1 1 / 38 Lectures about XRF (X-Ray Fluorescence) Advanced Physics Laboratory Laurea Magistrale in Fisica year Camerino
2 2 / 38 X-ray Fluorescence XRF is an acronym for X-Ray Fluorescence. The XRF technique is considered one of the most widespread and useful methods for the determination of presence and quantity of the elements in a given substance. X-ray fluorescence is a consequence of the ionization of atoms, for which the excited state returns to the ground state emitting x-ray photons of well-defined energies. The characteristic x-ray emission can be induced either by interaction with charged particles or with x-ray photons generated by a Coolidge tube. The XRF technique here discussed is based on the primary excitation of a x-ray beam, followed by the secondary x-ray fluorescence emission. The x-ray emission is then detected and analysed by a photon detector, usually a solid state device characterized by a sufficient energy resolution.
3 3 / 38 X-ray Fluorescence As an example, a typical tube used as a x-ray generator (known as Coolidge tube) is described in the following figure: There are of course many alternative ways of producing x-rays: rotating anodes, radioisotopes (Fe 55, Cd 109, Am 241 ), storage rings.
4 This device was designed by William Coolidge in The Coolidge tube, also called hot cathode tube, is the most widely used. It works with a good quality vacuum (about 10 4 Pa, or 10 9 Atm). In the Coolidge tube, the electrons are produced by thermionic effect from a tungsten filament heated by an electric current. The filament is the cathode of the tube. The high voltage potential is between the cathode and the anode, the electrons are thus accelerated, and then hit the anode. 4 / 38
5 Usually in this (sidewindow) tubes we find an Electrostatic Lens to focus the beam onto a very small spot on the anode. The anode is specially designed to dissipate the heat and wear resulting from this intense focused barrage of electrons: Mechanically spun to increase the area heated by the beam. Cooled by circulating coolant.the anode is angled at 120 degrees off perpendicular to the electron current so as to allow escape of some of the X-ray photons which are emitted essentially perpendicular to the direction of the electron current. The anode is usually made out of W, Mo, or Cu. The tube has a window designed for escape of the generated Xray photons. The power of a Coolidge tube usually ranges from 1 to 4 kw. 5 / 38
6 The x-ray emission spectrum at the anode consists basically in two components: the Bremsstrahlung (deceleration) radiation and the x-ray fluorescence pattern. The first is a continuous spectrum extending up to the limiting electron energy (for example 20 kev if the potential difference between the anode and the cathode is 20 kv). The x-ray fluorescent pattern is instead composed by well-defined peaks located at energy (or wavelengths) typical of the atomic constituents of the anode. These energies reflects the electronic structure of the atoms. The origin of the Bremsstrahlung is due to the deceleration imposed to the electrons arriving inside the anode from the surface, while the xray fluorescence pattern is due to the emission of photons as a result of a de-excitation of the electrons of the atomic core levels (originally excited as a result of the collisions with the electrons of the incoming beam. 6 / 38
7 X-ray Fluorescence: the x-ray source 7 / 38
8 8 / 38 X-ray Fluorescence XRF is a non-invasive technique, leaving the sample unchanged after application. In fact, the electrons involved in the x-ray emission processes are those related to the core levels and for this reason chemical bonds, structure and morphology are not affected. Moreover, the number of excitation processes is low as compared to the available electrons (weak field approximation, valid for all of the x-ray sources except free-electron lasers). A notable exception for the application of XRF is on bio-materials, for which changes can be observed. The samples do not usually need any specific preparation because the energy of the exciting photons is usually high enough (10-30 kev) allowing us to analyse depths of the order of 10 microns or more.
9 9 / 38 X-ray Fluorescence The examination of the XRF spectra allows us to identify the atomic species present in a given sample. In fact, the energy of the x-ray emission lines is a well-defined function of the atomic number Z (see following slides). For its powerful information content and simple application methodology, the XRF technique is applied in many different contexts and fields, from basic physics to chemistry, material science and mineralogy.
10 10 / 38 X-ray Fluorescence Atomic levels involved in the emission of x-ray emission according to the accepted terminology (line K α1 corresponds to a transition from the L 3 (2p 3/2 ) to the K (1s) level).
11 11 / 38 X-ray Fluorescence X-ray emission spectra (K, L, M) ordered according to the atomic number of the elements.
12 12 / 38 X-ray Fluorescence Typical Fluorescence pattern obtained using an energy-dispersive detector.
13 13 / 38 X-ray Fluorescence For light (Mg, Al, Si, Ca..) or medium-z (Cr, Fe, Cu, Ag..) elements the x-ray emission lines of the K series are used preferentially (those involving the 1s core electrons). For heavier elements (like Au, Pb..) the L emission lines, falling in the useful x-ray range under investigation, are used. Those lines involved transitions to the upper levels 2s, 2p 3/2, 2p 1/2. There are different methods for collecting XRF spectra depending on the particular experimental needs, basically divided into energy and angular dispersive techniques.
14 14 / 38 X-ray Fluorescence The XRF collected in Wavelength Dispersive mode (WDX, or WDAX, or WDS) the spectrum of emitted photons is measured after a monochromator (crystal). This method is based on a complex experimental set-up and usually requires longer collection times but can give an excellent energy resolution. The rapid collection of x-ray emission patterns is allowed by the use of modern solid-state detectors coupled with standard multi-channel electronics for photon counting. Those detectors can have sufficient energy resolution to identify different emission lines and are at the basis of the energy dispersive spectrometry (EDS, EDX or Energy Dispersive Analysis X-EDAX)). The experimental set-up may be portable.
16 16 / 38 X-ray Fluorescence: detectors There are several limitations of the solid-state detectors including their resolution and energy-dependent efficiency, which is also limited for low energies by the Be window protecting the detector (light elements can not be detected). The Be window is used for two main reasons: 1. Protection of the solid-state (Si or Ge typically) 2. Preservation of the internal vacuum conditions, in which the sensor is cooled down (using liquid nitrogen or thermoelectric devices) avoiding condensation of water onto the detector.
17 17 / 38 X-ray Fluorescence: detectors Transmission of a Be window (thickness of 10 µm)
18 X-ray Fluorescence: detectors Good solid-state detectors consist essentially of a microns thick silicon junction type p-i-n diode (PIN) with a bias up to 1000 V across it. The lithium-drifted centre part forms the non-conducting i-layer, where Li compensates the residual acceptors which would otherwise make the layer p-type. When an X-ray photon passes through, it causes a swarm of electron-hole pairs to form, and this causes a voltage pulse. To obtain sufficiently low conductivity, the detector must be maintained at low temperature, and liquid-nitrogen must be used for the best resolution. With some loss of resolution, the much more convenient Peltier cooling can be employed. 18 / 38
19 X-ray Fluorescence: detectors Ordinary Silicon PIN photodiodes can serve as detectors for X-ray and gamma ray photons. The detection efficiency is a function of the thickness of the silicon wafer. For a wafer thickness of 300 microns (ignoring attenuation in the diode window and/or package) the detection efficiency can is close to 100% at 10 KeV, falling to approximately 1% at 150 KeV(3). For energies above approximately 60 KeV, photons interact almost entirely through Compton scattering. Moreover, the active region of the diode is in electronic equilibrium with the surrounding medium the diode package, substrate, window and outer coating, etc., so that Compton recoil electrons which are produced near and close enough to penetrate the active volume of the diode, are also detected. For this reason the overall detection efficiency at 150 KeV and above is maintained fairly constant (approximately 1%) over a wide range of photon energies. A silicon PIN diode can be thought of as a solid-state equivalent to an ionization-chamber radiation detector. 19 / 38
20 X-ray Fluorescence: detectors 20 / 38
21 X-ray Fluorescence: detectors Knowledge of the photon collection yield of the detector is important for a quantitative analysis of the XRF spectra. 21 / 38
22 22 / 38 X-ray Fluorescence An important feature of the XRF is the possibility of performing a chemical analysis in a rapid and non-distructive way. In particular this technique allows us: a) Qualitative analysis for the identification of the elements through the position of the typical x-ray emission lines in the XRF spectra. b) Quantitative analysis that needs an accurate data-analysis accounting for the intensity of the different x-ray emission lines. The measured intensity depends in fact on a number of factors including the characteristic of the source, of the detector, of the geometry, and sample morphology. The use of standards with known concentration of the elements under investigation is particularly important for precise determinations.
23 23 / 38 X-ray Fluorescence Qualitative analysis, limited to the identification of the atomic species, is relatively simple. Nevertheless, in some cases there is an almost perfect overlap of emission lines of two successive series (L e K, M ed L,... ) hindering the identification of some elements. In those cases, a better energy resolution of the detector (obtained for example using an angular dispersive technique) may be necessary. When the emission lines are overlapped and/or the concetration of one of the elements is very low, the identification of the presence of a selected atomic species may be not possible and further inspections may be needed.
24 24 / 38 X-ray Fluorescence Fluorescence yield for K and L levels. The excited states (ionization of core electrons) created by the interaction with the primary photons (or by other processes) are followed by a recombination process following different routes (schematically divided into radiative and non-radiative processes). The Fluorescence yield indicates the fraction of de-excitation processes involving the emission of fluorescence photons.
25 Fluorescence yield references 25 / 38
26 The x-ray data booklet 26 / 38
27 X-ray Fluorescence: quantitative analysis An important feature of the XRF technique is that it is possible, under certain conditions, obtain quantitative information about the concentration of the chemical species present in a given sample (stoichiometry). Theoretical background and methods for performing a reliable quantitative analysis are discussed in various textbooks and specialized articles. Here we shall refer our discussion to the book Quantitative X-ray Spectrometry by Ron Jenkins, R. W. Gould, and Dale Gedcke (2nd edition, 1995). The theoretical framework is fully understood, but the technique is more easily applied to samples with specified thickness and morphology (grain size, texture, surface) and the use of standards of known concentration is often necessary for reliable determinations. The application is vast, from mineralogy to material science, from chemistry to metallurgy also using electron microscopy equipments. 27 / 38
31 XRF - Quant. anal.: excitation of specific (K) core levels 31 / 38
32 XRF - Quant. anal.: inclusion of specific contributions 32 / 38
33 33 / 38 XRF - Quant. anal.: monochromatic source The integral disappears if we can consider the source monochromatic (energy E 0 ) reducing considerably several problems in the data-analysis.
34 34 / 38 XRF - Quant. anal.: infinite thickness If the thickness of the sample can be considered infinite (with respect to the typical photon absorption lenghts, the inverse of the absorption coefficient) further simplifications are obtained. Typical absorption lengths in the x-ray range ( 10 kev) are of the order of 10 µm.
35 35 / 38 XRF - Quant. anal.: secondary absorption The actual intensity of the fluorescence lines depends also on the secondary self-absorption of the sample. So quantitative analysis must take account of both primary and secondary absorption terms, acting at different photon energies.
36 36 / 38 XRF: Quantitative analysis Quantitative XRF data-analysis is possible within the above-mentioned theoretical framework. Geometry of the experiment, source and detector features must be taken into account. Practical data-analysis is based on a peak-fitting technique, in which several fluorescence peaks are identified above a measured background. The intensity of each peak is given by the integrated number of counts related to each peak (after background subtraction). Concentration (Wi) of element i can be obtained by solving the corresponding set of equations for each observed fluorescence peak. Full statistical analysis can be done taking into account the experimental noise.
37 37 / 38 X-ray Fluorescence Typical shape of a XRF spectrum. Peaks indicate the presence of well-defined chemical species.
38 38 / 38 X-ray Fluorescence Typical analysis of XRF spectra: removal of background, peak-fitting parameters.
What are x-rays and how are they generated? What are x-rays? X-rays are part of the electromagnetic radiation spectrum. This spectrum includes radio waves, microwaves, infrared, the visible spectrum, ultra
EDS system Most common X-Ray measurement system in the SEM lab. Major elements (10 wt% or greater) identified in ~10 secs. Minor elements identifiable in ~100 secs. Rapid qualitative and accurate quantitative
Module 9 This module presents information on what X-rays are and how they are produced. Introduction Module 9, Page 2 X-rays are a type of electromagnetic radiation. Other types of electromagnetic radiation
Energy Dispersive X-ray X Analysis in the TEM Lecture 19 Example x-ray x spectra (EDS) Several examples of EDS spectra Note: Relative # of counts Energy range of each spectrum Pure Ge Al film on Si Silica
- X-ray vision http://www.uab.edu/surgonc/cases/gi/case2/ctscanof.htm http://www.museumboerhaave.nl/aacollection/aajpegs/m22/9955.jpg - motivation and outline X-rays have been known for over 110 years.
Copyright (C) JCPDS-International Centre for Diffraction Data 1999 393 Spectral distribution from end window X-ray tubes N. Broll 1, P. de Chateaubourg 2 1 FORTEX - E.N.S.A.I.S. 24, bld de la Victoire,
46 Spectroscopy 22(5) May 2007 Basic Fundamental Parameters in X-Ray Fluorescence The fundamental parameters approach to calibration in X-ray fluorescence is unique because it is based upon the theoretical
X-Ray Fluorescence (XRF): Understanding Characteristic X-Rays What is XRF? X-Ray Fluorescence is defined as The emission of characteristic "secondary" (or fluorescent) X-rays from a material that has been
Production of X-Rays Yoichi Watanabe, Ph.D. Masonic Memorial Building M10-M (612)626-6708 firstname.lastname@example.org MPHY 5170/TRAD 7170, Fall semester Contents 1) Physics of X-ray production 2) The X-ray tube 3)
Physics 441/2: Transmission Electron Microscope Introduction In this experiment we will explore the use of transmission electron microscopy (TEM) to take us into the world of ultrasmall structures. This
X-RAY FLUORESCENCE SPECTROSCOPY IN PLASTICS RECYCLING Brian L. Riise and Michael B. Biddle MBA Polymers, Inc., Richmond, CA, USA Michael M. Fisher American Plastics Council, Arlington, VA, USA X-Ray Fluorescence
1 DETECTION OF GAMMA RADIATION FYSZ460 Syventävien opintojen laboratoriotyöt 1. Introduction Detection of gamma radiation is one of the most important research tools in nuclear physics. Detection of gamma
243 CALCULATION METHODS OF X-RAY SPECTRA: A COMPARATIVE STUDY B. Chyba, M. Mantler, H. Ebel, R. Svagera Technische Universit Vienna, Austria ABSTRACT The accurate characterization of the spectral distribution
Outline Exponential Attenuation Chapter 3 F.A. Attix, Introduction to Radiological Physics and Radiation Dosimetry Simple exponential attenuation and plural modes of absorption arrow-beam vs. broad-beam
Problem Set 6 UV-Vis Absorption Spectroscopy 13-1. Express the following absorbances in terms of percent transmittance: a 0.051 b 0.918 c 0.379 d 0.261 e 0.485 f 0.072 A = log P o /P = log1/t = - log T
Fundamentals of modern UV-visible spectroscopy Presentation Materials The Electromagnetic Spectrum E = hν ν = c / λ 1 Electronic Transitions in Formaldehyde 2 Electronic Transitions and Spectra of Atoms
Raman spectroscopy Lecture Licentiate course in measurement science and technology Spring 2008 10.04.2008 Antti Kivioja Contents - Introduction - What is Raman spectroscopy? - The theory of Raman spectroscopy
The Physics of Energy sources Renewable sources of energy Solar Energy B. Maffei Bruno.email@example.com Renewable sources 1 Solar power! There are basically two ways of using directly the radiative
Introduction to Geiger Counters A Geiger counter (Geiger-Muller tube) is a device used for the detection and measurement of all types of radiation: alpha, beta and gamma radiation. Basically it consists
THEORY OF XRF Getting acquainted with the principles Peter Brouwer THEORY OF XRF Getting acquainted with the principles Peter Brouwer First published in The Netherlands under the title Theory of XRF. Copyright
Production of X-rays and Interactions of X-rays with Matter Goaz and Pharoah. Pages 11-20. Neill Serman Electrons traveling from the filament ( cathode) to the target (anode) convert a small percentage
Experiment N3 X-RAY AND GAMMA RAY SPECTRA References: Handbook of Chemistry and Physics Nuclear Level Schemes, A=45 Through A=257, QC795.8.E5.N8 Background: As in case of atomic physics, much of what we
EDXRF of Used Automotive Catalytic Converters Energy Dispersive X-Ray Fluorescence (EDXRF) is a very powerful technique for measuring the concentration of elements in a sample. It is fast, nondestructive,
COLLEGE PHYSICS Chapter 29 INTRODUCTION TO QUANTUM PHYSICS Quantization: Planck s Hypothesis An ideal blackbody absorbs all incoming radiation and re-emits it in a spectrum that depends only on temperature.
Introduction Until the early 20 th century physicists used to explain the phenomena in the physical world around them using theories such a mechanics, electromagnetism, thermodynamics and statistical physics
X-Ray Photoelectron Spectroscopy Demonstrations Aim: (1) To understand basic XPS theory and experimental practice; (2) To appreciate the analytical strengths of XPS through simple data collection and analysis.
Copyright JCPDS - International Centre for Diffraction Data 24, Advances in X-ray Analysis, Volume 47. 64 ABSTRACT IMPROVEMENTS IN LOW POWER, END-WINDOW, TRANSMISSION-TARGET X-RAY TUBES Charles Jensen,
Basic Concepts of X-ray X Fluorescence by Miguel Santiago, Scientific Instrumentation Specialist University of Puerto Rico Mayagüez Campus Department of Geology X-ray / Microanalysis Laboratory F - 304
Principles of Imaging Science I (RAD119) X-ray Production & Emission X-ray Production X-rays are produced inside the x-ray tube when high energy projectile electrons from the filament interact with the
contamination analysis for compound semiconductors ANALYTICAL SERVICES B u r i e d d e f e c t s, E v a n s A n a l y t i c a l g r o u p h e l p s y o u C O N T R O L C O N T A M I N A T I O N Contamination
Direct Determination of Phosphorus Content in Electroless Nickel Plating Using X-ray Fluorescence (XRF) Spectroscopy Michael Haller Jim Bogert, Ryan Boyle Fischer Technology, Windsor CT, USA Volker Rößiger,
Electron Beam Specimen Interaction The interaction of a high energy electron beam with the specimen will produce various effects resulting in a range of signals being emitted. The incident electrons interact
Application Note: XRF Coating Thickness and Composition Analysis by Micro-EDXRF www.edax.com Coating Thickness and Composition Analysis by Micro-EDXRF Introduction: The use of coatings in the modern manufacturing
Radiology Physics Just take a deep breath OR: I DIDN T SIGN UP TO LEARN THIS STUFF Chris Ober, DVM, PhD, DACVR 7 February 2011 Why worry about physics? Know what the system can give you Know what the system
Gamma Ray Attenuation Properties of Common Shielding Materials Daniel R. McAlister, Ph.D. PG Research Foundation, Inc. 955 University Lane Lisle, IL 60532, USA Introduction In areas where people are likely
Part 3 tracking detectors material effects track models fitting with momentum passage of particles through matter particles traversing through a medium interact with that medium they loose energy ionization:
Scanning Electron Microscopy (SEM) Do it with electrons! Microscopy Structure determines properties We have discussed crystal structure (x-ray diffraction) But consider now different level of structure
A Bit of History A good discussion of the early x-ray discoveries may be found in Chapter 1 of Moore and Reynolds (1997). I have borrowed freely from a variety of sources for this section. An online sketch
L 36 Modern Physics  How lasers work Medical applications of lasers Applications of high power lasers Medical imaging techniques CAT scans MRI s How lasers work First we must understand the difference
ATOMS: PHOTONS AND THE PHOTOELECTRIC EFFECT QUESTIONS SODIUM LAMPS (2012;2) Low pressure sodium lamps are widely used in street lighting. The lamps produce light when an electric current is passed through
CHEMICAL ANALYSIS OF MINERALS Quantitative methodology in mineralogy and mineral physics CHEMICAL ANALYSIS OF MINERALS There are many different types of both qualitative ti and quantitative analysis available
Portable X-ray fluorescence Spectroscopy Michael A. Wilson Research Soil Scientist USDA-NRCS National Soil Survey Center Lincoln, NE OBJECTIVES Background of the method Features of the instrument Applications
The Diode The diode is a two terminal semiconductor device that allows current to flow in only one direction. It is constructed of a P and an N junction connected together. Diode Operation No current flows
CHAPTER 30 THE NATURE OF THE ATOM CONCEPTUAL QUESTIONS 1. REASONING AND SOLUTION A tube is filled with atomic hydrogen at room temperature. Electromagnetic radiation with a continuous spectrum of wavelengths,
X-ray Analysis Chris Boothroyd Center for electron nanoscopy, Technical University of Denmark Advanced TEM course, Thurs 16 Sept 2010 Overview X-ray Analysis in TEM 1. Electron beam interactions and X-ray
Modern Physics Lab Introduction to Nuclear Radiation 9/04 Purpose of the Experiment - become familiar with detectors for radioactive decay products - apply statistical analysis techniques to data - understand
is any technique that uses light to measure chemical concentrations. Theory When a sample absorbs electromagnetic radiation it undergoes a change in energy. The interaction happens between the sample and
Sputtering Vacuum Evaporation Recap Use high temperatures at high vacuum to evaporate (eject) atoms or molecules off a material surface. Use ballistic flow to transport them to a substrate and deposit.
PHOTOELECTRIC EFFECT AND DUAL NATURE OF MATTER AND RADIATIONS 1. Photons 2. Photoelectric Effect 3. Experimental Set-up to study Photoelectric Effect 4. Effect of Intensity, Frequency, Potential on P.E.
Chapter 5 X-ray imaging 5.1 The physics of diagnostic X-rays Learning objectives: How are X-rays produced in an X-ray tube? How is the energy of the photons from an X-ray tube controlled? Why are photons
SURFACE CHARACTERIZATION Marie-Geneviève Centre d Etudes de Chimie Métallurgique, CNRS, Vitry-sur-Seine, France Keywords: Surface Analysis, Surface imaging, Surface composition, Surface chemical analysis,
Electron Beam Lithography in Nanofabrication Electron Solid Interaction Electron Beam Lithography Lee Chow Department of Physics University of Central Florida Some Applications 4/7/2011 Lecture 8 2 Electron
Tutorial 4.6 Gamma Spectrum Analysis Slide 1. Gamma Spectrum Analysis In this module, we will apply the concepts that were discussed in Tutorial 4.1, Interactions of Radiation with Matter. Slide 2. Learning
CHAPTER 2 DRUG SUBSTANCES AND INSTRUMENTS EMPLOYED 2.1 Preparation of Pure polymorphic forms Lamivudine polymorphic Form I and Form II drug substances were prepared in Aurobindo Pharma Limited Research
Name: Class: Date: ID: A Practice questions for Ch. 7 Multiple Choice Identify the choice that best completes the statement or answers the question. 1. When ignited, a uranium compound burns with a green
CHEM 343: Problem Set #4 (Spectroscopy) 1) What is the energy, in ev, of UV radiation at 250 nm? What about Visible radiation at 550 nm? hc a) Use the expression E = = hv. Where c is the speed of light,
Electron Microscopy 3. SEM Image formation, detection, resolution, signal to noise ratio, interaction volume, contrasts 3-1 SEM is easy! Just focus and shoot "Photo"!!! Please comment this picture... Any
Energy < 380 ev Resolution LYNXEYE XE-T High-Resolution Position Sensitive Detector with Superb Energy Resolution The LYNXEYE XE-T is the next generation "Compound Silicon Strip" detector with superb energy
6 Luminescence quantum efficiency and local optical density of states in thin film ruby made by ion implantation Single crystal, (0001) oriented, sapphire samples were implanted with 150 kev Cr ions at
Electron Microscopy SEM and TEM Content 1. Introduction: Motivation for electron microscopy 2. Interaction with matter 3. SEM: Scanning Electron Microscopy 3.1 Functional Principle 3.2 Examples 3.3 EDX
Gamma and X-Ray Detection DETECTOR OVERVIEW The kinds of detectors commonly used can be categorized as: a. Gas-filled Detectors b. Scintillation Detectors c. Semiconductor Detectors The choice of a particular
Select the best charged particle detector for your application PIPS (Passivated Implanted Planar Silicon) detectors CANBERRA offers two types of silicon charged particle detectors to cover a wide range
Abstract EFFICIENCY FOR CLOSE GEOMETRIES AND EXTENDED SOURCES OF A P-TYPE GERMANIUM DETECTOR WITH LOW-ENERGY SENSITIVITY R.M. Keyser (1) and T.R. Twomey (1) (1) ORTEC, 801 South Illinois Avenue, Oak Ridge,
Spectroscopy II - Transmission, Absorption, Fluorescence, Beer s law and Scattering. Spectrophotometry and Fluorometry In this experiment you will use the Vernier SpectroVis Plus, a small, computer controlled
Types of X-ray Detectors C. Segre (IIT) PHYS 570 - Fall 2010 September 21, 2010 1 / 10 Types of X-ray Detectors Gas detectors C. Segre (IIT) PHYS 570 - Fall 2010 September 21, 2010 1 / 10 Types of X-ray
Copyright JCPDS - International Centre for Diffraction Data 2003, Advances in X-ray Analysis, Volume 46. 38 SPECTRAL INTERFERENCE IN X-RAY FLUORESCENCE ANALYSIS OF COMMON MATERIALS Frank R. Feret and Hafida
Chapter 7 Photon Counting and Spectroscopy in the Gamma Ray Energy Range 7.1 Objective Determine whether a radioactive source or set of sources obey Poisson statistics, and whether the radiation is isotropic.
PUMPED Nd:YAG LASER Last Revision: August 21, 2007 QUESTION TO BE INVESTIGATED: How can an efficient atomic transition laser be constructed and characterized? INTRODUCTION: This lab exercise will allow
DETECTION OF CHARGED PARTICLES WITH SILICON DETECTORS A) DESCRIPTION OF SILICON CHARGED PARTICLE DETECTOR The silicon charged particle detector is a wafer of silicon having surface contacts forming a p-n
Ion Beam Sputtering: Practical Applications to Electron Microscopy Applications Laboratory Report Introduction Electron microscope specimens, both scanning (SEM) and transmission (TEM), often require a
Introduction to Spectroscopy. ARCHIMEJ TECHNOLOGY The SPECTROSCOPY 2.0 Company To understand what the core of our project is about, you need to grasp some basic notions of optical spectroscopy. This lesson
Solid State Detectors = Semi-Conductor based Detectors Materials and their properties Energy bands and electronic structure Charge transport and conductivity Boundaries: the p-n junction Charge collection
CHAPTER 4 Structure of the Atom 4.1 The Atomic Models of Thomson and Rutherford 4.2 Rutherford Scattering 4.3 The Classic Atomic Model 4.4 The Bohr Model of the Hydrogen Atom 4.5 Successes and Failures
NIU Radiation Safety Research X-ray Safety Fundamentals Goals of this training presentation: 1. Explain what are X-rays. 2. Explain the hazards of X-ray devices used in NIU research. 3. Explain NIU requirements
ADVANCED FE APPLICATIONS COURSE Theory of Microscopy Thermionic Emitters» Boil electrons over the top of the energy barrier» The current density depends on work the temperature and the cathode function
What does DXA mean and why is it important? Principles of operation of DXA systems Dr Wil Evans Head of Medical Physics University Hospital of Wales, Cardiff DXA stands for Dual energy X-ray Absorptiometry
X-RAY TUBE SELECTION CRITERIA FOR BGA / CSP X-RAY INSPECTION David Bernard Dage Precision Industries Inc. Fremont, California firstname.lastname@example.org ABSTRACT The x-ray inspection of PCB assembly processes