of direct intermediate frequency (IF) sampling is not
More informationLecture 16: Noise and Filters
Lecture 16: Noise and Filters Overview 1. Periodic and Aperiodic Signals Review: by periodic signals, we mean signals that have a waveform shape that repeats. The time taken for the waveform to repeat
More informationModule 4. Contents. Digital Filters  Implementation and Design. Signal Flow Graphs. Digital Filter Structures. FIR and IIR Filter Design Techniques
Module 4 Digital Filters  Implementation and Design Digital Signal Processing. Slide 4.1 Contents Signal Flow Graphs Basic filtering operations Digital Filter Structures Direct form FIR and IIR filters
More informationApplication Note 9. Digital FIR Decimator & Analog Lowpass
Application Note 9 App Note Application Note 9 Highlights Multirate FIR Design Cascade Analog Lowpass Circuit Optimization Comb Filter Correction Sin(x)/x Correction n Design Objective 16:1 FIR Decimation
More informationThe Calculation of G rms
The Calculation of G rms QualMark Corp. Neill Doertenbach The metric of G rms is typically used to specify and compare the energy in repetitive shock vibration systems. However, the method of arriving
More informationSHOCK AND VIBRATION RESPONSE SPECTRA COURSE Unit 18. Filtering
SHOCK AND VIBRATION RESPONSE SPECTRA COURSE Unit 18. Filtering By Tom Irvine Email: tomirvine@aol.com Introduction Filtering is a tool for resolving signals. Filtering can be performed on either analog
More informationLABVIEW DSP TEST TOOLKIT FOR TI DSP
LABVIEW DSP TEST TOOLKIT FOR TI DSP Contents The LabVIEW DSP Test Toolkit for TI DSP uses LabVIEW to automate TI s Code Composer Studio (CCS) Integrated Development Environment (IDE) so you can create
More informationDigital Filter Plus User's Guide. Version January, 2015
Digital Filter Plus User's Guide Version 2.50 3 January, 2015 2014 Numerix Ltd. Email : mailto:numerix@numerixdsp.com WWW : http://www.numerixdsp.com/ INTRODUCTION 3 INSTALLATION 4 USING DIGITAL FILTER
More informationIntroduction to Digital Audio
Introduction to Digital Audio Before the development of highspeed, lowcost digital computers and analogtodigital conversion circuits, all recording and manipulation of sound was done using analog techniques.
More informationCHAPTER 6 Frequency Response, Bode Plots, and Resonance
ELECTRICAL CHAPTER 6 Frequency Response, Bode Plots, and Resonance 1. State the fundamental concepts of Fourier analysis. 2. Determine the output of a filter for a given input consisting of sinusoidal
More informationThree Levels of DataLogging Software from National Instruments
Three Levels of DataLogging Software from National Instruments One of the most common data acquisition applications is simply logging acquired data to disk or a database for future analysis. Most National
More informationLabVIEW Signal Processing Course Manual
LabVIEW Signal Processing Course Manual LabVIEW Signal Processing Course Manual Course Software Version 1.0 September 1997 Edition Part Number 321569A01 Copyright Copyright 1997 by National Instruments
More informationSUMMARY. Additional Digital/Software filters are included in Chart and filter the data after it has been sampled and recorded by the PowerLab.
This technique note was compiled by ADInstruments Pty Ltd. It includes figures and tables from S.S. Young (2001): Computerized data acquisition and analysis for the life sciences. For further information
More informationLab 5 Getting started with analogdigital conversion
Lab 5 Getting started with analogdigital conversion Achievements in this experiment Practical knowledge of coding of an analog signal into a train of digital codewords in binary format using pulse code
More informationTeaching Speech and Audio Processing Implementations Using LabView Program and DAQ Boards
Paper ID #5984 Teaching Speech and Audio Processing Implementations Using LabView Program and DAQ Boards Prof. Jean Jiang, Purdue University, North Central DR. JEAN JIANG is currently with the College
More informationLAB 12: ACTIVE FILTERS
A. INTRODUCTION LAB 12: ACTIVE FILTERS After last week s encounter with op amps we will use them to build active filters. B. ABOUT FILTERS An electric filter is a frequencyselecting circuit designed
More informationSignaling is the way data is communicated. This type of signal used can be either analog or digital
3.1 Analog vs. Digital Signaling is the way data is communicated. This type of signal used can be either analog or digital 1 3.1 Analog vs. Digital 2 WCB/McGrawHill The McGrawHill Companies, Inc., 1998
More informationLabview VI Example Virtual Filters Written by: Dan Lankow 2014
PH315 Portland State University Labview VI Example Virtual Filters Written by: Dan Lankow 2014 1. ABSTRAT For this lab, you will be introduced to Labview. You will be implementing a Low Pass, High Pass,
More informationGETTING STARTED WITH LABVIEW POINTBYPOINT VIS
USER GUIDE GETTING STARTED WITH LABVIEW POINTBYPOINT VIS Contents Using the LabVIEW PointByPoint VI Libraries... 2 Initializing PointByPoint VIs... 3 Frequently Asked Questions... 5 What Are the
More informationDevelopment of MATLABBased Digital Signal Processing Teaching Module with Graphical User Interface Environment for Nigerian University
International Journal of Engineering and Technology, 2 (3) (2013) 230238 Science Publishing Corporation www.sciencepubco.com/index.php/ijet Development of MATLABBased Digital Signal Processing Teaching
More informationFILTER CIRCUITS. A filter is a circuit whose transfer function, that is the ratio of its output to its input, depends upon frequency.
FILTER CIRCUITS Introduction Circuits with a response that depends upon the frequency of the input voltage are known as filters. Filter circuits can be used to perform a number of important functions in
More informationPeakVue Analysis for Antifriction Bearing Fault Detection
August 2011 PeakVue Analysis for Antifriction Bearing Fault Detection Peak values (PeakVue) are observed over sequential discrete time intervals, captured, and analyzed. The analyses are the (a) peak values
More informationData Acquisition in LabVIEW
Telemark University College Department of Electrical Engineering, Information Technology and Cybernetics Data Acquisition in LabVIEW HANS PETTER HALVORSEN, 2013.08.16 Faculty of Technology, Postboks 203,
More informationIMPLEMENTATION OF FIR FILTER USING EFFICIENT WINDOW FUNCTION AND ITS APPLICATION IN FILTERING A SPEECH SIGNAL
IMPLEMENTATION OF FIR FILTER USING EFFICIENT WINDOW FUNCTION AND ITS APPLICATION IN FILTERING A SPEECH SIGNAL Saurabh Singh Rajput, Dr.S.S. Bhadauria Department of Electronics, Madhav Institute of Technology
More informationAnalog and Digital Interfacing
Analog and Digital Interfacing BJ Furman Spring, 1999 DAC & ADC DigitaltoAnalog Conversion (DAC) Converts a binary word from computer to a scaled analog voltage Used for controlling systems that require
More informationLab 2: Filter Design and Implementation
http://www.comm.utoronto.ca/~dkundur/course/realtimedigitalsignalprocessing/ Page 1 of 1 Lab 2: Filter Design and Implementation Objectives of this Lab This lab Professor Deepa Kundur introduces you
More informationContinuous time vs Discrete time. Lecture 13. Sampling & Discrete signals (Lathi ) Sampling Process. Sampling Theorem. f s. = 500Hz ELECTRONICS
Lecture 13 Sampling & Discrete signals (Lathi 8.18.2) Peter Cheung Department of Electrical & Electronic Engineering Imperial College London URL: www.ee.imperial.ac.uk/pcheung/teaching/ee2_signals Email:
More informationHarmonics and Noise in Photovoltaic (PV) Inverter and the Mitigation Strategies
Soonwook Hong, Ph. D. Michael Zuercher Martinson Harmonics and Noise in Photovoltaic (PV) Inverter and the Mitigation Strategies 1. Introduction PV inverters use semiconductor devices to transform the
More informationDigital image processing
Digital image processing The twodimensional discrete Fourier transform and applications: image filtering in the frequency domain Introduction Frequency domain filtering modifies the brightness values
More informationExperiment 5 A Fourier series analyzer
Experiment 5 A Fourier series analyzer Achievements in this experiment Compose arbitrary periodic signals from a series of sine and cosine waves. Confirm the Fourier Series equation. Compute fourier coefficients
More informationThe Effective Number of Bits (ENOB) of my R&S Digital Oscilloscope Technical Paper
The Effective Number of Bits (ENOB) of my R&S Digital Oscilloscope Technical Paper Products: R&S RTO1012 R&S RTO1014 R&S RTO1022 R&S RTO1024 This technical paper provides an introduction to the signal
More informationANTIALIASING FILTERS WITH ZERO PHASE DISTORTION
ANTIALIASING FILTERS WITH ZERO PHASE DISTORTION Howto, version: 3.1, Date: 04.05.2016 DEWESoft d.o.o. Gabrsko 11a, 1420 Trbovlje, Slovenia support@dewesoft.com Dewesoft has a possibility to set different
More informationManual Analysis Software AFD 1201
AFD 1200  AcoustiTube Manual Analysis Software AFD 1201 Measurement of Transmission loss acc. to Song and Bolton 1 Table of Contents Introduction  Analysis Software AFD 1201... 3 AFD 1200  AcoustiTube
More information9: FILTERS, ANALYSIS OF PERIODIC SOUNDS
9: FILTERS, ANALYSIS OF PERIODIC SOUNDS A. Introduction Some weeks you synthesized (put together) complex periodic signals with harmonic components. It is also possible to analyze (take apart) a complex
More informationLab 4 Sampling, Aliasing, FIR Filtering
47 Lab 4 Sampling, Aliasing, FIR Filtering This is a software lab. In your report, please include all Matlab code, numerical results, plots, and your explanations of the theoretical questions. The due
More informationAn Introduction to Modular Arbitrary Function Generators
An Introduction to Modular Arbitrary Function Generators Electronic test and measurements equipment can be classified into two major categories; measurement instruments and signal sources. Instruments
More informationA Primer for the ESyst Analog Systems ilab James Hardison  Document version 1.0
A Primer for the ESyst Analog Systems ilab James Hardison  hardison@mit.edu Document version 1.0 This document is intended as an introduction to the ESyst Analog Systems ilab. The functionality of this
More informationCTA300. Communication Trainer Analog RELATED PRODUCTS. Communication Trainer kit
Communication Trainer kit Communication Trainer RELATED PRODUCTS v Digital Communication Trainers v Optical Fibers Communication Trainers v Digital and Communication Trainers v Communication Electronic
More informationControl Analog Out, Digital Out, and Pulse Out
Control Analog Out, Digital Out, and Pulse Out Exercise 8 Completed front panel and block diagram In this exercise, you will create a program to control the SensorDAQ s analog out terminal (this is pin
More informationANALYZER BASICS WHAT IS AN FFT SPECTRUM ANALYZER? 21
WHAT IS AN FFT SPECTRUM ANALYZER? ANALYZER BASICS The SR760 FFT Spectrum Analyzer takes a time varying input signal, like you would see on an oscilloscope trace, and computes its frequency spectrum. Fourier's
More informationUSING THE ANALOG DEVICES ACTIVE FILTER DESIGN TOOL
USING THE ANALOG DEVICES ACTIVE FILTER DESIGN TOOL INTRODUCTION The Analog Devices Active Filter Design Tool is designed to aid the engineer in designing allpole active filters. The filter design process
More informationAnalog to Digital and Digital to Analog Conversion 2013 by the Regents of the University of California. All rights reserved.
University of California at Berkeley Physics 111 Laboratory Basic Semiconductor Circuits (BSC) Lab 10 Analog to Digital and Digital to Analog Conversion References: Wells & Wells Entire Book on LabView.
More informationImage Enhancement in the Frequency Domain
Image Enhancement in the Frequency Domain Jesus J. Caban Outline! Assignment #! Paper Presentation & Schedule! Frequency Domain! Mathematical Morphology %& Assignment #! Questions?! How s OpenCV?! You
More informationEELE445  Lab 2 Pulse Signals
EELE445  Lab 2 Pulse Signals PURPOSE The purpose of the lab is to examine the characteristics of some common pulsed waveforms in the time and frequency domain. The repetitive pulsed waveforms used are
More information1.3 Expand ANSI C and.net Development with Express Technology. Bridging Validation and Test with Common Software Technology
1.3 Expand ANSI C and.net Development with Express Technology Bridging Validation and Test with Common Software Technology 47 Interactive versus Automated Measurements/Tests Interactive Measurements Benchtop
More informationDigital Signal Processing Fundamentals
Application Note 023 Digital Signal Processing Fundamentals D. Koenig Introduction 1 Recent advances in digital signal processing (DSP) technology make it easier for scientists to develop powerful personal
More informationDSP Laboratory Work S. Laboratory exercises with TMS320C5510 DSK
DSP Laboratory Work 521485S Laboratory exercises with TMS320C5510 DSK Jari Hannuksela Information Processing Laboratory Dept. of Electrical and Information Engineering, University of Oulu ovember 14, 2008
More informationMatlab GUI for WFB spectral analysis
Matlab GUI for WFB spectral analysis Jan Nováček Department of Radio Engineering K13137, CTU FEE Prague Abstract In the case of the sound signals analysis we usually use logarithmic scale on the frequency
More informationInternational Journal of Advanced Research in Computer Science and Software Engineering
Volume 2, Issue 8, August 2012 ISSN: 2277 128X International Journal of Advanced Research in Computer Science and Software Engineering Research Paper Available online at: www.ijarcsse.com Development of
More informationEE133 Winter 2002 Cookbook Filter Guide Welcome to the Cookbook Filter Guide!
Welcome to the! Don t have enough time to spice out that perfect filter before Aunt Thelma comes down for dinner? Well this handout is for you! The following pages detail a fast set of steps towards the
More informationNational Instruments: LabVIEW
National Instruments: LabVIEW LabVIEW is a graphical programming language that uses icons instead of lines of text to create applications. In contrast to textbased programming languages, where instructions
More informationFourier Analysis. Nikki Truss. Abstract:
Fourier Analysis Nikki Truss 09369481 Abstract: The aim of this experiment was to investigate the Fourier transforms of periodic waveforms, and using harmonic analysis of Fourier transforms to gain information
More informationFFT Frequency Detection on the dspic
FFT Frequency Detection on the dspic by Jac Kersing Nov 2012 Abstract In this article Jac explains how he solved a practical challenge using powerful dspic devices and Fast Fourier Transform algorithms.
More information4.3 AnalogtoDigital Conversion
4.3 AnalogtoDigital Conversion overview including timing considerations block diagram of a device using a DAC and comparator example of a digitized spectrum number of data points required to describe
More informationFourier Transform and Image Filtering. CS/BIOEN 6640 Lecture Marcel Prastawa Fall 2010
Fourier Transform and Image Filtering CS/BIOEN 6640 Lecture Marcel Prastawa Fall 2010 The Fourier Transform Fourier Transform Forward, mapping to frequency domain: Backward, inverse mapping to time domain:
More informationNonData Aided Carrier Offset Compensation for SDR Implementation
NonData Aided Carrier Offset Compensation for SDR Implementation Anders Riis Jensen 1, Niels Terp Kjeldgaard Jørgensen 1 Kim Laugesen 1, Yannick Le Moullec 1,2 1 Department of Electronic Systems, 2 Center
More informationMATLAB for Audio Signal Processing. MATLAB for Audio Signal Processing. Aliasing. Getting data into the computer
MATLAB for Audio Signal Processing P. Professorson UT Arlington Night School MATLAB for Audio Signal Processing Getting real world data into your computer Analysis based on frequency content Fourier analysis
More informationFrequency Response and Continuoustime Fourier Transform
Frequency Response and Continuoustime Fourier Transform Goals Signals and Systems in the FDpart II I. (Finiteenergy) signals in the Frequency Domain  The Fourier Transform of a signal  Classification
More informationNEP Noise Equivalent Power
NEP Noise Equivalent Power Verena Mackowiak (GmbH), Jens Peupelmann (GmbH), Yi Ma (USA), and Anthony Gorges (USA) Thorlabs, Inc., 56 Sparta Avenue, Newton, NJ 07860, USA Thorlabs GmbH, HansBocklerStr.
More informationFundamental Concepts in EMG Signal Acquisition
Fundamental Concepts in EMG Signal Acquisition Gianluca De Luca Copyright DelSys Inc, 2 Rev.2., March 23 The information contained in this document is presented free of charge, and can only be used for
More informationω = 1/R 1 C 1, which is called the 3dB cut off frequency.
Lab 2 Linear Networks and Modular Construction Networks which contain only inductor, capacitors and resistors are called linear networks This means that if A and B are two sine wave inputs then the output
More informationAnalog to Digital, A/D, Digital to Analog, D/A Converters. An electronic circuit to convert the analog voltage to a digital computer value
Analog to Digital, A/D, Digital to Analog, D/A Converters An electronic circuit to convert the analog voltage to a digital computer value Best understood by understanding Digital to Analog first. A fundamental
More informationFourier Analysis Last Modified 9/5/06
Measurement Lab Fourier Analysis Last Modified 9/5/06 Any timevarying signal can be constructed by adding together sine waves of appropriate frequency, amplitude, and phase. Fourier analysis is a technique
More informationHideo Okawara s Mixed Signal Lecture Series. DSPBased Testing Fundamentals 46 Perpin Signal Generator
Hideo Okawara s Mixed Signal Lecture Series DSPBased Testing Fundamentals 46 Perpin Signal Generator Advantest Corporation, Tokyo Japan August 2012 Preface to the Series ADC and DAC are the most typical
More informationLecture  4 Diode Rectifier Circuits
Basic Electronics (Module 1 Semiconductor Diodes) Dr. Chitralekha Mahanta Department of Electronics and Communication Engineering Indian Institute of Technology, Guwahati Lecture  4 Diode Rectifier Circuits
More informationSIGNAL PROCESSING FOR EFFECTIVE VIBRATION ANALYSIS
SIGNAL PROCESSING FOR EFFECTIVE VIBRATION ANALYSIS Dennis H. Shreve IRD Mechanalysis, Inc Columbus, Ohio November 1995 ABSTRACT Effective vibration analysis first begins with acquiring an accurate timevarying
More informationLab 8: Basic Filters: Low Pass and High Pass
Lab 8: Basic Filters: Low Pass and High Pass Names: 1.) 2.) 3.) Beginning Challenge: Build the following circuit. Charge the capacitor by itself, and then discharge it through the inductor. Measure the
More informationAnalog and Digital Filters Anthony Garvert November 13, 2015
Analog and Digital Filters Anthony Garvert November 13, 2015 Abstract In circuit analysis and performance, a signal transmits some form of information, such as a voltage or current. However, over a range
More informationFrequency Domain Analysis
Exercise 4. Frequency Domain Analysis Required knowledge Fourierseries and Fouriertransform. Measurement and interpretation of transfer function of linear systems. Calculation of transfer function of
More informationImpedance 50 (75 connectors via adapters)
VECTOR NETWORK ANALYZER PLANAR TR1300/1 DATA SHEET Frequency range: 300 khz to 1.3 GHz Measured parameters: S11, S21 Dynamic range of transmission measurement magnitude: 130 db Measurement time per point:
More informationTechniques for Extending RealTime Oscilloscope Bandwidth
Techniques for Extending RealTime Oscilloscope Bandwidth Over the past decade, data communication rates have increased by a factor well over 10X. Data rates that were once 1Gb/sec and below are now routinely
More informationRelative Position Tracking for User Interface. Controls Using Doppler Shifted Ultrasound
Relative Position Tracking for User Interface Controls Using Doppler Shifted Ultrasound Justin Bradfield and Alex Mudd Cornell College Phy312, November 2012, Professor Derin Sherman Introduction As computers
More informationSGN1158 Introduction to Signal Processing Test. Solutions
SGN1158 Introduction to Signal Processing Test. Solutions 1. Convolve the function ( ) with itself and show that the Fourier transform of the result is the square of the Fourier transform of ( ). (Hints:
More informationBasics of Digital Recording
Basics of Digital Recording CONVERTING SOUND INTO NUMBERS In a digital recording system, sound is stored and manipulated as a stream of discrete numbers, each number representing the air pressure at a
More informationIntroduction to Digital Filters
CHAPTER 14 Introduction to Digital Filters Digital filters are used for two general purposes: (1) separation of signals that have been combined, and (2) restoration of signals that have been distorted
More information