# EE 442. Lab Experiment No. 1 1/12/2007. Introduction to the Function Generator and the Oscilloscope

Save this PDF as:

Size: px
Start display at page:

Download "EE 442. Lab Experiment No. 1 1/12/2007. Introduction to the Function Generator and the Oscilloscope"

## Transcription

1 EE 442 Lab Experiment No. 1 1/12/2007 Introduction to the Function Generator and the Oscilloscope 1

2 I. INTRODUCTION EE 442 Laboratory Experiment 1 The purpose of this lab is to learn the basic operation of a function generator and an oscilloscope. II. THE AGILENT 33220A FUNCTION GENERATOR (Image courtesy Figure 1 Agilent 33220A function generator PRELAB READING: A function generator is a two terminal piece of test equipment that produces a time varying voltage signal at its output terminals. The Agilent (Figure 1) is capable of producing standard waveforms, as well as a number of arbitrary user defined waveforms. Only the square wave, ramp, and sinusoidal waveforms will be used in this class. When the function generator is energized, it powers up in a default state. This default state is defined by a sinusoidal function at a frequency of 1 khz with a voltage amplitude of 100mV peak to peak (shorthand notation is pp, so this voltage would be 100mV pp.) The generator powers up with its output off. 2

3 PRELAB QUESTIONS: 1. What is the Agilent function generator capable of producing? 2. What is the default frequency and output voltage when the function generator is powered up? 3. Based on Figure 1, what is the frequency range of the Agilent 33220? EXPERIMENT: Upon power-up, the default frequency will appear on the display panel as 1.000,000,0 khz (an 8-digit number.) Press the button under the Ampl menu and the 100mV pp output voltage value will be displayed on the panel. Now press the Freq button to return to the frequency display. The purpose of this exercise is to learn to perform the following necessary operations: 1. Change the frequency of the waveform from 1 khz to whatever value is desired. 2. Change the voltage amplitude from 100 mv pp to whatever value is desired. The user has four options for displaying the voltage: Volts peak-to-peak (V pp), milli-volts peak-to-peak (mv pp), Volts rms (V rms) and milli-volts rms (mv rms). In this class, V pp will be used most of the time. 3. Change the time-varying function from a sinusoid to a square or triangular wave. To change the frequency, the knob may be used in conjunction with the arrow buttons. The arrow buttons change the highlighted digit position of the frequency value. The value of that highlighted digit is controlled by the knob. The arrow buttons move the blinking digit one space to the right or left. EXPERIMENT: 1. From the default frequency, use the knob to obtain a frequency of 3.2 khz. 2. Use the arrow buttons and the knob to set a frequency of khz. 3. Use the arrow buttons and the knob to set a frequency of Hz. Perhaps the easiest way to set a particular frequency (e.g., 1.234,567,8 khz) is to punch it in directly. 3

4 EXPERIMENT: 1. Press the ten buttons and the button under khz, in that order. The unit (button under the khz unit) button is the last button pressed when setting a new frequency value. Pressing this button also causes the generator to take in this new information and output a waveform at the new frequency. Another useful button is the left arrow button. This button allows you to correct a mistake or change your mind in the middle of entering a number. 2. Try another frequency: set the frequency to Hz. The next task to explore is the method of changing the value of the output voltage. The same knob and buttons are used to set the voltage as were used to set the frequency. Also, as was true for the frequency case, when establishing a new voltage value with the direct punch-in method, the last button to be pressed is one of the unit buttons. When a unit button is pressed, the generator will output a waveform with the (new) displayed voltage value. The options are: 1. milli-volts peak-to-peak (mv pp) 2. Volts peak-to-peak (V pp) 3. milli-volts rms (mv rms) 4. Volts rms (V rms) The voltage units can be changed, without changing the voltage value, by pressing a number on the keypad, using the arrow button to delete the number, and then pressing the button under the appropriate unit. Another method of establishing a voltage value is to use the arrow buttons to define the highlighted digit. Then set the value of that highlighted digit using the knob. Note that when the unit position is highlighted, the knob allows adjustment of both the units and the decimal point (they re linked together). EXPERIMENT: 1. Press the button under the Ampl menu. The voltage amplitude of the sinusoidal waveform should now be displayed. (The waveform should still be at its default value of 100 mv pp). 2. Change the voltage of the sinusoidal waveform to V pp using the knob and the arrow buttons. 3. Change the voltage to read an equivalent value in Volts rms. (Answer = mv rms.) 4. Change the voltage in step 3 to V pp by pressing the right arrow button to cause the mv pp to be highlighted. Then rotate the 4

5 knob clockwise. Note the relationship between the units and the decimal point. 5. Set the voltage to V pp by punching-in the value in directly 6. Also remember that to check on the frequency of the voltage being generated just press the button under Freq. To return to the voltage mode, press the button under Ampl. To select the function to be generated, merely press the button associated with the square wave, ramp, or the noise function. Finally, in order to insure that the same voltage value appears at the output terminals of the function generator that appears on the display panel, perform the following sequence: 1. Press the Utility button. 2. Press the button under Output Setup. 3. Press the button under Load. High Z should be highlighted. 4. Press the button under Done. This procedure must be performed every time the function generator is turned on. A failure to complete this sequence may cause the voltage appearing at the terminals of the function generator (for certain load conditions) to be different from the value shown on the display panel. One of the consequences of this procedure is that the default voltage amplitude changes from 100 mv pp to 200 mv pp. (Checking these values is a handy test to see if this task was performed.) III. INTRODUCTION TO THE AGILENT DSO3202A OSCILLOSCOPE (Image courtesy Figure 2 Agilent DSO3202A oscilloscope 5

6 PRELAB READING: In its basic (and normal) mode of operation, an oscilloscope (Figure 2) is nothing more than a very sophisticated voltmeter. It is basically a two input-terminal instrument providing a two dimensional visual display of a time dependent signal voltage waveform. This display, which appears on the screen, makes possible the observation and measurement of voltage signals which can be very complex functions of time. The following paragraphs will describe how an analog oscilloscope works (As far as the user is concerned, the operation of a digital scope is the same but has some extra features). The primary component of an analog oscilloscope is a cathode ray tube (Figure 3). Inside the cathode ray tube (CRT) electrons are generated from a heated cathode and are accelerated towards the front of the screen. When these electrons strike the screen they cause the phosphorescent coating covering the screen to glow, thus creating a white dot at their point of impact. (Image courtesy Figure 3 Basic operation of an analog oscilloscope Also inside the CRT are two sets of plates, a horizontal set and a vertical set. An electric field is created between the horizontal plates and is proportional to the magnitude and polarity of the input voltage signal. This electric field causes the electron beam to be deflected proportionally to the intensity and polarity of the electric field. In this manner the beam is swept between the top and bottom of the screen. Another electric field is created between the vertical set of plates which sweeps the beam from left to right (facing the screen). 6

7 Before the input voltage can be used to deflect the electron beam, it must be scaled to an appropriate level to create an electric field between the horizontal plates of sufficient strength to deflect the electron beam a desired amount. The user can control this scaling with the volts/division knob. This knob is usually adjusted so that the waveform takes up as much of the screen as possible without running into another waveform (multiple trace operation) and without running off the screen. The voltage signal which creates the electric field that sweeps the signal from left to right can come from one of two places: internal or external to the oscilloscope. If the horizontal deflection voltage is derived internally, it will come from a sweep generator. The sweep generator creates a saw tooth shaped voltage waveform that draws the trace across the screen. The rate of the sweep generator is set by the user and is chosen to display the appropriate amount of the input waveform. This control is the seconds/division knob on the front panel of the scope. Each cycle of the sweep generator is initiated by a trigger signal which will be discussed in a later laboratory experiment. The horizontal deflection voltage can also come from an external input signal. Basically another set of input voltage terminals is connected to the horizontal deflection plates. In this manner, the user can graph a y voltage as a function of an x voltage rather than a y voltage as a function of time. EXPERIMENT: Turn the oscilloscope on and set the above controls to the following settings (The scope will take a minute or two to boot up). Horizontal System Controls 500ms/div To change this setting, turn the big knob on the Horizontal section of the front panel (Figure 4). The sweep rate (s/) is displayed at the bottom right corner of the screen next to the sampling rate (Sa/s). (Image courtesy Figure 4 Horizontal system controls 7

8 Vertical System Controls Channel 1 coupling to ground 2 V/div (1X probe) To change the coupling, push the 1 button to bring up the menu for channel 1 (Figure 5). This menu will allow you to change the coupling. While in this menu change the probe attenuation factor from 10X to 1X. This change will make the voltage scales match for the activities we are going to do later. To change the volts per division setting, rotate the large knob above the 1 button. The volts per division (V/) setting is displayed in the lower left hand corner of the scope s screen. (Image courtesy Figure 5 Vertical system controls The sweep rate was set to 500ms/div and there are 10 divisions on the screen, so it should take about 5 s for the dot to move from the left corner of the screen to the right corner of the screen. What you are seeing is the sweep generator voltage being applied to the horizontal deflection plates. This voltage is a piecewise linear function with respect to time as shown in the Figure 6. 8

9 Figure 6 Simulated sweep and trigger Waveforms Waveform 1 is a representation of the voltage from the sweep generator. Waveform 2 is a representation of the trigger waveform. Note how both the trigger pulse and the sweep start at the same time. PRELAB READING: Since the sweep portion of the saw-tooth voltage waveform is linear in time, when it is applied to the horizontal deflection plates, it produces a corresponding horizontal (x) deflection of the electron beam which is also linear in time. The beam (spot) therefore travels across the screen with constant velocity. Combining this linear time-vs.-x dependence, with linear voltage-vs.-y dependence, results in a time dependant voltage waveform that can be visually displayed, as a two dimensional graph, where the vertical axis is defined by a voltage scale, and the horizontal axis is defined by a time scale. There are a couple of other features present in the saw-tooth wave, shown in Figure 6, that should be mentioned. That portion of the waveform with a negative slope defines the retrace period during which the voltage of the saw-tooth wave returns to zero to prepare for another sweep. The reset (standby) period (where the sweep is flat-lined) is when the sweep generator and trigger circuits reset themselves in preparation for the next sweep cycle. During the retrace and reset period the electron beam is turned off. A trigger pulse waveform corresponding to the saw-tooth waveform is also shown in the figure. The relationship between these two waveforms will be discussed in the triggering section of this experiment. 9

10 PRELAB QUESTION: 1. How many channels does the oscilloscope have? Where is the auto-scale button? EXPERIMENT: Change the sweep rate from 500ms/div to 500μs/div. Rotate the knob one click at a time while noting the change of velocity of the trace on the screen. Notice that after a few clicks of the knob, the sweep generator is sweeping so fast that the dot appears to change into a line. The dot is moving so fast that the phosphorescent coating is still glowing when the dot returns. This persistence of the phosphorescent coating is what allows us to see a voltage signal waveform that repeats many tens to millions of times per second. IV. DC VOLTAGE MEASUREMENTS PRELAB READING: To perform this part of the experiment, keep the controls as they were in the previous procedure, with the CH1 vertical scale factor at 2 V/div and the sweep rate at 500μs/div. Note that the input coupling was set to the ground position. The grounded input setting and the CH1 vertical position control (little knob under the 1 button), provide a means for the user to establish a zero voltage reference trace location anywhere on the screen. Convenience and/or practical reasons can suggest the best location. The internal implementation of the coupling setting can be understood by referring to Figure 7. With the input coupling set to ground, any signal voltage connected to the CH1 input terminal would be disconnected (open circuited) from the oscilloscope. At the same time, zero volts (by a short circuit to ground) is applied to the input to the vertical attenuator/amplifier circuits. Note that if a voltage signal were connected to the CH1 input terminal, it would NOT be short circuited to ground. It is good experimental practice to always check (and reposition if necessary) the location (on the screen aka. graticule) of the zero volt trace prior to making a voltage related measurement. Note that if the coupling feature were not available, the alternative procedure of establishing a zero volt trace would be to disconnect the probe from the circuit being measured and then connect it to a ground terminal (to simulate zero volts). In time this method could become a very inconvenient process. PRELAB QUESTION: 1. How many coupling modes does the oscilloscope have? What are they? 10

12 voltage measurements, at various scale factor settings, and verify with the digital multimeter. When you are finished, disconnect the meter from the supply and turn it off. 8. Reduce the DC supply voltage to zero and set the sweep rate to 0.1 s/div. Set the vertical scale to 2 V/div and zero the trace one division from the bottom of the graticule. Now increase the voltage output of the DC supply in such a manner that you can control the vertical displacement of the beam as it moves horizontally across the screen. Vary this voltage up and down rapidly and slowly, and note the relationship between voltage, beam (spot) displacement, and time. 9. Reduce the DC supply voltage to zero and turn it off. Disconnect the scope from the supply and turn it and the meter off. V. AC VOLTAGE MEASUREMENTS PRELAB READING: In order to use the scope to make AC voltage measurements, which means the observation and measurement of time varying voltage signals, we first need to define some of the properties of periodic waveforms. This task is done in Figure 8 for a sinusoid. One defining property of a sinusoid is its peak amplitude (value). This value can be obtained on the scope by measuring the vertical (voltage) distance from the (user set) zero voltage (horizontal) axis to either the positive peak or the negative peak. In the case of a sinusoid, these two peaks are equal in amplitude. Another method, which is often quicker since the zero volt reference does not have to be used, is to measure the peak to peak amplitude and then divide by two. 12

13 Figure 8 Components of a sinusoid PRELAB QUESTION: 1. What is the peak voltage, and peak-to-peak voltage of the waveform shown in Figure 8? EXPERIMENT: Set the scope to measure a 2 V pp, 1 khz (2 V/div and 200 µs/div) signal and set the zero volt reference level. Make sure that the input coupling is set to DC before you proceed. Connect the scope to the signal generator with a BNC to BNC cable. Turn the signal generator on and set its output impedance to HIGH Z. If you connect the scope to the signal generator using any other method, be sure that you observe polarity vary carefully as both the scope and the signal generator are earth grounded. A mistake in polarity will mean a short circuit and probably a blown fuse in the signal generator. With the signal generator powered up and connected, set it to output a 2 V pp, 1 khz sinusoid and turn the output on (push the Output button). When everything is set up correctly you should see about 2 sinusoidal 13

14 cycles on the screen. You may have to play with the trigger level knob to obtain a stable display. Change the amplitude of the output from the signal generator and note the effects on the scope. Set the signal generator to output 8 V pp. The waveform on the graticule of the scope should be just superimposed between the +4 and -4 volt lines. Change to several different scale factors and make sure the output voltage is still 8 V pp. You may loose the signal as you change scales, if this problem occurs you will need to re-adjust the trigger level. A point can be made here in regards to basic voltage and time measurement techniques. In general, if the particular situation permits, it is always a good practice to enlarge the waveform as much as possible. This practice allows a more accurate measurement to be made. VI. MEASUREMENT OF FREQUENCY PRELAB READING: To determine the frequency of a sinusoidal waveform, using a scope, it is first necessary to measure the period T of that waveform. Figure 9 suggests two of the most common methods of measuring T. The first method is performed by measuring the time between two alternate zero crossings. The second method is performed by measuring the time between two consecutive (equal polarity) peaks. Both measurements should give the same result, although the zero crossing technique is probably the easiest to use and the most accurate. If the zero crossing method is used, it is imperative that the zero voltage reference be accurately set. Once the period T is found, the frequency can be 1 determined from the relation f =. T 14

15 Figure 9 Frequency measurement techniques PRELAB QUESTION: 1. What are the two methods for measuring the period T? EXPERIMENT: 1 We will make use of this expression f = by examining the relationship T between the frequency of the signal to be displayed and the sweep rate of the scope. The signal generator should still be set at 8V pp, 1 khz. Set the vertical scale of the scope to 2 V/div and 200 µs/div. Reduce the frequency and note what happens on the graticule. The waveform begins to spread out and pretty soon there is only one cycle of the waveform instead of two. Increase the frequency of the signal generator past 1 khz and note how the waveform gets compressed and more cycles of the sinusoid appear on the screen. From this exercise you should see that the period of the waveform decreases with increasing frequency. The next task will be to take a simple frequency measurement with the scope. Make sure the signal generator is still set on 1 khz. Re-establish the zero volt reference on the scope. Once this step is completed, the 15

16 waveform from the signal generator should be seen on the graticule. Before proceeding answer the questions below. What should the period of the waveform be (from the signal generator)? T = Based on the above answer, about how many waveform periods should be seen on the graticule if the sweep rate is 200 µs/div? Number of Periods = How many periods do you see on the scope? Number of Periods = How many divisions (estimate to the nearest tenth of a division) on the graticule does one period of the waveform occupy? Number of Divisions = What is the period of the measured waveform (number of divisions multiplied by the sweep rate setting)? T = What is the frequency of the waveform? F = With a 1 khz signal still displayed on the scope, change the sweep rate and note the effects of this change. When you increase the sweep rate to 100 µs/div you should see only one period of the waveform. When you decrease the sweep rate to 500 µs/div you should see about 5 periods of the waveform displayed on the graticule. Note that for the above sweep rate changes the frequency of the input signal remained constant. Leave everything set up as it is for the next part of this experiment. 16

17 VII. TRIGGERING PHENOMENON PRELAB READING: A. General Discussion In its simplest sense, the primary role of the trigger circuit is to determine where the trace will begin on the waveform of the signal to be displayed. This definition assumes that were the trace starts is important to the user! To accomplish the task of triggering, the following series of events take place. When certain conditions are satisfied, the trigger circuit will send a turn-on message (trigger pulse) to the sweep generator. Upon receiving this message, the sweep generator will turn on, causing the electron beam to begin its horizontal sweep across the screen. The conditions mentioned above are the voltage level and the slope polarity of the signal being triggered off of, both of which are pre-selected by the user. Another condition for the trigger pulse to be accepted is that the sweep generator can t be in the middle of a sweep when the trigger pulse arrives, i.e., a sweep can t be initiated in the middle of another sweep. It should be mentioned that the signal being triggered off of (trigger signal), whose waveform provides the basis for a trigger pulse, does not have to be the same signal as connected to the vertical input. However the frequencies of the two signals should be integrally related (one frequency should be an integer multiple of the other) if a stable waveform display is desired. In general, trigger signals are derived from one of two basic source locations: external or internal. An external trigger signal comes from a voltage not being measured by the scope. If this type of signal is used, it is connected to the external input terminal of the scope. The trigger source should be set to external and the trigger signal input coupling should be set to whatever coupling fits the situation (usually DC). An internal trigger signal is derived from voltage signals being measured by the scope. Examples of measured voltages are the voltages applied to either the channel 1 (CH1) or channel 2 (CH2) input terminals. If this trigger option is chosen, the trigger source should be set to CH1 or CH2 which ever input you want to trigger off of. 17

18 EXPERIMENT: B. Trigger Level Control Make the following settings on the scope. Set the trigger level to 0V (use the double arrow knob in the Triggering section), slope to positive, trigger mode to auto, sweep rate to 0.2 ms/div (push the Mode/Coupling button to bring up the appropriate menu) and the CH1 vertical scale set to 2 V/div. Also set the scope to trigger off of CH1 (trigger source). Establish a zero volt trace and make sure the scope is connected to the signal generator. Adjust the signal generator output waveform to 10 V pp, 1 khz. The trace should be starting at 0 V on the left side of the screen and have positive slope where it starts. Once the scope and signal generator are set up correctly, rotate the trigger level control counterclockwise and watch what happens to the beginning of the trace (at the left side of the screen). The trace should be starting at a negative value. As the control is rotated further, the starting value of the trace should become more negative. If the trigger level is reduced below the negative peak of the signal, the waveform will become unstable. Rotate the trigger level control the other way and you will notice that the trace starts from a positive value. The trace will start from an increasingly positive value as the trigger level is made more positive. Again, if you exceed the peak voltage of the waveform the trace will cease to be stable. C. Trigger Slope Control Adjust the trigger level control so the trace is starting at about zero volts from the left edge of the screen. Watch what happens to the beginning of the trace when you change the slope from positive to negative. Wiggle the trigger level control button a little to note the effects of changing the slope. D. Trigger Modes We have been using the automatic triggering mode for the last experiments. In this mode of operation triggering will always occur and a trace will always exist even in the absence of a trigger signal. It is a very convenient feature since it allows a trace to exist even when zero volts is applied to the input that is being triggered off of. This condition happens every time you establish your zero reference trace. The problem with the automatic triggering mode is that it will not create a stable trace at frequencies less than about 20 Hz. For these frequencies the normal mode of triggering must be used. 18

19 The problems with the normal mode of triggering are that triggering will occur only when triggering off a non DC signal and triggering will not occur if the magnitude of the trigger level exceeds the amplitude of the signal being triggered off of. Both the normal and automatic triggering modes have their attributes and drawbacks, but the automatic mode will be the most versatile for our needs. (To change the trigger mode press the Mode/Coupling button and then press the button next to the sweep option.) VIII. MEASUREMENT OF SIGNALS WITH AC AND DC COMPONENTS EXERCISE (THIS SECTION IS OPTIONAL) The voltage signals studied so far in this experiment have been either AC or DC. Many signals in practical circuits have both an AC component and a DC (time average value) component. For example, the voltage v= 4+ 2sin( ωt) V has a DC component of 4 V and an AC component of 2sin( ω t) V. The AC portion has a peak value of 2 V and a frequency of ω f = Hz. The next task of this experiment is to connect the function 2π generator in series with the DC power supply to create the above expression. Before energizing the DC supply (Agilent E3630A, use the +6 V tap) turn it voltage control all the way down (fully ccw). Connect the two sources in series and then connect the scope across them as shown in Figure 10. It is imperative that the DC supply and the function generator be properly connected together. If, for instance, the two devices were connected parallel, damage would be done to the signal generator. Ask your lab instructor to check your wiring before energizing either source. 19

20 Red/+ DC Power Supply Red/+ Oscilloscope Black/- Black/- Black/- Red/+ Signal Generator Figure 10 Circuit diagram for connecting the power supply and the signal generator to the oscilloscope Set the scope to 200 μs/div, DC coupling, CH1 to 1 V/div, to trigger automatically, and trigger off of CH1. Turn on the signal generator and set its output to 0 V pp, 1 khz. Position the zero volt reference trace 2 major divisions from the bottom of the graticule. Turn on the DC supply and increase the voltage until the trace is at the 4 volt line on the graticule. Now use the knob to sweep the signal generator s output to 4 V pp, 1 khz. Sketch the resultant composite waveform on the attached graph paper. The average value of this composite waveform is 4 volts and should remain that value for any amplitude sinusoid. This characteristic should have been observed as the amplitude of the signal generator was being increased to 2 volts. It is time to examine AC coupling of the CH1 input. From Figure 11, note that when AC coupling is selected, a capacitor is placed in the path between the input terminal to the amplifier/attenuator circuits of the scope (the switch will connected to point 2). The purpose of the capacitor is to block the DC component while allowing the AC component to pass. In essence, the average value of the waveform will be reduced to zero, leaving only the time varying portion of the waveform. This capability allows accurate measurement of the AC component of a signal when it is dwarfed in magnitude by the DC component. 20

21 DC Blocking Capacitor 2 Selector Switch Channel 1 Input 3 1 Channel 1 Amplif ier 4 Figure 11 Input circuitry between the probe and channel 1 amplifier 0 Change the input coupling from DC to AC and sketch the new waveform on the graph paper, use the same coordinates as you used for your last sketch. Note that the waveform dropped by a level equal to the amount of the average value of the waveform. The ability to have both AC and DC coupling allows for an easy method to measure the average of a waveform. Turn off the function generator and DC supply and remove them from the circuit. IX. EXTERNAL HORIZONTAL INPUT (X-Y OPERATION) (THIS SECTION IS OPTIONAL) Thus far in this experiment we have been using the internal sweep system as our only means of generating a horizontal deflection voltage. To help in understanding the horizontal deflection system we will now perform an experiment employing an externally applied horizontal voltage, using the DC supply. The function generator will still be used to supply the vertical input voltage. First, establish the zero volt reference trace at the center of the graticule. Then connect the function generator with its output set to 4 V pp, 1 khz. Set the scope to read this 4 V pp, 1 khz signal. Once the signal is displayed on the scope, slow the sweep rate to 100 ms/div. Notice how the display on the scope changed from a couple of cycles of sinusoid to a bar across the screen 4 volts high. Reduce the signal generator frequency to 1 Hz. To maintain a stable waveform, switch the trigger settings from automatic to normal (see part V for instructions if needed). Adjust the trigger level until the trace starts at zero volts. You should see a 4 V pp, 1 Hz sine wave on the screen. Next, change the scope from Y-T mode to X-Y mode (push the Main/Delayed button and then push the button next to the Time Base block). This mode bypasses the internal sweep circuit and connects the horizontal deflection plates to the CH1 input terminal. With nothing applied to the CH1 input, the display on the scope should resemble a 4 volt high vertical line in the center of the screen. Use the horizontal position control to move this line to the left side of the screen. 21

22 Connect the DC power supply to the CH1 input of the scope. Set the CH1 scale factor to 500 mv/div. Turn the voltage control knob all the way down and then energize the power supply. Now, slowly increase the DC voltage on the power supply and watch what happens on the screen of the scope. Play with this knob, turning it back and forth. With a little practice, you should be able to generate a decent sine wave. The X input capability allows the scope to be used as an X-Y plotter. One example of this mode of operation is the observation of i-v terminal characteristics of electronic devices. Since the scope is a voltage sensitive instrument, the current axis values on the i-v plot are obtained by connecting the scope across a resistor R in series with the device under test. A 1/R scaling factor is then used to obtain the i axis values. This example is mentioned only to illustrate a point, and is not a performance part of this experiment. Turn off the function generator and the DC supply and disconnect the scope probes. To prepare for the next experiment, set the sweep rate of the scope to 20 μs/div. X. AN ADDITIONAL FEATURE OF THE OSCILLOSCOPE A final feature worth mentioning is the auto-scale feature. Pushing this button will automatically set the volts/div, s/div, and triggering for most of the waveforms you will need to measure. After you use this button, you can use the Measure menus to make the scope show the measurements on the screen, rather than reading them from the graticule. XI. CONCLUSION Now you should be able to make a variety of AC, DC, and combination of AC and DC measurements. Save your notes from this section as the oscilloscope will an invaluable tool in subsequent experiments. 22

### The RC Circuit. Pre-lab questions. Introduction. The RC Circuit

The RC Circuit Pre-lab questions 1. What is the meaning of the time constant, RC? 2. Show that RC has units of time. 3. Why isn t the time constant defined to be the time it takes the capacitor to become

### EXPERIMENT NUMBER 5 BASIC OSCILLOSCOPE OPERATIONS

1 EXPERIMENT NUMBER 5 BASIC OSCILLOSCOPE OPERATIONS The oscilloscope is the most versatile and most important tool in this lab and is probably the best tool an electrical engineer uses. This outline guides

### The Oscilloscope. Vision is the art of seeing things invisible. J. Swift ( ) OBJECTIVE To learn to operate a cathode ray oscilloscope.

The Oscilloscope Vision is the art of seeing things invisible. J. Swift (1667-1745) OBJECTIVE To learn to operate a cathode ray oscilloscope. THEORY The oscilloscope, or scope for short, is a device for

### The Oscilloscope and the Function Generator:

The Oscilloscope and the Function Generator: Some introductory exercises for students in the advanced labs Introduction So many of the experiments in the advanced labs make use of oscilloscopes and function

### Lab 1: The Digital Oscilloscope

PHYSICS 220 Physical Electronics Lab 1: The Digital Oscilloscope Object: To become familiar with the oscilloscope, a ubiquitous instrument for observing and measuring electronic signals. Apparatus: Tektronix

### The Oscilloscope, the Signal Generator and Your Filter s Test Setup SGM 5/29/2013

The Oscilloscope, the Signal Generator and Your Filter s Test Setup SGM 5/29/2013 1. Oscilloscope A multimeter is an appropriate device to measure DC voltages, however, when a signal alternates at relatively

### Lab E1: Introduction to Circuits

E1.1 Lab E1: Introduction to Circuits The purpose of the this lab is to introduce you to some basic instrumentation used in electrical circuits. You will learn to use a DC power supply, a digital multimeter

### Electrical Resonance

Electrical Resonance (R-L-C series circuit) APPARATUS 1. R-L-C Circuit board 2. Signal generator 3. Oscilloscope Tektronix TDS1002 with two sets of leads (see Introduction to the Oscilloscope ) INTRODUCTION

### Physics 2306 Experiment 7: Time-dependent Circuits, Part 1

Name ID number Date Lab CRN Lab partner Lab instructor Objectives Physics 2306 Experiment 7: Time-dependent Circuits, Part 1 To study the time dependent behavior of the voltage and current in circuits

### EXPERIMENT NUMBER 8 CAPACITOR CURRENT-VOLTAGE RELATIONSHIP

1 EXPERIMENT NUMBER 8 CAPACITOR CURRENT-VOLTAGE RELATIONSHIP Purpose: To demonstrate the relationship between the voltage and current of a capacitor. Theory: A capacitor is a linear circuit element whose

### 1. Oscilloscope is basically a graph-displaying device-it draws a graph of an electrical signal.

CHAPTER 3: OSCILLOSCOPE AND SIGNAL GENERATOR 3.1 Introduction to oscilloscope 1. Oscilloscope is basically a graph-displaying device-it draws a graph of an electrical signal. 2. The graph show signal change

### PHYS 2426 Engineering Physics II (Revised July 7, 2011) AC CIRCUITS: RLC SERIES CIRCUIT

PHYS 2426 Engineering Physics II (Revised July 7, 2011) AC CIRCUITS: RLC SERIES CIRCUIT INTRODUCTION The objective of this experiment is to study the behavior of an RLC series circuit subject to an AC

### Charge and Discharge of a Capacitor

Charge and Discharge of a Capacitor INTRODUCTION Capacitors 1 are devices that can store electric charge and energy. Capacitors have several uses, such as filters in DC power supplies and as energy storage

### RC Circuits and The Oscilloscope Physics Lab X

Objective RC Circuits and The Oscilloscope Physics Lab X In this series of experiments, the time constant of an RC circuit will be measured experimentally and compared with the theoretical expression for

### EXPERIMENT 6 CLIPPING AND CLAMPING DIODE CIRCUITS

EXPERIMENT 6 CLIPPING AND CLAMPING DIODE CIRCUITS OBJECTIVES To understand the theory of operation of the clipping and clamping diode circuits. To design wave shapes that meet different circuits needs.

### Oscilloscope, Function Generator, and Voltage Division

1. Introduction Oscilloscope, Function Generator, and Voltage Division In this lab the student will learn to use the oscilloscope and function generator. The student will also verify the concept of voltage

### Beginners Guide to the TDS 210 and TDS 220 Oscilloscopes

Beginners Guide to the TDS 210 and TDS 220 Oscilloscopes By David S. Lay P. Eng Foreword This guide contains information to help you become familiar with using digital oscilloscopes. You should work through

### AC Measurements Using the Oscilloscope and Multimeter by Mr. David Fritz

AC Measurements Using the Oscilloscope and Multimeter by Mr. David Fritz 1 Sine wave with a DC offset f = frequency in Hz A = DC offset voltage (average voltage) B = Sine amplitude Vpp = 2B Vmax = A +

### MAKE SURE TA & TI STAMPS EVERY PAGE BEFORE YOU START

Laboratory Section: Last Revised on December 15, 2014 Partners Names Grade EXPERIMENT 10 Electronic Circuits 0. Pre-Laboratory Work [2 pts] 1. How are you going to determine the capacitance of the unknown

### RLC Series Resonance

RLC Series Resonance 11EM Object: The purpose of this laboratory activity is to study resonance in a resistor-inductor-capacitor (RLC) circuit by examining the current through the circuit as a function

### DIODE CIRCUITS LABORATORY. Fig. 8.1a Fig 8.1b

DIODE CIRCUITS LABORATORY A solid state diode consists of a junction of either dissimilar semiconductors (pn junction diode) or a metal and a semiconductor (Schottky barrier diode). Regardless of the type,

### Fundamentals of Signature Analysis

Fundamentals of Signature Analysis An In-depth Overview of Power-off Testing Using Analog Signature Analysis www.huntron.com 1 www.huntron.com 2 Table of Contents SECTION 1. INTRODUCTION... 7 PURPOSE...

### Experiment 2 Diode Applications: Rectifiers

ECE 3550 - Practicum Fall 2007 Experiment 2 Diode Applications: Rectifiers Objectives 1. To investigate the characteristics of half-wave and full-wave rectifier circuits. 2. To recognize the usefulness

### Annex: VISIR Remote Laboratory

Open Learning Approach with Remote Experiments 518987-LLP-1-2011-1-ES-KA3-KA3MP Multilateral Projects UNIVERSITY OF DEUSTO Annex: VISIR Remote Laboratory OLAREX project report Olga Dziabenko, Unai Hernandez

### MATERIALS. Multisim screen shots sent to TA.

Page 1/8 Revision 0 9-Jun-10 OBJECTIVES Learn new Multisim components and instruments. Conduct a Multisim transient analysis. Gain proficiency in the function generator and oscilloscope. MATERIALS Multisim

### PSPICE tutorial: a simple DC circuit Getting started

PSPICE tutorial: a simple DC circuit We will learn some of the basic maneuvers of using the Cadence schematic capture program and PSPice engine through a simple example -- a diode rectifier circuit. The

### ε: Voltage output of Signal Generator (also called the Source voltage or Applied

Experiment #10: LR & RC Circuits Frequency Response EQUIPMENT NEEDED Science Workshop Interface Power Amplifier (2) Voltage Sensor graph paper (optional) (3) Patch Cords Decade resistor, capacitor, and

### Designing a Poor Man s Square Wave Signal Generator. EE-100 Lab: Designing a Poor Man s Square Wave Signal Generator - Theory

EE-100 Lab: - Theory 1. Objective The purpose of this laboratory is to introduce nonlinear circuit measurement and analysis. Your measurements will focus mainly on limiters and clamping amplifiers. During

### Class #12: Experiment The Exponential Function in Circuits, Pt 1

Class #12: Experiment The Exponential Function in Circuits, Pt 1 Purpose: The objective of this experiment is to begin to become familiar with the properties and uses of the exponential function in circuits

### Using an Oscilloscope

Using an Oscilloscope The oscilloscope is used to measure a voltage that changes in time. It has two probes, like a voltmeter. You put these probes on either side of the thing that you want to measure

### Name Date Day/Time of Lab Partner(s) Lab TA

Name Date Day/Time of Lab Partner(s) Lab TA Objectives LAB 7: AC CIRCUITS To understand the behavior of resistors, capacitors, and inductors in AC Circuits To understand the physical basis of frequency-dependent

### EE 1202 Experiment #2 Resistor Circuits

EE 1202 Experiment #2 Resistor Circuits 1. ntroduction and Goals: Demonstrates the voltage-current relationships in DC and AC resistor circuits. Providing experience in using DC power supply, digital multimeter,

### CHARGING and DISCHARGING CAPACITORS

Lab 4 Oscilloscopes, Electrocardiograms, Discharging Capacitors LAB 4a LAB 4b Lab 4c THE OSCILLOSCOPE ELECTROCARDIOGRAMS (ECG or EKG) CHARGING and DISCHARGING CAPACITORS EXPERIMENTAL QUESTION In this lab,

### Inductors in AC Circuits

Inductors in AC Circuits Name Section Resistors, inductors, and capacitors all have the effect of modifying the size of the current in an AC circuit and the time at which the current reaches its maximum

### Lab Exercise 1: Acoustic Waves

Lab Exercise 1: Acoustic Waves Contents 1-1 PRE-LAB ASSIGNMENT................. 2 1-3.1 Spreading Factor: Spherical Waves........ 2 1-3.2 Interference In 3-D................. 3 1-4 EQUIPMENT........................

### SERIES-PARALLEL DC CIRCUITS

Name: Date: Course and Section: Instructor: EXPERIMENT 1 SERIES-PARALLEL DC CIRCUITS OBJECTIVES 1. Test the theoretical analysis of series-parallel networks through direct measurements. 2. Improve skills

### Overview: The purpose of this experiment is to introduce diode rectifier circuits used in DC power supplies.

UNIVERSITY OF NORTH CAROLINA AT CHARLOTTE Department of Electrical and Computer Engineering Experiment No. 3 Diodes and Bridge Rectifiers Overview: The purpose of this experiment is to introduce diode

### Lab 3 - DC Circuits and Ohm s Law

Lab 3 DC Circuits and Ohm s Law L3-1 Name Date Partners Lab 3 - DC Circuits and Ohm s Law OBJECTIES To learn to apply the concept of potential difference (voltage) to explain the action of a battery in

### EE320L Electronics I. Laboratory. Laboratory Exercise #5. Clipping and Clamping Circuits. Angsuman Roy

EE320L Electronics I Laboratory Laboratory Exercise #5 Clipping and Clamping Circuits By Angsuman Roy Department of Electrical and Computer Engineering University of Nevada, Las Vegas Objective: The purpose

### AC CIRCUITS - CAPACITORS AND INDUCTORS

EXPRIMENT#8 AC CIRCUITS - CAPACITORS AND INDUCTORS NOTE: Two weeks are allocated for this experiment. Before performing this experiment, review the Proper Oscilloscope Use section of Experiment #7. Objective

### RC Circuit (Power amplifier, Voltage Sensor)

Object: RC Circuit (Power amplifier, Voltage Sensor) To investigate how the voltage across a capacitor varies as it charges and to find its capacitive time constant. Apparatus: Science Workshop, Power

### Magnetism. ***WARNING: Keep magnets away from computers and any computer disks!***

Magnetism This lab is a series of experiments investigating the properties of the magnetic field. First we will investigate the polarity of magnets and the shape of their field. Then we will explore the

10. A ircuits* Objective: To learn how to analyze current and voltage relationships in alternating current (a.c.) circuits. You will use the method of phasors, or the vector addition of rotating vectors

### Experiment #9: RC and LR Circuits Time Constants

Experiment #9: RC and LR Circuits Time Constants Purpose: To study the charging and discharging of capacitors in RC circuits and the growth and decay of current in LR circuits. Part 1 Charging RC Circuits

### Altoids Tin Headphone Amplifier Lab

Altoids Tin Headphone Amplifier Lab Michigan State University AEE/IEEE Step 1: Required Parts Table 1 shows a complete listing of the parts required to complete this project. Figure 1 shows a picture of

### Figure 1: Multiple unsynchronized snapshots of the same sinusoidal signal.

1 Oscilloscope Guide Introduction An oscilloscope is a device used to observe and measure time-dependent electronic signals. It is essentially an enhanced voltmeter which displays a graph of potential

### ENGR 210 Lab 11 Frequency Response of Passive RC Filters

ENGR 210 Lab 11 Response of Passive RC Filters The objective of this lab is to introduce you to the frequency-dependent nature of the impedance of a capacitor and the impact of that frequency dependence

### Lab 3 Rectifier Circuits

ECET 242 Electronic Circuits Lab 3 Rectifier Circuits Page 1 of 5 Name: Objective: Students successfully completing this lab exercise will accomplish the following objectives: 1. Learn how to construct

### CHARGE TO MASS RATIO OF THE ELECTRON

CHARGE TO MASS RATIO OF THE ELECTRON In solving many physics problems, it is necessary to use the value of one or more physical constants. Examples are the velocity of light, c, and mass of the electron,

### APPLICATION NOTE 29 Testing Capacitors with High DC Bias

APPLICATION NOTE 29 Testing Capacitors with High DC Bias This application note will describe the process of analysing the impedance of a capacitor when subjected to high DC bias voltages. This particular

### " = R # C. Create your sketch so that Q(t=τ) is sketched above the delineated tic mark. Your sketch. 1" e " t & (t) = Q max

Physics 241 Lab: Circuits DC Source http://bohr.physics.arizona.edu/~leone/ua/ua_spring_2010/phys241lab.html Name: Section 1: 1.1. Today you will investigate two similar circuits. The first circuit is

### Ph 3504 Nuclear Magnetic Resonance and Electron Spin Resonance

Ph 3504 Nuclear Magnetic Resonance and Electron Spin Resonance Required background reading Tipler, Llewellyn, section 12-3 (you only need to read the part labeled Nuclear Magnetic Resonance on pages 596-597

### See Horenstein 4.3 and 4.4

EE 462: Laboratory # 4 DC Power Supply Circuits Using Diodes by Drs. A.V. Radun and K.D. Donohue (2/14/07) Department of Electrical and Computer Engineering University of Kentucky Lexington, KY 40506 Updated

### Magnetic Fields and Their Effects

Name Date Time to Complete h m Partner Course/ Section / Grade Magnetic Fields and Their Effects This experiment is intended to give you some hands-on experience with the effects of, and in some cases

### Lab #9: AC Steady State Analysis

Theory & Introduction Lab #9: AC Steady State Analysis Goals for Lab #9 The main goal for lab 9 is to make the students familar with AC steady state analysis, db scale and the NI ELVIS frequency analyzer.

### Transistor Amplifiers

Physics 3330 Experiment #7 Fall 1999 Transistor Amplifiers Purpose The aim of this experiment is to develop a bipolar transistor amplifier with a voltage gain of minus 25. The amplifier must accept input

### UNIVERSITY OF NORTH CAROLINA AT CHARLOTTE. Department of Electrical and Computer Engineering

UNIVERSITY OF NORTH CAROLINA AT CHARLOTTE Department of Electrical and Computer Engineering Experiment No. 5 - Gain-Bandwidth Product and Slew Rate Overview: In this laboratory the student will explore

### E X P E R I M E N T 7

E X P E R I M E N T 7 The RC Circuit Produced by the Physics Staff at Collin College Copyright Collin College Physics Department. All Rights Reserved. University Physics II, Exp 7: The RC Circuit Page

### Lab #4 Capacitors and Inductors. Capacitor and Inductor Transient Response

Capacitor and Inductor Transient Response Capacitor Theory Like resistors, capacitors are also basic circuit elements. Capacitors come in a seemingly endless variety of shapes and sizes, and they can all

### Class A Amplifier Design

Module 2 Amplifiers Introduction to Amplifier Design What you ll learn in Module 2. Basic design process. Section 2.0 Introduction to Amplifier Design. Section 2.1 DC Conditions. Design a BJT class A common

### Basic oscilloscope operation

asic oscilloscope operation This worksheet and all related files are licensed under the Creative Commons ttribution License, version.0. To view a copy of this license, visit http://creativecommons.org/licenses/by/.0/,

### Measuring Electric Phenomena: the Ammeter and Voltmeter

Measuring Electric Phenomena: the Ammeter and Voltmeter 1 Objectives 1. To understand the use and operation of the Ammeter and Voltmeter in a simple direct current circuit, and 2. To verify Ohm s Law for

### Date: Deflection of an Electron in a Magnetic Field

Name: Partners: Date: Deflection of an Electron in a Magnetic Field Purpose In this lab, we use a Cathode Ray Tube (CRT) to measure the effects of an electric and magnetic field on the motion of a charged

### The R-C series circuit

School of Engineering Department of Electrical and Computer Engineering 332:224 Principles of Electrical Engineering II Laboratory Experiment 4 The C series circuit 1 Introduction Objectives To study the

### EE 1202 Experiment #4 Capacitors, Inductors, and Transient Circuits

EE 1202 Experiment #4 Capacitors, Inductors, and Transient Circuits 1. Introduction and Goal: Exploring transient behavior due to inductors and capacitors in DC circuits; gaining experience with lab instruments.

### ECEN 1400, Introduction to Analog and Digital Electronics

ECEN 1400, Introduction to Analog and Digital Electronics Lab 4: Power supply 1 INTRODUCTION This lab will span two lab periods. In this lab, you will create the power supply that transforms the AC wall

### Experiment A5. Hysteresis in Magnetic Materials

HYSTERESIS IN MAGNETIC MATERIALS A5 1 Experiment A5. Hysteresis in Magnetic Materials Objectives This experiment illustrates energy losses in a transformer by using hysteresis curves. The difference betwen

### LABORATORY 10 TIME AVERAGES, RMS VALUES AND THE BRIDGE RECTIFIER. Bridge Rectifier

LABORATORY 10 TIME AVERAGES, RMS VALUES AND THE BRIDGE RECTIFIER Full-wave Rectification: Bridge Rectifier For many electronic circuits, DC supply voltages are required but only AC voltages are available.

### Experiment 3: Magnetic Fields of a Bar Magnet and Helmholtz Coil

MASSACHUSETTS INSTITUTE OF TECHNOLOGY Department of Physics 8.02 Spring 2009 Experiment 3: Magnetic Fields of a Bar Magnet and Helmholtz Coil OBJECTIVES 1. To learn how to visualize magnetic field lines

### Bipolar Transistor Amplifiers

Physics 3330 Experiment #7 Fall 2005 Bipolar Transistor Amplifiers Purpose The aim of this experiment is to construct a bipolar transistor amplifier with a voltage gain of minus 25. The amplifier must

### The Oscilloscope and the Function Generator:

The Oscilloscope and the Function Generator: Some introductory exercises for students in the advanced labs Introduction So many of the experiments in the advanced labs make use of oscilloscopes and function

### Deflection of Electrons in an Electric Field

Name: Partners: Date: Deflection of Electrons in an Electric Field Purpose In this lab, we use a Cathode Ray Tube (CRT) to measure the effects of an electric field on the motion of a charged particle,

### Lab 5: Simple Electrical Circuits

Lab 5: Simple Electrical Circuits Introduction: In this laboratory you will explore simple DC (direct current) electrical circuits. The primary goal of the lab will be to develop a model for electricity.

### very small Ohm s Law and DC Circuits Purpose: Students will become familiar with DC potentiometers circuits and Ohm s Law. Introduction: P31220 Lab

Ohm s Law and DC Circuits Purpose: Students will become familiar with DC potentiometers circuits and Ohm s Law. Introduction: Ohm s Law for electrical resistance, V = IR, states the relationship between

### Lab 4 - Data Acquisition

Spring 11 Lab 4 - Data Acquisition Lab 4-1 Lab 4 - Data Acquisition Format This lab will be conducted during your regularly scheduled lab time in a group format. Each student is responsible for learning

### FREQUENCY RESPONSE OF AN AUDIO AMPLIFIER

2014 Amplifier - 1 FREQUENCY RESPONSE OF AN AUDIO AMPLIFIER The objectives of this experiment are: To understand the concept of HI-FI audio equipment To generate a frequency response curve for an audio

### GENERAL SCIENCE LABORATORY 1110L Lab Experiment 6: Ohm s Law

GENERAL SCIENCE LABORATORY 1110L Lab Experiment 6: Ohm s Law OBJECTIVES: To verify Ohm s law, the mathematical relationship among current, voltage or potential difference, and resistance, in a simple circuit.

### The purposes of this experiment are to test Faraday's Law qualitatively and to test Lenz's Law.

260 17-1 I. THEORY EXPERIMENT 17 QUALITATIVE STUDY OF INDUCED EMF Along the extended central axis of a bar magnet, the magnetic field vector B r, on the side nearer the North pole, points away from this

### Electronic WorkBench tutorial

Electronic WorkBench tutorial Introduction Electronic WorkBench (EWB) is a simulation package for electronic circuits. It allows you to design and analyze circuits without using breadboards, real components

### ANADOLU UNIVERSITY DEPARTMENT OF ELECTRICAL AND ELECTRONICS ENGINEERING

ANADOLU UNIVERSITY DEPARTMENT OF ELECTRICAL AND ELECTRONICS ENGINEERING EEM 102 INTRODUCTION TO ELECTRICAL ENGINEERING EXPERIMENT 9: DIODES AND DC POWER SUPPLY OBJECTIVE: To observe how a diode functions

### Lab 3 Ohm s Law and Resistors

` Lab 3 Ohm s Law and Resistors What You Need To Know: The Physics One of the things that students have a difficult time with when they first learn about circuits is the electronics lingo. The lingo and

### = V peak 2 = 0.707V peak

BASIC ELECTRONICS - RECTIFICATION AND FILTERING PURPOSE Suppose that you wanted to build a simple DC electronic power supply, which operated off of an AC input (e.g., something you might plug into a standard

### Experiment 3: Magnetic Fields of a Bar Magnet and Helmholtz Coil

MASSACHUSETTS INSTITUTE OF TECHNOLOGY Department of Physics 8.02 Spring 2006 Experiment 3: Magnetic Fields of a Bar Magnet and Helmholtz Coil OBJECTIVES 1. To learn how to visualize magnetic field lines

### Capacitors. We charge a capacitor by connecting the two plates to a potential difference, such as a battery:

RC Circuits PHYS 1112L Capacitors A capacitor is an electrical component that stores charge. The simplest capacitor is just two charged metal plates separated by a non-conducting material: In the diagram

### Lab 5 Operational Amplifiers

Lab 5 Operational Amplifiers By: Gary A. Ybarra Christopher E. Cramer Duke University Department of Electrical and Computer Engineering Durham, NC. Purpose The purpose of this lab is to examine the properties

### University of Alberta Department of Electrical and Computer Engineering. EE 250 Laboratory Experiment #5 Diodes

University of Alberta Department of Electrical and Computer Engineering EE 250 Laboratory Experiment #5 Diodes Objective: To introduce basic diode concepts. Introduction: The diode is the most fundamental

### Digital Fundamentals

Digital Fundamentals Tenth Edition Floyd Chapter 1 2009 Pearson Education, Upper 2008 Pearson Saddle River, Education NJ 07458. All Rights Reserved Analog Quantities Most natural quantities that we see

### CIRCUITS LABORATORY EXPERIMENT 9. Operational Amplifiers

CIRCUITS LABORATORY EXPERIMENT 9 Operational Amplifiers 9.1 INTRODUCTION An operational amplifier ("op amp") is a direct-coupled, differential-input, highgain voltage amplifier, usually packaged in the

### PCM Encoding and Decoding:

PCM Encoding and Decoding: Aim: Introduction to PCM encoding and decoding. Introduction: PCM Encoding: The input to the PCM ENCODER module is an analog message. This must be constrained to a defined bandwidth

### EE 242 EXPERIMENT 5: COMPUTER SIMULATION OF THREE-PHASE CIRCUITS USING PSPICE SCHEMATICS 1

EE 242 EXPERIMENT 5: COMPUTER SIMULATION OF THREE-PHASE CIRCUITS USING PSPICE SCHEMATICS 1 Objective: To build, simulate, and analyze three-phase circuits using OrCAD Capture Pspice Schematics under balanced

### Physics 42 Lab 4 Fall 2012 Cathode Ray Tube (CRT)

Physics 42 Lab 4 Fall 202 Cathode Ray Tube (CRT) PRE-LAB Read the background information in the lab below and then derive this formula for the deflection. D = LPV defl 2 SV accel () Redraw the diagram

### Capacitors & RC Circuits

Capacitors & C Circuits Name: EQUIPMENT NEEDED: Circuits Experiment Board One D-cell Battery Wire leads Multimeter Capacitors(100 F, 330 F) esistors(1k, 4.7k ) Logger Pro Software, ULI Purpose The purpose

### The George Washington University School of Engineering and Applied Science Department of Electrical and Computer Engineering

The George Washington University School of Engineering and Applied Science Department of Electrical and Computer Engineering Final Design Report Dual Channel Stereo Amplifier By: Kristen Gunia Prepared

### Lab 3: Introduction to Data Acquisition Cards

Lab 3: Introduction to Data Acquisition Cards INTRODUCTION: In this lab, you will be building a VI to display the input measured on a channel. However, within your own VI you will use LabVIEW supplied

### Calibration and Use of a Strain-Gage-Instrumented Beam: Density Determination and Weight-Flow-Rate Measurement

Chapter 2 Calibration and Use of a Strain-Gage-Instrumented Beam: Density Determination and Weight-Flow-Rate Measurement 2.1 Introduction and Objectives This laboratory exercise involves the static calibration

### Experiment #11: LRC Circuit (Power Amplifier, Voltage Sensor)

Experiment #11: LRC Circuit (Power Amplifier, Voltage Sensor) Concept: circuits Time: 30 m SW Interface: 750 Windows file: RLC.SWS EQUIPMENT NEEDED Science Workshop Interface Power Amplifier (2) Voltage

### University of Pennsylvania Department of Electrical and Systems Engineering Digital Audio Basics

University of Pennsylvania Department of Electrical and Systems Engineering Digital Audio Basics ESE250 Spring 2013 Lab 2: Sampling Friday, January 18, 2013 For Lab Session: Thursday, January 24, 2013

### LAB 7 MOSFET CHARACTERISTICS AND APPLICATIONS

LAB 7 MOSFET CHARACTERISTICS AND APPLICATIONS Objective In this experiment you will study the i-v characteristics of an MOS transistor. You will use the MOSFET as a variable resistor and as a switch. BACKGROUND