# Application Note Noise Frequently Asked Questions

Size: px
Start display at page:

Transcription

1 : What is? is a random signal inherent in all physical components. It directly limits the detection and processing of all information. The common form of noise is white Gaussian due to the many random processes that make up electric currents or thermal agitation of conductive elements. Why is it Important? Because electronic noise is ubiquitous, present in all passive and active components, it is critical for engineers to characterize and understand how it limits the transmission of information. What does the term Gaussian signify? The term Gaussian refers to the voltage distribution of the source of noise. Due to its random nature the noise voltage of a component is usually a Gaussian distribution. This is characterized by its mean value and random voltage excursions that follow a bell shaped Gaussian curve Plot of voltage distribution of noise source µ=0 V Voltage of noise source versus time What does the term White signify? White refers to the noise source power spectral density, which is ideally flat with frequency. In reality at some point often due to mismatch there is a reduction in the measurable noise level What is AWGN? Additive White Gaussian (AWGN) is a term to refer to the fact that noise eventually combines with the desired signal and is a major limiting factor in the transmission of information.

2 What are the common usages of a noise source? Sources are used to measure Figure, provide a source of AWGN to generate CNR or EbNo to measure error rates, and are used as an economical source of broad band power for built in test applications such as signal strength calibrators and radar applications. They can be used to increase the dynamic range of Analog to Digital converters by dithering and reducing correlated noise. They are often found in Disk drive testing, wireless testing, CATV both analog and DOCSYS, jamming, SATCOM for BER and NF, as well employed as a source of jitter. What types of noise sources are available? sources can be a simple noise diode which generates a low level of noise, to amplified noise sources supplied in multiple form factors to instrumentation grade noise generators which amplify, attenuate and process both the noise and a user added signal. diodes come in a variety of packages and can be surface mount or DIP for PCB mount or coaxial for system integration. What is ENR? ENR refers to Excess Ratio which is 10 log {(Th-290)/290} essentially a normalized measure of how much the noise source is above thermal in its power. At high ENRs > 15 db the density of power can be approximated by adding the ENR to -174 dbm/hz. How are noise sources tested and specified? sources are typically tested depending on their output level. Sources that are used in Figure applications are typically 6 to 30 db in ENR. These usually require a Figure Meter or a dedicated noise radiometer due to their low power levels. Since the ENR value is used to calculate the Figure directly low power noise sources typically are supplied with calibrated ENR values. Higher power noise sources typically are supplied with aggregate power measurements such as on a power meter with spectral flatness observed on a spectrum analyzer. What is the crest factor of a noise source? Sources are characterized by their crest factor, which is the peak to average ratio of the noise. For example a 5:1 crest factor of the noise voltage is 20 log5 =14 db. This is a measure of the quality of the noise distributions and one way to measure its Gaussian nature. theoretically has an unbounded distribution so that it should have an infinite crest factor but the physical realization of the noise generator will limit the output excursions, via the amplifiers, diode junctions etc. Why and when is crest factor important? Crest factor is important primarily in Bit Error Rate applications. In low power applications in which noise powers are being compared such as NF, it is largely insignificant. In BER applications it is important because the BER being measured is a direct function of the Carrier to ratio, and if these noise excursions do not occur as expected the errors will fall off and the erroneous results will occur. One important note is that it is the crest factor of the resultant noise in the receiver and its bandwidth that

3 will determine the resultant crest factor. This is significant because often times the required noise is a much larger BW than the receiver, for example a tuned receiver operating over a wide BW requires the noise to cover the entire RF BW. This can put a strain on the realizable crest factor because the wide band high power amplifiers required to cover the entire BW can be cost prohibitive and degrade system accuracy in other ways such as excess current or reliability. Since the noise is often filtered in the receiver, the crest factor of the resultant noise is improved as the excess BW is stripped away reducing the noise power and leaving the noise farther from the clipping point. In the limit clipped noise becomes Gaussian as the measurement BW is reduced. The required crest factor should take into consideration all of the above. What is BITE? BITE stands for Built-In-Test and refers to the utilization of a noise source as a source of power to test a system. For example the noise source may be put on a PCB via a TO-8 can, dip package or surface mount with a coupler or a switch to selectively inject the noise into the circuit. By turning on the noise source and detecting the system output power the system can be verified automatically and remotely. The noise source can also be used to calibrate the receivers noise figure by comparing its know value to the receiver. temperature, Figure, frequency response, sensitivity and gain are among the parameters that can be measured. What is Eb/No? Eb/No stands for Energy per bit divided by Density. It is essentially a normalized Carrier to ratio for digital systems. Typically Eb/No is plotted versus BER to measure the effectiveness of the information transfer. What is CNR? CNR is Carrier to Ratio and it is the relative power level of the carrier signal to the noise level in a system. It typically determines the quality of the system and BER is plotted against CNR. Carrier refers to the information signal in this case. What is BER? Bit error rate is the frequency of errors that occur when bits are transmitted in a digital system. Critically it is a function of signal to noise ratio or carrier to noise ratio. What is Figure? Figure is defined as the ratio of the signal to noise power at the input to the signal to noise power at the output of a device, in other words the degradation of the signal to noise ratio as the signal passes through the device. Since the input noise level is usually thermal noise from the source the convention is to adopt a reference temperature of 290deg K. The noise figure becomes the ratio of the total noise power output to that portion of the noise power output due to noise at input when the source is 290 deg K. Hewlett Packard App Note 57-1 describes these definitions and calculations.

4 How is noise figure measured and calculated? figure is typically determined by using a calibrated noise source which is traceable to international standards. This noise source is essentially compared to the unknown noise figure and by measuring this difference noise figure is computed: NF= ENR db-10 log (Y-1) + Tcorr Tcorr is a temperature correction factor that can be applied if the temperature deviates significantly from 290 K. Y is the Y factor which is the ratio of the output power with the noise on to the output power with the noise off. By employing this method of measuring the Y factor, only relative accuracies are significant which makes the measurement easier than attempting to measure exact powers which can be quite low and tough to measure. How do noise powers add? powers add as incoherent signals which means that their powers must be added. For example if your inject a noise source into a spectrum analyzer and see that the noise floor increases 3 db, then the actual noise source power is at the original noise floor level. This relationship allows you to calculate the noise power of signals below the measurement noise floor: 10 log [{Inverse log (diff/10)} -1)] Where diff is the db difference in measured powers. Of course small changes in power occur as the unknown noise is far below the known and this results in increasing inaccuracy as the power goes much lower. Why can t I see my noise source on a spectrum analyzer? If you are attempting to measure a lower power noise source, < 30 db ENR, in all probability the spectrum analyzer Figure, which usually is at a minimum of 25 db and many times is 35 db, is above the noise level of noise source. At these levels we can approximate Figure and ENR and compare directly to see if the noise source will be detectable. This source could be measured with an LNA in front of the Spectrum Analyzer although to get an exact ENR we would need to know the NF of the LNA and its gain but we can see if the approximate deflection occurs. For example a 15 db ENR noise source should change the noise level about 10 db if the noise figure of the LNA is about 5 db, as long as the LNA gain is sufficient to overcome the figure of the analyzer. Higher power noise sources can be measured on a spectrum analyzer for flatness and on a power meter for output power. Why test at high power levels? Sometimes it is convenient to test at higher power levels. For example BER measurements are a function of Carrier to Ratio and they can be quite sensitive with large changes in BER resulting with small changes in CNR. Rather than test at low power levels that are very difficult to measure, often times it is easier to inject more noise power and test at levels that are easier to establish what is the actual CNR.

5 Also often times tests are done at lower CNR so that the BER is higher and the low BER results are extrapolated, which saves test time because the errors come so infrequently at high CNR. What is noise power spectral density? Typically referred to as No, this is the amount of power the source will output in a one hertz bandwidth. It is essentially a normalized output power. Since noise power is proportional to bandwidth, No is used to compute the power in any bandwidth. What is -174 dbm/hz? This is a convenient number to use, it represents the amount of power in a one hertz bandwidth that a thermal noise source has at the reference temperature of 290K, which is close to room temperature. This results from the equation P= ktb where k= Boltzmann s constant, T is temp in degrees K and B is the bandwidth. For example the available thermal noise power in a resistor in a 1 MHz bandwidth would be -114 dbm because 10 log (1MHz) =60 db is added to the -174 dbm/ Hz. What is No and how is it used to calculate noise output power? No is the noise density of the Source. It is the output power per hertz that the source provides. To calculate the power that the source will have in a BW the No is added to the db (BW). For example a -80 dbm/hz amplified noise module with 1 GHz BW will have a minimum of -80 dbm/hz + 10 log (1 GHz) = -80 dbm/hz + 90 db = +10 dbm. If this source is measured on a spectrum analyzer with the Resolution BW set to 1 MHz then -80 dbm/hz + 10 log (1 MHz) = -20 dbm will be displayed. In actuality the noise source will have some out of band noise and the resolution BW has a noise equivalent BW greater than its setting so some adjustment of these numbers will be needed for a more accurate number. For many applications this first order approximation will suffice. Aggregate output power should be measured on a power meter, although it could be approximated by adding 10 log (BWns/RBW) to the number on the spectral analyzer. Also when performing power calculations on Sources if the ENR is known the output power density can be approximately calculated by adding the ENR to -174 dbm/hz. This is accurate to less than 0.2 db at 15 db ENR and less than.01 db for ENRs greater than 30 db. For example a 34 db ENR noise source would have a noise spectral density of -174 dbm/hz + 34 db = -140 dbm/hz. In a 10 MHz BW this would result in -140 dbm/hz + 70 db = -70 dbm. For lower ENRs the Th has to be solved for directly from the definition of ENR then the noise density 10 log kth would be computed. Why is Power proportional to BW? Since noise is a random signal its power is distributed over its usable bandwidth BW, and the noise source is considered white due to its constant spectral density. This results in the power measured being proportional to BW. If a certain power is measured in X BW, then if the BW is increased to 2X the power measured is double or 3 db higher. This should be noted when measuring high level noise sources on a spectrum analyzer. This is critical because as a system s BW increases to allow for more information to be processed, this will also introduce more noise power and reduce the CNR and potentially reduce the dynamic range of the system. This is a major trade off in all communication systems. How do I calculate the overall output power of my noise source?

7 is on, the approximate output power will be -174dBm/Hz + 65 db which is -109 dbm/hz. The aggregate power out of the amplifier is then -109 dbm/hz + 10 log (18 GHz) = - 7dBm. This is too close to the P1dB of the preamp as it will begin to compress the amplifier, which due to the large peaks of the noise which will effectively reduce the head room to less than this 7 db difference. In this case a preamp with at least +10 dbm P1dB would be needed, or a noise source of 15 db would be a better selection because the relative differences in the Y factor are reduced, resulting in less stress on the measurement system. Also the power into the spectrum analyzer should be noted as this can compress the analyzer response also. This example is meant to demonstrate the potential care that must be used for this measurement, all these possible errors can be accounted for and various analyzers do some of this but if the operator is not aware noise figure errors can result which may not always be easy to detect. What are some useful conversions? Output Power (dbm) = No + 10 Log (BW) No = ENR for ENRs > 15 db No= Output Power(dBm) 10 Log (BW) What type of noise source should I choose for my application.? If you are attempting to make a noise figure measurement typically you should choose a calibrated coaxial noise source, with either 6, 15 or 30 db ENR. This will allow you to measure Figure using the cal points provided with a noise figure meter or a spectrum analyzer. 15 db is the most common as it can comfortably measure high and low noise figures. Very low noise figures can use a 6 db source which will have reduced VSWR uncertainty and reduced Y factors. 30 db sources are used in high noise figure applications or when the noise maybe injected via a coupler, or high loss device. If you are looking to make BER measurements typically you would want to choose a higher power noise source like an amplified module or an instrument. This will allow you to set Carrier to Ratios easier. Although it can be done with a low power noise source the measurement is difficult at these low powers. Since the BER depends primarily on the ratio of the Carrier to, typically the CNR is set at higher powers with a power meter and a calibrated filter or by measuring on a spectrum analyzer. If you have any additional questions about noise, please do not hesitate to ask us at Wave!

### Optimizing IP3 and ACPR Measurements

Optimizing IP3 and ACPR Measurements Table of Contents 1. Overview... 2 2. Theory of Intermodulation Distortion... 2 3. Optimizing IP3 Measurements... 4 4. Theory of Adjacent Channel Power Ratio... 9 5.

### Visual System Simulator White Paper

Visual System Simulator White Paper UNDERSTANDING AND CORRECTLY PREDICTING CRITICAL METRICS FOR WIRELESS RF LINKS Understanding and correctly predicting cellular, radar, or satellite RF link performance

### Agilent AN 1315 Optimizing RF and Microwave Spectrum Analyzer Dynamic Range. Application Note

Agilent AN 1315 Optimizing RF and Microwave Spectrum Analyzer Dynamic Range Application Note Table of Contents 3 3 3 4 4 4 5 6 7 7 7 7 9 10 10 11 11 12 12 13 13 14 15 1. Introduction What is dynamic range?

### F = S i /N i S o /N o

Noise figure Many receiver manufacturers specify the performance of their receivers in terms of noise figure, rather than sensitivity. As we shall see, the two can be equated. A spectrum analyzer is a

### Agilent AN 1316 Optimizing Spectrum Analyzer Amplitude Accuracy

Agilent AN 1316 Optimizing Spectrum Analyzer Amplitude Accuracy Application Note RF & Microwave Spectrum Analyzers Table of Contents 3 3 4 4 5 7 8 8 13 13 14 16 16 Introduction Absolute versus relative

### VCO Phase noise. Characterizing Phase Noise

VCO Phase noise Characterizing Phase Noise The term phase noise is widely used for describing short term random frequency fluctuations of a signal. Frequency stability is a measure of the degree to which

### MEASUREMENT UNCERTAINTY IN VECTOR NETWORK ANALYZER

MEASUREMENT UNCERTAINTY IN VECTOR NETWORK ANALYZER W. Li, J. Vandewege Department of Information Technology (INTEC) University of Gent, St.Pietersnieuwstaat 41, B-9000, Gent, Belgium Abstract: Precision

### RF Network Analyzer Basics

RF Network Analyzer Basics A tutorial, information and overview about the basics of the RF Network Analyzer. What is a Network Analyzer and how to use them, to include the Scalar Network Analyzer (SNA),

### MATRIX TECHNICAL NOTES

200 WOOD AVENUE, MIDDLESEX, NJ 08846 PHONE (732) 469-9510 FAX (732) 469-0418 MATRIX TECHNICAL NOTES MTN-107 TEST SETUP FOR THE MEASUREMENT OF X-MOD, CTB, AND CSO USING A MEAN SQUARE CIRCUIT AS A DETECTOR

### Power Amplifier Gain Compression Measurements

Technical Brief Power Amplifier Gain Compression Measurements GPIB Private Bus Sweep Out Sweep In Pulse In AC Mod Out Blank/Marker Out Blanking In Overview The 1 db gain compression of an amplifier describes

### 1. The Slotted Line. ECE 584 Microwave Engineering Laboratory Experiments. Introduction:

ECE 584 Microwave Engineering Laboratory Experiments 1. The Slotted Line Introduction: In this experiment we will use a waveguide slotted line to study the basic behavior of standing waves and to measure

### Conquering Noise for Accurate RF and Microwave Signal Measurements. Presented by: Ernie Jackson

Conquering Noise for Accurate RF and Microwave Signal Measurements Presented by: Ernie Jackson The Noise Presentation Review of Basics, Some Advanced & Newer Approaches Noise in Signal Measurements-Summary

### Making Accurate Voltage Noise and Current Noise Measurements on Operational Amplifiers Down to 0.1Hz

Author: Don LaFontaine Making Accurate Voltage Noise and Current Noise Measurements on Operational Amplifiers Down to 0.1Hz Abstract Making accurate voltage and current noise measurements on op amps in

### Experiment 7: Familiarization with the Network Analyzer

Experiment 7: Familiarization with the Network Analyzer Measurements to characterize networks at high frequencies (RF and microwave frequencies) are usually done in terms of scattering parameters (S parameters).

### Understanding Noise Figure

Understanding Noise Figure Iulian Rosu, YO3DAC / VA3IUL, http://www.qsl.net/va3iul One of the most frequently discussed forms of noise is known as Thermal Noise. Thermal noise is a random fluctuation in

### Technical Datasheet Scalar Network Analyzer Model 8003-10 MHz to 40 GHz

Technical Datasheet Scalar Network Analyzer Model 8003-10 MHz to 40 GHz The Giga-tronics Model 8003 Precision Scalar Network Analyzer combines a 90 db wide dynamic range with the accuracy and linearity

### Jeff Thomas Tom Holmes Terri Hightower. Learn RF Spectrum Analysis Basics

Jeff Thomas Tom Holmes Terri Hightower Learn RF Spectrum Analysis Basics Learning Objectives Name the major measurement strengths of a swept-tuned spectrum analyzer Explain the importance of frequency

### Impedance 50 (75 connectors via adapters)

VECTOR NETWORK ANALYZER PLANAR TR1300/1 DATA SHEET Frequency range: 300 khz to 1.3 GHz Measured parameters: S11, S21 Dynamic range of transmission measurement magnitude: 130 db Measurement time per point:

### Timing Errors and Jitter

Timing Errors and Jitter Background Mike Story In a sampled (digital) system, samples have to be accurate in level and time. The digital system uses the two bits of information the signal was this big

### Michael Hiebel. Fundamentals of Vector Network Analysis

Michael Hiebel Fundamentals of Vector Network Analysis TABIH OF CONTENTS Table of contents 1 Introduction 12 1.1 What is a network analyzer? 12 1.2 Wave quantities and S-parameters 13 1.3 Why vector network

### Network Analyzer Operation

Network Analyzer Operation 2004 ITTC Summer Lecture Series John Paden Purposes of a Network Analyzer Network analyzers are not about computer networks! Purposes of a Network Analyzer Measures S-parameters

### Department of Electrical and Computer Engineering Ben-Gurion University of the Negev. LAB 1 - Introduction to USRP

Department of Electrical and Computer Engineering Ben-Gurion University of the Negev LAB 1 - Introduction to USRP - 1-1 Introduction In this lab you will use software reconfigurable RF hardware from National

### RF Communication System. EE 172 Systems Group Presentation

RF Communication System EE 172 Systems Group Presentation RF System Outline Transmitter Components Receiver Components Noise Figure Link Budget Test Equipment System Success Design Remedy Transmitter Components

### SIGNAL GENERATORS and OSCILLOSCOPE CALIBRATION

1 SIGNAL GENERATORS and OSCILLOSCOPE CALIBRATION By Lannes S. Purnell FLUKE CORPORATION 2 This paper shows how standard signal generators can be used as leveled sine wave sources for calibrating oscilloscopes.

### Antenna Properties and their impact on Wireless System Performance. Dr. Steven R. Best. Cushcraft Corporation 48 Perimeter Road Manchester, NH 03013

Antenna Properties and their impact on Wireless System Performance Dr. Steven R. Best Cushcraft Corporation 48 Perimeter Road Manchester, NH 03013 Phone (603) 627-7877 FAX: (603) 627-1764 Email: sbest@cushcraft.com

### CANFI: Cheap Automatic Noise Figure Indicator. Frank Schmäling DL2ALF Wolf-Henning Rech DF9IC Alexander Kurpiers DL8AAU

CANFI: Cheap Automatic Noise Figure Indicator Frank Schmäling DL2ALF Wolf-Henning Rech DF9IC Alexander Kurpiers DL8AAU What is a noise figure meter? Noise figure is defined by source ENR and a measured

### How To Use A Sound Card With A Subsonic Sound Card

!"## \$#!%!"# &"#' ( "#' )*! #+ #,# "##!\$ -+./0 1" 1! 2"# # -&1!"#" (2345-&1 #\$6.7 -&89\$## ' 6! #* #!"#" +" 1##6\$ "#+# #-& :1# # \$ #\$#;1)+#1#+

### Multi-Carrier GSM with State of the Art ADC technology

Multi-Carrier GSM with State of the Art ADC technology Analog Devices, October 2002 revised August 29, 2005, May 1, 2006, May 10, 2006, November 30, 2006, June 19, 2007, October 3, 2007, November 12, 2007

### EE 186 LAB 2 FALL 2004. Network Analyzer Fundamentals and Two Tone Linearity

Network Analyzer Fundamentals and Two Tone Linearity Name: Name: Name: Objective: To become familiar with the basic operation of a network analyzer To use the network analyzer to characterize the in-band

### RFSPACE CLOUD-IQ #CONNECTED SOFTWARE DEFINED RADIO

CLOUD-IQ #CONNECTED SOFTWARE DEFINED RADIO 1 - SPECIFICATIONS Cloud-IQ INTRODUCTION The Cloud-IQ is a high performance, direct sampling software radio with an ethernet interface. It offers outstanding

### Jeff Thomas Tom Holmes Terri Hightower. Learn RF Spectrum Analysis Basics

Jeff Thomas Tom Holmes Terri Hightower Learn RF Spectrum Analysis Basics Agenda Overview: Spectrum analysis and its measurements Theory of Operation: Spectrum analyzer hardware Frequency Specifications

### UNDERSTANDING NOISE PARAMETER MEASUREMENTS (AN-60-040)

UNDERSTANDING NOISE PARAMETER MEASUREMENTS (AN-60-040 Overview This application note reviews noise theory & measurements and S-parameter measurements used to characterize transistors and amplifiers at

### Agilent PN 8753-1 RF Component Measurements: Amplifier Measurements Using the Agilent 8753 Network Analyzer. Product Note

Agilent PN 8753-1 RF Component Measurements: Amplifier Measurements Using the Agilent 8753 Network Analyzer Product Note 2 3 4 4 4 4 6 7 8 8 10 10 11 12 12 12 13 15 15 Introduction Table of contents Introduction

Introduction to Receivers Purpose: translate RF signals to baseband Shift frequency Amplify Filter Demodulate Why is this a challenge? Interference (selectivity, images and distortion) Large dynamic range

### 102 26-m Antenna Subnet Telecommunications Interfaces

DSMS Telecommunications Link Design Handbook 26-m Antenna Subnet Telecommunications Interfaces Effective November 30, 2000 Document Owner: Approved by: Released by: [Signature on file in TMOD Library]

### Network analyzer and spectrum analyzer two in one

R&S ZVL Vector Network Analyzer Network analyzer and spectrum analyzer two in one The R&S ZVL is the lightest and smallest vector network analyzer in its class. On top of this, it can be used as a full-featured

### Automatic compression measurement using network analyzers

Automatic compression measurement using network analyzers Introduction The dynamic range of an amplifier is determined by noise figure and compression. In multi carrier applications third order intercept

### E190Q Lecture 5 Autonomous Robot Navigation

E190Q Lecture 5 Autonomous Robot Navigation Instructor: Chris Clark Semester: Spring 2014 1 Figures courtesy of Siegwart & Nourbakhsh Control Structures Planning Based Control Prior Knowledge Operator

### A Guide to Calibrating Your Spectrum Analyzer

A Guide to Calibrating Your Application Note Introduction As a technician or engineer who works with electronics, you rely on your spectrum analyzer to verify that the devices you design, manufacture,

### 1 Multi-channel frequency division multiplex frequency modulation (FDM-FM) emissions

Rec. ITU-R SM.853-1 1 RECOMMENDATION ITU-R SM.853-1 NECESSARY BANDWIDTH (Question ITU-R 77/1) Rec. ITU-R SM.853-1 (1992-1997) The ITU Radiocommunication Assembly, considering a) that the concept of necessary

### Noise Figure Measurement without a Noise Source on a Vector Network Analyzer

Noise Figure Measurement without a Noise Source on a Vector Network Analyzer Application Note Products: &S ZVA &S ZVT &S ZVAB-K30 &S NP-Z55 This application note describes the noise figure measurement

### Understanding Mixers Terms Defined, and Measuring Performance

Understanding Mixers Terms Defined, and Measuring Performance Mixer Terms Defined Statistical Processing Applied to Mixers Today's stringent demands for precise electronic systems place a heavy burden

### DT3: RF On/Off Remote Control Technology. Rodney Singleton Joe Larsen Luis Garcia Rafael Ocampo Mike Moulton Eric Hatch

DT3: RF On/Off Remote Control Technology Rodney Singleton Joe Larsen Luis Garcia Rafael Ocampo Mike Moulton Eric Hatch Agenda Radio Frequency Overview Frequency Selection Signals Methods Modulation Methods

### Agilent N8973A, N8974A, N8975A NFA Series Noise Figure Analyzers. Data Sheet

Agilent N8973A, N8974A, N8975A NFA Series Noise Figure Analyzers Data Sheet Specifications Specifications are only valid for the stated operating frequency, and apply over 0 C to +55 C unless otherwise

### Amplifier for Small Magnetic and Electric Wideband Receiving Antennas (model AAA-1B)

Amplifier for Small Magnetic and Electric Wideband Receiving Antennas (model AAA-1B) 1. Description and Specifications Contents 1.1 Description 1.2 1.2 Specifications 1.3 1.3 Tested parameters in production

### Measurement of Adjacent Channel Leakage Power on 3GPP W-CDMA Signals with the FSP

Products: Spectrum Analyzer FSP Measurement of Adjacent Channel Leakage Power on 3GPP W-CDMA Signals with the FSP This application note explains the concept of Adjacent Channel Leakage Ratio (ACLR) measurement

### Propagation Channel Emulator ECP_V3

Navigation simulators Propagation Channel Emulator ECP_V3 1 Product Description The ECP (Propagation Channel Emulator V3) synthesizes the principal phenomena of propagation occurring on RF signal links

### 0HDVXULQJWKHHOHFWULFDOSHUIRUPDQFH FKDUDFWHULVWLFVRI5),)DQGPLFURZDYHVLJQDO SURFHVVLQJFRPSRQHQWV

0HDVXULQJWKHHOHFWULFDOSHUIRUPDQFH FKDUDFWHULVWLFVRI5),)DQGPLFURZDYHVLJQDO SURFHVVLQJFRPSRQHQWV The treatment given here is introductory, and will assist the reader who wishes to consult the standard texts

### is the power reference: Specifically, power in db is represented by the following equation, where P0 P db = 10 log 10

RF Basics - Part 1 This is the first article in the multi-part series on RF Basics. We start the series by reviewing some basic RF concepts: Decibels (db), Antenna Gain, Free-space RF Propagation, RF Attenuation,

### RF Measurements Using a Modular Digitizer

RF Measurements Using a Modular Digitizer Modern modular digitizers, like the Spectrum M4i series PCIe digitizers, offer greater bandwidth and higher resolution at any given bandwidth than ever before.

### AMICSA 2012. Integrated SAR Receiver/Converter for L, C and X bands Markku Åberg VTT Technical Research Centre of Finland

AMICSA 2012 Integrated SAR Receiver/Converter for L, C and X bands Markku Åberg VTT Technical Research Centre of Finland 2 The Team Markku Åberg (1), Jan Holmberg (1), Faizah Abu Bakar (2), Tero Nieminen

### Calculating Antenna System Return Loss As Viewed Through The RF Path

Calculating Antenna System Return Loss As Viewed Through The RF Path Lou Meyer, Director of Technical Marketing December 2009 1. Introduction Return loss (RL), reflection coefficient (Γ) and voltage standing

### Phase Noise Measurement Methods and Techniques

Phase Noise Measurement Methods and Techniques Presented by: Kay Gheen, Agilent Technologies Introduction Extracting electronic signals from noise is a challenge for most electronics engineers. As engineers

### Performing Amplifier Measurements with the Vector Network Analyzer ZVB

Product: Vector Network Analyzer R&S ZVB Performing Amplifier Measurements with the Vector Network Analyzer ZVB Application Note This document describes typical measurements that are required to be made

### Engineering Sciences 151. Electromagnetic Communication Laboratory Assignment 3 Fall Term 1998-99

Engineering Sciences 151 Electromagnetic Communication Laboratory Assignment 3 Fall Term 1998-99 WAVE PROPAGATION II: HIGH FREQUENCY SLOTTED LINE AND REFLECTOMETER MEASUREMENTS OBJECTIVES: To build greater

### ANALYZER BASICS WHAT IS AN FFT SPECTRUM ANALYZER? 2-1

WHAT IS AN FFT SPECTRUM ANALYZER? ANALYZER BASICS The SR760 FFT Spectrum Analyzer takes a time varying input signal, like you would see on an oscilloscope trace, and computes its frequency spectrum. Fourier's

### THERMAL ANEMOMETRY ELECTRONICS, SOFTWARE AND ACCESSORIES

TSI and TSI logo are registered trademarks of TSI Incorporated. SmartTune is a trademark of TSI Incorporated. THERMAL ANEMOMETRY ELECTRONICS, SOFTWARE AND ACCESSORIES IFA 300 Constant Temperature Anemometry

### ECE 435 INTRODUCTION TO THE MICROWAVE NETWORK ANALYZER

ECE 435 INTRODUCTION TO THE MICROWAVE NETWORK ANALYZER Latest revision: October 1999 Introduction A vector network analyzer (VNA) is a device capable of measuring both the magnitude and phase of a sinusoidal

### Achieving New Levels of Channel Density in Downstream Cable Transmitter Systems: RF DACs Deliver Smaller Size and Lower Power Consumption

Achieving New Levels of Channel Density in Downstream Cable Transmitter Systems: RF DACs Deliver Smaller Size and Lower Power Consumption Introduction By: Analog Devices, Inc. (ADI) Daniel E. Fague, Applications

### ETSI EN 302 774 V1.2.1 (2012-02)

EN 302 774 V1.2.1 (2012-02) Harmonized European Standard Broadband Wireless Access Systems (BWA) in the 3 400 MHz to 3 800 MHz frequency band; Base Stations; Harmonized EN covering the essential requirements

### T = 1 f. Phase. Measure of relative position in time within a single period of a signal For a periodic signal f(t), phase is fractional part t p

Data Transmission Concepts and terminology Transmission terminology Transmission from transmitter to receiver goes over some transmission medium using electromagnetic waves Guided media. Waves are guided

### ISC Dual Frequency WFS RFPD Test Procedure:

LASER INTERFEROMETER GRAVITATIONAL WAVE OBSERVATORY LIGO Laboratory / LIGO Scientific Collaboration LIGO 08 January 2013 ISC Dual Frequency WFS RFPD Test Procedure: Richard Abbott Distribution of this

### Keysight Technologies Understanding the Fundamental Principles of Vector Network Analysis. Application Note

Keysight Technologies Understanding the Fundamental Principles of Vector Network Analysis Application Note Introduction Network analysis is the process by which designers and manufacturers measure the

Applying Error Correction to Network Analyzer Measurements Application Note 287-3 Table of Contents Page Introduction 2 Sources of Errors and Types of Errors 3 Types of Error Correction 4 One-Port 4 The

RX-AM4SF Receiver The super-heterodyne receiver RX-AM4SF can provide a RSSI output indicating the amplitude of the received signal: this output can be used to create a field-strength meter capable to indicate

### RF and Noise Math Library. By Joseph L. Cahak Copyright 2013 Sunshine Design Engineering Services

RF and Noise Math Library By Joseph L. Cahak Copyright 2013 Sunshine Design Engineering Services Contents RF Math, Power and Spectral Analysis Library... 6 Software... 6 DLL... 6 LLB Labview Library...

### EE4367 Telecom. Switching & Transmission. Prof. Murat Torlak

Path Loss Radio Wave Propagation The wireless radio channel puts fundamental limitations to the performance of wireless communications systems Radio channels are extremely random, and are not easily analyzed

### Reading: HH Sections 4.11 4.13, 4.19 4.20 (pgs. 189-212, 222 224)

6 OP AMPS II 6 Op Amps II In the previous lab, you explored several applications of op amps. In this exercise, you will look at some of their limitations. You will also examine the op amp integrator and

### The front end of the receiver performs the frequency translation, channel selection and amplification of the signal.

Many receivers must be capable of handling a very wide range of signal powers at the input while still producing the correct output. This must be done in the presence of noise and interference which occasionally

### How PLL Performances Affect Wireless Systems

May 2010 Issue: Tutorial Phase Locked Loop Systems Design for Wireless Infrastructure Applications Use of linear models of phase noise analysis in a closed loop to predict the baseline performance of various

### APPLICATION NOTES POWER DIVIDERS. Things to consider

Internet Copy Rev A Overview Various RF applications require power to be distributed among various paths. The simplest way this can be done is by using a power splitter/divider. Power dividers are reciprocal

### APSYN420A/B Specification 1.24. 0.65-20.0 GHz Low Phase Noise Synthesizer

APSYN420A/B Specification 1.24 0.65-20.0 GHz Low Phase Noise Synthesizer 1 Introduction The APSYN420 is a wideband low phase-noise synthesizer operating from 0.65 to 20 GHz. The nominal output power is

### Advanced Photon Source. RF Beam Position Monitor Upgrade Robert M. Lill

Advanced Photon Source RF Beam Position Monitor Upgrade Robert M. Lill Filter Comparator (original design) to ATT LPF A+B ATT BPF S bo ATT ATT LPF 180 A-B 180 ATT BPF D x ti ATT LPF C+D ATT BPF D y 180

### Making Spectrum Measurements with Rohde & Schwarz Network Analyzers

Making Spectrum Measurements with Rohde & Schwarz Network Analyzers Application Note Products: R&S ZVA R&S ZVB R&S ZVT R&S ZNB This application note describes how to configure a Rohde & Schwarz Network

### Vector Network Analyzer Techniques to Measure WR340 Waveguide Windows

LS-296 Vector Network Analyzer Techniques to Measure WR340 Waveguide Windows T. L. Smith ASD / RF Group Advanced Photon Source Argonne National Laboratory June 26, 2002 Table of Contents 1) Introduction

### Understanding Dynamic Range in Acceleration Measurement Systems. February 2013 By: Bruce Lent

in Acceleration Measurement Systems February 2013 By: Bruce Lent Topics to discuss Definition of dynamic range The effective range Making full use of the high level Using filters to improve dynamic range

MAINTENANCE & ADJUSTMENT Circuit Theory The concept of PLL system frequency synthesization is not of recent development, however, it has not been a long age since the digital theory has been couplet with

### 14.5GHZ 2.2KW CW GENERATOR. GKP 22KP 14.5GHz WR62 3x400V

14.5GHZ 2.2KW CW GENERATOR GKP 22KP 14.5GHz WR62 3x400V UTILIZATION OF GKP 22KP GENERATOR With its characteristics of power stability whatever the load, very fast response time at a pulse, low ripple,

### Four quadrant diode front end module for the Virgo Linear Alignment 3/ 30 mw, plus configuration

NI K HEF NATIONAL INSTITUTE FOR NUCLEAR AND HIGH ENERGY PHYSICS Four quadrant diode front end module for the Virgo Linear Alignment 3/ 30 mw, plus configuration Find the most recent files and related files

### Improving Network Analyzer Measurements of Frequency-translating Devices Application Note 1287-7

Improving Network Analyzer Measurements of Frequency-translating Devices Application Note 1287-7 - + - + - + - + Table of Contents Page Introduction......................................................................

### DRM compatible RF Tuner Unit DRT1

FEATURES DRM compatible RF Tuner Unit DRT1 High- Performance RF Tuner Frequency Range: 10 KHz to 30 MHz Input ICP3: +13,5dBm, typ. Noise Figure @ full gain: 14dB, typ. Receiver Factor: -0,5dB, typ. Input

### ETSI EN 300 328-1 V1.2.2 (2000-07)

EN 300 328-1 V1.2.2 (2000-07) European Standard (Telecommunications series) Electromagnetic compatibility and Radio spectrum Matters (ERM); Wideband Transmission systems; data transmission equipment operating

### Understand the effects of clock jitter and phase noise on sampled systems A s higher resolution data converters that can

designfeature By Brad Brannon, Analog Devices Inc MUCH OF YOUR SYSTEM S PERFORMANCE DEPENDS ON JITTER SPECIFICATIONS, SO CAREFUL ASSESSMENT IS CRITICAL. Understand the effects of clock jitter and phase

### Agilent Fundamentals of RF and Microwave Noise Figure Measurements. Application Note 57-1

Agilent Fundamentals of RF and Microwave Noise Figure Measurements Application Note 57-1 Table of Contents 1. What is Noise Figure?...3 Introduction...3 The importance of noise in communication systems...4

### A Simple Model of Relative Power vs. Reach

A Simple Model of Relative Power vs. Reach, Broadcom Corp. Sept. 17, 2012 for IEEE 802.3 Next Generation BASE-T study group Geneva 0 Supporters Brad Booth Valerie Maguire Rick Rabinovitch Dave Chalupsky

### Power meter. Caratterizzazione di amplificatori a RF. Amplificatori a RF. Incertezza 8-03-2012. Corso di Laboratorio di misure ad alta frequenza

Corso di Laboratorio di misure ad alta frequenza Caratterizzazione di amplificatori a RF Docenti: Andrea Mostacci; Alessandra Paffi andrea.mostacci@uniroma1.it; paffi@die.uniroma1.it A.A. 2011-2012 Sensori:

### GSM frequency planning

GSM frequency planning Band : 890-915 and 935-960 MHz Channel spacing: 200 khz (but signal bandwidth = 400 khz) Absolute Radio Frequency Channel Number (ARFCN) lower band: upper band: F l (n) = 890.2 +

### GSM/EDGE Output RF Spectrum on the V93000 Joe Kelly and Max Seminario, Verigy

GSM/EDGE Output RF Spectrum on the V93000 Joe Kelly and Max Seminario, Verigy Introduction A key transmitter measurement for GSM and EDGE is the Output RF Spectrum, or ORFS. The basis of this measurement

### Agilent De-embedding and Embedding S-Parameter Networks Using a Vector Network Analyzer. Application Note 1364-1

Agilent De-embedding and Embedding S-Parameter Networks Using a Vector Network Analyzer Application Note 1364-1 Introduction Traditionally RF and microwave components have been designed in packages with

### AN-837 APPLICATION NOTE

APPLICATION NOTE One Technology Way P.O. Box 916 Norwood, MA 262-916, U.S.A. Tel: 781.329.47 Fax: 781.461.3113 www.analog.com DDS-Based Clock Jitter Performance vs. DAC Reconstruction Filter Performance

### Interpreting the Information Element C/I

Prepared Date Rev Document no pproved File/reference 1(17) 2000-04-11 Interpreting the Information Element C/I This document primarily addresses users of TEMS Investigation. 2(17) 1 Introduction Why is

### DESIGN AND IMPLMENTATION OF INTELLIGENT MOBILE PHONE DETECTOR

DESIGN AND IMPLMENTATION OF INTELLIGENT MOBILE PHONE DETECTOR Christian C. Mbaocha Department of Electrical/Electronic Engineering, Federal University of Technology, NIGERIA. christian_ij@yahoo.com ABSTRACT

### Using Spectrum Analyzers For Signal Monitoring

Using Spectrum Analyzers For Signal Monitoring Presented by: 3 March 004 1 1 At the completion of this module you will have an understanding of how commercially available spectrum analyzers can be used

### DSA800 Series Spectrum Analyzer

DSA800 Series Spectrum Analyzer Configuration Guide This guide is used to help users to configure DSA800 series spectrum analyzer according to their requirements. You can get an overall understanding of

### 22.302 Experiment 5. Strain Gage Measurements

22.302 Experiment 5 Strain Gage Measurements Introduction The design of components for many engineering systems is based on the application of theoretical models. The accuracy of these models can be verified

### Optical Fibres. Introduction. Safety precautions. For your safety. For the safety of the apparatus

Please do not remove this manual from from the lab. It is available at www.cm.ph.bham.ac.uk/y2lab Optics Introduction Optical fibres are widely used for transmitting data at high speeds. In this experiment,