Meters, Power Supplies and Generators


 Loraine Cole
 1 years ago
 Views:
Transcription
1 1. Meters Meters, Power Supplies and Generators Generally analog meters respond to the average of the signal being measured. This is due to the mechanical mass of the pointer and the RC response time of the electronics. Since the average of most ac signals is zero, ac meters require more circuitry to measure values of time varying voltages. Many meters can function as both current and voltage meters. Usually there are separate inputs for the voltage and current functions. Trying to use the instrument as a voltmeter, when the leads are connected to the current input will either destroy the meter or blow a fuse. Voltage Measurements Voltage is analogous to pressure in a water pipe. And like pressure, it has to be measured relative to a reference. A voltage is measured across a circuit component, to indicate the voltage difference or voltage drop. The dashed lines in figure 1 indicate that all parts are contained within one physical unit. In this case, the resistance is a part of the meter and there is no way to separate it from the meter. An ideal voltmeter has a resistance of 4S. Figure 1. The Voltmeter To measure voltage electronically, a voltmeter uses a known, stable, resistor. All voltage meters draw current, because of that resistor. A typical digital volt meter (DVM) has a resistance of 10 MS while some older analog meters may be as low as 3.2 ks. Current Measurements Current is the rate change of charge and is, therefore, a measure of the number of electrons per second moving through a circuit. Since the current into and the current out of a lumped component are equal, you cannot measure a current drop across a component. To measure current you have to measure the actual charge flow through the wire. This measurement is done by inserting relatively small resistor in series with the circuit and measuring the voltage drop across the resistor. This is normally shown transformed into a series resistor and an ideal (resistance of 0S) current meter. The series model is shown in figure 2. In the typical digital multimeter (DMM), on a 2A range, the resistance is 100 ms. Because a current meter has resistance, it will cause a voltage drop and may affect the operation of the circuit. The actual amount of resistance depends upon the sensitivity of the meter. For 2009 E&CE Dept. University of Waterloo page1
2 example, the DMM on a 200 ua range has a resistance of 1 ks, while a nanoampere meter will be more than 100 ks. Figure 2. The Ammeter Resistance Measurements An ohmmeter applies Ohm s Law to determine resistarnce. Ohm s Law states the voltage = resistance x current. By applying a known voltage (produced by an internal battery or transformer) to a resistor and measuring the current flow, the resistance can be determined. Power Measurements A wattmeter measures the product of voltage and current to determine the power in watts. Measurement Errors This is a complex issue and we will only cover the basics. An error causes the measured value to be different from the actual value. Human, or gross, errors are usually due to mistakes in using or reading an instrument. Two examples are viewing an analog meter needle at an angle rather than straight on (parallax error), and reading the wrong meter scale. Other errors may be caused by the instruments themselves. Plots of data (figure 3) typically include error bars, which indicate the accuracy of the measurement, the effect of the meter loading upon the circuit, etc. Plots of theoretical calculations can also include a tolerance to account for the nonperfect circuit components (ie ±1%, ±5%, etc), temperature variations, etc. Figure 3. A Plot Illustrating Error Bars Accuracy The accuracy of a measurement is how close the value came to the actual value that exists, not the theoretical one. Instrument accuracy is usually broken down into systematic errors and random errors. The systematic errors are either instrument related (ie friction in a mechanical meter pointer bearing) or environmental. The method by which the meter works (mechanical, electrical,..), and the quality of design and construction, greatly affect how large these errors are. Quite often both of these are further broken down into static and dynamic errors. Dynamic errors would be caused by the meter not being able to respond fast enough. Precision Precision pertains to the number of significant figures and the repeatability of a measurement. High precision instruments provide more significant figures and have greater repeatability E&CE Dept. University of Waterloo page2
3 Resolution This is the smallest change in the quantity being measured that an instrument will indicate. For a digital meter, this is simply one of the smallest digits. With an analog meter, this is usually ½ of the smallest divisions on the meter face. Repeatability The repeatability is a measure of the fluctuations which are due to noise and errors that are not repeatable. When a measurement is repeated several times, and then statistically analysed, the repeatability can be expressed as the standard deviation. Uncertainty This is the combined effect of accuracy, precision, repeatability, noise, and errors on the measurement. It represents how uncertain the measurement is. 1.1 Calculating Meter Error Analog Meters The error for an analog meter is usually not explained in detail. However, most of the error is due to the mechanical meter movement. This error is constant, across a given meter range, regardless of the magnitude of the variable being measured. An analog meter stamped as being accurate to 2%, is accurate to 2% of the full scale deflection (FSD). For example, suppose a reading of 35.0 V (dc) was obtained on the 50 V range of a dc voltmeter, that is accurate to 2%. The error is: 2% * 50 V FSD = ±1 V Therefore V = 35.0 ± 1 V (dc) or expressed as a percent V = 35.0 V (dc) ± 2.9% Since the absolute error of ±1 V is constant for the range, the smaller the measured voltage is in relation to full scale, the larger the percent error becomes. If this meter range was used to measure a 9V battery the error would be ± 11%. For analog meters, it is more accurate to measure values with the needle close to full scale. Below is a table of the accuracy for various measurements made with the Simpson 260 meter. Measurement Accuracy: ± (% of full scale deflection) DC voltage ranges DC current (>50uA) ranges AC voltage ranges 2% 2% 3% Digital Multimeters The DMM is the most common instrument you will encounter. The precision of a DMM is usually expressed in digits. The most common model has 3 ½ digits. That means that you have three full digits that are capable of indicating from 0 to 9 and one ½ digit. The ½ means that a +1, blank ie. 0, or 1 can be displayed. There are also 4 ½ and 6 ½ digit meters as well as 3/4 digit meters. A 3/4 digit meter can display +2, +1, blank (0), 1 or 2 as the leading digit. A DMM has two sources of systematic error. One part of the error is due to tolerances in resistors. It is expressed as a percentage of the input and therefore varies with the magnitude of 2009 E&CE Dept. University of Waterloo page3
4 the voltage being measured. The second part of the error is due to the digital conversion process. It is fixed for a given meter range and is expressed in the number of digits to which the reading is accurate. Each digit, has the value of a single count of the rightmost digit on the display. For example, consider a reading of 1.345V on a 2 V dc range. The meter displays a maximum of V on this range. The smallest change on the meter display is V. Then the errors are, percent error: 0.1% * V = mv digit error: + 1 digit = 1 * V = 1 mv. Therefore, V = V ±2.345 mv or V = V ±0.17%. Now consider the same measurement on the 20 V range. The reading would have been 1.34 V and, percent error: 0.1% * 1.34 V = 1.34 mv digit error: + 1 digit = 1 * 0.01 V = 10 mv. Therefore, V = 1.34 V ±11.34 mv or V = 1.34 V ± 0.85%. This shows that the most accurate measurements are made on the lowest meter range that will accommodate the measured value. If the meter were to read V (rms) while measuring an 8 khz triangle wave on a V RMS scale; how large is the error? The meter measures True RMS and we can trust a reading done on a triangular wave. On the AC range the error is 0.5% + 2 digits. Then the errors are, percent error: 0.5% * V = 80.6 mv (rms) digit error: + 2 digits = 2 * 10 mv = 20 mv (rms) Therefore V = V RMS ± 101 mv RMS or V = V RMS ± 0.62%. Below is a table of the accuracy for various measurements made with the Fluke 8010A model. Measurement Accuracy: ± (% of reading + # least significant digits) DC DC current (up to 2A) AC, 45 Hz to 10 khz AC current (up to 2A range, 45 Hz to 2 khz) Resistance (up to 200 ks range) Resistance (2000 ks and 20 MS ranges) Meter Loading Every meter affects the circuit it is connected to by drawing current (voltage meter) or having a voltage drop (current meter). In later years you will learn more about other ways in which a meter affects a circuit (capacitance, inductance, adding noise,...). Analog Meters You can calculate the resistance between the terminals of a voltmeter, and determine if it will affect your circuit significantly. An analog voltmeter s resistance is usually specified as a sensitivity in ohms/volt. The resistance for a particular voltage range is determined by multiplying the sensitivity by the full scale voltage on that range. Some meters may give the 2009 E&CE Dept. University of Waterloo page4
5 total resistance while others give the current required for full scale deflection. As long as the meter draws much less current than the component across which it is connected, you can say that its effect will be small. A better technique is to calculate the error introduced, by the current that flows through the meter, and see if it is less than the accuracy to which you want to make a measurement. Ammeters are more complex. The standard current shunt (precision resistor in the meter) has a 50 mv drop at the maximum current rating. Since this voltage is usually small, an ammeter should not affect the circuit too much. However, an ammeter does affect a circuit by adding inductance. This will be explored more later. Remember that every time you change the range, the shunt resistance changes. Since the shunt resistance, in a normal meter, can vary from milliohms to kiloohms, using the wrong current range can easily affect your circuit. In comparison, a voltmeter s resistance will typically only vary from 100 ks to 10 MS. Digital Meters For all ac and dc voltage ranges, the DMM has a fixed input resistance which is typically 10MS,. On the current range, the meter will usually have a 200 mv drop at the maximum current, but the resistance is different for each range. Digital meters are usually more accurate, since they remove parallax and other mechanical errors. However, if you do not know how to select the range properly, it is very easy to obtain errors as large as any analog meter. This happens quite often on the current ranges. 1.3 Meter Waveform Shape and Frequency Dependence The more common meters measure the rectified average voltage, while better ones measure the root mean square (RMS). Unless you know what the shape of the waveform is, and understand how the meter measures a voltage, the number that you read will be useless. Each of the different methods gives the same voltage for a sinewave in a certain frequency range. However, they will all be different if given another waveshape. Ac meters are calibrated at one sinusoidal frequency to read the RMS voltage or current. However, most measure the rectified average of the ac component of a signal. The rectifier circuit acts like an absolute mathematical function. The ac component is what you have left when you remove the average or dc voltage from a signal. In real life, when you set a meter to read an ac signal you not only remove the dc signal but you also remove some of the low frequencies of a waveform (ie. the meter acts like a high pass filter). Meter bandwidth is the frequency range over which measurements are accurate. It more generally refers to the frequency range over which a circuit or instrument functions properly. For example, a Fluke 8010A digital meter should not be used below 45 Hz or above 50 khz. The error goes from 0.5% to 1.0% error when the frequency is over 10 khz, and 5.0% when the frequency is between 20 khz and 50 khz. Oscilloscopes and all other instruments also suffer from similar effects. Before you can use a meter, you must consider the Crest Factor of the waveform being measured. It is the ratio of the peak voltage to the RMS voltage. Mechanical meters are slow to respond because the needle, as a physical system with mass, takes time to accelerate. Electronic meters also have similar problems and this is responsible for the use of the Crest Factor. Some examples are given in the following table E&CE Dept. University of Waterloo page5
6 Waveform Square wave Sine wave Triangular wave White noise Crest Factor to 4.0 The Fluke 8010A meters only measure the correct rms value if the Crest Factor is under 3.0. So if, for a sinewave, you measure the rms voltage and measure the peak voltage on an oscilloscope (that can handle the waveform without introducing more errors), you can see if the meter is reading the rms value properly. The Crest Factor is more difficult to calculate for rectangular waveforms. A good instrumentation book or instrument manual will explain this. An rms reading meter measures the root of the mean of the square of the signal (proportional to the power) or: The purpose of an rms equivalent value of a waveform, is to simplify power calculations by providing an equivalent dc voltage, that would develop the same average power in a resistive load. If a voltage waveform is not described, but you know that it is equivalent to 10 V (rms), then you will have 100W of average power dissipated in a 1S load (ie.p AVERAGE = V RMS *V RMS /R). If you did not know the rms voltage, the only way to calculate the power would be to do an integration, from an approximation of the waveshape as seen on an oscilloscope. This only works for resistive loads because rms value is a magnitude and does not consider phase information. A meter that reads an average rectified value for a waveform of period T, indicates a value Therefore meter scales are calibrated to read 1.11 * V average, and since 1.11 * = 0.707, the reading is the rms value for a sinewave and a sinewave only. Notes: 1) As V PK is the average value of fullwave rectified sinusoids only, a rectifying averaging meter is only accurate for sinusoidal waveforms. The multiplier 1.11, known as the form factor, is specific to a pure sinusoidal waveform. Averaging ac meters are therefore, only accurate for sinusoidal waveforms. 2) The above discussion assumes that the meter uses a fullwave rectifier. A second type, called a halfwave rectifier, exists and would read a different voltage because it only responds to the positive parts of a waveform. Because of this, a halfwave rectifier can give very strange measurements if the waveform is not symmetrical about 0V. Here is a table of how some meters would respond to common signals of 1 V (peak) or 2 V (peaktopeak). The dc average value will usually be 0V for a sinewave since sinewave generators usually have no dc voltage while square or triangular wave usually start from 0V and go to some positive or negative voltage (single ended signal generator) E&CE Dept. University of Waterloo page6
7 Waveform DC average 1 V PK voltage AC rectified average voltage * 1.11 (A) RMS volts (B) % difference (A), (B) Square Sine Triangular ±1 or 0 0 ±0.5 or % +0% 3.8% 2. DC Power Supplies A dc power supply may have a fixed voltage output (usually accurate to ± 5%) or may be adjustable. Adjustable supplies have a control which allows one to change the dc voltage. Both types have two output leads where one is indicated as being positive (with respect to the other lead). The Goldstar GP105 power supply also has an adjustable current limit. This feature, shuts off the output voltage (and produces a beeping alarm until the power supply is turned off) when the current goes beyond a certain limit. To resume operation, the external circuit should be checked and/or the power supply s current limit needs to be raised (turning dial clockwise) before turning the power supply on again. Because of this, the power supply will act like a voltage source until the current limit is exceeded. The current limit control should be set before you perform the lab experiment. The best way to prevent circuit parts from exploding due to overheating is to limit the current to 100 ma. Most components will not explode, or be destroyed, with 100 ma the way that they would with 1.5A. Press the current limit button so that the present current limit is displayed on the power supply s meter. Adjust the current limiting knob to the maximum current you require. Power Supply Connections Most power supplies are floating. This means that neither the positive, nor negative outputs are connected to ground or any common point. This means that you can connect the power supply in any way, even several hundred volts above ground. A nonfloating power supply has one lead grounded or connected to a common point. This means that there is only one lead which can be connected anywhere in a circuit since the common or grounded one cannot be moved. These power supplies may be referred to as single ended. The word single means that you can only have a positive or negative voltage from the supply. If the supply is double ended, you can obtain either a positive or negative voltage from it. A Goldstar GP105 variable DC power supply is shown in figure 4 and its front panel controls are described below E&CE Dept. University of Waterloo page7
8 Î. POWER: Push on. Ï. LED power on indicator. Ð. Panel METER ZERO ADJUST. Ñ. METER SELECTION SWITCH V for voltage. A for current. CURRENT LIMIT switch: when pushed in the panel meter indicates the output currentlimiting value set. Ò. PANEL METER. Use as an indicator. No precision given for this meter. Ó. VOLTAGE CONTROL: Adjust VOLTAGE of the output at the + and  terminals. Ô. CURRENT CONTROL: Adjust CURRENT control to desired maximum output current allowable. Õ. OUTPUT TERMINALS: Connect the load across the + and  terminals. The symbol designates the common terminal which is connected to earth via the power cord. Figure 4. Power Supply Front Panel NOTE: The + or the  can be connected to the common. Hence you can have a positive common (or ground) or you can have a negative common (or ground) E&CE Dept. University of Waterloo page8
9 3. Signal Generators A signal generator provides a timevarying signal to test circuits. This can be anything from a simple sine, square or triangular wave up to a TV station signal with different types of noise added. This symbol and basic circuit model for a singleended (notice the grounded output) generator are shown in figure 5. Figure 5. Signal Generator R OUT is the output resistance and may sometimes be called R INT or internal resistance. Every electronic circuit has a certain amount of resistance. Signal generators are usually 600S, 75S, 50S or some other value due to varying industry standards. You have to keep the output resistance in mind. Whenever you draw current from a signal generator, the voltage at the output terminals will change due to the voltage drop across the internal resistance. Signal generators, like dc power supplies, can have different properties. They may have one of the outputs connected to ground or have a circuit common point. This is a singleended generator as opposed to a floating one, where neither of the outputs is connected to a common or ground. The generator may be unipolar, where the voltage is always of one polarity, or it may be bipolar, where the voltage is both positive and negative at different times. Generally, sinewave generators are bipolar and squarewave generators are unipolar. Unipolar generators are further broken down into positive and negative. A positive, unipolar, squarewave generator produces squarewaves that start at 0V, go up to some positive peak, and then return to 0V in each cycle. Signal generators have imperfections. The waveforms are not pure. There can be noise or distortion in the output. Noise is a random signal and distortion is a change in or distortion of a 'pure' signal. An ideal sinewave generator should produce v(t) = sin(t), while a distorted signal could be v(t) = sin(t) * sin(3t). The frequency or amplitude of a generator output voltage will not be stable. The user is expected to keep this in mind. Each generator has an amplitude and frequency control. The controls generally work in ranges so you will usually see a range control that limits the minimum and maximum frequency or voltage and a fine control to adjust within the range. With time, and as the instrument heats up, the frequency and amplitude of the signal generated will change E&CE Dept. University of Waterloo page9
10 Function Generators A function generator is a signal generator that is capable of producing more than one waveform shape. The B&K Precision model 3010 function generator is shown in figure 6. Figure 6. B & K Precision Function Generator Î. Power Switch Ï. Frequency Range Selection Buttons Ð. Waveform Function Selection Buttons Ñ. Amplitude Dial Ò. DC Offset Dial Ó. Frequency Dial marked from 0.1 to 10 (multiplies by range) Ô. TTL Level Output Connector Õ. Common Connector Ö. Function Generator Output Connector 2009 E&CE Dept. University of Waterloo page10
DIODE CIRCUITS LABORATORY. Fig. 8.1a Fig 8.1b
DIODE CIRCUITS LABORATORY A solid state diode consists of a junction of either dissimilar semiconductors (pn junction diode) or a metal and a semiconductor (Schottky barrier diode). Regardless of the type,
More informationLab E1: Introduction to Circuits
E1.1 Lab E1: Introduction to Circuits The purpose of the this lab is to introduce you to some basic instrumentation used in electrical circuits. You will learn to use a DC power supply, a digital multimeter
More informationThe RC Circuit. Prelab questions. Introduction. The RC Circuit
The RC Circuit Prelab questions 1. What is the meaning of the time constant, RC? 2. Show that RC has units of time. 3. Why isn t the time constant defined to be the time it takes the capacitor to become
More informationAnalog and Digital Meters
Analog and Digital Meters Devices and Measurements Objective At the conclusion of this presentation the student will describe and identify: Safety precautions when using test equipment Analog Multimeters
More informationExperiment 2 Diode Applications: Rectifiers
ECE 3550  Practicum Fall 2007 Experiment 2 Diode Applications: Rectifiers Objectives 1. To investigate the characteristics of halfwave and fullwave rectifier circuits. 2. To recognize the usefulness
More informationEXPERIMENT NUMBER 5 BASIC OSCILLOSCOPE OPERATIONS
1 EXPERIMENT NUMBER 5 BASIC OSCILLOSCOPE OPERATIONS The oscilloscope is the most versatile and most important tool in this lab and is probably the best tool an electrical engineer uses. This outline guides
More information= V peak 2 = 0.707V peak
BASIC ELECTRONICS  RECTIFICATION AND FILTERING PURPOSE Suppose that you wanted to build a simple DC electronic power supply, which operated off of an AC input (e.g., something you might plug into a standard
More informationTable of Contents. The Basics of Electricity 2. Using a Digital Multimeter 4. Testing Voltage 8. Testing Current 10. Testing Resistance 12
Table of Contents The Basics of Electricity 2 Using a Digital Multimeter 4 IDEAL Digital Multimeters An Introduction The Basics of Digital Multimeters is designed to give you a fundamental knowledge of
More informationName Date Day/Time of Lab Partner(s) Lab TA
Name Date Day/Time of Lab Partner(s) Lab TA Objectives LAB 7: AC CIRCUITS To understand the behavior of resistors, capacitors, and inductors in AC Circuits To understand the physical basis of frequencydependent
More informationLABORATORY 10 TIME AVERAGES, RMS VALUES AND THE BRIDGE RECTIFIER. Bridge Rectifier
LABORATORY 10 TIME AVERAGES, RMS VALUES AND THE BRIDGE RECTIFIER Fullwave Rectification: Bridge Rectifier For many electronic circuits, DC supply voltages are required but only AC voltages are available.
More informationPrecise Measurement FAQs
Precise Measurement FAQs Here are some Frequently Asked Questions and Answers associated with low level measurements. If you don t find answers to your questions here, you can send a specific question
More informationBSNL TTA Question PaperInstruments and Measurement Specialization 2007
BSNL TTA Question PaperInstruments and Measurement Specialization 2007 (1) Instrument is a device for determining (a) the magnitude of a quantity (b) the physics of a variable (c) either of the above
More informationPHYSICS 111 LABORATORY Experiment #3 Current, Voltage and Resistance in Series and Parallel Circuits
PHYSCS 111 LABORATORY Experiment #3 Current, Voltage and Resistance in Series and Parallel Circuits This experiment is designed to investigate the relationship between current and potential in simple series
More informationWHY DIFFERENTIAL? instruments connected to the circuit under test and results in V COMMON.
WHY DIFFERENTIAL? Voltage, The Difference Whether aware of it or not, a person using an oscilloscope to make any voltage measurement is actually making a differential voltage measurement. By definition,
More informationFig. 1 Analogue Multimeter Fig.2 Digital Multimeter
ELECTRICAL INSTRUMENT AND MEASUREMENT Electrical measuring instruments are devices used to measure electrical quantities such as electric current, voltage, resistance, electrical power and energy. MULTIMETERS
More informationEDEXCEL NATIONAL CERTIFICATE/DIPLOMA UNIT 5  ELECTRICAL AND ELECTRONIC PRINCIPLES NQF LEVEL 3 OUTCOME 4  ALTERNATING CURRENT
EDEXCEL NATIONAL CERTIFICATE/DIPLOMA UNIT 5  ELECTRICAL AND ELECTRONIC PRINCIPLES NQF LEVEL 3 OUTCOME 4  ALTERNATING CURRENT 4 Understand singlephase alternating current (ac) theory Single phase AC
More informationThe Oscilloscope, the Signal Generator and Your Filter s Test Setup SGM 5/29/2013
The Oscilloscope, the Signal Generator and Your Filter s Test Setup SGM 5/29/2013 1. Oscilloscope A multimeter is an appropriate device to measure DC voltages, however, when a signal alternates at relatively
More informationProperties of electrical signals
DC Voltage Component (Average voltage) Properties of electrical signals v(t) = V DC + v ac (t) V DC is the voltage value displayed on a DC voltmeter Triangular waveform DC component Halfwave rectifier
More informationCurrent and Temperature Ratings
Document 3611 Current and Temperature Ratings Introduction This application note describes: How to interpret Coilcraft inductor current and temperature ratings Our current ratings measurement method and
More informationInductors in AC Circuits
Inductors in AC Circuits Name Section Resistors, inductors, and capacitors all have the effect of modifying the size of the current in an AC circuit and the time at which the current reaches its maximum
More informationOverview: The purpose of this experiment is to introduce diode rectifier circuits used in DC power supplies.
UNIVERSITY OF NORTH CAROLINA AT CHARLOTTE Department of Electrical and Computer Engineering Experiment No. 3 Diodes and Bridge Rectifiers Overview: The purpose of this experiment is to introduce diode
More informationPrecision Diode Rectifiers
by Kenneth A. Kuhn March 21, 2013 Precision halfwave rectifiers An operational amplifier can be used to linearize a nonlinear function such as the transfer function of a semiconductor diode. The classic
More informationConstructing a precision SWR meter and antenna analyzer. Mike Brink HNF, Design Technologist.
Constructing a precision SWR meter and antenna analyzer. Mike Brink HNF, Design Technologist. Abstract. I have been asked to put together a detailed article on a SWR meter. In this article I will deal
More informationLab #4 Capacitors and Inductors. Capacitor and Inductor Transient Response
Capacitor and Inductor Transient Response Capacitor Theory Like resistors, capacitors are also basic circuit elements. Capacitors come in a seemingly endless variety of shapes and sizes, and they can all
More informationABCs of DMMs Multimeter features and functions explained Application Note
ABCs of DMMs Multimeter features and functions explained Application Note Digital multimeters offer a wide selection of features. Choosing the right meter for the job can be challenging unless you know
More informationEXPERIMENT 6 CLIPPING AND CLAMPING DIODE CIRCUITS
EXPERIMENT 6 CLIPPING AND CLAMPING DIODE CIRCUITS OBJECTIVES To understand the theory of operation of the clipping and clamping diode circuits. To design wave shapes that meet different circuits needs.
More informationSeries and Parallel Circuits
Direct Current (DC) Direct current (DC) is the unidirectional flow of electric charge. The term DC is used to refer to power systems that use refer to the constant (not changing with time), mean (average)
More informationElectrical Resonance
Electrical Resonance (RLC series circuit) APPARATUS 1. RLC Circuit board 2. Signal generator 3. Oscilloscope Tektronix TDS1002 with two sets of leads (see Introduction to the Oscilloscope ) INTRODUCTION
More informationEXPERIMENT NUMBER 8 CAPACITOR CURRENTVOLTAGE RELATIONSHIP
1 EXPERIMENT NUMBER 8 CAPACITOR CURRENTVOLTAGE RELATIONSHIP Purpose: To demonstrate the relationship between the voltage and current of a capacitor. Theory: A capacitor is a linear circuit element whose
More informationPower Supplies. 1.0 Power Supply Basics. www.learnaboutelectronics.org. Module
Module 1 www.learnaboutelectronics.org Power Supplies 1.0 Power Supply Basics What you ll learn in Module 1 Section 1.0 Power Supply Basics. Basic functions of a power supply. Safety aspects of working
More informationThe George Washington University School of Engineering and Applied Science Department of Electrical and Computer Engineering
The George Washington University School of Engineering and Applied Science Department of Electrical and Computer Engineering Final Design Report Dual Channel Stereo Amplifier By: Kristen Gunia Prepared
More informationElectrical Measurements
Electrical Measurements Experimental Objective The objective of this experiment is to become familiar with some of the electrical instruments. You will gain experience by wiring a simple electrical circuit
More informationResistance, Ohm s Law, and the Temperature of a Light Bulb Filament
Resistance, Ohm s Law, and the Temperature of a Light Bulb Filament Name Partner Date Introduction Carbon resistors are the kind typically used in wiring circuits. They are made from a small cylinder of
More informationThe full wave rectifier consists of two diodes and a resister as shown in Figure
The FullWave Rectifier The full wave rectifier consists of two diodes and a resister as shown in Figure The transformer has a centretapped secondary winding. This secondary winding has a lead attached
More informationThe OP AMP , Figure 1
The OP AMP Amplifiers, in general, taking as input, one or more electrical signals, and produce as output, one or more variations of these signals. The most common use of an amplifier is to accept a small
More informationPHYS 2426 Engineering Physics II (Revised July 7, 2011) AC CIRCUITS: RLC SERIES CIRCUIT
PHYS 2426 Engineering Physics II (Revised July 7, 2011) AC CIRCUITS: RLC SERIES CIRCUIT INTRODUCTION The objective of this experiment is to study the behavior of an RLC series circuit subject to an AC
More informationAnalog Electronics. Module 1: Semiconductor Diodes
Analog Electronics s PREPARED BY Academic Services Unit August 2011 Applied Technology High Schools, 2011 s Module Objectives Upon successful completion of this module, students should be able to: 1. Identify
More informationLab 9: Op Amps Lab Assignment
3 class days 1. Differential Amplifier Source: HandsOn chapter 8 (~HH 6.1) Lab 9: Op Amps Lab Assignment Difference amplifier. The parts of the pot on either side of the slider serve as R3 and R4. The
More informationLab 3 Rectifier Circuits
ECET 242 Electronic Circuits Lab 3 Rectifier Circuits Page 1 of 5 Name: Objective: Students successfully completing this lab exercise will accomplish the following objectives: 1. Learn how to construct
More informationClass A Amplifier Design
Module 2 Amplifiers Introduction to Amplifier Design What you ll learn in Module 2. Basic design process. Section 2.0 Introduction to Amplifier Design. Section 2.1 DC Conditions. Design a BJT class A common
More informationStep Response of RC Circuits
Step Response of RC Circuits 1. OBJECTIVES...2 2. REFERENCE...2 3. CIRCUITS...2 4. COMPONENTS AND SPECIFICATIONS...3 QUANTITY...3 DESCRIPTION...3 COMMENTS...3 5. DISCUSSION...3 5.1 SOURCE RESISTANCE...3
More informationEssential Electrical Concepts
Essential Electrical Concepts Introduction Modern vehicles incorporate many electrical and electronic components and systems: Audio Lights Navigation Engine control Transmission control Braking and traction
More informationLab 1: The Digital Oscilloscope
PHYSICS 220 Physical Electronics Lab 1: The Digital Oscilloscope Object: To become familiar with the oscilloscope, a ubiquitous instrument for observing and measuring electronic signals. Apparatus: Tektronix
More informationCircuits and Resistivity
Circuits and Resistivity Look for knowledge not in books but in things themselves. W. Gilbert OBJECTIVES To learn the use of several types of electrical measuring instruments in DC circuits. To observe
More informationRectifier circuits & DC power supplies
Rectifier circuits & DC power supplies Goal: Generate the DC voltages needed for most electronics starting with the AC power that comes through the power line? 120 V RMS f = 60 Hz T = 1667 ms) = )sin How
More informationAC Measurements Using the Oscilloscope and Multimeter by Mr. David Fritz
AC Measurements Using the Oscilloscope and Multimeter by Mr. David Fritz 1 Sine wave with a DC offset f = frequency in Hz A = DC offset voltage (average voltage) B = Sine amplitude Vpp = 2B Vmax = A +
More informationCharge and Discharge of a Capacitor
Charge and Discharge of a Capacitor INTRODUCTION Capacitors 1 are devices that can store electric charge and energy. Capacitors have several uses, such as filters in DC power supplies and as energy storage
More informationFluke 8808A Digital Multimeter Extended Specifications
Fluke 8808A Digital Multimeter Extended Specifications Making measurements is as simple as pushing a button The Fluke 8808A 5.5 digit multimeter has a broad range of functions, measuring volts, ohms and
More informationLab 3  DC Circuits and Ohm s Law
Lab 3 DC Circuits and Ohm s Law L31 Name Date Partners Lab 3  DC Circuits and Ohm s Law OBJECTIES To learn to apply the concept of potential difference (voltage) to explain the action of a battery in
More informationLAB 7 MOSFET CHARACTERISTICS AND APPLICATIONS
LAB 7 MOSFET CHARACTERISTICS AND APPLICATIONS Objective In this experiment you will study the iv characteristics of an MOS transistor. You will use the MOSFET as a variable resistor and as a switch. BACKGROUND
More informationTransistor Amplifiers
Physics 3330 Experiment #7 Fall 1999 Transistor Amplifiers Purpose The aim of this experiment is to develop a bipolar transistor amplifier with a voltage gain of minus 25. The amplifier must accept input
More informationChapter 13: Electric Circuits
Chapter 13: Electric Circuits 1. A household circuit rated at 120 Volts is protected by a fuse rated at 15 amps. What is the maximum number of 100 watt light bulbs which can be lit simultaneously in parallel
More informationChapter 22 Further Electronics
hapter 22 Further Electronics washing machine has a delay on the door opening after a cycle of washing. Part of this circuit is shown below. s the cycle ends, switch S closes. t this stage the capacitor
More informationEE 1202 Experiment #2 Resistor Circuits
EE 1202 Experiment #2 Resistor Circuits 1. ntroduction and Goals: Demonstrates the voltagecurrent relationships in DC and AC resistor circuits. Providing experience in using DC power supply, digital multimeter,
More informationRectifier: It is a circuit which employs one or more diodes to convert ac voltage into pulsating dc voltage. We will consider the following circuits:
Rectifier: It is a circuit which employs one or more diodes to convert ac voltage into pulsating dc voltage. We will consider the following circuits: (i) Half wave rectifier. (ii) Full wave rectifier.
More informationIDEAL INDUSTRIES, INC. TECHNICAL MANUAL MODEL: MODEL: Multimeter Service Information
IDEAL INDUSTRIES, INC. TECHNICAL MANUAL MODEL: 61340 MODEL: 61342 Multimeter Service Information The Service Information provides the following information: Precautions and safety information Specifications
More informationε: Voltage output of Signal Generator (also called the Source voltage or Applied
Experiment #10: LR & RC Circuits Frequency Response EQUIPMENT NEEDED Science Workshop Interface Power Amplifier (2) Voltage Sensor graph paper (optional) (3) Patch Cords Decade resistor, capacitor, and
More informationEXPERIMENT 7 OHM S LAW, RESISTORS IN SERIES AND PARALLEL
260 7 I. THEOY EXPEIMENT 7 OHM S LAW, ESISTOS IN SEIES AND PAALLEL The purposes of this experiment are to test Ohm's Law, to study resistors in series and parallel, and to learn the correct use of ammeters
More informationThe Oscilloscope. Vision is the art of seeing things invisible. J. Swift ( ) OBJECTIVE To learn to operate a cathode ray oscilloscope.
The Oscilloscope Vision is the art of seeing things invisible. J. Swift (16671745) OBJECTIVE To learn to operate a cathode ray oscilloscope. THEORY The oscilloscope, or scope for short, is a device for
More informationENGR 210 Lab 11 Frequency Response of Passive RC Filters
ENGR 210 Lab 11 Response of Passive RC Filters The objective of this lab is to introduce you to the frequencydependent nature of the impedance of a capacitor and the impact of that frequency dependence
More informationLock  in Amplifier and Applications
Lock  in Amplifier and Applications What is a Lock in Amplifier? In a nut shell, what a lockin amplifier does is measure the amplitude V o of a sinusoidal voltage, V in (t) = V o cos(ω o t) where ω o
More informationThe 555 timer: Waveform shaping
The 555 timer: Waveform shaping Introduction: Oscillators are ubiquitous not only in research environments but also in our modern, technologyoriented society. Oscillators are used for communication, timing
More informationStudent Name Instructor Name. High School or Vocational Center Grade. COMPETENCY RECORD FOR ARTICULATION Muskegon Community College Electronics
Student Name Instructor Name High School or Vocational Center Grade COMPETENCY RECORD FOR ARTICULATION Muskegon Community College Electronics Please check below each skill the student has mastered as described,
More informationThe Basics of Digital Multimeters
IDEAL INDUSTRIES INC. The Basics of Digital Multimeters A guide to help you understand the basic Features and Functions of a Digital Multimeter. Author: Patrick C Elliott Field Sales Engineer IDEAL Industries,
More informationTESTS OF 1 MHZ SIGNAL SOURCE FOR SPECTRUM ANALYZER CALIBRATION 7/8/08 Sam Wetterlin
TESTS OF 1 MHZ SIGNAL SOURCE FOR SPECTRUM ANALYZER CALIBRATION 7/8/08 Sam Wetterlin (Updated 7/19/08 to delete sine wave output) I constructed the 1 MHz square wave generator shown in the Appendix. This
More informationMATERIALS. Multisim screen shots sent to TA.
Page 1/8 Revision 0 9Jun10 OBJECTIVES Learn new Multisim components and instruments. Conduct a Multisim transient analysis. Gain proficiency in the function generator and oscilloscope. MATERIALS Multisim
More informationECEN 1400, Introduction to Analog and Digital Electronics
ECEN 1400, Introduction to Analog and Digital Electronics Lab 4: Power supply 1 INTRODUCTION This lab will span two lab periods. In this lab, you will create the power supply that transforms the AC wall
More informationExperiment 3 ~ Ohm's Law, Measurement of Voltage, Current and Resistance
Experiment 3 ~ Ohm's Law, Measurement of Voltage, Current and Resistance Objective: In this experiment you will learn to use the multimeter to measure voltage, current and resistance. Equipment: Bread
More informationThe W5JCK Guide to the Mathematic Equations Required for the Amateur Extra Class Exam
The W5JCK Guide to the Mathematic Equations Required for the Amateur Extra Class Exam This document contains every question from the Extra Class (Element 4) Question Pool* that requires one or more mathematical
More informationR f. V i. ET 438a Automatic Control Systems Technology Laboratory 4 Practical Differentiator Response
ET 438a Automatic Control Systems Technology Laboratory 4 Practical Differentiator Response Objective: Design a practical differentiator circuit using common OP AMP circuits. Test the frequency response
More informationBipolar Transistor Amplifiers
Physics 3330 Experiment #7 Fall 2005 Bipolar Transistor Amplifiers Purpose The aim of this experiment is to construct a bipolar transistor amplifier with a voltage gain of minus 25. The amplifier must
More informationSERIESPARALLEL DC CIRCUITS
Name: Date: Course and Section: Instructor: EXPERIMENT 1 SERIESPARALLEL DC CIRCUITS OBJECTIVES 1. Test the theoretical analysis of seriesparallel networks through direct measurements. 2. Improve skills
More informationLecture  4 Diode Rectifier Circuits
Basic Electronics (Module 1 Semiconductor Diodes) Dr. Chitralekha Mahanta Department of Electronics and Communication Engineering Indian Institute of Technology, Guwahati Lecture  4 Diode Rectifier Circuits
More informationPhysics 2306 Experiment 7: Timedependent Circuits, Part 1
Name ID number Date Lab CRN Lab partner Lab instructor Objectives Physics 2306 Experiment 7: Timedependent Circuits, Part 1 To study the time dependent behavior of the voltage and current in circuits
More informationDirect versus Alternating Current Things We Can Measure
Phil Sherrod W4PHS Direct versus Alternating Current Things We Can Measure Direct Current (DC) Alternating Current (AC) Voltage Voltage (peak, RMS) Current Current (peak, effective) Power True power, Apparent
More informationENGR4300 Electronic Instrumentation Quiz 4 Spring 2011 Name Section
ENGR4300 Electronic Instrumentation Quiz 4 Spring 2011 Name Section Question I (20 points) Question II (20 points) Question III (20 points) Question IV (20 points) Question V (20 points) Total (100 points)
More informationUsing an Oscilloscope
Using an Oscilloscope The oscilloscope is used to measure a voltage that changes in time. It has two probes, like a voltmeter. You put these probes on either side of the thing that you want to measure
More informationAC Direct OffLine Power Supplies
AC Direct OffLine Power Supplies r Introduction Many DC power supplies found in electronic systems, including those in this Tech School, rectify the 120 volts available at an electric outlet. The initial
More informationDiode Applications. by Kenneth A. Kuhn Sept. 1, 2008. This note illustrates some common applications of diodes.
by Kenneth A. Kuhn Sept. 1, 2008 This note illustrates some common applications of diodes. Power supply applications A common application for diodes is converting AC to DC. Although halfwave rectification
More informationExperiment #3, Ohm s Law
Experiment #3, Ohm s Law 1 Purpose Physics 182  Summer 2013  Experiment #3 1 To investigate the oltage, , characteristics of a carbon resistor at room temperature and at liquid nitrogen temperature,
More informationvery small Ohm s Law and DC Circuits Purpose: Students will become familiar with DC potentiometers circuits and Ohm s Law. Introduction: P31220 Lab
Ohm s Law and DC Circuits Purpose: Students will become familiar with DC potentiometers circuits and Ohm s Law. Introduction: Ohm s Law for electrical resistance, V = IR, states the relationship between
More informationSIGNAL GENERATORS and OSCILLOSCOPE CALIBRATION
1 SIGNAL GENERATORS and OSCILLOSCOPE CALIBRATION By Lannes S. Purnell FLUKE CORPORATION 2 This paper shows how standard signal generators can be used as leveled sine wave sources for calibrating oscilloscopes.
More informationElectronics. Basic Concepts. Yrd. Doç. Dr. Aytaç GÖREN Yrd. Doç. Dr. Levent ÇETİN
Electronics Basic Concepts Electric charge Ordinary matter is made up of atoms which have positively charged nuclei and negatively charged electrons surrounding them. Charge is quantized as the subtraction
More informationRESONANCE AND FILTERS
14221 RESONANCE AND FILTERS Experiment 1, Resonant Frequency and Circuit Impedance For more courses visit www.ciewc.edu OBJECTIVES 1. To verify experimentally our theoretical predictions concerning the
More informationEXERCISES in ELECTRONICS and SEMICONDUCTOR ENGINEERING
Department of Electrical Drives and Power Electronics EXERCISES in ELECTRONICS and SEMICONDUCTOR ENGINEERING Valery Vodovozov and Zoja Raud http://learnelectronics.narod.ru Tallinn 2012 2 Contents Introduction...
More informationRC Circuits. The purpose of this lab is to understand how capacitors charge and discharge.
Department of Physics and Geology Purpose Circuits Physics 2402 The purpose of this lab is to understand how capacitors charge and discharge. Materials Decade Resistance Box (CENCO), 0.1 µf, 0.5µF, and
More informationKirchhoff's Rules and Applying Them
[ Assignment View ] [ Eðlisfræði 2, vor 2007 26. DC Circuits Assignment is due at 2:00am on Wednesday, February 21, 2007 Credit for problems submitted late will decrease to 0% after the deadline has passed.
More informationEE 1202 Experiment #4 Capacitors, Inductors, and Transient Circuits
EE 1202 Experiment #4 Capacitors, Inductors, and Transient Circuits 1. Introduction and Goal: Exploring transient behavior due to inductors and capacitors in DC circuits; gaining experience with lab instruments.
More informationThe current that flows is determined by the potential difference across the conductor and the resistance of the conductor (Ohm s law): V = IR P = VI
PHYS1000 DC electric circuits 1 Electric circuits Electric current Charge can move freely in a conductor if an electric field is present; the moving charge is an electric current (SI unit is the ampere
More informationThe D.C Power Supply
The D.C Power Supply Voltage Step Down Electrical Isolation Converts Bipolar signal to Unipolar Half or Full wave Smoothes the voltage variation Still has some ripples Reduce ripples Stabilize the output
More informationLABORATORY 2 THE DIFFERENTIAL AMPLIFIER
LABORATORY 2 THE DIFFERENTIAL AMPLIFIER OBJECTIVES 1. To understand how to amplify weak (small) signals in the presence of noise. 1. To understand how a differential amplifier rejects noise and common
More information73 Series III Multimeter
73 Series III Multimeter Instruction Sheet Read First: Safety Information W To avoid damage to the meter, electric shock or injury: Never use the meter if the meter or test leads look damaged. Be sure
More informationUniversity of Alberta Department of Electrical and Computer Engineering. EE 250 Laboratory Experiment #5 Diodes
University of Alberta Department of Electrical and Computer Engineering EE 250 Laboratory Experiment #5 Diodes Objective: To introduce basic diode concepts. Introduction: The diode is the most fundamental
More informationELEC 435 ELECTRONICS I. Rectifier Circuits
ELEC 435 ELECTRONICS I Rectifier Circuits Common types of Transformers The Rectifier Rectification is the conversion of an alternating current to a pulsating direct current. Rectification occurs in both
More informationHalfWave Rectifiers
HalfWave Rectifiers Important Points of This Lecture Calculation of output voltage using appropriate piecewise models for diode for simple (unfiltered) halfwave rectifier Differences between calculations
More informationEE320L Electronics I. Laboratory. Laboratory Exercise #5. Clipping and Clamping Circuits. Angsuman Roy
EE320L Electronics I Laboratory Laboratory Exercise #5 Clipping and Clamping Circuits By Angsuman Roy Department of Electrical and Computer Engineering University of Nevada, Las Vegas Objective: The purpose
More informationDiode Applications. As we have already seen the diode can act as a switch Forward biased or reverse biased  On or Off.
Diode Applications Diode Switching As we have already seen the diode can act as a switch Forward biased or reverse biased  On or Off. Voltage Rectifier A voltage rectifier is a circuit that converts an
More informationPH 210 Electronics Laboratory I Instruction Manual
PH 210 Electronics Laboratory I Instruction Manual Index Page No General Instructions 2 Experiment 1 Common Emitter (CE) Amplifier 4 Experiment 2 Multistage amplifier: Cascade of two CE stages 7 Experiment
More informationPart 2: Receiver and Demodulator
University of Pennsylvania Department of Electrical and Systems Engineering ESE06: Electrical Circuits and Systems II Lab Amplitude Modulated Radio Frequency Transmission System MiniProject Part : Receiver
More informationExperiment A5. Hysteresis in Magnetic Materials
HYSTERESIS IN MAGNETIC MATERIALS A5 1 Experiment A5. Hysteresis in Magnetic Materials Objectives This experiment illustrates energy losses in a transformer by using hysteresis curves. The difference betwen
More information