# Experiment 5. Strain Gage Measurements

 To view this video please enable JavaScript, and consider upgrading to a web browser that supports HTML5 video
Save this PDF as:

Size: px
Start display at page:

## Transcription

1 Experiment 5 Strain Gage Measurements Introduction The design of components for many engineering systems is based on the application of theoretical models. The accuracy of these models can be verified through testing. One way of validating a math model is to use a strain gage measurement system to measure the strain at the surface of a structure. The sensor used to transfer a mechanical strain to a quantifiable output is the strain gage. The strain gage has a resistance which changes as a function of mechanical strain. Although the change in resistance can be quantified using an ohm meter, signal conditioning is used to convert the change in resistance to a voltage. The voltage may then be amplified, thus reducing quantization errors. The relationship between the input strain and the output voltage may be determined using the system sensitivity. The objective of this assignment is to experimentally determine the system sensitivity and compare it to the ideal sensitivity of a strain gage measurement system. The actual system sensitivity will then be used to determine the modulus of elasticity of a cantilever beam. Pre-Lab Assignment 1. Determine the ideal system sensitivity of a strain gage system expressed in millivolts per microstrain, given the following: Sg = 2.085, Ei = 2 Vdc, G = Also, determine the ideal slope of the calibration equation in microstrain per millivolt. 2. Calculate the deflection required to produce 100, 200, 300, 400, 500, 600, 700, 800, and 900 microstrain at a strain gage mounted to a cantilever beam. The geometry of the beam and location of the gage is as follows: b = 1.0" t = 0.125" L = 10.0" x = 9.0" (beam width) (beam thickness) (length of cantilever beam) (distance from the strain gage to the micrometer) 3. Determine ten different beam loading values that will be used in lab to end load a cantilever beam using a platform and weights. Load values should increase by 100 gram intervals with an initial load of approximately 173 grams. Calculate the stress levels generated at the strain gage location for each load. The weights available are as follows: gram platform gram weight gram weights1-500 gram weight 4. Read Section 8.1 and 8.6 in text. ME Strain Gage Measurement Lab 1 Rev

2 Assignment 1: Equipment Configuration & Initialization Prior to calibrating the strain gage measurement system, the signal conditioning module must be connected to the strain gage and initialized. The four major components of the 2311 Signal Conditioner Amplifier are the variable DC power supply, wheatstone bridge, low pass filter, and amplifier. The power supply is used to excite the wheatstone bridge which converts a change in the strain gage resistance to a corresponding change in voltage. The amplifier is used to amplify the output signal from the wheatstone bridge reducing possible quantization errors. The lowpass filter is used to filter out undesired frequencies associated with background noise. Procedure 1. Install the cantilever beam into the Flexor. The beam should be placed with the strain gage on the upper surface closest to the clamp. Adjust the free end of the beam so that motion is not impeded by the sides of the Flexor. 2. Connect the lead wires of the strain gage to the binding posts of the Flexor as shown in figure 1. Connect leads numbered 1, 2, and 3 of the Flexor to the input cable of the 2311 Signal Conditioning Amplifier at points "S-", "D-120", and "P+" respectively. 3. Measure and record the following cantilever beam dimensions: x - distance from the center of the strain gage to the micrometer load point. L - distance from the clamped end of the beam to the micrometer load point. b - beam width t - beam thickness Figure 1 - Strain Gage System Initialization 4. Compare the measured values with dimensions used in the pre-lab assignment. Adjust the length of the cantilever beam so that it coincides with the pre-lab value used. If the values of x, b, and t differ, pre-lab calculations must be recalculated. ME Strain Gage Measurement Lab 2 Rev

3 5. Check that the power to the 2311 Signal Conditioner is shut off. Plug the power cord from the bottom rear of the 2311 into the bench power strip. 6. Connect the Bendix connector attached to the wire junction to the rear of the The +/- 10 volt BNC output on the top rear of the 2311 should be connected to the oscilloscope and digital multimeter. 7. Set the LOW PASS FILTER to 10 Hz. Shut the EXCITATION switch to OFF. 8. Turn the 2311 on using the POWER button at the bottom of the module. 9. Balance the power amplifier by setting the gain to X100. Check that the dial to the left of the gain buttons is set to 1. Adjustment will result in a fractional value of the set gain. 10. Examine the two lights at the top of the front panel. If either is lit, turn the AMP BAL adjustment screw located below the EXCITATION toggle switch so that both lamps are fully extinguished. 11. Balance the internal wheatstone bridge by first setting the EXCITATION voltage to 2 Vdc. Turn the EXCITATION switch to ON. One of the two output lamps at the top of the module should light indicating that the wheatstone bridge is out of balance. Press the AUTO BAL switch to the RESET position for a second then release. After a few seconds the lamps should extinguish indicating the bridge is balanced. 12. Set the amplifier GAIN to X1000. If either of the output lamps are lit, turn the TRIM knob until the lamps are extinguished. The multimeter should be reading zero volts. The strain gage system is now initialized and ready to make measurements. 13. Manually load the beam. Note the change in output voltage on the multimeter as well as the two output lamps on the front of the As the strain gage is mechanically strained, the change in resistance is sensed by the wheatstone bridge and a resultant output voltage is generated. Figure 2 shows the major components of the 2311 Signal Conditioning Amplifier used to convert the input resistance to an output voltage. ME Strain Gage Measurement Lab 3 Rev

4 Figure Signal Conditioning Amplifier Assignment 2: Determine the System Sensitivity The system sensitivity of the strain gage measurement system can be determined by displacing the free end of the cantilever beam a known distance and recording the output voltage generated by the signal conditioning module. The strain induced in the beam at the strain gage location is determined using the following equation: where is the strain at the surface of the beam at the gage location, is the deflection at the free end of the beam, is the beam's thickness, is the distance from the strain gage to the point where the load is applied, and is the cantilever beam length. Procedure 1. With the wheatstone bridge balanced, an output voltage of 0 Volts DC should be displayed on the multimeter. Adjust the TRIM knob if necessary. 2. Adjust the micrometer on the Flexor so that the tip is just touching the cantilever beam. The multimeter and output lamps can be used to determine when contact occurs. Record the reading on the micrometer and the multimeter as the initial displacement and corresponding output voltage for the undeflected beam. 3. Using the micrometer, deflect the beam an amount that will produce a strain of 100 microstrain at the strain gage. Record the ideal strain level, the micrometer setting, and the output voltage indicated by the multimeter. 4. Deflect the beam an amount that will produce strain levels of 200, 300, 400, 500, 600, 700, 800, and 900 microstrain at the strain gage. Record the ideal strain level, beam deflection, and output voltage from the system. 5. Unload the beam and record the final output voltage. The output should return to ME Strain Gage Measurement Lab 4 Rev

5 the initial value with no load on the beam. If the two values do not coincide, contact the instructor. Assignment 3: Young's Modulus Determination The Modulus of Elasticity of the cantilever beam can be determined by hanging known weights from the free end of the beam and recording the corresponding output voltage from the 2311 Signal Conditioning Amplifier. The flexure formula can be used to calculate the stress level at the gage, while the strain gage system sensitivity is used to quantify the strain. Plotting stress and strain data and performing a regression analysis on the linear elastic range will result in the determination of the Modulus of Elasticity of the beam. Procedure 1. Initialize the 2311 Signal Conditioning Amplifier. 2. After weighing the platform, hang the hook and platform from the end of the cantilever beam as seen in figure #3. Record the platform weight and output voltage from the strain gage system. 3. Add the weight necessary to produce the 2 nd stress level calculated in the pre-lab assignment question #3. Record the weight including the weight of the platform and the corresponding output voltage. Repeat this process for a total of ten loading values calculated in the pre-lab. 4. Unload the beam at the same increments and record the output voltage to verify that significant hysteresis errors have not occurred. Figure 3 - Young's Modulus Determination ME Strain Gage Measurement Lab 5 Rev

6 Post-lab Analysis 1. Determine the strain gage system sensitivity. Compare the actual system sensitivity to the ideal value determined from the pre-lab assignment. Discuss possible sources of error. 2. Plot the strain produced at the strain gage and the corresponding output voltage from the 2311 Signal Conditioning Amplifier. Properly distinguish experimental data from curve fit data. 3. Using the calibrated system sensitivity, determine the strain produced at the strain gage in assignment #3. 4. Compare the ten values of strain obtained while loading the beam to those obtained during unloading. Discuss why differences may have occurred. Are these differences acceptable? 5. Calculate the stress levels generated from end loading the cantilever beam in assignment #3. 6. Plot the stress vs. strain from assignment #3. Perform a regression analysis on the data to determine Young's Modulus. 7. What type of material is the beam made of? Support conclusions with theoretical and experimental proof as well as listing any publications referenced. 8. Explain to methods of increasing the strain gage system sensitivity. Why would the strain gage measurement system sensitivity be increased. ME Strain Gage Measurement Lab 6 Rev

7 Strain Gage Experiment #5 for ME Lab I (scanned document) 8 Department of Mechanical Engineering

8 Strain Gage Experiment #5 for ME Lab I (scanned document) 9 Department of Mechanical Engineering

9 Strain Gage Experiment #5 for ME Lab I (scanned document) 10 Department of Mechanical Engineering

10 Strain Gage Experiment #5 for ME Lab I (scanned document) 11 Department of Mechanical Engineering

11 Strain Gage Experiment #5 for ME Lab I (scanned document) 12 Department of Mechanical Engineering

12 Strain Gage Experiment #5 for ME Lab I (scanned document) 13 Department of Mechanical Engineering

13 Strain Gage Experiment #5 for ME Lab I (scanned document) 14 Department of Mechanical Engineering

14 Strain Gage Experiment #5 for ME Lab I (scanned document) 15 Department of Mechanical Engineering

15 Strain Gage Experiment #5 for ME Lab I (scanned document) 16 Department of Mechanical Engineering

16 Strain Gage Experiment #5 for ME Lab I (scanned document) 17 Department of Mechanical Engineering

17 Strain Gage Experiment #5 for ME Lab I (scanned document) 18 Department of Mechanical Engineering

18 Strain Gage Experiment #5 for ME Lab I (scanned document) 19 Department of Mechanical Engineering

19 Strain Gage Experiment #5 for ME Lab I (scanned document) 20 Department of Mechanical Engineering

20 Strain Gage Experiment #5 for ME Lab I (scanned document) 21 Department of Mechanical Engineering

21 Strain Gage Experiment #5 for ME Lab I (scanned document) 22 Department of Mechanical Engineering

22 Strain Gage Experiment #5 for ME Lab I (scanned document) 23 Department of Mechanical Engineering

23 Strain Gage Experiment #5 for ME Lab I (scanned document) 24 Department of Mechanical Engineering

24 Strain Gage Experiment #5 for ME Lab I (scanned document) 25 Department of Mechanical Engineering

25 Strain Gage Experiment #5 for ME Lab I (scanned document) 26 Department of Mechanical Engineering

26 Strain Gage Experiment #5 for ME Lab I (scanned document) 27 Department of Mechanical Engineering

27 Strain Gage Experiment #5 for ME Lab I (scanned document) 28 Department of Mechanical Engineering

28 Strain Gage Experiment #5 for ME Lab I (scanned document) 29 Department of Mechanical Engineering

29 Strain Gage Experiment #5 for ME Lab I (scanned document) 30 Department of Mechanical Engineering

30 Strain Gage Experiment #5 for ME Lab I (scanned document) 31 Department of Mechanical Engineering

31 Strain Gage Experiment #5 for ME Lab I (scanned document) 32 Department of Mechanical Engineering

32 Strain Gage Experiment #5 for ME Lab I (scanned document) 33 Department of Mechanical Engineering

33 Strain Gage Experiment #5 for ME Lab I (scanned document) 34 Department of Mechanical Engineering

### Calibration and Use of a Strain-Gage-Instrumented Beam: Density Determination and Weight-Flow-Rate Measurement

Chapter 2 Calibration and Use of a Strain-Gage-Instrumented Beam: Density Determination and Weight-Flow-Rate Measurement 2.1 Introduction and Objectives This laboratory exercise involves the static calibration

### MENG 302L Lab 4: Modulus of Elasticity and Poisson s Ratio

MENG 302L Lab 4: Modulus of Elasticity and Poisson s Ratio Introduction: In Lab 4 we will measure the two fundamental elastic constants relating stress to strain: Modulus of Elasticity and Poisson s Ratio.

### Resistivity of a Superconductor

Resistivity of a Superconductor PURPOSE: To investigate the resistive properties of a superconductor as a function of temperature. To gain experience doing measurements at cryogenic temperatures using

### UNIVERSITY OF NORTH CAROLINA AT CHARLOTTE. Department of Electrical and Computer Engineering

UNIVERSITY OF NORTH CAROLINA AT CHARLOTTE Department of Electrical and Computer Engineering Experiment No. 6 Precision Resistance Measurements Introduction: It is sometimes necessary to make resistance

### Lab E1: Introduction to Circuits

E1.1 Lab E1: Introduction to Circuits The purpose of the this lab is to introduce you to some basic instrumentation used in electrical circuits. You will learn to use a DC power supply, a digital multimeter

### Strain gauge lab Part 1

Strain gauge lab Part 1 In this lab, you will make measurements of mechanical strain in small aluminum beams as you bend them. We will continue this lab next week, so you will want to be careful to maintain

### The Electronic Scale

The Electronic Scale Learning Objectives By the end of this laboratory experiment, the experimenter should be able to: Explain what an operational amplifier is and how it can be used in amplifying signal

### THE STRAIN GAGE PRESSURE TRANSDUCER

THE STRAIN GAGE PRESSURE TRANSDUCER Pressure transducers use a variety of sensing devices to provide an electrical output proportional to applied pressure. The sensing device employed in the transducers

### EECS 100/43 Lab 2 Function Generator and Oscilloscope

1. Objective EECS 100/43 Lab 2 Function Generator and Oscilloscope In this lab you learn how to use the oscilloscope and function generator 2. Equipment a. Breadboard b. Wire cutters c. Wires d. Oscilloscope

### The RC Circuit. Pre-lab questions. Introduction. The RC Circuit

The RC Circuit Pre-lab questions 1. What is the meaning of the time constant, RC? 2. Show that RC has units of time. 3. Why isn t the time constant defined to be the time it takes the capacitor to become

### THERMAL ANEMOMETRY ELECTRONICS, SOFTWARE AND ACCESSORIES

TSI and TSI logo are registered trademarks of TSI Incorporated. SmartTune is a trademark of TSI Incorporated. THERMAL ANEMOMETRY ELECTRONICS, SOFTWARE AND ACCESSORIES IFA 300 Constant Temperature Anemometry

### MCE380: Measurements and Instrumentation Lab. Chapter 9: Force, Torque and Strain Measurements

MCE380: Measurements and Instrumentation Lab Chapter 9: Force, Torque and Strain Measurements Topics: Elastic Elements for Force Measurement Dynamometers and Brakes Resistance Strain Gages Holman, Ch.

### PHYS-2212 LAB Ohm s Law and Measurement of Resistance

Objectives PHYS-2212 LAB Ohm s Law and Measurement of Resistance Part I: Comparing the relationship between electric current and potential difference (voltage) across an ohmic resistor with the voltage-current

### Inductors in AC Circuits

Inductors in AC Circuits Name Section Resistors, inductors, and capacitors all have the effect of modifying the size of the current in an AC circuit and the time at which the current reaches its maximum

### Overview: The purpose of this experiment is to introduce diode rectifier circuits used in DC power supplies.

UNIVERSITY OF NORTH CAROLINA AT CHARLOTTE Department of Electrical and Computer Engineering Experiment No. 3 Diodes and Bridge Rectifiers Overview: The purpose of this experiment is to introduce diode

### Digital to Analog Conversion Using Pulse Width Modulation

Digital to Analog Conversion Using Pulse Width Modulation Samer El-Haj-Mahmoud Electronics Engineering Technology Program Texas A&M University Instructor s Portion Summary The purpose of this lab is to

### RLC Series Resonance

RLC Series Resonance 11EM Object: The purpose of this laboratory activity is to study resonance in a resistor-inductor-capacitor (RLC) circuit by examining the current through the circuit as a function

### Chapter 5: Tool Dynamometers

Chapter 5: Tool Dynamometers LEARNING OBJECTIVES Different types of transducers used in Dynamometers Design Requirements Types of Dynamometers ---------------------------------------------------------------------------------------------------------------------

### R A _ + Figure 2: DC circuit to verify Ohm s Law. R is the resistor, A is the Ammeter, and V is the Voltmeter. A R _ +

Physics 221 Experiment 3: Simple DC Circuits and Resistors October 1, 2008 ntroduction n this experiment, we will investigate Ohm s Law, and study how resistors behave in various combinations. Along the

### Environmental Monitoring with Sensors: Hands-on Exercise

Environmental Monitoring with Sensors: Hands-on Exercise Now that you ve seen a few types of sensors, along with some circuits that can be developed to condition their responses, let s spend a bit of time

### RESONANCE AND FILTERS

1422-1 RESONANCE AND FILTERS Experiment 1, Resonant Frequency and Circuit Impedance For more courses visit www.cie-wc.edu OBJECTIVES 1. To verify experimentally our theoretical predictions concerning the

### Experiment 10 Inductors in AC Circuits

Experiment 1 Inductors in AC Circuits Preparation Prepare for this week's experiment by looking up inductors, self inductance, enz's aw, inductive reactance, and R circuits Principles An inductor is made

LABORATORY EXPERIMENT Infrared Transmitter/Receiver (Note to Teaching Assistant: The week before this experiment is performed, place students into groups of two and assign each group a specific frequency

### Physics 2306 Experiment 7: Time-dependent Circuits, Part 1

Name ID number Date Lab CRN Lab partner Lab instructor Objectives Physics 2306 Experiment 7: Time-dependent Circuits, Part 1 To study the time dependent behavior of the voltage and current in circuits

### Oscilloscope, Function Generator, and Voltage Division

1. Introduction Oscilloscope, Function Generator, and Voltage Division In this lab the student will learn to use the oscilloscope and function generator. The student will also verify the concept of voltage

### Evaluation copy. Ohm s Law. Computer

Ohm s Law Computer 22 The fundamental relationship among the three important electrical quantities current, voltage, and resistance was discovered by Georg Simon Ohm. The relationship and the unit of electrical

### APC International, Ltd. Piezo d 33 Test System

APC International, Ltd. Piezo d 33 Test System APC International, Ltd APC Part number: 90-2030 Print in USA APC International, Ltd. All rights reserved 1. DESCRIPTIONS The part number 90-2030 (d 33 meter)

### ELEC 2020 EXPERIMENT 6 Zener Diodes and LED's

ELEC 2020 EXPERIMENT 6 Zener Diodes and LED's Objectives: The experiments in this laboratory exercise will provide an introduction to diodes. You will use the Bit Bucket breadboarding system to build and

### Preview of Period 12: Electric Circuits

Preview of Period 2: Electric Circuits 2. Voltage, Current, and esistance How are voltage, current, and resistance related? 2.2 esistance and Voltage of esistors in Connected in Series How does current

### Section 6 - Electronics

Section 6 - Electronics 6.1. Power for Excitation Piezoresistive transducers are passive devices and require an external power supply to provide the necessary current (I x ) or voltage excitation (E x

### Wilcoxon Research PR710 Series Signal conditioner operating guide

Wilcoxon Research PR710 Series Signal conditioner operating guide Caution: This manual should be read carefully before operating any of the PR710 series signal conditioners. Meggitt Sensing Systems 20511

### Electrical Resonance

Electrical Resonance (R-L-C series circuit) APPARATUS 1. R-L-C Circuit board 2. Signal generator 3. Oscilloscope Tektronix TDS1002 with two sets of leads (see Introduction to the Oscilloscope ) INTRODUCTION

### RC Circuits and The Oscilloscope Physics Lab X

Objective RC Circuits and The Oscilloscope Physics Lab X In this series of experiments, the time constant of an RC circuit will be measured experimentally and compared with the theoretical expression for

### Using a Multimeter to Analyze a Circuit: Measuring Current and Voltage Calculating Power and Resistance

Name: Date: Using a Multimeter to Analyze a Circuit: Measuring Current and Voltage Calculating Power and Resistance Background Information and Pre-Lab Activity Materials: One solar module One small DC

### EE 320L Electronics I Laboratory. Laboratory Exercise #4 Diode and Power Supply Circuit

EE 320L Electronics I Laboratory Laboratory Exercise #4 Diode and Power Supply Circuit Department of Electrical and Computer Engineering University of Nevada, at Las Vegas Objective: The purpose of this

### Electric Circuits II. Physics 133 Experiments Electric Circuits II 1

Physics 133 Experiments Electric Circuits II 1 Electric Circuits II GOALS To examine Ohm's Law: the pivotal relationship between voltage and current for resistors To closely study what current does when

### Resistors in Series and Parallel

Resistors in Series and Parallel INTRODUCTION Direct current (DC) circuits are characterized by the quantities current, voltage and resistance. Current is the rate of flow of charge. The SI unit is the

### Ohm s Law. Electrical Quantity Description Unit Water Analogy Voltage or Potential Difference

Ohm s Law Experiment 25 The fundamental relationship among the three important electrical quantities current, voltage, and resistance was discovered by Georg Simon Ohm. The relationship and the unit of

### EE 311: Electrical Engineering Junior Lab Active Filter Design (Sallen-Key Filter)

EE 311: Electrical Engineering Junior Lab Active Filter Design (Sallen-Key Filter) Objective The purpose of this experiment is to design a set of second-order Sallen-Key active filters and to investigate

### Lab 4: A Variety of Sensors for the Measurement of Temperature, Strain, Pressure, and Displacement

Chapter 5 Lab 4: A Variety of Sensors for the Measurement of Temperature, Strain, Pressure, and Displacement Objectives: Understand and build Wheatstone Bridges Use strain gauges to measure the deformation

### Filters and Waveform Shaping

Physics 333 Experiment #3 Fall 211 Filters and Waveform Shaping Purpose The aim of this experiment is to study the frequency filtering properties of passive (R, C, and L) circuits for sine waves, and the

### Loop Calibration and Maintenance

Loop Calibration and Maintenance Application Note Introduction Process instrumentation requires periodic calibration and maintenance to ensure that it is operating correctly. This application note contains

### Isolated AC Sine Wave Input 3B42 / 3B43 / 3B44 FEATURES APPLICATIONS PRODUCT OVERVIEW FUNCTIONAL BLOCK DIAGRAM

Isolated AC Sine Wave Input 3B42 / 3B43 / 3B44 FEATURES AC averaging technique used to rectify, amplify, and filter 50 Hz to 400 Hz sine-wave signals. Accepts inputs of between 20 mv to 550 V rms to give

### Experiment-5 Study of I-V Characteristics of Gunn Diodes

OBJECTIVES EC 341 Microwave Laboratory Experiment-5 Study of I-V Characteristics of Gunn Diodes 1. V-I Characteristics 2. Output Power and frequency as a function of voltage. 3. Square wave modulation

### PIEZO FILM LAB AMPLIFIER

SPECIFICATIONS Charge or Voltage Mode Operation BNC Input and Output 0.01 to 1000 mv/pc Sensitivity Range in Charge Mode 1M to 1G Input Resistance, -40 to 40dB Gain in Voltage Mode Multi-Pole, Low-Pass

### LABORATORY 1 WRITEUP - PHYSICS 517/617. Prof. L. S. Durkin July 5, 1992

LABORATORY 1 WRITEUP - PHYSICS 517/617 Prof. L. S. Durkin July 5, 1992 DISCLAIMER: There are many ways to write up a lab report, none of them superior to any other. Below is an example of an acceptable

### MODEL 2202IQ (1991-MSRP \$549.00)

F O R T H E L O V E O F M U S I C F O R T H E L O V E O F M U S I C MODEL 2202IQ (1991-MSRP \$549.00) OWNER'S MANUAL AND INSTALLATION GUIDE INTRODUCTION Congratulations on your decision to purchase a LINEAR

### LAB 4 DIGITAL FILTERING

LAB 4 DIGITAL FILTERING 1. LAB OBJECTIVE The purpose of this lab is to design and implement digital filters. You will design and implement a first order IIR (Infinite Impulse Response) filter and a second

### MAKE SURE TA & TI STAMPS EVERY PAGE BEFORE YOU START

Laboratory Section: Last Revised on December 15, 2014 Partners Names Grade EXPERIMENT 10 Electronic Circuits 0. Pre-Laboratory Work [2 pts] 1. How are you going to determine the capacitance of the unknown

### Experiment P19: Simple Harmonic Motion - Mass on a Spring (Force Sensor, Motion Sensor)

PASCO scientific Physics Lab Manual: P19-1 Science Workshop S. H. M. Mass on a Spring Experiment P19: Simple Harmonic Motion - Mass on a Spring (Force Sensor, Motion Sensor) Concept Time SW Interface Macintosh

### Measuring Electric Phenomena: the Ammeter and Voltmeter

Measuring Electric Phenomena: the Ammeter and Voltmeter 1 Objectives 1. To understand the use and operation of the Ammeter and Voltmeter in a simple direct current circuit, and 2. To verify Ohm s Law for

### LAB ELEC 25.CALC From Physics with Calculators, Vernier Software & Technology, 2003

OHM S LAW LAB ELEC 25.CALC From Physics with Calculators, Vernier Software & Technology, 2003 INTRODUCTION The fundamental relationship among the three important electrical quantities current, voltage,

### COURSE NOTES Techniques for Measuring Drain Voltage and Current

COURSE NOTES Techniques for Measuring Drain Voltage and Current Introduction These course notes are to be read in association with the PI University video course, Techniques for Measuring Drain Voltage

### ε: Voltage output of Signal Generator (also called the Source voltage or Applied

Experiment #10: LR & RC Circuits Frequency Response EQUIPMENT NEEDED Science Workshop Interface Power Amplifier (2) Voltage Sensor graph paper (optional) (3) Patch Cords Decade resistor, capacitor, and

### PHYS 130 Laboratory Experiment 11 Hooke s Law & Simple Harmonic Motion

PHYS 130 Laboratory Experiment 11 Hooke s Law & Simple Harmonic Motion NAME: DATE: SECTION: PARTNERS: OBJECTIVES 1. Verify Hooke s Law and use it to measure the force constant of a spring. 2. Investigate

### Experiment #11: LRC Circuit (Power Amplifier, Voltage Sensor)

Experiment #11: LRC Circuit (Power Amplifier, Voltage Sensor) Concept: circuits Time: 30 m SW Interface: 750 Windows file: RLC.SWS EQUIPMENT NEEDED Science Workshop Interface Power Amplifier (2) Voltage

### PHYSICS 176 UNIVERSITY PHYSICS LAB II. Experiment 4. Alternating Current Measurement

PHYSICS 176 UNIVERSITY PHYSICS LAB II Experiment 4 Alternating Current Measurement Equipment: Supplies: Oscilloscope, Function Generator. Filament Transformer. A sine wave A.C. signal has three basic properties:

### Basic DC Circuits. Electrical Quantity Description Unit Water Analogy Voltage or Potential Difference

Basic DC Circuits Current and voltage can be difficult to understand, because the flow of electrons and potential differences cannot be observed by the unaided human eye. To clarify these terms, some people

### LRC Circuits. Purpose. Principles PHYS 2211L LAB 7

Purpose This experiment is an introduction to alternating current (AC) circuits. Using the oscilloscope, we will examine the voltage response of inductors, resistors and capacitors in series circuits driven

### Product Data. A/RH Series Relative Humidity Transmitters

Relative Humidity DISPLAYS A CI Product Data A/RH Series Relative Humidity Transmitters First in Sensor Quality The A/RH Series Relative Humidity transmitters, convert a resistance change to a linear to

### EXPERIMENT 7 OHM S LAW, RESISTORS IN SERIES AND PARALLEL

260 7- I. THEOY EXPEIMENT 7 OHM S LAW, ESISTOS IN SEIES AND PAALLEL The purposes of this experiment are to test Ohm's Law, to study resistors in series and parallel, and to learn the correct use of ammeters

### Frequency Response of Filters

School of Engineering Department of Electrical and Computer Engineering 332:224 Principles of Electrical Engineering II Laboratory Experiment 2 Frequency Response of Filters 1 Introduction Objectives To

### Lab 9 Kirchhoff s Laws and Wheatstone Bridge

Lab 9 Kirchhoff s Laws and Wheatstone Bridge Learning Goals to explore Kirchhoff s two laws of electrical circuits to use a voltage sensor, current sensor, and the DataStudio software to measure the voltage

### International Journal of Engineering Research-Online A Peer Reviewed International Journal Articles available online http://www.ijoer.

RESEARCH ARTICLE ISSN: 2321-7758 DESIGN AND DEVELOPMENT OF A DYNAMOMETER FOR MEASURING THRUST AND TORQUE IN DRILLING APPLICATION SREEJITH C 1,MANU RAJ K R 2 1 PG Scholar, M.Tech Machine Design, Nehru College

### CIRCUIT DIAGRAMS AND COMPONENT DRAWINGS

CIRCUIT DIAGRAMS AND COMPONENT DRAWINGS Dr. Victor Giurgiutiu Page 70 1/17/01 BASIC MULTIMIETER OPERATION BASIC MULTIMETER INFORMATION Multimeter is a measuring instrument. It can be used to measure voltage,

### RESONANCE AND FILTERS

1422-1 RESONANCE AND FILTERS Experiment 4, Resonant Frequency and Impedance of a Parallel Circuit, For more courses visit www.cie-wc.edu OBJECTIVES: 1. To demonstrate the resonant frequency of a parallel

### P150A Experimental Lab #7 Ohm s Law (Ver A 10/12)

Ohm s Law The fundamental relationship among the three important electrical quantities current, voltage, and resistance was discovered by Georg Simon Ohm. The relationship and the unit of electrical resistance

### Physics 3330 Experiment #2 Fall 1999. DC techniques, dividers, and bridges R 2 =(1-S)R P R 1 =SR P. R P =10kΩ 10-turn pot.

Physics 3330 Experiment #2 Fall 1999 DC techniques, dividers, and bridges Purpose You will gain a familiarity with the circuit board and work with a variety of DC techniques, including voltage dividers,

### Piezoelectric Lab. Module 3: Dr. Yirong Lin. MECH 4395/5390 and IE 4395 Conducted: 7/29/14. Group 4. Jose Luis Coronel Jr. Italia Valles-Ochoa

Piezoelectric Lab Module 3: Dr. Yirong Lin MECH 4395/5390 and IE 4395 Conducted: 7/29/14 Group 4 Jose Luis Coronel Jr. Italia Valles-Ochoa Abstract The purpose of this lab report is to specify the process

### EXPERIMENT NUMBER 5 BASIC OSCILLOSCOPE OPERATIONS

1 EXPERIMENT NUMBER 5 BASIC OSCILLOSCOPE OPERATIONS The oscilloscope is the most versatile and most important tool in this lab and is probably the best tool an electrical engineer uses. This outline guides

### Faculty of Engineering and Information Technology. Lab 2 Diode Circuits

Faculty of Engineering and Information Technology Subject: 48521 Fundamentals of Electrical Engineering Assessment Number: 2 Assessment Title: Lab 2 Diode Circuits Tutorial Group: Students Name(s) and

### Bipolar Transistor Amplifiers

Physics 3330 Experiment #7 Fall 2005 Bipolar Transistor Amplifiers Purpose The aim of this experiment is to construct a bipolar transistor amplifier with a voltage gain of minus 25. The amplifier must

### Series and Parallel Circuits

Series and Parallel Circuits Ver. 1.2 In this experiment we will investigate the properties of several resistors connected in series and parallel. Our purpose is to verify the simple equations for the

### There are many different type of accelerometers and each has unique characteristics, advantages and disadvantages. The different types include:

How does a piezo-electric accelerometer work? Piezo-electric crystals are man-made or naturally occurring crystals that produce a charge output when they are compressed, flexed or subjected to shear forces.

### ME : Electronics and Instrumentation Lab 1: Basic Resistor Circuits and DC Power

ME 530.241: Electronics and Instrumentation Lab 1: Basic Resistor Circuits and DC Power Louis L. Whitcomb and Kyle B. Reed Department of Mechanical Engineering The Johns Hopkins University Spring 2009

### Preview of Period 13: Electrical Resistance and Joule Heating

Preview of Period 13: Electrical Resistance and Joule Heating 13.1 Electrical Resistance of a Wire What does the resistance of a wire depend upon? 13.2 Resistance and Joule Heating What effect does resistance

### AMPLIFIED HIGH SPEED FIBER PHOTODETECTOR USER S GUIDE

AMPLIFIED HIGH SPEED FIBER PHOTODETECTOR USER S GUIDE Thank you for purchasing your Amplified High Speed Fiber Photodetector. This user s guide will help answer any questions you may have regarding the

### User's Manual. Research Isometric Transducer 0 to 5 grams and 0 to 50 grams

Research Isometric Transducer 0 to 5 grams and 0 to 50 grams User's Manual Research Isometric Transducer 0 to 5 g and 0 to 50 g, 110 VAC/60 Hz MA1 72-4481 Research Isometric Transducer 0 to 5 g and 0 to

### Meters, Power Supplies and Generators

1. Meters Meters, Power Supplies and Generators Generally analog meters respond to the average of the signal being measured. This is due to the mechanical mass of the pointer and the RC response time of

### Table of Contents. The Basics of Electricity 2. Using a Digital Multimeter 4. Testing Voltage 8. Testing Current 10. Testing Resistance 12

Table of Contents The Basics of Electricity 2 Using a Digital Multimeter 4 IDEAL Digital Multimeters An Introduction The Basics of Digital Multimeters is designed to give you a fundamental knowledge of

### How Does it Flow? Electricity, Circuits, and Motors

How Does it Flow? Electricity, Circuits, and Motors Introduction In this lab, we will investigate the behavior of some direct current (DC) electrical circuits. These circuits are the same ones that move

### Mechanical Pressure Measurement David Gardellin, P.E.

Mechanical Pressure Measurement David Gardellin, P.E. Bourdon tube-type pressure sensors are the most common for industrial use in the family of elastic pressure elements. They have been in use for over

### Step Response of RC Circuits

Step Response of RC Circuits 1. OBJECTIVES...2 2. REFERENCE...2 3. CIRCUITS...2 4. COMPONENTS AND SPECIFICATIONS...3 QUANTITY...3 DESCRIPTION...3 COMMENTS...3 5. DISCUSSION...3 5.1 SOURCE RESISTANCE...3

### PHYS 2426 Engineering Physics II (Revised July 7, 2011) AC CIRCUITS: RLC SERIES CIRCUIT

PHYS 2426 Engineering Physics II (Revised July 7, 2011) AC CIRCUITS: RLC SERIES CIRCUIT INTRODUCTION The objective of this experiment is to study the behavior of an RLC series circuit subject to an AC

### FREQUENCY RESPONSE AND PASSIVE FILTERS LABORATORY. We start with examples of a few filter circuits to illustrate the concept.

FREQUENCY RESPONSE AND PASSIVE FILTERS LABORATORY In this experiment we will analytically determine and measure the frequency response of networks containing resistors, AC source/sources, and energy storage

### FUTEK USB Sensor Technology Learn how Analog and Digital Output can Seamlessly Integrate

FUTEK USB Sensor Technology Learn how Analog and Digital Output can Seamlessly Integrate Purpose The purpose of this presentation is to demonstrate the functionality of USB Sensor Technology Advantages

### Transistor Amplifiers

Physics 3330 Experiment #7 Fall 1999 Transistor Amplifiers Purpose The aim of this experiment is to develop a bipolar transistor amplifier with a voltage gain of minus 25. The amplifier must accept input

### G11 GSDR TX Calibration

G11 GSDR TX Calibration Step 1. Setting the Soundcard Output voltage. Step 2. Adjusting the Max Power Output. Step 3. Calibrate G11 Power/SWR measurement head. Step 4. Adjusting FWD Power meter in GSDR.

### PHYS245 Lab: Light bulb and resistor ΙΙ: Current voltage (I-V) curves

Purpose: PHYS245 Lab: Light bulb and resistor ΙΙ: Current voltage (I-V) curves Measure the current voltage curve of a light bulb and a resistor using a variable d.c. power supply. Understanding of Ohm

### The Time Constant of an RC Circuit

The Time Constant of an RC Circuit 1 Objectives 1. To determine the time constant of an RC Circuit, and 2. To determine the capacitance of an unknown capacitor. 2 Introduction What the heck is a capacitor?

### Contents. Document information

User Manual Contents Document information... 2 Introduction... 3 Warnings... 3 Manufacturer... 3 Description... Installation... Configuration... Troubleshooting...11 Technical data...12 Device Scope: PCB

### Instruction Manual. 2in1 LAN Tester & Multimeter. Model: LA-1011

Instruction Manual 2in1 LAN Tester & Multimeter Model: LA-1011 1 Contents Introduction... Features... Safety Precautions.. Meter Description... Electrical Specification... Operation.. AutoRanging Multimeter.

### CIRCUITS AND SYSTEMS LABORATORY EXERCISE 6 TRANSIENT STATES IN RLC CIRCUITS AT DC EXCITATION

CIRCUITS AND SYSTEMS LABORATORY EXERCISE 6 TRANSIENT STATES IN RLC CIRCUITS AT DC EXCITATION 1. DEVICES AND PANELS USED IN EXERCISE The following devices are to be used in this exercise: oscilloscope HP

### Teacher s Guide Physics Labs with Computers, Vol C P52: LRC Circuit. Teacher s Guide - Activity P52: LRC Circuit (Voltage Sensor)

Teacher s Guide Physics Labs with Computers, Vol. 2 012-06101C P52: LRC Circuit Teacher s Guide - Activity P52: LRC Circuit (Voltage Sensor) Concept DataStudio ScienceWorkshop (Mac) ScienceWorkshop (Win)