Sensor Performance Metrics


 Maurice Lambert
 2 years ago
 Views:
Transcription
1 Sensor Performance Metrics Michael Todd Professor and Vice Chair Dept. of Structural Engineering University of California, San Diego me if you want a copy.
2 Outline Sensors as dynamic systems Zeroorder sensors Firstorder sensors Secondorder sensors Fundamental sensor performance metrics sensitivity bandwidth/transmission band(s) crossaxis sensitivity sensitivity to extraneous measurands
3 The Process of Getting An Acceleration Measurement plate vibration epoxy temperature viscoelastic effects housing selfdynamics (e.g., resonances) boundary conditions binding post manufacturing tolerances selfdynamics (e.g., crossaxis sensitivity) environment EMI cabling contact leads contact wire stray capacitance coupling piezoelectric element boundary conditions seismic mass seating signal conditioning DAQ board DSP data presentation environment human error (e.g., filter settings) quantization error human error software glitches (e.g., roundoff) human error Microsoft observer
4 Input/Output Representation of Sensors What we WANT to measure is some kind of response of the structure (e.g., acceleration at point q, Y q ( ) ), to some kind of excitation of the plate (e.g., impact hammer strike at point p, X p ( )); we usually describe the response in terms of an FRF (frequency response function): H pq () = Y q ()/ X p () Just as the plate transfers an excitation into an acceleration, the accelerometer subsequently transfers this acceleration into a measured representation of itself. Thus, what we ACTUALLY measure, Y q,obesrved, is NOT necessarily equal to what we WANT to measure, : ( ) Y q ( ) ( ) Y q,observed () = H epoxy ()H hou sin g ()H post ()H mass ()H piezo ()H wires () ########### "########### $ Y () q H sensor ( ) And this doesn t include all data acquisition and signal conditioning response functions We need to understand the behavior of the sensor transfer function in order to choose the right sensor characteristics for an application and calibrate it.
5 A Failure Example...Measuring Pendulation Loads Sea state can lead to dangerous load pendulation during LOTS exercises Design fiber optic accelerometer array to monitor load and ship motions Also asked to monitor centripetal acceleration (has DC component) Actual signal What we measured accelerometers + electronics had lowfrequency attenuation
6 General Input/Output Sensor Model: Accelerometer inertial mass housing piezoelectric material contact plate binding post inertial mass/ damping block housing (base) mass k m c effective damping/ stiffness x y input motion input motion Newton s 2nd Law on m: m d2 y dt 2 + c dy dt + ky ##" ## $ "output" = c dx dt + kx # "# $ "input" And, in general, d n y a n dt n + a d n1 y n1 dt n1 + + a dy 1 dt + a 0y = F(t) Sensors are themselves dynamical systems
7 ZeroOrder Sensors These sensors don t behave as dynamic systems and can only measure static signals sensor model is a 0 y(t) = F(t) so the output is easily obtained y(t) = (1/a 0 ) " # K static sensitivity The sensor s static sensitivity K determines the simple proportional input/output relationship. The sensor has no relevant dynamics, so it responds instantly to changes in F(t), provided the changes aren t too fast Sensor ideally contains no storage or inertial elements (or, more accurately, the behavior of these elements is negligible) There are no sensors that are truly zeroorder, but in some applications, sensors act like zeroorder P plenum spring y piston P P atm = ka piston "y, or intake valve force due to pressure difference P atm "P = ka # " piston # $ static sensitivity, K F(t) "y
8 FirstOrder Sensors These sensors contain storage elements which cannot respond immediately to inputs Modeled as firstorder dynamical system time constant a 1 dy dt + a 0y = F(t) dy dt + y = KF(t) static sensitivity The zerostate response (with no input) of this system shows the influence of : y(t) =5 =20 =0.5 t The larger the sensor s time constant, the longer it takes to respond to input changes.
9 FirstOrder Sensors (Cont d) We know from dynamic systems theory that we can utilize the powerful kernel method based on zerostate impulseresponse to get the formula for the response to ANY input. The impulse response for an impulseloaded firstorder system is dy dt + y = K"(t) h 1 (t) = Ke t /" /" For any general input F(t), then, we know the sensor s response to it: y(t) = y 0 e t /" + t /" t # F(T )h 1 (t T )dt T =0 = y 0 e " # + F(T )(K /" )e(tt )/" dt T =0$ $ $ " $ $ $ $ # transient response t # steadystate response
10 control volume Q E Example: Bulb Thermometer The change in energy stored in the bulb, E, must balance the exchange of heat between sensor and environment: E = Q mc dt dt = ha(t "T ) mc dt ha dt +T = T # # dt +T = F(t) dt m = liquid mass in the sensor c = specific heat of liquid h = convective heat transfer coefficient A = sensor surface area Standard from for firstorder sensor Note that the static sensitivity, K, is 1
11 FirstOrder Sensors: Step Input Consider the sensor response to a step change in the input, say 0 to A, so F(t)=A: y(t) = y 0 e t /" t + # (tt )/" A(K /" )e dt = KA + (KA y0 )e T =0 Our steady response (what we want to measure) is KA, which is the actual measurement A modified by the constant static sensitivity of the sensor K Our current measurement (initial condition) is y 0 We can define an error fraction function as t /" (t) = y(t) " y ss "t /# = e y 0 " y ss 0.37 When t=, the sensor has responded to 63% of the real measurement; When t=2.3, the sensor has responded to 90% of the real measurement; When t=5, the sensor has responded to 99% of the real measurement t / One may experimentally find by plotting the ln of error function during test data collection, since ln = "(1/# )t, and finding the slope of the bestfit line.
12 FirstOrder Sensors: Periodic Input Consider an input that is periodic in time, so F(t) = Asint y(t) = y 0 e t /" t + $ (tt )/" Asin#T (K /" )e dt = Ce t /" + T =0 where C = y 0 + KA" 1 + " 2, tan# = $" 2 KA sin(#t +%) 1 + " 2 2 # Once the transient portion dies out, we have the firstorder sensor frequency response to a periodic input The steadystate sensor response matches the input in frequency, but it lags the input by a time delay and gets attenuated: / " / " input output
13 FirstOrder Sensors: Periodic Input (Cont d) This attenuation and lag, as a function of parameters," are shown here M phase shift (deg) Attenuation factor M is the ratio of the dynamic amplitude to the static amplitude: KA M = " 2 KA 1 = " 2 " " Phase shift was defined previously, tan 1 "# to measure high frequencies, need very small or get too much attenuation (filtering) to measure lowfrequency (towards static) signals, can use systems with large
14 SecondOrder Sensors These sensors contain both storage elements (in the form of damping) as well as inertia Modeled as secondorder dynamic systems: d 2 y a 2 dt 2 + a dy 1 dt + a 0 y = F(t) 1 n 2 d 2 y dt 2 + 2" dy n dt + y = KF(t) n = a 0 /a 2 # natural frequency " = a 1 /2 a 0 a 2 # damping ratio The natural frequency corresponds to a time scale that the sensor wants to respond at for any inputs it sees The damping ratio determines the characteristic manner in which the sensor response approaches the steadystate measurement: 0 < < 1 (underdamped) sensor transient responds will oscillate (at ) n > 1 sensor transient response will monotonically (not oscillate) approach steadystate; = 1, called critical damping, represents the quickest monotonic approach (optimal)
15 SecondOrder Sensors (Cont d) The vast majority of sensor designs result in subcritical damping (oscillations), so we will proceed with our discussion with that assumption. The transient response of a secondorder sensor system with subcritical damping is ( " y trans (t) = e " nt y 0 cos" n 1 2 t + y 0 + y 0 n " % + * $ ' cos # n 1" 2 n 1" 2 t )* &, sensor initial conditions We can again use the kernel method to get the general sensor response: 1 n 2 d 2 y dt 2 + 2" dy n dt + y = K#(t) results in h 2 (t) = K n 1"# 2 e"# nt sin n 1"# 2 t For any general input F(t), then, we know the sensor s response to it: y(t) = y trans (t)+ t " T =0 t " T =0 F(T )h 2 (t T )dt = y trans (t) # "# $ + F(T )(K# n / 1$ 2 )e $# n (tt ) sin [# n 1$ 2 (t T )]dt ##########"########## $ transient response steadystate response
16 SecondOrder Sensors: Step Input Consider the sensor response to a step change in the input, say 0 to A, so F(t)=A: t $ T =0 y(t) = y trans (t)+ A(K n / 1"# 2 )e "# n (t"t ) sin [ n 1"# 2 (t "T )]dt % = KA " KAe "# nt ' cos d t + & # 1"# sin 2 d t ( *, where d = n 1"# 2 ) KA = 0 if < 1, sensor rings at d = 0.25 = 0.5 = 1 = 2 n t these oscillations die out and the expected measurement, KA, is eventually achieved system performance can be characterized by rise time and settling time
17 SecondOrder Sensors: More Step Input As with the firstorder sensor, we can compute an error function (t) : (t) n t % (t) = e "#$ nt ' cos$ d t + & # 1 " # sin$ ( dt* 2 ) rise time: time it takes to first achieve 90% of steadystate (or, error is 10%) settling time: time it takes for oscillations to settle within ±10% of steadystate n t settling time rise time rise time decreases with decreased, but the tradeoff is that settling time increases with decreased the region near the intersection of these curves is an optimal design point for the sensor
18 SecondOrder Sensors: Periodic Input Consider an input that is periodic in time, so F(t) = Asint t % T =0 y(t) = y trans + AsinT (K n / 1"# 2 )e "$ n (t"t ) sin d (t "T ) dt = y trans + KAsin(t +&), where tan& = "2# / n ( 1" ( / n ) 2 ) 2 + ( 2# / n ) 2 1" ( / n ) 2 After the transient portion (which oscillates at sensor time scale n ) dies out, the sensor response oscillates at the same frequency as the input,, but with a phase shift and an amplitude modification, both of which depend on damping and the ratio / n = 0.5 / n =1.5 / n output lags the input and is magnified for <1 / n output is attenuated for / n >1, and the lag continues to increase
19 SecondOrder Sensors: Periodic Input (Cont d) This amplitude effect M and phase shift are shown here; define = " / " n M = 0.01 = 0.25 = 0.5 = 1 = 2 amplification (resonance) transmission attenuation phase shift (deg) = 0.01 = 0.25 = 1 = 0.5 = 2 M = 1 ( 1 " 2 ) 2 + ( 2#" ) 2 resonance occurs if input frequency approaches the sensor s natural frequency (large attenuation, especially for small ) M is near unity for small and goes to zero for large (regardless of ) phase shift increasingly lags as increases, with the most abrupt changes occurring at resonance (especially for small ) output is in phase with input for small and out of phase for large (regardless of )
20 Transmission Band: Unity Gain and Linear Phase Regime As shown previously, there is a frequency regime for secondorder sensors where transmission is near unity and phase shifts are near zero. M " " = 2# +# 2 1 +# error tolerance = 0.7 corresponds to the widest band for retaining a unity transfer function this is also the regime where the phase is most closely linearly related to the frequency nonlinear phase shifts cause a nonuniform phase lead/lag in the original signal and the measured signal (signal distortion)
21 Signal Distortion Suppose have original signal to be measured: x(t) = sin2t + sin 4t A linear phase shift means that the measured signal may be represented by: y(t) = sin 2t +" ( ) + sin( 4t + 2" ) = sin# + sin2# which by inspection is the same as the original signal but timeshifted Consider a nonlinearly phase shifted measured signal: y(t) = sin 2t ( ) + sin( 4t + 2) Near sensor resonance, phase distortion is maximized nonlinear phase measured linear phase measured original
22 Accelerometers: Case of SecondOrder Sensor housing binding post inertial mass m y piezoelectric material contact plate inertial mass/ damping block housing (base) mass k c effective damping/ stiffness x input motion direct force input motion direct force, F(t) Newton s 2nd Law on m: m d2 y dt 2 + c dy dt + ky = c dx dt + kx + F(t) Accelerometers are designed so that the mass is protected from direct forces, so F(t)=0; if we define z=yx, we can rewrite this equation and normalize as before for secondorder systems: 1 n 2 d 2 z dt 2 + 2" dz n dt + z = # 1 2 n d 2 x dt 2 input x output z accelerometers measured acceleration of input by relative motion response of the inertial mass
23 Let s Consider a Periodic Input to this Accelerometer... If x = Asint, then x = "A 2 sint, and z(t) = KA2 sin(t "#) ( 1" 2 ) 2 + 2$ ( ) 2, tan# = " 2$ Very response to previous secondorder 1" 2 response, but with an extra 2 in the numerator; phase shift is the same We can define M for this system depending on what we consider the input: displacement, velocity, or acceleration Displacement input amplitude: A good trans. Velocity input amplitude: Acceleration input amplitude: A A 2 pretty much stinks good trans. M = 2 ( 1" 2 ) 2 + 2# ( ) 2 M = ( 1" 2 ) 2 + 2# ( ) 2 M = 1 ( 1 " 2 ) 2 + ( 2#" ) 2
24 Demonstration of Accelerometer in Acceleration Mode Consider a periodic acceleration input labeled in black) and the corresponding measured output (labeled in red) for three different damping ratios over a swept frequency range (0< <8) = 0.1 = 0.7 = 2.0 In acceleration mode, output signal is in phase and in nearunity transmission for low and then enters attenuated, outofphase behavior for postresonant Very low or very high damping shrink the useful transmission band ( = 0.7 is optimal)
25 Measuring Wideband Signals Beyond Transmission Band We will assume that the sensor behaves as a linear dynamic system, so multiple inputs may be treated by superposition; suppose a sensor s resonance is at 3 Hz, and we try to measure a signal composed of 1 Hz and 5 Hz components: x(t) = sin2t + sin10t Input components are in black, outputs in red: 1 Hz component + 5 Hz component = total input and response Not very good signal representation due to distortions induced by resonance
26 An Accelerometer CutAway Let s look at a cutaway view of a seismic accelerometer: contains seismic mass, as expected utilizes elastic beams as springs in this example, uses fiber Bragg gratings to transduce bending of beams into displacement/acceleration seismic mass housing fiber optic cable fiber Bragg grating (fiber optic transducer, attached to beam) elastic beams (springs)
27 Sensor Performance Properties We will utilize this example of an accelerometer to explore several sensor performance properties: sensitivity: What is the response of the sensor to inputs, usually over a range of time scales? (Usually given in terms of the transfer function) resolution: What is the minimum detectable value of the intended input? (Usually given in terms of power or amplitude spectral density) crossaxis sensitivity: How much does the sensor respond to inputs not aligned with the primary sensing direction? (Usually expressed as a fraction of the main sensitivity) multiple resonances: Does the sensor have multiple nonlinear (resonant) areas that affect sensitivity and response? (The answer is typically, yes ). sensitivity to extraneous measurands: Does the sensor respond to unintended inputs? (Does an accelerometer, for example, also respond to strain or temperature inputs yielding false signals?)
MCE603: Interfacing and Control of Mechatronic Systems. Topic 7c: Piezoelectric Accelerometers and Strain Sensors. Charge Amplifiers.
MCE603: Interfacing and Control of Mechatronic Systems Chapter 7: Actuators and Sensors Topic 7c: Piezoelectric Accelerometers and Strain Sensors. Charge Amplifiers. Reference: Gautschi, Ch.9. Cleveland
More informationSteps to selecting the right accelerometer TP 327
Steps to selecting the right accelerometer TP 327 Steps to selecting the right accelerometer At first look, an accelerometer manufacturer s catalog or web site can be intimidating to both the novice and
More informationStructural Dynamics, Dynamic Force and Dynamic System
Structural Dynamics, Dynamic Force and Dynamic System Structural Dynamics Conventional structural analysis is based on the concept of statics, which can be derived from Newton s 1 st law of motion. This
More informationChapter 5: Tool Dynamometers
Chapter 5: Tool Dynamometers LEARNING OBJECTIVES Different types of transducers used in Dynamometers Design Requirements Types of Dynamometers 
More information1.10 Using Figure 1.6, verify that equation (1.10) satisfies the initial velocity condition. t + ") # x (t) = A! n. t + ") # v(0) = A!
1.1 Using Figure 1.6, verify that equation (1.1) satisfies the initial velocity condition. Solution: Following the lead given in Example 1.1., write down the general expression of the velocity by differentiating
More informationLesson 11. Luis Anchordoqui. Physics 168. Tuesday, December 8, 15
Lesson 11 Physics 168 1 Oscillations and Waves 2 Simple harmonic motion If an object vibrates or oscillates back and forth over same path each cycle taking same amount of time motion is called periodic
More informationModeling Mechanical Systems
chp3 1 Modeling Mechanical Systems Dr. Nhut Ho ME584 chp3 2 Agenda Idealized Modeling Elements Modeling Method and Examples Lagrange s Equation Case study: Feasibility Study of a Mobile Robot Design Matlab
More informationModal Excitation. D. L. Brown University of Cincinnati Structural Dynamics Research Laboratory. M. A. Peres The Modal Shop Inc Cincinnati, OH
D. L. Brown University of Cincinnati Structural Dynamics Research Laboratory M. A. Peres The Modal Shop Inc Cincinnati, OH Intoduction The presentation is concerned with a short tutorial on modal excitation.
More informationApplications of SecondOrder Differential Equations
Applications of SecondOrder Differential Equations Secondorder linear differential equations have a variety of applications in science and engineering. In this section we explore two of them: the vibration
More informationLecture L19  Vibration, Normal Modes, Natural Frequencies, Instability
S. Widnall 16.07 Dynamics Fall 2009 Version 1.0 Lecture L19  Vibration, Normal Modes, Natural Frequencies, Instability Vibration, Instability An important class of problems in dynamics concerns the free
More informationThere are many different type of accelerometers and each has unique characteristics, advantages and disadvantages. The different types include:
How does a piezoelectric accelerometer work? Piezoelectric crystals are manmade or naturally occurring crystals that produce a charge output when they are compressed, flexed or subjected to shear forces.
More informationAUTODESK SIMULATION MULTIPHYSICS 2013
AUTODESK SIMULATION MULTIPHYSICS 2013 Which Analysis to Use? FANKOM MÜHENDİSLİK 2/4/2013 AUTODESK SIMULATION MULTIPHYSICS Which Analysis to Use? Use the following guidelines to help choose the correct
More informationRANDOM VIBRATION AN OVERVIEW by Barry Controls, Hopkinton, MA
RANDOM VIBRATION AN OVERVIEW by Barry Controls, Hopkinton, MA ABSTRACT Random vibration is becoming increasingly recognized as the most realistic method of simulating the dynamic environment of military
More informationUnderstanding Dynamic Range in Acceleration Measurement Systems. February 2013 By: Bruce Lent
in Acceleration Measurement Systems February 2013 By: Bruce Lent Topics to discuss Definition of dynamic range The effective range Making full use of the high level Using filters to improve dynamic range
More informationRLC Circuits. OBJECTIVES To observe free and driven oscillations of an RLC circuit.
ircuits It doesn t matter how beautiful your theory is, it doesn t matter how smart you are. If it doesn t agree with experiment, it s wrong. ichard Feynman (19181988) OBJETIVES To observe free and driven
More informationMounting instructions. Acceleration Transducer B12. B 26.B12.10 en
Mounting instructions Acceleration Transducer B12 B 26.B12.10 en B12 3 Contents Page Safety instructions.............................................. 4 1 Scope of supply..............................................
More informationCHAPTER 3 MODAL ANALYSIS OF A PRINTED CIRCUIT BOARD
45 CHAPTER 3 MODAL ANALYSIS OF A PRINTED CIRCUIT BOARD 3.1 INTRODUCTION This chapter describes the methodology for performing the modal analysis of a printed circuit board used in a hand held electronic
More information4 SENSORS. Example. A force of 1 N is exerted on a PZT5A disc of diameter 10 mm and thickness 1 mm. The resulting mechanical stress is:
4 SENSORS The modern technical world demands the availability of sensors to measure and convert a variety of physical quantities into electrical signals. These signals can then be fed into data processing
More informationLaminar and Turbulent flow. Flow Sensors. Reynolds Number. Thermal flow Sensor. Flow and Flow rate. R = Mass Flow controllers
Flow and Flow rate. Laminar and Turbulent flow Laminar flow: smooth, orderly and regular Mechanical sensors have inertia, which can integrate out small variations due to turbulence Turbulent flow: chaotic
More informationExperimental Modal Analysis
Experimental Analysis A Simple NonMathematical Presentation Peter Avitabile, University of Massachusetts Lowell, Lowell, Massachusetts Often times, people ask some simple questions regarding modal analysis
More informationSoil Dynamics Prof. Deepankar Choudhury Department of Civil Engineering Indian Institute of Technology, Bombay
Soil Dynamics Prof. Deepankar Choudhury Department of Civil Engineering Indian Institute of Technology, Bombay Module  2 Vibration Theory Lecture  8 Forced Vibrations, Dynamic Magnification Factor Let
More information22.302 Experiment 5. Strain Gage Measurements
22.302 Experiment 5 Strain Gage Measurements Introduction The design of components for many engineering systems is based on the application of theoretical models. The accuracy of these models can be verified
More informationA C O U S T I C S of W O O D Lecture 3
Jan Tippner, Dep. of Wood Science, FFWT MU Brno jan. tippner@mendelu. cz Content of lecture 3: 1. Damping 2. Internal friction in the wood Content of lecture 3: 1. Damping 2. Internal friction in the wood
More informationDYNAMIC ANALYSIS ON STEEL FIBRE
International Journal of Civil Engineering and Technology (IJCIET) Volume 7, Issue 2, MarchApril 2016, pp. 179 184, Article ID: IJCIET_07_02_015 Available online at http://www.iaeme.com/ijciet/issues.asp?jtype=ijciet&vtype=7&itype=2
More informationUnsteady Pressure Measurements
Quite often the measurements of pressures has to be conducted in unsteady conditions. Typical cases are those of the measurement of timevarying pressure (with periodic oscillations or step changes) the
More informationThe Sonometer The Resonant String and Timbre Change after plucking
The Sonometer The Resonant String and Timbre Change after plucking EQUIPMENT Pasco sonometers (pick up 5 from teaching lab) and 5 kits to go with them BK Precision function generators and Tenma oscilloscopes
More information19.7. Applications of Differential Equations. Introduction. Prerequisites. Learning Outcomes. Learning Style
Applications of Differential Equations 19.7 Introduction Blocks 19.2 to 19.6 have introduced several techniques for solving commonlyoccurring firstorder and secondorder ordinary differential equations.
More informationCoupled spring equations
int. j. math. educ. sci. technol., 2003 vol. 34, no. 1, 65 79 Coupled spring equations TEMPLE H. FAY* Technikon Pretoria and Mathematics, University of Southern Mississippi, Box 5045, Hattiesburg, MS 394065045,
More informationResponse to Harmonic Excitation Part 2: Damped Systems
Response to Harmonic Excitation Part 2: Damped Systems Part 1 covered the response of a single degree of freedom system to harmonic excitation without considering the effects of damping. However, almost
More informationDynamic Process Modeling. Process Dynamics and Control
Dynamic Process Modeling Process Dynamics and Control 1 Description of process dynamics Classes of models What do we need for control? Modeling for control Mechanical Systems Modeling Electrical circuits
More informationHooke s Law and Simple Harmonic Motion
Hooke s Law and Simple Harmonic Motion OBJECTIVE to measure the spring constant of the springs using Hooke s Law to explore the static properties of springy objects and springs, connected in series and
More informationMath 267  Practice exam 2  solutions
C Roettger, Fall 13 Math 267  Practice exam 2  solutions Problem 1 A solution of 10% perchlorate in water flows at a rate of 8 L/min into a tank holding 200L pure water. The solution is kept well stirred
More informationPID Control. Chapter 10
Chapter PID Control Based on a survey of over eleven thousand controllers in the refining, chemicals and pulp and paper industries, 97% of regulatory controllers utilize PID feedback. Desborough Honeywell,
More information3.2 Sources, Sinks, Saddles, and Spirals
3.2. Sources, Sinks, Saddles, and Spirals 6 3.2 Sources, Sinks, Saddles, and Spirals The pictures in this section show solutions to Ay 00 C By 0 C Cy D 0. These are linear equations with constant coefficients
More informationINTERNOISE 2007 2831 AUGUST 2007 ISTANBUL, TURKEY
INTERNOISE 2007 2831 AUGUST 2007 ISTANBU, TURKEY Force estimation using vibration data Ahmet Ali Uslu a, Kenan Y. Sanliturk b, etin Gül Istanbul Technical University Faculty of echanical Engineering
More informationUNIVERSITY OF CALIFORNIA, SAN DIEGO Electrical & Computer Engineering Department ECE 101  Fall 2009 Linear Systems Fundamentals
UNIVERSITY OF CALIFORNIA, SAN DIEGO Electrical & Computer Engineering Department ECE 101  Fall 2009 Linear Systems Fundamentals MIDTERM EXAM You are allowed one 2sided sheet of notes. No books, no other
More informationLab M1: The Simple Pendulum
Lab M1: The Simple Pendulum Introduction. The simple pendulum is a favorite introductory exercise because Galileo's experiments on pendulums in the early 1600s are usually regarded as the beginning of
More informationAP Physics C. Oscillations/SHM Review Packet
AP Physics C Oscillations/SHM Review Packet 1. A 0.5 kg mass on a spring has a displacement as a function of time given by the equation x(t) = 0.8Cos(πt). Find the following: a. The time for one complete
More informationMethods for Vibration Analysis
. 17 Methods for Vibration Analysis 17 1 Chapter 17: METHODS FOR VIBRATION ANALYSIS 17 2 17.1 PROBLEM CLASSIFICATION According to S. H. Krandall (1956), engineering problems can be classified into three
More informationMinimizing measurement uncertainty in calibration and use of accelerometers TP 299
Minimizing measurement uncertainty in calibration and use of accelerometers TP 299 Minimizing measurement uncertainty in calibration and use of accelerometers ABSTRACT This paper summarizes techniques
More informationManufacturing Equipment Modeling
QUESTION 1 For a linear axis actuated by an electric motor complete the following: a. Derive a differential equation for the linear axis velocity assuming viscous friction acts on the DC motor shaft, leadscrew,
More informationAcceleration levels of dropped objects
Acceleration levels of dropped objects cmyk Acceleration levels of dropped objects Introduction his paper is intended to provide an overview of drop shock testing, which is defined as the acceleration
More informationReview of First and SecondOrder System Response 1
MASSACHUSETTS INSTITUTE OF TECHNOLOGY DEPARTMENT OF MECHANICAL ENGINEERING 2.151 Advanced System Dynamics and Control Review of First and SecondOrder System Response 1 1 FirstOrder Linear System Transient
More informationSpring Simple Harmonic Oscillator. Spring constant. Potential Energy stored in a Spring. Understanding oscillations. Understanding oscillations
Spring Simple Harmonic Oscillator Simple Harmonic Oscillations and Resonance We have an object attached to a spring. The object is on a horizontal frictionless surface. We move the object so the spring
More informationPractical Application of Industrial Fiber Optic Sensing Systems
Practical Application of Industrial Fiber Optic Sensing Systems John W. Berthold and David B. Needham Davidson Instruments, Inc. P.O. Box 130100, The Woodlands, TX 77393 ABSTRACT In this presentation,
More informationVibration Course Enhancement through a Dynamic MATLAB Graphic User Interface
Vibration Course Enhancement through a Dynamic MATLAB Graphic User Interface Elizabeth K. Ervin 1 [Weiping Xu 2 ] Abstract From the string of a guitar to the radio wave, vibration occurs all the time and
More informationState Newton's second law of motion for a particle, defining carefully each term used.
5 Question 1. [Marks 28] An unmarked police car P is, travelling at the legal speed limit, v P, on a straight section of highway. At time t = 0, the police car is overtaken by a car C, which is speeding
More informationAnalysis of seismic response control for longspan cablestayed. bridge under traveling wave input *
Analysis of seismic response control for longspan cablestayed bridge under traveling wave input * QI ingjun, LI iaojun 2 ( Associate Professor, School of Civil Engineering, Shandong Jianzhu University,
More informationFrequency response of a general purpose singlesided OpAmp amplifier
Frequency response of a general purpose singlesided OpAmp amplifier One configuration for a general purpose amplifier using an operational amplifier is the following. The circuit is characterized by:
More informationFiber Optic High Density ACC Sensor System Faser Optisches BeschleunigungsMesssystem
End Winding Remote unit PNR: 9600.30.180 Remote unit and 6 x OneAxis optical, MICAccelerometer Frequency response 1 1000 Hz Easy installation on the endwinding High voltage electrical insulation between
More informationFirst Order System. Transfer function: Response to a unit step input is: Partial Fraction Expansion leads to: Inverse Laplace transform leads to:
First Order System Transfer function: Response to a unit step input is: Partial Fraction Expansion leads to: Inverse Laplace transform leads to: First Order System At t = T, the output is: T represents
More informationNotice numbers may change randomly in your assignments and you may have to recalculate solutions for your specific case.
HW1 Possible Solutions Notice numbers may change randomly in your assignments and you may have to recalculate solutions for your specific case. Tipler 14.P.003 An object attached to a spring has simple
More informationTransducer Mounting and Test Setup Configurations. Rick Bono The Modal Shop
Transducer Mounting and Test Setup Configurations Rick Bono The Modal Shop 1 Transducer Mounting Mechanical connection method Stud mount Adhesive mount Magnetic mount Pressfit friction mount Test parameter
More informationStructural Dynamics of Linear Elastic SingleDegreeofFreedom (SDOF) Systems
Structural Dynamics of Linear Elastic SingleDegreeofFreedom (SDOF) Systems SDOF Dynamics 31 This set of slides covers the fundamental concepts of structural dynamics of linear elastic singledegreeoffreedom
More informationTCOM 370 NOTES 994 BANDWIDTH, FREQUENCY RESPONSE, AND CAPACITY OF COMMUNICATION LINKS
TCOM 370 NOTES 994 BANDWIDTH, FREQUENCY RESPONSE, AND CAPACITY OF COMMUNICATION LINKS 1. Bandwidth: The bandwidth of a communication link, or in general any system, was loosely defined as the width of
More informationLab 5: Conservation of Energy
Lab 5: Conservation of Energy Equipment SWS, 1meter stick, 2meter stick, heavy duty bench clamp, 90cm rod, 40cm rod, 2 double clamps, brass spring, 100g mass, 500g mass with 5cm cardboard square
More informationSGN1158 Introduction to Signal Processing Test. Solutions
SGN1158 Introduction to Signal Processing Test. Solutions 1. Convolve the function ( ) with itself and show that the Fourier transform of the result is the square of the Fourier transform of ( ). (Hints:
More informationDrive marketwinning innovation, without risk
SOLIDWORKS SIMULATION Drive marketwinning innovation, without risk Ensure adequate exit duct sizing with Flow simulation Check resonance between the fan and the motor Truck Mount SnowBlower Wausau Everest
More informationBIOMEDICAL ULTRASOUND
BIOMEDICAL ULTRASOUND Goals: To become familiar with: Ultrasound wave Wave propagation and Scattering Mechanisms of Tissue Damage Biomedical Ultrasound Transducers Biomedical Ultrasound Imaging Ultrasonic
More informationUpdated 2013 (Mathematica Version) M1.1. Lab M1: The Simple Pendulum
Updated 2013 (Mathematica Version) M1.1 Introduction. Lab M1: The Simple Pendulum The simple pendulum is a favorite introductory exercise because Galileo's experiments on pendulums in the early 1600s are
More informationDYNAMICAL ANALYSIS OF SILO SURFACE CLEANING ROBOT USING FINITE ELEMENT METHOD
International Journal of Mechanical Engineering and Technology (IJMET) Volume 7, Issue 1, JanFeb 2016, pp. 190202, Article ID: IJMET_07_01_020 Available online at http://www.iaeme.com/ijmet/issues.asp?jtype=ijmet&vtype=7&itype=1
More informationMechanical Vibrations
Mechanical Vibrations A mass m is suspended at the end of a spring, its weight stretches the spring by a length L to reach a static state (the equilibrium position of the system). Let u(t) denote the displacement,
More informationOscillations. Vern Lindberg. June 10, 2010
Oscillations Vern Lindberg June 10, 2010 You have discussed oscillations in Vibs and Waves: we will therefore touch lightly on Chapter 3, mainly trying to refresh your memory and extend the concepts. 1
More informationSection 5.0 : Horn Physics. By Martin J. King, 6/29/08 Copyright 2008 by Martin J. King. All Rights Reserved.
Section 5. : Horn Physics Section 5. : Horn Physics By Martin J. King, 6/29/8 Copyright 28 by Martin J. King. All Rights Reserved. Before discussing the design of a horn loaded loudspeaker system, it is
More informationOPTIMAL ANALYSIS, DESIGN AND TESTING OF AN ELEC TROMAGNETIC DAMPER WITH RESONANT SHUNT CIRCUIT FOR VIBRATION CONTROL OF A CIVIL STRUCTURE
OPTIMAL ANALYSIS, DESIGN AND TESTING OF AN ELEC TROMAGNETIC DAMPER WITH RESONANT SHUNT CIRCUIT FOR VIBRATION CONTROL OF A CIVIL STRUCTURE Wai Kei Ao, Paul Reynolds University of Exeter, Vibration Engineering
More informationSecond Order Systems
Second Order Systems Second Order Equations Standard Form G () s = τ s K + ζτs + 1 K = Gain τ = Natural Period of Oscillation ζ = Damping Factor (zeta) Note: this has to be 1.0!!! Corresponding Differential
More informationNumerical Solution of Differential Equations
Numerical Solution of Differential Equations Dr. Alvaro Islas Applications of Calculus I Spring 2008 We live in a world in constant change We live in a world in constant change We live in a world in constant
More informationSafakcan Tuncdemir 1, William M. Bradley *2. 1. Introduction
Modeling and Experimental Verification of the Power Transfer and Thermal Characteristics of Piezoelectric Transformers Subjected to Combined Mechanical and Electrical Loading Safakcan Tuncdemir 1, William
More informationTransient Mass Transfer
Lecture T1 Transient Mass Transfer Up to now, we have considered either processes applied to closed systems or processes involving steadystate flows. In this lecture we turn our attention to transient
More informationcorrectchoice plot f(x) and draw an approximate tangent line at x = a and use geometry to estimate its slope comment The choices were:
Topic 1 2.1 mode MultipleSelection text How can we approximate the slope of the tangent line to f(x) at a point x = a? This is a Multiple selection question, so you need to check all of the answers that
More information1. Fluids Mechanics and Fluid Properties. 1.1 Objectives of this section. 1.2 Fluids
1. Fluids Mechanics and Fluid Properties What is fluid mechanics? As its name suggests it is the branch of applied mechanics concerned with the statics and dynamics of fluids  both liquids and gases.
More informationTransmission Line and Back Loaded Horn Physics
Introduction By Martin J. King, 3/29/3 Copyright 23 by Martin J. King. All Rights Reserved. In order to differentiate between a transmission line and a back loaded horn, it is really important to understand
More information2.6 The driven oscillator
2.6. THE DRIVEN OSCILLATOR 131 2.6 The driven oscillator We would like to understand what happens when we apply forces to the harmonic oscillator. That is, we want to solve the equation M d2 x(t) 2 + γ
More informationBlackbody radiation derivation of Planck s radiation low
Blackbody radiation derivation of Planck s radiation low 1 Classical theories of Lorentz and Debye: Lorentz (oscillator model): Electrons and ions of matter were treated as a simple harmonic oscillators
More informationInput Shaping for Sway Control in Gantry Cranes
IOSR Journal of Mechanical and Civil Engineering (IOSRJMCE) ISSN : 22781684 Volume 1, Issue 2 (MayJune 2012), PP 3646 Input Shaping for Sway Control in Gantry Cranes Jeslin Thalapil 1 1 Department of
More informationTHE NOT SO SIMPLE PENDULUM
INTRODUCTION: THE NOT SO SIMPLE PENDULUM This laboratory experiment is used to study a wide range of topics in mechanics like velocity, acceleration, forces and their components, the gravitational force,
More informationActive Vibration Isolation of an Unbalanced Machine Spindle
UCRLCONF206108 Active Vibration Isolation of an Unbalanced Machine Spindle D. J. Hopkins, P. Geraghty August 18, 2004 American Society of Precision Engineering Annual Conference Orlando, FL, United States
More informationDecoding an Accelerometer Specification. What Sensor Manufacturer s Don t Tell You! David Lally VP Engineering PCB Piezotronics, Inc.
Decoding an Accelerometer Specification Sheet What Sensor Manufacturer s Don t Tell You! David Lally VP Engineering PCB Piezotronics, Inc. IMACXXVII February 2008 1 Overview Specification Sheet Provides
More informationIMU Components An IMU is typically composed of the following components:
APN064 IMU Errors and Their Effects Rev A Introduction An Inertial Navigation System (INS) uses the output from an Inertial Measurement Unit (IMU), and combines the information on acceleration and rotation
More informationPiezo Technologies  Technical Resource Paper
An Overview of the Properties of Different Piezoceramic Materials Material Families Four of the more important types of piezoceramic materials are introduced below. Lead ZirconateTitanate  Because of
More informationPhysics 41 HW Set 1 Chapter 15
Physics 4 HW Set Chapter 5 Serway 8 th OC:, 4, 7 CQ: 4, 8 P: 4, 5, 8, 8, 0, 9,, 4, 9, 4, 5, 5 Discussion Problems:, 57, 59, 67, 74 OC CQ P: 4, 5, 8, 8, 0, 9,, 4, 9, 4, 5, 5 Discussion Problems:, 57, 59,
More informationPhysics 6C, Summer 2006 Homework 2 Solutions
Physics 6C, Summer 006 Homework Solutions All problems are from the nd edition of Walker. Numerical values are different for each student. Chapter 3 Problems. Figure 330 below shows a circuit containing
More informationThe accelerometer designed and realized so far is intended for an. aerospace application. Detailed testing and analysis needs to be
86 Chapter 4 Accelerometer Testing 4.1 Introduction The accelerometer designed and realized so far is intended for an aerospace application. Detailed testing and analysis needs to be conducted to qualify
More informationDispersion diagrams of a waterloaded cylindrical shell obtained from the structural and acoustic responses of the sensor array along the shell
Dispersion diagrams of a waterloaded cylindrical shell obtained from the structural and acoustic responses of the sensor array along the shell B.K. Jung ; J. Ryue ; C.S. Hong 3 ; W.B. Jeong ; K.K. Shin
More informationAP1 Oscillations. 1. Which of the following statements about a springblock oscillator in simple harmonic motion about its equilibrium point is false?
1. Which of the following statements about a springblock oscillator in simple harmonic motion about its equilibrium point is false? (A) The displacement is directly related to the acceleration. (B) The
More informationReview of Fourier series formulas. Representation of nonperiodic functions. ECE 3640 Lecture 5 Fourier Transforms and their properties
ECE 3640 Lecture 5 Fourier Transforms and their properties Objective: To learn about Fourier transforms, which are a representation of nonperiodic functions in terms of trigonometric functions. Also, to
More informationPipe Vibration and Pressure Detection
Briiel & Kjaer ffi]j W i l Pipe Vibration and Pressure Detection 13069 I CONTENTS Page Pipe Vibration Literature 1 Detection of Pressure Variations in Thin Walled Pipes by Vibration Measurement 3 Pipe
More informationIntroduction To Materials Science FOR ENGINEERS, Ch. 5. Diffusion. MSE 201 Callister Chapter 5
Diffusion MSE 21 Callister Chapter 5 1 Goals: Diffusion  how do atoms move through solids? Fundamental concepts and language Diffusion mechanisms Vacancy diffusion Interstitial diffusion Impurities Diffusion
More information1.(6pts) Find symmetric equations of the line L passing through the point (2, 5, 1) and perpendicular to the plane x + 3y z = 9.
.(6pts Find symmetric equations of the line L passing through the point (, 5, and perpendicular to the plane x + 3y z = 9. (a x = y + 5 3 = z (b x (c (x = ( 5(y 3 = z + (d x (e (x + 3(y 3 (z = 9 = y 3
More informationCASE STUDY. University of Windsor, Canada Modal Analysis and PULSE Reflex. Automotive Audio Systems PULSE Software including PULSE Reflex, Transducers
CASE STUDY University of Windsor, Canada Modal Analysis and PULSE Reflex Automotive Audio Systems PULSE Software including PULSE Reflex, Transducers The automotive glass subwoofer technology has been in
More informationAdvanced Measurements
Albaha University Faculty of Engineering Mechanical Engineering Department Lecture 9: Pressure measurement Ossama Abouelatta o_abouelatta@yahoo.com Mechanical Engineering Department Faculty of Engineering
More informationMethods to predict fatigue in CubeSat structures and mechanisms
Methods to predict fatigue in CubeSat structures and mechanisms By Walter Holemans (PSC), Floyd Azure (PSC) and Ryan Hevner (PSC) Page 1 Outline Problem Statement What is fatigue? Cyclic loading and strength
More informationPrelab Exercises: Hooke's Law and the Behavior of Springs
59 Prelab Exercises: Hooke's Law and the Behavior of Springs Study the description of the experiment that follows and answer the following questions.. (3 marks) Explain why a mass suspended vertically
More informationAn Electromagnetic Micro Power Generator Based on Mechanical Frequency UpConversion
International Journal of Materials Science and Engineering Vol. 1, No. December 013 An Electromagnetic Micro Power Generator Based on Mechanical Frequency UpConversion Vida Pashaei and Manouchehr Bahrami
More informationUnderstanding Poles and Zeros
MASSACHUSETTS INSTITUTE OF TECHNOLOGY DEPARTMENT OF MECHANICAL ENGINEERING 2.14 Analysis and Design of Feedback Control Systems Understanding Poles and Zeros 1 System Poles and Zeros The transfer function
More informationAlternative Linear Motion Systems. Iron Core Linear Motors
Alternative Linear Motion Systems ME EN 7960 Precision Machine Design Topic 5 ME EN 7960 Precision Machine Design Alternative Linear Motion Systems 51 Iron Core Linear Motors Provide actuation forces
More informationComparison of the Response of a Simple Structure to Single Axis and Multiple Axis Random Vibration Inputs
Comparison of the Response of a Simple Structure to Single Axis and Multiple Axis Random Vibration Inputs Dan Gregory Sandia National Laboratories Albuquerque NM 87185 (505) 8449743 Fernando Bitsie Sandia
More informationFluids and Solids: Fundamentals
Fluids and Solids: Fundamentals We normally recognize three states of matter: solid; liquid and gas. However, liquid and gas are both fluids: in contrast to solids they lack the ability to resist deformation.
More informationENERGY TRANSFER SYSTEMS AND THEIR DYNAMIC ANALYSIS
ENERGY TRANSFER SYSTEMS AND THEIR DYNAMIC ANALYSIS Many mechanical energy systems are devoted to transfer of energy between two points: the source or prime mover (input) and the load (output). For chemical
More informationENGR 210 Lab 11 Frequency Response of Passive RC Filters
ENGR 210 Lab 11 Response of Passive RC Filters The objective of this lab is to introduce you to the frequencydependent nature of the impedance of a capacitor and the impact of that frequency dependence
More information