2 Quite often the measurements of pressures has to be conducted in unsteady conditions. Typical cases are those of -the measurement of time-varying pressure (with periodic oscillations or step changes) -the pneumatic scanning of several different steady state pressures (using a single pressure transducer) In these cases, phenomena like the time or frequency response of the pressure measuring system (consisting of pressure tap or probe, pressure line, valves, transducer) and the effect of various parameters of the measuring system have to be considered. This is necessary in order to avoid measuring errors.
3 An enclosure where pressure is varying, as a result of any process (moving boundaries, mass-flow input, chemical reaction, heat transfer, phase change, etc ) Different cases of occurrence of unsteady pressures
4 An airfoil (or any other body) immersed in an unsteady mainstream, or a steady mainstream but subjected t of change of incidence, periodic or not (oscillating airfoil, or subjected to a pitch-up maneuver) Different cases of occurrence of unsteady pressures
5 The case of a test model (aircraft, ground vehicle, airfoil or turbomachinery blade, internal flow system) with a large number of pressure tabs, each one subjected to a steady pressure, and using a pneumatic switching device (scanning valve) to connect in sequence every tap to the same pressure transducer, which his therefore subjected dto a succession of pressure steps. Different cases of occurrence of unsteady pressures
6 In all the previous cases represented, the measured time-dependent pressure signal P will in general be different from the real time-dependent pressure P true. For example, in the case of a set of pressures measured using a scanning valve, different possible results of measurements could be obtained, characterized by an oscillating or a non-oscillatingoscillating response, with a small or a large damping. Possible results of scanning valve pressure measurements
7 In general, the response of a pressure measuring system can be examined: -in the time domain or -in the frequency domain, following a usual practice, adopted in the study of the transfer function of any mechanical or electrical system.
8 When working in time domain, a pressure step is applied to the system and the output is recorded. Several possible output shapes are shown in Figure characterized by an oscillatory or non-oscillatory behavior. It is important in all these cases to define the response time, i.e. that time interval after application of the pressure step after which the pressure transducer output has reasonably reached the true final steady-state value. Time response to a pressure step input
9 When working in frequency domain, a sinusoidal pressure variation of a given frequency is applied to the measuring system. The system output will in general also be periodic (or sinusoidal in the particular case of a linear system). One can then use the ratio of the output to input oscillation amplitudes and an output to input phase lag, plotted versus frequency to characterize the frequency response of the system. Amplification and phase lag versus frequency
10 Practical configurations of unsteady pressure measurements systems: Flush-mounted pressure transducers: The ideal configuration of a pressure transducer for measuring rapidly varying pressures is that of the flush-mounted transducer. In this arrangement the sensing membrane of the transducer is located directly on the surface where the pressure has to be measured, thus eliminating the need for a pressure tap, and for a plastic or metal tube connecting the tap to the inner cavity of the transducer.
11 Practical configurations of unsteady pressure measurements systems: Flush-mounted pressure transducers: The tap, the tubing and the internal volume of the transducer, which are present in any other arrangement, generally cause a substantial increase of the response time of the pressure measuring system. Therefore, the flush-mounted transducers generally yield the minimal response time, limited only by the mechanical response of the membrane for piezo-resistive transducers or of the sensing crystal for piezo-electricelectric transducers. Examples of pressure transducers that can be flush-mounted: - Kistler and PCB piezoelectric transducers (tip diameter of about 5mm) - Kulite and Endevco piezosensitive transducers, with semiconductor strain gages directly bounded on the sensing membrane (total tip diameter of about 2mm)
12 Practical configurations of unsteady pressure measurements systems: Pressure transducer connected to a pressure tap: The usual type of pressure transducers is characterized by the existence of two internal volumes, separated by the sensing membrane, and connected to the outside by two ports. In some cases the transducer is symmetrical (e.g. Validyne ) in other cases it is not (e.g. Statham, with two different internal cavity volumes) Validyne DP45 pressure transducer
13 Practical configurations of unsteady pressure measurements systems: Pressure transducer connected to a pressure tap: One of the two ports can be the measurement port, the other being a reference port left open -to ambient pressure( gage pressure measurement) -to vacuum (absolute pressure measurement) or both can be used as measurement ports (differential pressure measurements) Schematic representation of pressure transducer (for gage or absolute pressure measurement)
14 Practical configurations of unsteady pressure measurements systems: Pressure transducer connected to a pressure tap: In the case of the differential pressure measurement, a second connecting line and pressure tap must be added to the right port. Schematic representation of pressure transducer (for gage or absolute pressure measurement)
15 Practical configurations of unsteady pressure measurements systems: Pressure transducer connected to a pressure tap: When a pressure is applied, the membrane deforms, causing a variation of both internal cavity volumes. This variation is defined as the displacement volume of the pressure transducer, and together with the cavity volume (in the nondeformed membrane situation) characterizes the unsteady response capability of the specific pressure transducer. However, these two volume parameters alone DO NOT define the unsteady response of a pressure measuring system. This response also depends on the connecting line length and diameter, on the pressure tap dimensions, on the absolute pressure level over which a pressure variation is considered.
16 Pressure transducer connected to a pressure tap: Physically, time response of the measuring system is limited by the following succession of phenomena: After a pressure step increase at the pressure tap location a pressure wave will form and will travel along the connecting line towards the pressure wave will be reflected (as a compression or expansion wave, depending on conditions: dead-end end line, or transducer with large inner volume) and will travel back and forth a number of times until it quickly damps out. In the same time, some flow will be initiated in the connecting line, and fluid will flow into the transducer, -to cause a pressure increase inside the transducer (the required amount of fluid, due to its compressibility, depends on the cavity volume and on the absolute pressure level) -to compensate for the membrane deformation (the required amount of fluid depends on the displacement volume and on the amplitude of pressure step) The rate at which this will occur will also depend on the pressure loss characteristics of the connecting line, wall pressure tapping, or type of pressure probe used (static, total, multipleholes). This is different from the steady state situation, in which there is no flow in the connecting line, and no need to consider its pressure loss characteristics.
17 Pressure transducer connected to a pressure tap: It is absolutely wrong to speak about the time, or frequency of a specific pressure transducer. Rather one has to speak about the time or frequency response of a measuring system (transducer, plus line, plus possible fittings, plus pressure tap, or internal flow characteristics of a pressure probe)
18 Practical configurations of unsteady pressure measurements systems: Multiple pressures measurement using scanning valve: When a large number of pressures has to be measured, the use of a scanning valve is a must, as it greatly reduces the required number of transducers, and therefore the total cost of instrumentation, and also simplifies the transducer calibration procedure. 12 position scanning valve
19 48 position scanning valve
20 Practical configurations of unsteady pressure measurements systems: Multiple pressures measurement using scanning valve: At the moment of the switching from one position to the next, the portion of the line connected to the next pressure tap is at the correct pressure, different from the pressure existing in the line connected to the pressure transducer. To minimize the response time of this kind of system, the length and volume of the line connecting the scanning valve to the pressure transducer have to be reduced as much as possible, and that the length and diameter of the lines to the pressure taps have a much smaller effect. Schematic representation of scanning valve/ pressure transducer
21 Practical configurations of unsteady pressure measurements systems: Multiple pressures measurement using electronic pressure scanner: An alternative to the use of scanning valves for the measurement of a large number of pressures is the use of an electronic pressure scanner. This consists in a number of individual pressure transducers, assembled together in a module (with 16, 32 or 64 transducers each), each transducer individual port being permanently connected to the tap where pressure has to be measured. The switching from one measurement to another is done electronically, by sampling in sequence the output of each pressure transducer. Electronic pressure scanner (32 transducer modules)
22 Multiple pressures measurement using electronic pressure scanner: Another alternative is the use of a large number of individual low cost pressure transducers, connected to a data acquisition system with a sufficiently large number or channels. Large number of individual transducers
23 Supplying a digital I2C output, the differential pressure sensor measures with highest sensitivity and accuracy even at low differential pressures (< 10 Pa). Version: Connection: Calibrated range: Accuracy: SDP600 Manifold connection -500 to 500 Pa 0.5 Pa + 3% of reading SDP610 Tube connection -500 to 500 Pa 0.5 Pa + 3% of reading SDP500 Manifold connection 0 to 500 Pa 05P 0.5 Pa + 45% 4.5% of reading SDP510 Tube connection 0 to 500 Pa 0.5 Pa + 4.5% of reading differential-pressure-sensors/00_sdp600-differential-pressure-sensors.htm
24 Practical configurations of unsteady pressure measurements systems: Practical tips for pressure transducer connecting: In practice, the pressure tap, or the holes in a pressure probe do not have the same diameter as the pressure line. The line could be made of a number of tubes with different diameter, connected together by adaptor fittings.. Using very small pressure tap are unavoidable for some applications.
25 Practical configurations of unsteady pressure measurements systems: Practical tips for pressure transducer connecting: Using large line and a very small internal volume transducer lead to a pneumatic scheme Using large line and a very small internal volume transducer lead to a pneumatic scheme resembling that of a dead-end tube.
26 Practical configurations of unsteady pressure measurements systems: Practical tips for pressure transducer connecting: An incorrect fitting between the line and the transducer severely compromising the system response time.
27 Practical configurations of unsteady pressure measurements systems: Most pressure transducers have input ports with diameters around 4 to 6 mm, and most pressure tappings in models are done using 1 mm or 1.5 mm tubing. For instance, below Figure depicts a common but completely wrong practice, if response time has to be minimized.
28 Practical configurations of unsteady pressure measurements systems: It is better to custom build a special adapter, with same diameter as the line, directly fitted to the pressure transducer.
29 The step response of the simplest line-cavity system: Figure shows schematically a pressure transducer connected to a pressure tap of same diameter as the connecting line. At time t=0, a pressure step of amplitude P 0, over an initial absolute pressure P a, is applied at tap location. At t=0, the pressure in the line (from locations 1 to 2) and in the cavity 3 of the transducer is at the initial value P a. The fluid in the line starts moving, and the pressure in cavity 3 will rise. line cavity Simplest line-cavity system
30 Divide problem into 2 parts Line analysis (Fluid motion) Cavity analysis (Pressure rise in the cavity) line cavity Simplest line-cavity system
31 Fluid behaviour in line: The whole amount of fluid contained in the line, of cross sectional area S, will be subjected to a force S(P 1 -P 2 ) causing it to accelerate. A dynamic balance, equating this force to the sum of fluid inertia and wall friction forces will yield:
32 Fluid behaviour in line: Assuming that the wall friction coefficient in unsteady conditions is the same as that in steady flow conditions (this is the weak point of this method), we can use the relations for steady pipe flow: where P, steady state pipe pressure loss is given by the sum of a distributed pressure loss and of localized pressure losses (e.g. at pipe inlet, outlet, bends, branchings )
33 Fluid behaviour in line: So the dynamic balance equation becomes: In the case of turbulent flow, and explicitly making the exit pressure loss term ξ=1: In the case of laminar flow:
34 Fluid behaviour in line: These first order, non-linear equations describe the unsteady flow in the line connecting pressure transducer to tap or probe. The choice between laminar and turbulent flow case is given by the The choice between laminar and turbulent flow case is given by the value of the pipe Reynolds number Re=U d/ ν (smaller or larger than about 2000)
35 Fluid behaviour in cavity: To describe the increase of pressure (P 3 - relative pressure with respect to absolute pressure P a a) in the transducer cavity, mass conservation equation is used equating the mass flow entering the cavity to the time variation of mass contained in the cavity: where volume of the cavity V ol must be considered as a variable since the pressure transducer membrane deforms in time. V ol is not directly a function of time but rather of pressure P 3, and this latter is a function of time.
36 Fluid behaviour in cavity: A polytropic compression takes place inside the cavity: The mass conservation equation then becomes: Assuming that the pressure step P 3 is small compared to initial pressure P a, as in most cases in practice
37 Fluid behaviour in cavity: As pressure transducers are designed to be linear instruments, the quantity dv ol /dp 3 is a constant given by dv ol /dp 3 = displacement volume / transducer pressure range and therefore integration ti of the expression, with initial iti condition P 3 =0 yields: Volume flow rate
38 Fluid behaviour in cavity: Analogy between fluid flow and electricity: Fluid Electricity P 3 = 1 C t 0 USdt V = 1 C t 0 Idt P, Pressure V, Voltage Volume flow rate I, Current Fluidic capacitance due to transducer inner volume and is the parameter fully charecterizing the behaviour of the transducer volume alone.
39 Fluid behaviour in cavity: C is not only dependent on the internal volume and on the displacement volume, but also in on the transducer pressure range (pressure variation over which the total displacement volume occurs) and on the absolute level of pressure at which the transducer is used. Thus physically the same transducer, with a given internal volume, displacement volume and differential pressure range, will exhibit a larger capacity (therefore increasing response time of the system) at low absolute pressures P a.
40 Combined line-cavity system: Eliminating the variable P 2 =P 3 in the line and cavity equations, we obtain: valid in the laminar flow case, but also representing the turbulent case (with k 1 =0 and k 2 =1+k) This is a first order integro-differential equation, with initial condition U=0 at t=0 The integral can be eliminated by differentiation, obtaining the non-linear, second order differential equation with two initial conditions U=0 and du/dt=p 1 /ρl at t=0
41 Combined line-cavity system: This equation can be solved numerically by a classical single-step time-marching technique ( e.g. 4th order Runge-Kutta scheme) Once U(t), i.e. the variation in time of the average velocity in the line is known, the pressure P 3 in the transducer cavity can be obtained from the equation derived previously: P 3 t 0 1 = USdt C System constants:
42 Combined line-cavity system: C l l t d ti f li it t ( ff t f ti li di t k i Calculated time response of line-cavity system (effect of connecting line diameter keeping constant its length)
43 Combined line-cavity system:
44 Combined line-cavity system: The behavior of the system changes from that of an oscillating second order system to that of a damped second order system, then to that of a first order system.
45 Combined line-cavity system: Figure below shows results of calculations for a system that can be considered of first order, as a result of the use of a transducer with a relatively large inner volume and of a line with a relatively small diameter. Calculated and measured time response of non-oscillating line-cavity system P= 200 mm Hg =6V, d=1 mm)
46 Combined line-cavity system: The figure shows the effects of varying line diameter and length, and also clearly shows the difference between the system time constant τ l (based on extrapolation of the curve slope at origin) and the system response time τ (when the transducer output reaches the true value). Calculated and measured time response of non-oscillating line-cavity system P= 200 mm Hg =6V, d=1 mm)
47 Combined line-cavity system: Calculated and measured time response of non-oscillating line-cavity system P= 200 mm Hg =6V, d=1 mm)
48 Simplified Linear Analysis of unsteady pressure measuring systems: In some cases non-linear terms of previous second order differential equation can be neglected. This happens when two conditions are both verified: Flow inside the line must be laminar and the velocity in the line must be sufficiently low or the line is very long, or the diameter is sufficiently small so thatt the laminar pressure drop (proportional to the velocity U) results to be much larger than the pressure drop associated with dynamic pressure losses (occurring at line exit, bends or section changes), which are proportional to U 2, also in the case of laminar flow. In such case, the term k 2 is small compared to k, and the equation can be linearized as:
49 Simplified Linear Analysis of unsteady pressure measuring systems: Fluid Electricity L: Inductance R: resistance C: Capacitance I: Current V: Voltage Poiseuille relation for the pressure drop associated with laminar flow in a pipe Ohm s law for an electrical resistance R Equivalent resistance of a line: According to this analogy, the resistive behaviour of a line is due to viscous pressure losses.
50 Simplified Linear Analysis of unsteady pressure measuring systems: Fluid Electricity du ΔP S = Force = ρ ls = inertia dt l dq ΔP = ρ S dt Dynamic equilibrium of the accelerating (decelerating) flow in a pipe di V = L dt L: Inductance R: resistance C: Capacitance I: Current V: Voltage Voltage across an electrical inductor, with inductance L Equivalent inductance of a line: L = l ρ S According to this analogy, the inductive behaviour of a line is due to inertia of the fluid contained in it.
51 Simplified Linear Analysis of unsteady pressure measuring systems: Fluid Electricity L: Inductance R: resistance C: Capacitance I: Current V: Voltage V = t 1 Idt C 0 P, Pressure V, Voltage Q=US, Volume flow rate I, Current
52 Simplified Linear Analysis of unsteady pressure measuring systems: To assess the behaviour of a linearizable pressure measuring system, the quantities R, L and C can be evaluated using the formulas and used to solve the second order linear equation. Depending on the natural frequency ωn = 1/ LC and on the damping coefficient of the system ζ = R / 2 L ω The solution of second order equation is given in Figure, showing oscillating and non oscillating behaviour. n General solutions of a linear second order equation
53 Simplified Linear Analysis of unsteady pressure measuring systems: The behaviour of a line may be predominantly resistive (due to viscous losses), or predominantly inductive (due to fluid inertia), depending on the relative importance of the effects of R and L. L R = l ρ S 128 l / μ π d / 4 = 2 d 32ν Therefore, the type of behaviour of a line will not depend on its length, but only on its diameter, and on the kinematic viscosity of the fluid contained in it. Dimensionally L/R is a time, and thus we can define the line time constant: L R = τ l
54 Simplified Linear Analysis of unsteady pressure measuring systems: Analogy with electric circuit with a resistance and an inductance in series: V RI + I di L dt = with I=0 at t=0 The solution is asymptotically approaching the final steady state value V/R: τ l = L R V = R ( 1 exp( t / τ ) ) 1 l where is the circuit time constant, characterizing the initial speed of variation of current. Therefore, to determine whether a given line has an inductive or a resistive behaviour, one has to compare the line time constant τ l with typical observation time.
55 Simplified Linear Analysis of unsteady pressure measuring systems: Thus if Observation time << τ l Observation time >> τ l inductive effects predominate resistive effects predominate and observation time is the response time to reach steady state value. In addition, when line resistive effects predominate, the inductive second order term can be neglected in the second order system equation, which is then reduced to a first order equation: R di dt + C I Characterized by a time constant τ 1 =RC and a response time (time to approach final state within 1%) τ r =4.6 RC = 0
56 Simplified Linear Analysis of unsteady pressure measuring systems: A quick estimation of the behaviour of a line and pressure transducer system, approximated by a linear equation, can be conducted as follows: First calculate the response time for a first order system τ 1 And the time line constant τ l Then if τ l << τ 1 inductive effects occur only at the very beginning of the phenomenon and can be neglected so that the response time τ r =4.6RC Otherwise, if τ l >> τ 1, inductive effects are present during all the time and the full second order equation must be used: 2 2 d Q dq Q d Q 1 dq Q L + R + = = dt dt C dt τ l dt τ lτ 1
57 Simplified Linear Analysis of unsteady pressure measuring systems: 2 d Q 2 dt 1 + τ l dq dt Q + τ l τ 1 = 0 Type of solution (oscillating or not) of this equation will be determined by the sign of discrement: 2 Δ τ l = 0 1 = 4 τ1 τ1 τ l > 0 τ1 τ l l Δ = b 4ac = 1 4 τ1 = 4τ l The system is critically damped Δ > 0 > 4 τ l The system is over-damped d (no oscillation) τ1 τ l Δ < < 0 τ < 4τ The system is under damped (oscillation) 1 l τ 1
TELEDYNE HASTINGS TECHNICAL PAPERS INSTRUMENTS A LAMINAR FLOW ELEMENT WITH A LINEAR PRESSURE DROP VERSUS VOLUMETRIC FLOW Proceedings of FEDSM 98: June -5, 998, Washington, DC FEDSM98 49 ABSTRACT The pressure
Applied Fluid Mechanics Sixth Edition Robert L. Mott University of Dayton PEARSON Prentkv Pearson Education International CHAPTER 1 THE NATURE OF FLUIDS AND THE STUDY OF FLUID MECHANICS 1.1 The Big Picture
ME 305 Fluid Mechanics I Part 8 Viscous Flow in Pipes and Ducts These presentations are prepared by Dr. Cüneyt Sert Mechanical Engineering Department Middle East Technical University Ankara, Turkey email@example.com
EML 316L Experiment 3 Pipe Friction Laboratory Manual Mechanical and Materials Engineering Department College of Engineering FLORIDA INTERNATIONAL UNIVERSITY Nomenclature Symbol Description Unit A cross-sectional
FLUID FLOW Introduction Fluid flow is an important part of many processes, including transporting materials from one point to another, mixing of materials, and chemical reactions. In this experiment, you
Dynamic Process Modeling Process Dynamics and Control 1 Description of process dynamics Classes of models What do we need for control? Modeling for control Mechanical Systems Modeling Electrical circuits
ENERGY TRANSFER SYSTEMS AND THEIR DYNAMIC ANALYSIS Many mechanical energy systems are devoted to transfer of energy between two points: the source or prime mover (input) and the load (output). For chemical
Applications of Second-Order Differential Equations Second-order linear differential equations have a variety of applications in science and engineering. In this section we explore two of them: the vibration
Objectives 1. Have a deeper understanding of laminar and turbulent flow in pipes and the analysis of fully developed flow 2. Calculate the major and minor losses associated with pipe flow in piping networks
Fluid Mechanics Prof. T. I. Eldho Department of Civil Engineering Indian Institute of Technology, Bombay Lecture No. # 36 Pipe Flow Systems Welcome back to the video course on Fluid Mechanics. In today
Chapter 10 Flow Measurements Material from Theory and Design for Mechanical Measurements; Figliola, Third Edition Flow Rate Flow rate can be expressed in terms of volume flow rate (volume/time) or mass
Measure Pressure and Flow Install the EMANT300 USB DAQ and Stepper Motor Training Kit 1. Connect up the EMANT300 USB DAQ and Pressure and Air pump Adaptor as shown. EMANT300 USB DAQ Pressure and Air pump
Chapter 5: Tool Dynamometers LEARNING OBJECTIVES Different types of transducers used in Dynamometers Design Requirements Types of Dynamometers ---------------------------------------------------------------------------------------------------------------------
AlBalqa Applied University تا سست عام 997 The curriculum of associate degree in Air Conditioning, Refrigeration and Heating Systems consists of (7 credit hours) as follows: Serial No. Requirements First
Differential Relations for Fluid Flow In this approach, we apply our four basic conservation laws to an infinitesimally small control volume. The differential approach provides point by point details of
Sensors 2010, 10, 10560-10570; doi:10.3390/s101210560 OPEN ACCESS sensors ISSN 1424-8220 www.mdpi.com/journal/sensors Article A New Approach to Laminar Flowmeters Fernando Lopez Pena *, Alvaro Deibe Diaz,
I. OBJECTIVE OF THE EXPERIMENT. Swissmetro travels at high speeds through a tunnel at low pressure. It will therefore undergo friction that can be due to: 1) Viscosity of gas (cf. "Viscosity of gas" experiment)
1 Lecture 5 Hemodynamics Description of fluid flow Hydrodynamics is the part of physics, which studies the motion of fluids. It is based on the laws of mechanics. Hemodynamics studies the motion of blood
DEVELOPMENT OF HIGH SPEED RESPONSE LAMINAR FLOW METER FOR AIR CONDITIONING Toshiharu Kagawa 1, Yukako Saisu 2, Riki Nishimura 3 and Chongho Youn 4 ABSTRACT In this paper, we developed a new laminar flow
Vol. XX 2012 No. 4 28 34 J. ŠIMIČEK O. HUBOVÁ NUMERICAL ANALYSIS OF THE EFFECTS OF WIND ON BUILDING STRUCTURES Jozef ŠIMIČEK email: firstname.lastname@example.org Research field: Statics and Dynamics Fluids mechanics
Definitions 9-1a1 Fluids Substances in either the liquid or gas phase Cannot support shear Density Mass per unit volume Specific Volume Specific Weight % " = lim g#m ( ' * = +g #V $0& #V ) Specific Gravity
1 RC Circuits Equipment DataStudio with 750 interface, RLC circuit board, 2 voltage sensors (no alligator clips), 2x35 in. leads, 12 in. lead Reading Review operation of DataStudio oscilloscope. Review
BASIC CONCEPTS/DEFINITIONS OF FLUID MECHANICS (by Marios M. Fyrillas) 1. Density (πυκνότητα) Symbol: 3 Units of measure: kg / m Equation: m ( m mass, V volume) V. Pressure (πίεση) Alternative definition:
ES250: Electrical Science HW7: Energy Storage Elements Introduction This chapter introduces two more circuit elements, the capacitor and the inductor whose elements laws involve integration or differentiation;
2. FEA and ANSYS FEA and ANSYS Overview In this chapter, we will define Finite Element Analysis and also give you an idea of ANSYS capabilities. Topics covered: A. What is FEA? B. About ANSYS FEA and ANSYS
[ Assignment View ] [ Print ] Eðlisfræði 2, vor 2007 30. Inductance Assignment is due at 2:00am on Wednesday, March 14, 2007 Credit for problems submitted late will decrease to 0% after the deadline has
Jan Tippner, Dep. of Wood Science, FFWT MU Brno jan. tippner@mendelu. cz Content of lecture 3: 1. Damping 2. Internal friction in the wood Content of lecture 3: 1. Damping 2. Internal friction in the wood
CHAPTER 08 1. What is most likely to be the main driving force in pipe flow? A. Gravity B. A pressure gradient C. Vacuum 2.What is a general description of the flow rate in laminar flow? A. Small B. Large
23 RD INTERNATIONAL SYMPOSIUM ON BALLISTICS TARRAGONA, SPAIN 16-2 APRIL 27 NOVEL EXAMINATION OF GUN BORE RESISTANCE ANALYSIS AND EXPERIMENTAL VALIDATION D. Carlucci 1, J. Vega 1, M. Pocock 2, S. Einstein
Flow and Flow rate. Laminar and Turbulent flow Laminar flow: smooth, orderly and regular Mechanical sensors have inertia, which can integrate out small variations due to turbulence Turbulent flow: chaotic
Lecture T1 Transient Mass Transfer Up to now, we have considered either processes applied to closed systems or processes involving steady-state flows. In this lecture we turn our attention to transient
Response to Harmonic Excitation Part 2: Damped Systems Part 1 covered the response of a single degree of freedom system to harmonic excitation without considering the effects of damping. However, almost
Trench Etched Resonant Pressure Sensor: TERPS Russell Craddock & Peter Kinnell GE Measurement & Control Solutions, Fir Tree lane, Groby, UK LE6 0FH INTRODUCTION High precision pressure sensors based on
Dimensional Analysis and Similarity Dimensional analysis is very useful for planning, presentation, and interpretation of experimental data. As discussed previously, most practical fluid mechanics problems
A Comparison of Analytical and Finite Element Solutions for Laminar Flow Conditions Near Gaussian Constrictions by Laura Noelle Race An Engineering Project Submitted to the Graduate Faculty of Rensselaer
Unit 41: Fluid Mechanics Unit code: T/601/1445 QCF Level: 4 Credit value: 15 OUTCOME 3 TUTORIAL 5 DIMENSIONAL ANALYSIS 3 Be able to determine the behavioural characteristics and parameters of real fluid
Albaha University Faculty of Engineering Mechanical Engineering Department Lecture 9: Pressure measurement Ossama Abouelatta email@example.com Mechanical Engineering Department Faculty of Engineering
ELEN236 Capacitors This worksheet and all related files are licensed under the Creative Commons Attribution License, version 1.0. To view a copy of this license, visit http://creativecommons.org/licenses/by/1.0/,
The First Law of Thermodynamics: Closed Systems The first law of thermodynamics can be simply stated as follows: during an interaction between a system and its surroundings, the amount of energy gained
Fluid Mechanics: Fundamentals and Applications, 2nd Edition Yunus A. Cengel, John M. Cimbala McGraw-Hill, 2010 Chapter 5 MASS, BERNOULLI AND ENERGY EQUATIONS Lecture slides by Hasan Hacışevki Copyright
Mechanical Pressure Measurement David Gardellin, P.E. Bourdon tube-type pressure sensors are the most common for industrial use in the family of elastic pressure elements. They have been in use for over
ME 05 Mechanical Engineering Lab Page ME 05 Mechanical Engineering Laboratory Spring Quarter 00 Experiment # 3: Pipe Flow Objectives: a) Calibrate a pressure transducer and two different flowmeters (paddlewheel
hapter 14 Homework Due: 9:00am on Thursday November 19 2009 Note: To understand how points are awarded read your instructor's Grading Policy. [Return to Standard Assignment View] Good Vibes: Introduction
VISHAY DALE www.vishay.com Magnetics Selecting IHLP Composite Inductors for Non-Isolated Converters INTRODUCTION This application note will provide information to assist in the specification of IHLP composite
Fluids and Solids: Fundamentals We normally recognize three states of matter: solid; liquid and gas. However, liquid and gas are both fluids: in contrast to solids they lack the ability to resist deformation.
Current Transducer LAX SERIES I PN = 16-1 A Ref: LAX 1-NP For the electronic measurement of currents: DC, AC, pulsed..., with a galvanic isolation between the primary circuit (high power) and the secondary
HEFAT2012 9 th International Conference on Heat Transfer, Fluid Mechanics and Thermodynamics 16 18 July 2012 Malta Turbulence Modeling in CFD Simulation of Intake Manifold for a 4 Cylinder Engine Dr MK
4 SENSORS The modern technical world demands the availability of sensors to measure and convert a variety of physical quantities into electrical signals. These signals can then be fed into data processing
1.0 Introduction Long term performance of polymers Polymer materials exhibit time dependent behavior. The stress and strain induced when a load is applied are a function of time. In the most general form
VISUAL PHYSICS School of Physics University of Sydney Australia FLUID FLOW STREAMLINE LAMINAR FLOW TURBULENT FLOW REYNOLDS NUMBER? What type of fluid flow is observed? The above pictures show how the effect
Head Loss in Pipe Flow ME 123: Mechanical Engineering Laboratory II: Fluids Dr. J. M. Meyers Dr. D. G. Fletcher Dr. Y. Dubief 1. Introduction Last lab you investigated flow loss in a pipe due to the roughness
Transport Phenomena Macroscopic Balances for Nonisothermal Systems 1 Macroscopic Balances for Nonisothermal Systems 1. The macroscopic energy balance 2. The macroscopic mechanical energy balance 3. Use
NEBB STANDARDS SECTION-8 AIR SYSTEM TAB PROCEDURES 8.1 INTRODUCTION Testing, adjusting, and balancing of HVAC systems can best be accomplished by following a series of systematic procedures. The NEBB TAB
Energy and Flow Measurement for Hydronic Systems Presented By: George Szyszko Applications Consultant MCR for ONICON Incorporated Why Measure Thermal Energy? To control something, you must first measure
ME6130 An introduction to CFD 1-1 What is CFD? Computational fluid dynamics (CFD) is the science of predicting fluid flow, heat and mass transfer, chemical reactions, and related phenomena by solving numerically
Application Report SLOA033A - September 2000 Signal Conditioning Piezoelectric Sensors James Karki Mixed Signal Products ABSTRACT Piezoelectric elements are used to construct transducers for a vast number
Three terminal adjustable current sources Features Operates from 1V to 40V 0.02%/V current regulation Programmable from 1µA to 10mA ±3% initial accuracy Description The LM134/LM234/LM334 are 3-terminal
INTRODUCTION TO FLUID MECHANICS SIXTH EDITION ROBERT W. FOX Purdue University ALAN T. MCDONALD Purdue University PHILIP J. PRITCHARD Manhattan College JOHN WILEY & SONS, INC. CONTENTS CHAPTER 1 INTRODUCTION
ET 438a Automatic Control Systems Technology Laboratory 4 Practical Differentiator Response Objective: Design a practical differentiator circuit using common OP AMP circuits. Test the frequency response
Sixth Edition. Fundamentals of Fluid Mechanics International Student Version BRUCE R. MUNSON DONALD F. YOUNG Department of Aerospace Engineering and Engineering Mechanics THEODORE H. OKIISHI Department
Comparison of Inductance Calculation Techniques Tony Morcos Magnequench Technology Center Research Triangle Park, North Carolina 1 VCM Baseline: Geometry Axially-magnetized MQ3-F 42 NdFeB disk Br = 131kG
Chapter 4 Vehicle Dynamics 4.. Introduction In order to design a controller, a good representative model of the system is needed. A vehicle mathematical model, which is appropriate for both acceleration
NUMERICAL ANALYSIS FOR TWO PHASE FLOW DISTRIBUTION HEADERS IN HEAT EXCHANGERS B.Babu 1, Florence.T 2, M.Punithavalli 3, B.R.Rohit 4 1 Assistant professor, Department of mechanical engineering, Rathinam
THE STRAIN GAGE PRESSURE TRANSDUCER Pressure transducers use a variety of sensing devices to provide an electrical output proportional to applied pressure. The sensing device employed in the transducers
In sensor technology several different methods are used to measure pressure. It is usually differentiated between the measurement of relative, differential, and absolute pressure. The following article
QUESTION 1 For a linear axis actuated by an electric motor complete the following: a. Derive a differential equation for the linear axis velocity assuming viscous friction acts on the DC motor shaft, leadscrew,
1. Which of the following statements about a spring-block oscillator in simple harmonic motion about its equilibrium point is false? (A) The displacement is directly related to the acceleration. (B) The
ADVANCED LEAK TEST METHODS Before new technologies were available to directly measure very small leaks, most leak measurements were done indirectly. Direct mass leak flow measurement has many advantages.
EDUMECH Mechatronic Instructional Systems Ball on Beam System Product of Shandor Motion Systems Written by Robert Hirsch Ph.D. 998-9 All Rights Reserved. 999 Shandor Motion Systems, Ball on Beam Instructional
Direct EGR Flow Meter MODEL TEG-F20 Specification Sheet TEG-F20 Specification Sheet Ver. 2011-0310 Copyright(C) 2011 Tsukasa Sokken Co., Ltd. 1. Overview Sokken model TEG series is an EGR Flow meter to
High Speed Aerodynamics Prof. K. P. Sinhamahapatra Department of Aerospace Engineering Indian Institute of Technology, Kharagpur Module No. # 01 Lecture No. # 06 One-dimensional Gas Dynamics (Contd.) We
Heat Transfer Prof. Dr. Aloke Kumar Ghosal Department of Chemical Engineering Indian Institute of Technology, Guwahati Module No. # 02 One Dimensional Steady State Heat Transfer Lecture No. # 05 Extended
Force measurement Forces VECTORIAL ISSUES In classical mechanics, a force is defined as "an action capable of modifying the quantity of movement of a material point". Therefore, a force has the attributes
Chapter 30 Inductance - Mutual Inductance - Self-Inductance and Inductors - Magnetic-Field Energy - The R- Circuit - The -C Circuit - The -R-C Series Circuit . Mutual Inductance - A changing current in
Damping in a variable mass on a spring pendulum Rafael M. Digilov, a M. Reiner, and Z. Weizman Department of Education in Technology and Science, Technion-Israel Institute of Technology, Haifa 32000, Israel
PTC Thermistors [From Philips Data Handbook PA02 1989] 1. GENERAL Positive Temperature Coefficient (PTC) thermistors exhibit a high positive temperature coefficient of resistance. They differ from Negative
Fluid Mechanics: Fundamentals and Applications, 2nd Edition Yunus A. Cengel, John M. Cimbala McGraw-Hill, 2010 Lecture slides by Mehmet Kanoglu Copyright The McGraw-Hill Companies, Inc. Permission required
Sample Questions for the AP Physics 1 Exam Sample Questions for the AP Physics 1 Exam Multiple-choice Questions Note: To simplify calculations, you may use g 5 10 m/s 2 in all problems. Directions: Each
Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and