# Agilent AN 1316 Optimizing Spectrum Analyzer Amplitude Accuracy

Size: px
Start display at page:

Transcription

1 Agilent AN 1316 Optimizing Spectrum Analyzer Amplitude Accuracy Application Note RF & Microwave Spectrum Analyzers

2 Table of Contents Introduction Absolute versus relative measurements Factors affecting measurement uncertainty Sources of uncertainty Relative amplitude uncertainty factors Absolute amplitude uncertainty factors Calculating total uncertainty Examples Errors Overall amplitude accuracy Reducing overall uncertainty Summary References

3 Introduction Absolute versus relative measurements This application note covers the factors affecting the accuracy (or uncertainty) of amplitude measurements made with spectrum analyzers, and explains how to calculate worst-case uncertainty in general situations. It will help you analyze a measurement and decide what procedure to follow to minimize overall uncertainty. Also, you will learn the most important accuracy-related features and specifications to consider when choosing a spectrum analyzer. This note does not cover specialized measurement procedures for certain types of signals that introduce additional sources of uncertainty, such as noise, pulsed RF, and EMI. Refer to the References for additional reading. Amplitude measurements are either absolute (Figure 1) or relative (Figure 2). Absolute amplitude is the power or voltage level of a signal. For example, measuring the power level of a carrier in dbm is an absolute measurement. Relative amplitude is the difference between two levels, typically expressed in db. Relative measurements use one signal or signal component as a reference. For example, a harmonic distortion measurement uses the fundamental as the reference, and the harmonic is usually measured in dbc (decibels relative to the carrier level). Most spectrum analyzer applications involve relative amplitude measurements. Figure 1. Absolute measurement (one signal level measured with a single marker) Figure 2. Relative measurement (difference between two signal levels measured with the delta marker) 3

4 Factors affecting measurement uncertainty Let s look first at the factors affecting measurement uncertainty. Later sections will show how to calculate the uncertainty of an amplitude measurement by considering each of these factors, and how to obtain the best accuracy for a measurement. Sources of uncertainty The simplified block diagram in Figure 3 shows the elements of a typical spectrum analyzer. Some of these elements can contribute uncertainty to amplitude measurements. These contributions are called amplitude uncertainty factors in this application note, and are specified for most spectrum analyzers. Table 1 lists the amplitude uncertainty factors of a typical spectrum analyzer. The range of values for each factor covers a variety of spectrum analyzers. Most spectrum analyzers have specifications for both absolute and relative uncertainties. Since relative uncertainties affect the accuracy of both relative and absolute measurements, let s look first at those factors affecting relative measurement uncertainty. Table 1. Amplitude Uncertainty Factors Relative ±db Frequency response (flatness) 0.5 to 4 Band switching 0.5 to 1 Scale fidelity 0.5 to 2 Reference level (IF gain) 0.1 to 1 Input attenuator switching 0.5 to 2 Resolution bandwidth switching 0.1 to 1 Display scale switching 0.0 to 1 Absolute Frequency response 0.5 to 4 Calibrator 0.2 to 1 Preselector or input filter Mixer Log amp Envelope detector Input attenuator IF gain Resolution bandwidth filter Video filter Local oscillator A/D Ramp generator Display Figure 3. Elements of a typical superheterodyne spectrum analyzer 4

5 Relative amplitude uncertainty factors Frequency response (flatness) Frequency response uncertainty, or flatness, represents a window of uncertainty over the specified frequency range (sometimes called a band). See Figure 4. Spectrum analyzer frequency response is often the single highest contributor to uncertainty. It is a function of input attenuator flatness, mixer conversion loss, and preselector flatness (if applicable). Frequency response affects the displayed amplitudes of signals at different frequencies. It is frequency range dependent and specified as ±n db over a frequency range (typically a harmonic band) at a given attenuator setting. Frequency response uncertainty is usually specified for both relative and absolute measurements. Relative frequency response uncertainty describes the largest possible amplitude uncertainty over a given frequency range relative to the midpoint between the amplitude response extremes within that frequency range. The relative frequency response specification for a given frequency range tends to be lower than the absolute frequency response specification (defined below) for the same frequency range. However, to obtain the frequency response uncertainty for relative amplitude measurements within a band, the relative frequency response specification must be doubled to reflect the peak-to-peak frequency response. This peak-to-peak frequency response uncertainty is often greater than the absolute frequency response specification. A low-frequency RF analyzer might have a relative frequency response uncertainty of ±0.5 db. A microwave spectrum analyzer tuning in the 20-GHz range could well have an uncertainty in excess of ±4 db. In our calculations, we assume the worstcase situation, where the frequency response varies over the full amplitude uncertainty range. As shown in Figure 4, the worst-case response varies from plus 1 db to minus 1 db for a specification of ±1 db. We will see practical application of these specifications later. Some spectrum analyzers require you to peak the preselector ( preselector centering in the Agilent ESA series analyzers) for the frequency response specifications to be valid. On spectrum analyzers that require peaking of the signal amplitude, such as the Agilent 8593E, 8595E, and 8596E, using preselector peak prior to measuring a signal yields the most accurate amplitude reading at the specified frequency. The preselector peak function automatically adjusts the preselector tracking to peak the signal at the active marker. While preselector peak maximizes the peak response of the signal of interest, it may degrade the frequency response at other frequencies. So, it is important to set the preselector-peak values to their defaults before measuring a signal at another frequency. Some spectrum analyzers, such as the Agilent 8560 series, do not require preselector peaking. +1 db 0 1 db Band 1 Specification: ± 1 db Uncertainty: ± 2 db Figure 4. Typical in-band frequency response uncertainty 5

6 Band switching Mixing the input signal with harmonics of the local oscillator (LO) allows measurements over a very wide frequency range. Each LO harmonic provides a different harmonic frequency band within the analyzer s overall frequency range. When signals in different harmonic bands are measured, additional uncertainties arise when the analyzer switches from one band to another. On some spectrum analyzers, the band-switching points are visible as discontinuities in the displayed noise floor. To see if your measurement involves more than one band, refer to your analyzer s specifications for the frequency ranges related to each harmonic band. Although not always specified, a typical bandswitching uncertainty is ±1 db. Scale fidelity Scale fidelity uncertainty applies when a signal at one vertical position on the display graticule is measured with respect to another signal at a different vertical position; for example, when delta markers are placed at the peaks of two signals of unequal amplitude. Scale fidelity depends upon the detector s linearity, digitizing circuit linearity, and the log/linear and vertical amplifiers ability to transform different signal voltages into their appropriate relative power (log) or voltage (linear) levels on the display. With most spectrum analyzers, the accuracy of the log amplifier degrades near the bottom of the measurement range. The display itself is not a factor for analyzers using digital techniques and offering digital markers because the marker information is taken from trace memory, not the display. For log mode, scale fidelity is better over small amplitude differences, and ranges from a few tenths of a db for signals close together to 2 db for large amplitude differences. A typical scale fidelity specification is ±0.4 db/4 db to a maximum of ±1.0 db. The ±0.4 db/4 db specification is applied when the two signals are close in amplitude; the cumulative specification applies to larger amplitude differences. The remaining relative factors in Table 1 involve controls that can be changed during a measurement. Reference level (IF gain) The amplitude represented by the top line of the graticule is the reference level. It is a function of the input attenuation and the IF gain. The reference level control determines the IF gain. Uncertainty in the amount of IF gain at a particular reference level setting affects the accuracy of the reference level amplitude. Any changes in the reference level control introduce uncertainty. On spectrum analyzers that can be calibrated in linear voltage, the reference level control determines the amplitude per division scaling from the bottom line of the graticule. For 8560 series portable spectrum analyzers, the specification for this factor is IF gain uncertainty, but for ESA series and 8590 series analyzers it is reference level accuracy. An example of a reference level uncertainty specification is ±0.3 db at 20 dbm, with some incremental uncertainty the further you are from 20 dbm. 6

7 Input attenuator switching Input attenuation has inherent uncertainty that reduces reference level accuracy only if you change the attenuator setting between reference level calibration and the measurement. Because the input attenuator operates over the entire frequency range of the analyzer, its step accuracy is a function of frequency. At low frequencies, expect the attenuator to be quite good; at 20 GHz, not as good. (Since the IF gain operates at only one frequency, it is not surprising that its uncertainty is less than that of the input attenuator.) A typical input attenuator switching uncertainty is ±1 db. Resolution bandwidth switching Different resolution bandwidth (RBW) settings have different insertion loss characteristics. This may cause amplitude changes when the same signal is measured with different settings. Changing the bandwidth setting between amplitude measurements degrades accuracy. A typical RBW switching uncertainty is ±0.4 db. Display scale switching Changing the per division scaling factor for example, from 10 db/division to 1 db/division or to linear introduces uncertainty associated with the relative calibration characteristics of the log/linear amplifiers. Avoid adding this uncertainty by not changing the scale. A typical linear-to-log switching uncertainty is ±0.25 db at the reference level. This may not be a factor for spectrum analyzers that store trace data into memory and then display data from the memory. Impedance mismatch Although not included in Table 1 (since it depends on the signal source as well as the analyzer), impedance mismatch is another factor in measurement uncertainty. The spectrum analyzer tries to measure the signal level across a 50 Ω load (or 75 Ω load if your analyzer has a 75 Ω input), but the spectrum analyzer input impedance is not exactly the characteristic impedance, so uncertainties may arise. In some cases, uncertainty due to mismatch is small. For example, if the analyzer s input VSWR is 1.3 and the source s VSWR is 1.2, the maximum uncertainty due to mismatch is ±0.1 db. On the other hand, if both the analyzer and the source have VSWRs of 2.0, mismatch uncertainty approaches ±1 db. Absolute amplitude uncertainty factors Frequency response Absolute frequency response uncertainty describes the largest possible amplitude uncertainty over a given frequency range relative to the amplitude of an amplitude reference signal (for example, the 50 MHz amplitude reference of Agilent ESA series analyzers or the 300 MHz reference of Agilent 8560 series and 8590 series analyzers). Calibrator Absolute measurements are measurements made relative to a calibration signal, which has a known amplitude. Most spectrum analyzers have a built-in calibrator ( amplitude reference on ESA series analyzers) that provides a signal with specified amplitude at one frequency. A typical calibrator has an uncertainty of ±0.3 db. The calibrator provides absolute calibration for the top line of the graticule. Since the calibrator source typically operates at a single frequency, we rely upon the relative accuracy of the analyzer to translate the absolute calibration to other frequencies and amplitudes. Since our signal to be measured is usually at a different frequency than the calibrator, we may need to change the frequency control. Since it s typically at a different amplitude as well, we may need to change the reference level to bring the signal to the top graticule line for best accuracy (if IF gain uncertainty is less than scale fidelity uncertainty). These changes contribute relative amplitude uncertainty to the measurement. 7

8 Calculating total uncertainty Let s see how to calculate measurement uncertainty for a few general situations. Examples Relative measurement (delta marker) within a band When comparing two signals in the same frequency band, the relevant uncertainties are frequency response (relative) and scale fidelity. Figure 5 shows a 900 MHz signal at the reference level and its second harmonic at 1.8 GHz with an amplitude below the reference level. If the signals are in the same band, the frequency response uncertainty when comparing them is two times the specification (for example, two times ±1 db = 2 db peak-to-peak), since the amplitude uncertainty at each signal s position could fall on the plus or minus extreme of the specification window (refer to Figure 4). Specification values for three different instruments are given in Table 2. Table 2. Accuracy Specifications for In-Band Relative Measurements 8593E 8563E E4407B Specification Specification Specification Factor (±db) (±db) (±db) Frequency 1.0 to 2.0 (relative) 1.0 to 3.3 (relative) 0.5 to 2.0 (relative) response Scale fidelity 1.1 max 0.85 max 1.15 max (over 70 db range) (over 90 db range) (over 85 db range; RBW 1 khz) 1.28 max (over 98 db range, RBW 300 Hz) Fundamental Second harmonic 900 MHz 1.8 GHz Figure 5. Fundamental and second harmonic in the same frequency band 8

9 In general, unless the signals you are measuring are above 20 GHz, frequency response uncertainty will not be significant unless the signals are separated by 50 MHz or more. As an example, let s apply these uncertainties to a TOI (third-order intercept) measurement for two 2 GHz signals separated by 50 khz with third-order distortion at 80 dbc (see Figure 6). The uncertainty calculation when making the measurement with an 8560E is shown in Table 3. Table 3. Uncertainty Calculation for an In-Band TOI Measurement 8560E Factor Specification (±db) Uncertainty (±db) Frequency response 1.0 N/A (input tone) Frequency response 1.0 N/A (distortion product) Scale fidelity 0.85 max 0.85 (over 90 db range) Total uncertainty 0.85 Test tones 50 khz 2 GHz Figure 6. TOI measurement 80 dbc Distortion product Relative measurement (delta marker) across two or more bands If comparing signals in different bands (betweenband measurements), add the frequency response uncertainties of each band (see Figure 7). If the bandswitching uncertainty is specified, add it to the frequency response. If band-switching uncertainty is not specified, use the absolute frequency response (relative to the calibrator) instead of the relative frequency response uncertainty for each band. +2 db +1 db db Band 1 Freq response: ±1 db 2 db Figure 7. Frequency response uncertainty for a between-band measurement Band switching: ±1 db Band 3 Freq response: ±2 db Uncertainty: ±4 db 9

10 The relevant specifications for three different analyzers are given in Table 4. Let s look at an example of measuring a second harmonic with two different instruments. The fundamental is at 10 GHz and the level of the second harmonic is 85 dbc. Using the Agilent 8563E specifications for the calculation: Frequency response at 10 GHz Frequency response at 20 GHz Band-switching uncertainty Scale fidelity Total uncertainty = ±2.2 db = ±2.5 db = ±1.0 db = ±0.85 db = ±6.55 db Let s make the calculation for the 8593E. Use absolute frequency response because this is a between-band measurement and band-switching uncertainty is not specified. Frequency response uncertainty is ±2.5 db at 10 GHz and ±3.0 db at 20 GHz. Notice the scale fidelity specification is for the range 0 to 70 db. Since we need to measure 85 dbc, we must use a different measurement procedure than the one used with the 8563E. After measuring the fundamental, change the reference level to 20 dbm (assume the fundamental was measured with 0 dbm reference level). Though this introduces IF gain uncertainty, we can now make the measurement. Let us calculate total uncertainty: Frequency response at 10 GHz (0 dbm reference level) Frequency response at 20 GHz ( 20 dbm reference level) Scale fidelity ( 65 dbc) Reference level uncertainty (at 0 dbm) = 0.3 db X 20 Total uncertainty = ±2.5 db = ±3.0 db = ±1.05 db = ±0.5 db = ±7.05 db See Table 5 for a summary of these calculations. Note that the 8593E can make the measurement, but the wider display range on the 8563E allows the measurement to be made with less uncertainty, since the reference level doesn t have to change. The Agilent E4407B can make the measurement with the least amount of uncertainty. Table 4. Accuracy Specifications for Between-Band Relative Measurements 8593E 8563E E4407B Factor Specification (±db) Specification (±db) Specification (±db) Band switching No specification 1.0 None Frequency response 1.5 to 5.0 (absolute) 1.0 to 3.3 (relative) to (absolute) Scale fidelity 1.1 max 0.85 max 1.15 max (over 70 db range) (over 90 db range) (over 85 db range; RBW 1 khz) 1.28 max (over 98 db range, RBW 300 Hz) to 30 C 2. 0 to 55 C 10

11 Table 5. Uncertainty Calculations for a Between-Band Relative Measurement 8593E 8563E E4407B Factor Specification Uncertainty Specification Uncertainty Specification Uncertainty Frequency response ±2.5 db ±2.5 db ±2.2 db ±2.2 db ±2.0 db ±2.0 db GHz) (absolute) (relative) (relative) (absolute) (absolute) Frequency response ±3.0 db ±3.0 db ±2.5 db ±2.5 db ±2.0 db 1 ±2.0 db 1 GHz) (absolute) (relative) (relative) (absolute) (absolute) Band switching No ±1.0 db ±1.0 db No specification specification Scale fidelity ±1.1 db max ±1.05 db ±0.85 db max ±0.85 db ±1.15 db max ±1.15 db (over 70 db range) ( 65 dbc) (over 90 db range) max (over 85 db range) ( 85 dbc) ±1.28 db (over 98 db range) Reference level ±0.5 db N/A N/A Total uncertainty ±7.05 db ±6.55 db ±5.15 db to 30 C (±2.5 db for 0 to 55 C) As another example of a between-band measurement, suppose you need to make a spur measurement from 22 GHz to 26.5 GHz with a carrier at 10 GHz. The spurs must be 70 dbc or lower (see Figure 8). You want to know how accurately you can measure the spurs from 22 GHz to 26.5 GHz relative to the 10 GHz carrier. The factors to consider are the frequency response (relative) for the two bands, band-switching uncertainty, and scale fidelity. Using the 8563E specifications in Table 6, the frequency response (relative) for the 6.46 GHz to 13.2 GHz band is ±2.2 db, the frequency response for the 22 to 26.5 GHz band is ±3.3 db, band-switching uncertainty is ±1 db, and scale fidelity (log) is ±0.85 db, for a total uncertainty of ±7.35 db. Table 6. Uncertainty Calculation for a Between-Band Spur Measurement 8563E Factor Specification (±db) Uncertainty (±db) Frequency response (6.46 to 13.2 GHz) Frequency response (22 to 26.5 GHz) Band switching Scale fidelity 0.85 max (over 90 db range) 0.85 Total uncertainty 7.35 Carrier 70 dbc Spur 10 GHz 22 to 26.5 GHz Figure 8. Spur measurement 11

12 Absolute measurement (single marker) If only frequency and amplitude controls are changed, absolute accuracy depends on these three factors: calibrator accuracy, absolute frequency response from the signal to the calibrator, and IF gain uncertainty ( reference level accuracy for Agilent ESA series and 8590 series analyzers) due to changes from the calibrator level. For a signal measured at the reference level of the analyzer, the absolute amplitude uncertainty equals the calibrator uncertainty plus the absolute frequency response uncertainty. For a signal at a level other than the reference level, change the reference level control to bring the signal to the reference level for best accuracy (if IF gain uncertainty is less than scale fidelity uncertainty). In this case, absolute amplitude uncertainty is the sum of the reference level, absolute frequency response, and calibrator uncertainties. Suppose you need to measure a 2 GHz signal at 10 dbm. Using the Agilent 8560E specifications in Table 7 as an example, the calibrator uncertainty is ±0.3 db. The IF gain uncertainty of ±1.0 db can be left out since the calibrator level is 10 dbm in the 8560E. The absolute frequency response uncertainty is ±1.5 db, for a total uncertainty of ±1.8 db. When choosing a spectrum analyzer, you need to determine if a particular instrument can make your measurement. Suppose you need to make a spur measurement at 10 GHz, where the spur is offset from the carrier by 10 khz; the spur must be at 70 dbc or lower and the carrier level must be measured to ±3 db. In measuring the absolute level of the carrier, the factors to consider are calibrator accuracy, IF gain uncertainty, and absolute frequency response uncertainty at 10 GHz. For the 8563E, the specifications are ±0.3, ±1.0 db, and ±2.9 db, respectively, for a total amplitude uncertainty of ±4.2 db. This is greater than the ±3 db requirement. However, Ampcor (see page 15) can reduce the frequency response uncertainty of this analyzer to an acceptable level. Since the largest uncertainty contributor is typically frequency response, steps taken to reduce uncertainty should first focus on reducing frequency response uncertainty, as described below. Table 7. Uncertainty Calculation for an Absolute Amplitude Measurement 8560E Factor Specification (±db) Uncertainty (±db) Calibrator Reference level 1.0 N/A at reference level Frequency response 1.5 (absolute) 1.5 Total uncertainty

13 Errors Up to now we have discussed uncertainties. Uncertainties differ from errors in that we can correct for errors in our measurement. The following is a source of error that you should be aware of when measuring amplitude. Signal plus noise The amplitude at any point on the spectrum analyzer display is the sum of all energy present in the IF passband. Therefore, the displayed amplitude of a signal is actually the signal level plus the noise level. That is, the displayed signal is higher than its true value because of the added noise. Measurements of signals close to the displayed average noise level (DANL) are subject to an amplitude error due to the addition of noise. This error may be large or insignificant depending on the signal-to-noise ratio, and can be corrected. See Reference 3 for a discussion of measuring signals near noise. For signals that appear 10 db or more above the noise, the error is negligible. A similar error can occur when measuring closely spaced signals of unequal amplitude. This happens when both signals pass through the analyzer s IF and are detected together. The displayed amplitude of the smaller signal may be incorrect. This error is less than ±0.5 db and occurs if the smaller signal is less than 10 db above the larger signal s skirt. Overall amplitude accuracy Some modern spectrum analyzers specify overall absolute amplitude accuracy. This specification provides a single number that you add to the absolute frequency response uncertainty to easily determine amplitude accuracy when you are making a measurement within certain boundary conditions. For example, the specification for overall absolute amplitude accuracy of the Agilent ESA-E series analyzers is (±0.54 db + absolute frequency response) 1. Table 8 shows the amplitude uncertainty of this analyzer family using the overall amplitude accuracy specification. Table 8. Uncertainty Calculations for Absolute Amplitude Measurements Using Overall Absolute Amplitude Accuracy Specification ESA-E series Absolute frequency Uncertainty 1 Frequency range response specification 2 (±db) (±db) 9 khz to 3.0 GHz GHz to 6.7 GHz GHz to 13.2 GHz GHz to 25 GHz GHz to 26.5 GHz For reference levels 0 to 50 dbm; RBW = 1 khz; video BW = 1 khz; log scale; 0 to 50 db log range; coupled sweep time; sample detector; 0 to 50 dbm input signal; span 20 khz; 20 to 30 C to 30 C 13

14 Reducing overall uncertainty Minimize changes Before taking any data, step through a measurement to see if any controls can be left unchanged. If a single input attenuator setting, resolution bandwidth, and display scaling can be used for the measurement, then the uncertainties associated with changing these controls can be ignored. For best accuracy, calibrate the analyzer with the controls in the same state in which the measurements will be made. Substitute IF gain for scale fidelity If the reference level (IF gain) uncertainty is less than the scale fidelity, you can minimize the uncertainty of a relative measurement by using the reference level control to measure both signals at the same vertical position, eliminating scale fidelity. Place the first signal at a reference amplitude using the reference level control, and use the marker to read the amplitude value. Then adjust the reference level to bring the second signal to the same reference amplitude (measured with the marker), and calculate the difference between the two reference level settings to obtain the relative amplitude. Self calibration Many of today s spectrum analyzers have selfcalibration routines (such as the auto-alignment function in the ESA series analyzers). These routines generate error coefficients (for example, amplitude changes versus resolution bandwidth) that the analyzer uses to correct measured data. These routines help to ensure accurate amplitude measurements and give more freedom to change controls during a measurement. However, the uncertainty specifications assume self calibration has been done; therefore, self calibration is required to achieve specified accuracy. Characterization Characterizing your analyzer can reduce frequency response and calibrator uncertainties. If you have a more accurate calibration signal, or one closer to the frequency of interest, you can use that instead of the built-in calibrator. To reduce calibrator uncertainty, use a signal source and a precision power meter to calibrate the reference level (top line of the graticule). To reduce frequency response uncertainty, use a signal source having the same frequency as the signal-under-test to calibrate the reference level. Use data from a point-to-point calibration of the spectrum analyzer s frequency response for measurement correction over a frequency range. Ampcor Some spectrum analyzers have a built-in amplitude correction routine (such as Ampcor) that stores point-to-point frequency response characterization data in a table and automatically applies it to the measured data to yield corrected display information. Using a spectrum analyzer with Ampcor, a signal source and a power meter, you can also correct the input signal for unwanted variations due to the cables and adapters between the device-under-test and the analyzer (see Reference 2). Ampcor uses the table of frequency-amplitude pairs that you enter to make a correction curve to offset the variations in the input signal. The resulting signal trace represents the actual signal amplitude at the output of the device-under-test (see Figure 9). 14

15 Figure 9a. Original signal Figure 9b. Signal corrected by Ampcor Improving sensitivity As mentioned earlier, displayed signal amplitude is the sum of the signal and noise present in the passband. For signals close to the analyzer s noise floor (within 10 db), the error between the displayed amplitude and the true amplitude is significant (Reference 3 gives the equation for correcting this error). Therefore, when measuring low-level signals, the noise level of the analyzer should be as low as possible. Since the noise level depends on bandwidth, make the measurement with as narrow a resolution bandwidth as possible. Of course, you may not always be able to use a narrow resolution bandwidth; you will have to weigh this against the longer sweep time. However, digital resolution bandwidth filters allow you to use very narrow resolution bandwidths with shorter sweep times than analog filters. When measuring signals close to the noise level, use video filtering for a more stable, and therefore more repeatable measurement. When narrower video bandwidths are not available, use video averaging to digitally smooth the trace over multiple sweeps. Any equipment connected between the deviceunder-test and the analyzer input (such as an external attenuator) may reduce the available sensitivity depending on its insertion loss and mismatch. Furthermore, the frequency response characteristics of the insertion loss/gain and mismatch will introduce additional flatness uncertainty into the measurement. Using Ampcor, you can correct for this uncertainty. Preamplification Because of limited IF gain, small signals cannot be raised to the reference level of a log display. Scale fidelity uncertainty will therefore be a factor when measuring these signals. Some spectrum analyzers have a built-in preamplifier that increases the signal level so that smaller signals can be measured at the top graticule line. However, the preamplifier s gain flatness and mismatch may reduce the overall flatness. Also, since the preamplifier s gain must be subtracted from the measured signal level to determine true signal level, preamplifier gain uncertainty further affects total measurement uncertainty. These added uncertainties can be greater than the scale fidelity uncertainty. Reduce mismatch Reduce mismatch uncertainty by attaching a wellmatched pad (attenuator) to the analyzer input. Since an analyzer s match is worst with its input attenuator set to 0 db, avoid the 0 db setting whenever possible. 15

16 Summary Amplitude measurements with spectrum analyzers are subject to both relative and absolute uncertainties. The overall uncertainty of a measurement can be calculated by referring to the analyzer s accuracy or uncertainty specifications and adding those factors that apply to that measurement. Total uncertainty can be minimized by careful attention to measurement technique. In general, this means limiting the number of changes to analyzer control settings and substituting smaller uncertainties for larger ones. Uncertainty can be further reduced by characterization and amplitude correction. References 1. Agilent Application Note 150: Spectrum Analysis Basics, literature number Agilent Application Note : 8 Hints for Making Better Spectrum Analyzer Measurements, literature number E. 3. Agilent Application Note 1303: Spectrum Analyzer Measurements and Noise, literature number E. 4. Amplitude Accuracy Enhancement Techniques for the Agilent 8566A/B and 8568A/B Spectrum Analyzers (Agilent white paper). By internet, phone, or fax, get assistance with all your test and measurement needs. Online Assistance Phone or Fax United States: (tel) Canada: (tel) (fax) (905) Europe: (tel) (31 20) (fax) (31 20) Japan: (tel) (81) (fax) (81) Latin America: (tel) (305) (fax) (305) Australia: (tel) (fax) (61 3) New Zealand: (tel) (fax) (64 4) Asia Pacific: (tel) (852) (fax) (852) Product specifications and descriptions in this document subject to change without notice. Copyright 1999, 2000 Agilent Technologies Printed in U.S.A. 8/ E 5. Spectrum Analyzer Basics (part of Agilent Back-to-Basics Seminar), literature number E. 6. Spectrum Analyzer Frequency and Amplitude Accuracy (Agilent SR112 Fundamentals Product Training). 7. Agilent ESA-E Series Spectrum Analyzers Data Sheet, literature number E. 8. Agilent 8560 E-Series Spectrum Analyzers Data Sheet, literature number E. 9. Agilent 8590 E-Series Portable Spectrum Analyzers Data Sheet, literature number E.

### Agilent AN 1315 Optimizing RF and Microwave Spectrum Analyzer Dynamic Range. Application Note

Agilent AN 1315 Optimizing RF and Microwave Spectrum Analyzer Dynamic Range Application Note Table of Contents 3 3 3 4 4 4 5 6 7 7 7 7 9 10 10 11 11 12 12 13 13 14 15 1. Introduction What is dynamic range?

### Jeff Thomas Tom Holmes Terri Hightower. Learn RF Spectrum Analysis Basics

Jeff Thomas Tom Holmes Terri Hightower Learn RF Spectrum Analysis Basics Learning Objectives Name the major measurement strengths of a swept-tuned spectrum analyzer Explain the importance of frequency

### Jeff Thomas Tom Holmes Terri Hightower. Learn RF Spectrum Analysis Basics

Jeff Thomas Tom Holmes Terri Hightower Learn RF Spectrum Analysis Basics Agenda Overview: Spectrum analysis and its measurements Theory of Operation: Spectrum analyzer hardware Frequency Specifications

### F = S i /N i S o /N o

Noise figure Many receiver manufacturers specify the performance of their receivers in terms of noise figure, rather than sensitivity. As we shall see, the two can be equated. A spectrum analyzer is a

### A Guide to Calibrating Your Spectrum Analyzer

A Guide to Calibrating Your Application Note Introduction As a technician or engineer who works with electronics, you rely on your spectrum analyzer to verify that the devices you design, manufacture,

### Keysight Technologies 8 Hints for Better Spectrum Analysis. Application Note

Keysight Technologies 8 Hints for Better Spectrum Analysis Application Note The Spectrum Analyzer The spectrum analyzer, like an oscilloscope, is a basic tool used for observing signals. Where the oscilloscope

### Agilent E4401B, E4402B, E4404B, E4405B, and E4407B ESA-E Series Spectrum Analyzers

Agilent, E4402B, E4404B, E4405B, and E4407B ESA-E Series Spectrum Analyzers Technical Specifications All specifications apply over 0 C to + 55 C unless otherwise noted. The analyzer will meet its specifications

### Conquering Noise for Accurate RF and Microwave Signal Measurements. Presented by: Ernie Jackson

Conquering Noise for Accurate RF and Microwave Signal Measurements Presented by: Ernie Jackson The Noise Presentation Review of Basics, Some Advanced & Newer Approaches Noise in Signal Measurements-Summary

### SIGNAL GENERATORS and OSCILLOSCOPE CALIBRATION

1 SIGNAL GENERATORS and OSCILLOSCOPE CALIBRATION By Lannes S. Purnell FLUKE CORPORATION 2 This paper shows how standard signal generators can be used as leveled sine wave sources for calibrating oscilloscopes.

### Spectrum Analyzer Basics www. agilent.com/find/backtobasics

www. agilent.com/find/backtobasics Abstract Learn why spectrum analysis is important for a variety of applications and how to measure system and device performance using a spectrum analyzer. To introduce

### Technical Datasheet Scalar Network Analyzer Model 8003-10 MHz to 40 GHz

Technical Datasheet Scalar Network Analyzer Model 8003-10 MHz to 40 GHz The Giga-tronics Model 8003 Precision Scalar Network Analyzer combines a 90 db wide dynamic range with the accuracy and linearity

### Agilent N2717A Service Software Performance Verification and Adjustment Software for the Agilent ESA Spectrum Analyzers Product Overview

Agilent N2717A Service Software Performance Verification and Adjustment Software for the Agilent ESA Spectrum Analyzers Product Overview Reduce your cost of ownership by minimizing time to calibrate and

### Agilent PN 8753-1 RF Component Measurements: Amplifier Measurements Using the Agilent 8753 Network Analyzer. Product Note

Agilent PN 8753-1 RF Component Measurements: Amplifier Measurements Using the Agilent 8753 Network Analyzer Product Note 2 3 4 4 4 4 6 7 8 8 10 10 11 12 12 12 13 15 15 Introduction Table of contents Introduction

### Agilent Spectrum Analysis Basics Application Note 150

Agilent Spectrum Analysis Basics Application Note 150 Contents Chapter 1 Introduction...3 What is a spectrum?...3 Why measure spectra?...4 Chapter 2 The superheterodyne spectrum analyzer...6 Tuning equation...8

### 0HDVXULQJWKHHOHFWULFDOSHUIRUPDQFH FKDUDFWHULVWLFVRI5),)DQGPLFURZDYHVLJQDO SURFHVVLQJFRPSRQHQWV

0HDVXULQJWKHHOHFWULFDOSHUIRUPDQFH FKDUDFWHULVWLFVRI5),)DQGPLFURZDYHVLJQDO SURFHVVLQJFRPSRQHQWV The treatment given here is introductory, and will assist the reader who wishes to consult the standard texts

### Agilent N8973A, N8974A, N8975A NFA Series Noise Figure Analyzers. Data Sheet

Agilent N8973A, N8974A, N8975A NFA Series Noise Figure Analyzers Data Sheet Specifications Specifications are only valid for the stated operating frequency, and apply over 0 C to +55 C unless otherwise

### Impedance 50 (75 connectors via adapters)

VECTOR NETWORK ANALYZER PLANAR TR1300/1 DATA SHEET Frequency range: 300 khz to 1.3 GHz Measured parameters: S11, S21 Dynamic range of transmission measurement magnitude: 130 db Measurement time per point:

Introduction to Receivers Purpose: translate RF signals to baseband Shift frequency Amplify Filter Demodulate Why is this a challenge? Interference (selectivity, images and distortion) Large dynamic range

### Application Note Noise Frequently Asked Questions

: What is? is a random signal inherent in all physical components. It directly limits the detection and processing of all information. The common form of noise is white Gaussian due to the many random

### Agilent Spectrum Analysis Basics. Application Note 150

Agilent Spectrum Analysis Basics Application Note 150 Table of Contents Chapter 1 Introduction.......................................................4 Frequency domain versus time domain.......................................4

### The front end of the receiver performs the frequency translation, channel selection and amplification of the signal.

Many receivers must be capable of handling a very wide range of signal powers at the input while still producing the correct output. This must be done in the presence of noise and interference which occasionally

### Spectrum Analysis Basics. Application Note 150

Spectrum Analysis Basics Application Note 150 Agilent Technologies dedicates this application note to Blake Peterson. Blake s outstanding service in technical support reached customers in all corners of

### MATRIX TECHNICAL NOTES

200 WOOD AVENUE, MIDDLESEX, NJ 08846 PHONE (732) 469-9510 FAX (732) 469-0418 MATRIX TECHNICAL NOTES MTN-107 TEST SETUP FOR THE MEASUREMENT OF X-MOD, CTB, AND CSO USING A MEAN SQUARE CIRCUIT AS A DETECTOR

### HP 8590 E-Series Portable Spectrum Analyzers Technical Specifications

HP 8590 E-Series Portable Spectrum Analyzers Technical Specifications Product Specifications and data These specifications apply to the, 8593E, 8594E, 8595E, and 8596E spectrum analyzers. Specifications

### Scalar Network Analysis with the HP 8590 Series Spectrum Analyzers Product Overview

This literature was published years prior to the establishment of Agilent Technologies as a company independent from Hewlett-Packard and describes products or services now available through Agilent. It

### Improving Network Analyzer Measurements of Frequency-translating Devices Application Note 1287-7

Improving Network Analyzer Measurements of Frequency-translating Devices Application Note 1287-7 - + - + - + - + Table of Contents Page Introduction......................................................................

### RF Network Analyzer Basics

RF Network Analyzer Basics A tutorial, information and overview about the basics of the RF Network Analyzer. What is a Network Analyzer and how to use them, to include the Scalar Network Analyzer (SNA),

### EE 186 LAB 2 FALL 2004. Network Analyzer Fundamentals and Two Tone Linearity

Network Analyzer Fundamentals and Two Tone Linearity Name: Name: Name: Objective: To become familiar with the basic operation of a network analyzer To use the network analyzer to characterize the in-band

### MEASUREMENT UNCERTAINTY IN VECTOR NETWORK ANALYZER

MEASUREMENT UNCERTAINTY IN VECTOR NETWORK ANALYZER W. Li, J. Vandewege Department of Information Technology (INTEC) University of Gent, St.Pietersnieuwstaat 41, B-9000, Gent, Belgium Abstract: Precision

### Measurement Guide and Programming Examples

Measurement Guide and Programming Examples PSA and ESA Series Spectrum Analyzers This manual provides documentation for the following instruments: Agilent Technologies PSA Series E4443A (3 Hz - 6.7 GHz)

### The Effect of Network Cabling on Bit Error Rate Performance. By Paul Kish NORDX/CDT

The Effect of Network Cabling on Bit Error Rate Performance By Paul Kish NORDX/CDT Table of Contents Introduction... 2 Probability of Causing Errors... 3 Noise Sources Contributing to Errors... 4 Bit Error

### R&S FSW signal and spectrum analyzer: best in class now up to 50 GHz

R&S FSW signal and spectrum analyzer: best in class now up to 50 GHz The new R&S FSW 43 and R&S FSW 50 signal and spectrum analyzers make the outstanding features of the R&S FSW family available now also

### GSM/EDGE Output RF Spectrum on the V93000 Joe Kelly and Max Seminario, Verigy

GSM/EDGE Output RF Spectrum on the V93000 Joe Kelly and Max Seminario, Verigy Introduction A key transmitter measurement for GSM and EDGE is the Output RF Spectrum, or ORFS. The basis of this measurement

### How To Measure Two Tone, Third Order Intermodulation Distortion

Improve Two-Tone, Third-Order Intermodulation Testing First, it's important to define the significance of input levels. Then, details on the measurement technique will be given. Two-tone, third-order intermodulation

### Optimizing IP3 and ACPR Measurements

Optimizing IP3 and ACPR Measurements Table of Contents 1. Overview... 2 2. Theory of Intermodulation Distortion... 2 3. Optimizing IP3 Measurements... 4 4. Theory of Adjacent Channel Power Ratio... 9 5.

### Network Analyzer Operation

Network Analyzer Operation 2004 ITTC Summer Lecture Series John Paden Purposes of a Network Analyzer Network analyzers are not about computer networks! Purposes of a Network Analyzer Measures S-parameters

### Measurement of Adjacent Channel Leakage Power on 3GPP W-CDMA Signals with the FSP

Products: Spectrum Analyzer FSP Measurement of Adjacent Channel Leakage Power on 3GPP W-CDMA Signals with the FSP This application note explains the concept of Adjacent Channel Leakage Ratio (ACLR) measurement

### 2398 9 khz to 2.7 GHz Spectrum Analyzer

Spectrum Analyzers 2398 9 khz to 2.7 GHz Spectrum Analyzer A breakthrough in high performance spectrum analysis, combining cost effectiveness and portability in a new lightweight instrument 9 khz to 2.7

### Power Amplifier Gain Compression Measurements

Technical Brief Power Amplifier Gain Compression Measurements GPIB Private Bus Sweep Out Sweep In Pulse In AC Mod Out Blank/Marker Out Blanking In Overview The 1 db gain compression of an amplifier describes

### HP 8970B Option 020. Service Manual Supplement

HP 8970B Option 020 Service Manual Supplement Service Manual Supplement HP 8970B Option 020 HP Part no. 08970-90115 Edition 1 May 1998 UNIX is a registered trademark of AT&T in the USA and other countries.

### Keysight N9320B RF Spectrum Analyzer

Keysight N9320B RF Spectrum Analyzer 9 khz to 3.0 GHz Data Sheet 02 Keysight N9320B RF Spectrum Analyzer - Data Sheet Definitions and Conditions Specifications describe the performance of parameters covered

### Performing Amplifier Measurements with the Vector Network Analyzer ZVB

Product: Vector Network Analyzer R&S ZVB Performing Amplifier Measurements with the Vector Network Analyzer ZVB Application Note This document describes typical measurements that are required to be made

### Understanding Mixers Terms Defined, and Measuring Performance

Understanding Mixers Terms Defined, and Measuring Performance Mixer Terms Defined Statistical Processing Applied to Mixers Today's stringent demands for precise electronic systems place a heavy burden

### HP 70950B OPTICAL SPECTRUM ANALYZER

HP 71450B, 71451B, and 71452B Optical Spectrum Analyzers Technical Specifications Spectral Measurements from 600 to 1700 nm HP 70950B OPTICAL SPECTRUM ANALYZER OPTICAL INPUT The HP 71450B, 71451B, and

### Keysight Technologies N9320B RF Spectrum Analyzer

Keysight Technologies N9320B RF Spectrum Analyzer 9 khz to 3.0 GHz Data Sheet Definitions and Conditions The spectrum analyzer will meet its specifications when: It is within its calibration cycle It has

### TESTS OF 1 MHZ SIGNAL SOURCE FOR SPECTRUM ANALYZER CALIBRATION 7/8/08 Sam Wetterlin

TESTS OF 1 MHZ SIGNAL SOURCE FOR SPECTRUM ANALYZER CALIBRATION 7/8/08 Sam Wetterlin (Updated 7/19/08 to delete sine wave output) I constructed the 1 MHz square wave generator shown in the Appendix. This

### Engineering Sciences 151. Electromagnetic Communication Laboratory Assignment 3 Fall Term 1998-99

Engineering Sciences 151 Electromagnetic Communication Laboratory Assignment 3 Fall Term 1998-99 WAVE PROPAGATION II: HIGH FREQUENCY SLOTTED LINE AND REFLECTOMETER MEASUREMENTS OBJECTIVES: To build greater

### Maximizing Receiver Dynamic Range for Spectrum Monitoring

Home Maximizing Receiver Dynamic Range for Spectrum Monitoring Brian Avenell, National Instruments Corp., Austin, TX October 15, 2012 As consumers continue to demand more data wirelessly through mobile

### RF Communication System. EE 172 Systems Group Presentation

RF Communication System EE 172 Systems Group Presentation RF System Outline Transmitter Components Receiver Components Noise Figure Link Budget Test Equipment System Success Design Remedy Transmitter Components

### Spectrum and Power Measurements Using the E6474A Wireless Network Optimization Platform

Application Note Spectrum and Power Measurements Using the E6474A Wireless Network Optimization Platform By: Richard Komar Introduction With the rapid development of wireless technologies, it has become

### Application Note: Spread Spectrum Oscillators Reduce EMI for High Speed Digital Systems

Application Note: Spread Spectrum Oscillators Reduce EMI for High Speed Digital Systems Introduction to Electro-magnetic Interference Design engineers seek to minimize harmful interference between components,

### Visual System Simulator White Paper

Visual System Simulator White Paper UNDERSTANDING AND CORRECTLY PREDICTING CRITICAL METRICS FOR WIRELESS RF LINKS Understanding and correctly predicting cellular, radar, or satellite RF link performance

### AN-837 APPLICATION NOTE

APPLICATION NOTE One Technology Way P.O. Box 916 Norwood, MA 262-916, U.S.A. Tel: 781.329.47 Fax: 781.461.3113 www.analog.com DDS-Based Clock Jitter Performance vs. DAC Reconstruction Filter Performance

### Agilent Creating Multi-tone Signals With the N7509A Waveform Generation Toolbox. Application Note

Agilent Creating Multi-tone Signals With the N7509A Waveform Generation Toolbox Application Note Introduction Of all the signal engines in the N7509A, the most complex is the multi-tone engine. This application

### Agilent E3830 Series Wide-bandwidth Signal Analyzer Custom Systems 100 MHz Bandwidth Microwave Vector Signal Analysis

Agilent E3830 Series Wide-bandwidth Signal Analyzer Custom Systems 100 MHz Bandwidth Microwave Vector Signal Analysis The measurement challenge 2 Many advanced microwave applications involve complex wideband

### Understanding Power Splitters

Understanding Power Splitters how they work, what parameters are critical, and how to select the best value for your application. Basically, a 0 splitter is a passive device which accepts an input signal

### ANALYZER BASICS WHAT IS AN FFT SPECTRUM ANALYZER? 2-1

WHAT IS AN FFT SPECTRUM ANALYZER? ANALYZER BASICS The SR760 FFT Spectrum Analyzer takes a time varying input signal, like you would see on an oscilloscope trace, and computes its frequency spectrum. Fourier's

### X-Series Signal Analysis. Future-ready instruments Consistent measurement framework Broadest set of applications and software

X-Series Signal Analysis Future-ready instruments Consistent measurement framework Broadest set of applications and software Arrive Ahead with X-Series We can t predict the future, but Agilent can help

### Tx/Rx A high-performance FM receiver for audio and digital applicatons

Tx/Rx A high-performance FM receiver for audio and digital applicatons This receiver design offers high sensitivity and low distortion for today s demanding high-signal environments. By Wayne C. Ryder

### Dithering in Analog-to-digital Conversion

Application Note 1. Introduction 2. What is Dither High-speed ADCs today offer higher dynamic performances and every effort is made to push these state-of-the art performances through design improvements

### Optimizing VCO PLL Evaluations & PLL Synthesizer Designs

Optimizing VCO PLL Evaluations & PLL Synthesizer Designs Today s mobile communications systems demand higher communication quality, higher data rates, higher operation, and more channels per unit bandwidth.

### Agilent N9342C Handheld Spectrum Analyzer (HSA)

Agilent N9342C Handheld Spectrum Analyzer (HSA) 7 GHz Data Sheet Field testing just got easier www.agilent.com/find/hsa If you are making measurements in the field, the Agilent N9342C handheld spectrum

### Cable Analysis and Fault Detection using the Bode 100

Cable Analysis and Fault Detection using the Bode 100 By Stephan Synkule 2014 by OMICRON Lab V1.3 Visit www.omicron-lab.com for more information. Contact support@omicron-lab.com for technical support.

### MXE Instrument Software Revision History (Windows 7 Only)

MXE Instrument Software Revision History (Windows 7 Only) It is recommended that all instruments be kept up to date by installing the most recent version of software available. The most recent version

### Microwave. 6200B series RF and Microwave Test sets. www.aeroflex.com

Microwave 6200B series RF and Microwave Test sets With compliments Helmut Singer Elektronik www.helmut-singer.de info@helmut-singer.de fon +49 241 155 315 fax +49 241 152 066 Feldchen 16-24 D-52070 Aachen

### Field Calibration Software

SIGNAL HOUND Field Calibration Software User s Manual Version 1.1.0 7/8/2016 This information is being released into the public domain in accordance with the Export Administration Regulations 15 CFR 734

### Agilent E5100A Network Analyzer

Agilent E5100A Network Analyzer Data Sheet These specifications are the performance standards or limits against which the instrument is tested. When shipped from the factory, the E5100A meets the specifications

### Basics of RF Amplifier Measurements with the E5072A ENA Series Network Analyzer

Basics of RF Amplifier Measurements with the E5072A ENA Series Network Analyzer Application Note Introduction The RF power amplifier is a key component used in a wide variety of industries such as wireless

### A Network Analyzer For Active Components

A Network Analyzer For Active Components EEEfCom 29-30 Juni ULM Marc Vanden Bossche, NMDG Engineering Remi Tuijtelaars, BSW Copyright 2005 NMDG Engineering Version 2 Outline Review of S-parameters Theory

### How To Use A Sound Card With A Subsonic Sound Card

!"## \$#!%!"# &"#' ( "#' )*! #+ #,# "##!\$ -+./0 1" 1! 2"# # -&1!"#" (2345-&1 #\$6.7 -&89\$## ' 6! #* #!"#" +" 1##6\$ "#+# #-& :1# # \$ #\$#;1)+#1#+

### Handheld Spectrum Analyzer

9101 Handheld Spectrum Analyzer The 9101 Handheld Spectrum Analyzer provides RF engineers with the excellent performance of a workbench analyzer in a handheld form, at a competitive price. Highlights Covering

### Understanding Power Splitters

Understanding Power Splitters How they work, what parameters are critical, and how to select the best value for your application. Basically, a 0 splitter is a passive device which accepts an input signal

### AM/FM/ϕM Measurement Demodulator FS-K7

Data sheet Version 02.00 AM/FM/ϕM Measurement Demodulator FS-K7 July 2005 for the Analyzers FSQ/FSU/FSP and the Test Receivers ESCI/ESPI AM/FM/ϕM demodulator for measuring analog modulation parameters

### DSA800 Series Spectrum Analyzer

DSA800 Series Spectrum Analyzer Configuration Guide This guide is used to help users to configure DSA800 series spectrum analyzer according to their requirements. You can get an overall understanding of

### Agilent 8904A Multifunction Synthesizer dc to 600 khz

Agilent 8904A Multifunction Synthesizer dc to 600 khz Technical Specifications Build complex waveforms from common signals The Agilent Technologies 8904A Multifunction Synthesizer uses VLSIC technology

### Lock - in Amplifier and Applications

Lock - in Amplifier and Applications What is a Lock in Amplifier? In a nut shell, what a lock-in amplifier does is measure the amplitude V o of a sinusoidal voltage, V in (t) = V o cos(ω o t) where ω o

### Keysight N9000A CXA Signal Analyzer

Keysight N9000A CXA Signal Analyzer Notice: This document contains references to Agilent. Please note that Agilent s Test and Measurement business has become Keysight Technologies. For more information,

### 1. The Slotted Line. ECE 584 Microwave Engineering Laboratory Experiments. Introduction:

ECE 584 Microwave Engineering Laboratory Experiments 1. The Slotted Line Introduction: In this experiment we will use a waveguide slotted line to study the basic behavior of standing waves and to measure

### INTELLIGENT INTERACTIVE SYNTHESIZER SURFACE MOUNT MODEL: MFSH615712-100

SURFACE MOUNT MODEL: MFSH6572 65-72 MHz FEATURES: Small Size, Surface Mount (.6" x.6") Low Phase Noise Standard Programming Interface Ultra Wide Tuning Range Lead Free - RoHS Compliant Patented REL-PRO

### Electronic Communications Committee (ECC) within the European Conference of Postal and Telecommunications Administrations (CEPT)

Page 1 Electronic Communications Committee (ECC) within the European Conference of Postal and Telecommunications Administrations (CEPT) ECC RECOMMENDATION (06)01 Bandwidth measurements using FFT techniques

### Agilent U2000 Series USB Power Sensors

Agilent U2000 Series USB Power Sensors GSM Timeslot Burst Power Measurement Product Note Table of Content Introduction 2 Measuring GSM Timeslot 4 Signal Overview of Agilent U2000 5 Series USB Power Sensors

### Agilent E6832A W-CDMA Calibration Application

Agilent E6832A W-CDMA Calibration Application For the E6601A Wireless Communications Test Set Data Sheet The next generation of mobile phone manufacturing test. E6601A is the newest test set from Agilent

### Agilent 8753ET/8753ES Network Analyzers

Agilent 8753ET/8753ES Network Analyzers 8753ET, 300 khz to 3 or 6 GHz 8753ES, 30 khz to 3 or 6 GHz Configuration Guide System configuration summary The following summary lists the main components required

### Propagation Channel Emulator ECP_V3

Navigation simulators Propagation Channel Emulator ECP_V3 1 Product Description The ECP (Propagation Channel Emulator V3) synthesizes the principal phenomena of propagation occurring on RF signal links

### Handheld Spectrum Analyzers, 3.3 GHz & 8.5 GHz Models 2650A, 2652A, 2658A

Data sheet Handheld Spectrum Analyzers, 3.3 GHz & 8.5 GHz Models 2650A, 2652A, 2658A Truely portable solution for quick and precise spectrum analysis B&K Precision s 2650A series handheld spectrum analyzers

### RF Level Measurement Uncertainties with the Measuring Receiver R&S FSMR

Products: R&S FSMR, R&S NRP-Z27/37, R&S NRP-Z51/55 RF Level Measurement Uncertainties with the Measuring Receiver R&S FSMR Application Note Measurement uncertainties that occur when the R&S FSMR measuring

### Automatic compression measurement using network analyzers

Automatic compression measurement using network analyzers Introduction The dynamic range of an amplifier is determined by noise figure and compression. In multi carrier applications third order intercept

Introduction to 3GHz Spectrum Analyzer of Instek Front side Features Rear side 1Frequency Adjustment 2Ref. Input 3Ref. Out 4Ext. Trigger 5GPIB 6RS232 11 12 13 7USB 8VGA Output 9Ear Phone DC Input 11 Battery

### Fast and Accurate Test of Mobile Phone Boards

Products: R&S FSP Fast and Accurate Test of Mobile Phone Boards Short test times in conjunction with accurate and repeatable measurement results are essential when testing and calibrating mobile phones

### Understand the effects of clock jitter and phase noise on sampled systems A s higher resolution data converters that can

designfeature By Brad Brannon, Analog Devices Inc MUCH OF YOUR SYSTEM S PERFORMANCE DEPENDS ON JITTER SPECIFICATIONS, SO CAREFUL ASSESSMENT IS CRITICAL. Understand the effects of clock jitter and phase

### Test Report: Yaesu FTDX-1200, S/N 3F02051 (loaned by Bill Trippett W7VP)

Test Report: Yaesu FTDX-1200, S/N 3F02051 (loaned by Bill Trippett W7VP) Adam M. Farson VA7OJ/AB4OJ, 19-21 July 2013 1. Introduction and Scope: The following tests were conducted on the FTDX-1200: A. Receiver

### FREQUENCY RESPONSE ANALYZERS

FREQUENCY RESPONSE ANALYZERS Dynamic Response Analyzers Servo analyzers When you need to stabilize feedback loops to measure hardware characteristics to measure system response BAFCO, INC. 717 Mearns Road

### High Power Amplifier Measurements Using Agilent s Nonlinear Vector Network Analyzer

High Power Amplifier Measurements Using Agilent s Nonlinear Vector Network Analyzer Application Note 1408-19 Table of Contents Introduction...2 PNA-X Performance...3 Hardware Setup...6 Setup Examples...12

### PSM 900 and PSM 1000 Personal Monitor Systems TWO SYSTEMS. INFINITE POSSIBILITIES.

PSM 900 and PSM 1000 Personal Monitor Systems TWO SYSTEMS. INFINITE POSSIBILITIES. PSM 900 AND PSM 1000 PERSONAL MONITOR SYSTEMS. ONLY FROM SHURE. PSM 900 PSM 1000 PSM 900 and PSM 1000 Personal Monitor

### The Phase Modulator In NBFM Voice Communication Systems

The Phase Modulator In NBFM Voice Communication Systems Virgil Leenerts 8 March 5 The phase modulator has been a point of discussion as to why it is used and not a frequency modulator in what are called

### Agilent 35670A - Dynamic Signal Analyzer

Agilent 35670A - Dynamic Signal Analyzer Two- or Four-channel high-performance FFTbased spectrum /network analyzer 122 µhz to 102.4 khz 16-bit ADC United States: Price for 35670A from USD 18,300.00 35670A