Theory of Chromatography

Save this PDF as:

Size: px
Start display at page:

Transcription

1 Theory of Chromatography

2 The Chromatogram A chromatogram is a graph showing the detector response as a function of elution time. The retention time, t R, for each component is the time needed after injection of the mixture until that component reaches the detector.

3

4 The Chromatogram (cont.) Retention volume, V R, is the volume of mobile phase required to elute a particular solute from the column: V R = t R F where F is the mobile phase flow rate The dead time, t m, is the time of travel of unretained mobile phase through the column.

5 The Chromatogram (cont.) The adjusted retention time, t R, for a solute is the additional time required for solute to travel the length of the column beyond the time required by unretained solvent: t R = t R t m In GC, t m is usually taken as the time needed for CH 4 to travel through the column.

6 The Chromatogram (cont.) For any two components 1 and, the relative retention, α, is the ratio of their adjusted retention times: where, so α > 1. ' 1 ' R R t t = α 1 R R t t = 1 R R t t = ' 1 ' R R t t = ' 1 ' R R t t >

7 The Capacity Factor For each peak in the chromatogram, the capacity factor, k, is defined as: k' = t R t m t m k'= time solute spends in stationary phase time solute spends in mobile phase

8 The Capacity Factor (cont.) time solute spends in stationary phase k' = = time solute spends in mobile phase moles of solute in stationary phase moles of solute in mobile phase k ' = C V C s m V s m k ' = K C C V V s m s m = K t = R t m t m = t t ' R m

9 Relative Retention Alternative Expressions α = t t ' R ' R1 = k k ' ' 1 = K K 1

10 Efficiency of Separation

11 Resolution Solute moving through a column spreads into a Gaussian shape with standard deviation σ. Common measures of breadth are: The width w ½ measured at half-height The width w at the baseline between tangents drawn to the steepest parts of the peak (inflection points).

12

13 Resolution (cont.) It can be shown that: and w ½ =.35σ w = 4σ

14 Resolution (cont.) In chromatography, the resolution of two peaks from each other is defined as R s = Δt w R av = ΔV w av R = 0.589Δt w 1 av R where Δt R or ΔV R is the separation between peaks and w av is the average width of the two peaks.

15

16 Resolution So, separation of mixtures depends on: width of solute peaks (want narrow) efficiency spacing between peaks (want large spacing) selectivity

17 Example What is the resolution of two Gaussian peaks of identical width (3.7 s) and height eluting at 67.3 s and 74.9 s, respectively? ANS: Resolution =.3

18 Diffusion A band of solute broadens as it moves through a column. Ideally, an infinitely narrow band applied to the inlet of the column emerges with a Gaussian shape at the outlet.

19

20 Diffusion (cont.) One main cause of band spreading is diffusion. The diffusion coefficient measures the rate at which a substance moves randomly from a region of high concentration to a region of lower concentration.

21 Diffusion (cont.) The number of moles crossing each square meter per second, called the flux, is proportional to the concentration gradient: flux mol m s J = D dc dx

22

23 Broadening of Chromatographic Band by Diffusion If solute begins to move through a column in an infinitely sharp layer with m moles per unit crosssectional area of the column and spreads by diffusion alone, then the Gaussian profile of the band is described by c = m e x ( 4Dt ) 4 π Dt

24 The standard deviation of the band is σ = Dt

25

26 The Theory of Chromatography: Column Efficiency Plate theory - older; developed by Martin & Synge Rate theory - currently in use

27 Plate Theory - Martin & Synge View column as divided into a number (N) of adjacent imaginary segments called theoretical plates within each theoretical plate complete equilibration of analytes between stationary and mobile phase occurs

28

29 Plate Theory - Martin & Synge Significance? Greater separation occurs with: greater number of theoretical plates (N) as plate height (H or HETP) becomes smaller L = N H or H = L / N where L is the length of column, N is the number of plates, and H is the plate height

30 First Important Prediction of Plate Theory Band spreading - the width of bands increases as their retention time (volume) increases: Plate height is the constant of proportionality between the variance of the band and the distance it has traveled: σ = Dt = D u x x D = x = u x Hx

31 Second Significant Prediction of Plate Theory The smaller HETP, the narrower the eluted peak

32 Plate Theory - Practical Considerations Not unusual for a chromatography column to have millions of theoretical plates Columns often behave as if they have different numbers of plates for different solutes present in same mixture

33 Number of plates on column: N V = 16 w w b base width of the peak R b This equation is a measure of the efficiency of a column.

34 Sometimes the number of plates is measured at the bandwidth at half-height w 1/ : N = 5.54 V w R 1

35 Estimating the Plate Number for Asymmetric Peaks The Dorsey-Foley equation: N = 41.7 t R w0.1 A B

36

37 N can be Estimated Experimentally from a Chromatogram Knowing the number of theoretical plates and the length of the column, we can determine the HETP, height equivalent to a theoretical plate: H = HETP = L N = L 16 w V b R = L 16 w t b R

38 Effective Number of Theoretical Plates Introduced to characterize open tubular columns uses adjusted retention volume V R in lieu of total retention volume V R : b ' b ' eff = = w t w V N R R

39 Effective Number of Theoretical Plates (cont.) The N eff value is useful for comparing a packed and an open tubular column when both are used for the same separation. N and N eff are related by the expression N eff = N k' k' + 1

40 Rate Theory Based on a random walk mechanism for the migration of molecules through a column Takes into account: mechanism of band broadening effect of rate of elution on band shape availability of different paths for different solute molecules to follow diffusion of solute along length

41 Van Deemter Equation for Plate Height B H = A + + u x Cu x Multiple paths Longitudinal diffusion Equilibration time

42 In packed columns, all three terms contribute to band broadening In open tubular columns, A is zero In capillary electrophoresis, both A and C go to zero

43 H = A + min B C H = A + min BC u opt = B C

44 Longitudinal Diffusion Gives rise to B/u x term Solute continuously diffuses away from the concentrated center of its zone The greater the flow rate, the less time is spent in the column and the less longitudinal diffusion occurs

45 Longitudinal Diffusion (cont.) The variance resulting from diffusion is σ = D m t = D u m x L Plate height due to longitudinal diffusion: H D = σ L = D u x m B u x

46 Longitudinal Diffusion (cont.) In packed columns, the tortuosity coefficient γ is used to account for irregular diffusion patterns and is usually less than unity (γ ~ 0.6), because molecular diffusivity is smaller in packed columns than in open tubes (γ = 1): H D = γd u x m

47

48 Longitudinal Diffusion (cont.) Because longitudinal diffusion in a gas is much faster than in a liquid, the optimum flow rate in gas chromatography is higher than in liquid chromatography

49 Nonequilibrium (Resistance to Mass Transfer Term) This term comes from the finite time required for the solute to equilibrate between the mobile and stationary phases Some solute is stuck in the stationary phase, but the remainder in the mobile phase moves forward resulting in spreading of the zone The slower the flow rate, the more complete equilibration is and the less band broadening occurs

50

51 Resistance to Mass Transfer (cont.) Plate height due to finite equilibration time: H = Cu = + mass transfer x ( C ) s Cm ux where C s describes the rate of mass transfer through the stationary phase and C m describes the rate of mass transfer through the mobile phase. Specific equations for C s and C m depend on the type of chromatography.

52 Resistance to Mass Transfer (cont.) For gas chromatography in an open tubular column, the terms are: Mass transfer in stationary phase: Mass transfer in mobile phase: ( ) s f s D d k k C 1 ' 3 ' + = ( ) m m D r k k k C 1 ' 4 ' 11 ' =

53 Resistance to Mass Transfer (cont.) k the capacity factor d f the thickness of stationary phase film D s the diffusion coefficient of solute in the stationary phase r the column radius D m the diffusion coefficient of solute in the mobile phase

54 Resistance to Mass Transfer (cont.) Efficiency is increased by: Decreasing stationary phase thickness Reducing column radius Increasing temperature

55

56 Multiple Flow Paths (Eddy Diffusion) The term A accounts for the multitude of pathways that could be followed through the column by one molecule: A = λd p λ dimensionless constant characteristic of packing d p the particle diameter

57

58 Multiple Flow Paths (Eddy Diffusion) (cont.) The A term can be minimized by: Eliminating column packing (open tubular columns) Reducing the particle size of the packing Packing the column more uniformly (spherical particles of similar size)

59

60

61 Alternative Plate Height Equation: The Knox Equation Used to compare column efficiencies Makes use of so-called reduced parameters (dimensionless quantities): Reduced plate height: h = H/d p Reduced velocity: v = ud p /D m 1 b 3 h = aν + + ν cν

62 The Knox Equation (cont.) For well-packed columns of varying particle size and differing conditions, the coefficients a, b and c will be roughly constant: e.g. a =1, b =, and c = 0.05 for porous particles.

63 Modification of the van Deemter Equation: the Giddings Equation Giddings realized that the eddy diffusion and resistance to mass transfer in the mobile phase must be treated dependently: 5 1 B H = + + Csu + Cmu + 1 i + 1 = 1 u A C u 1 H e

64 The Giddings Equation H e extracolumn band broadening C 1 = c b d D m p c b a proportionality constant; c b 1

65 The Giddings Equation (cont.) The five possible mechanisms of band broadening are: Through channels between particles Through particles Resulting from uneven flow channels Between inhomogeneous regions Throughout the entire column length

66

67 Factors Affecting Resolution A more usable expression for resolution is R s α 1 1 k' = N 4 α k' + 1 efficiency selectivity retention

68 Required Plate Number If k and α are known, the required number of plates can be calculated: The R s value is set at the 6σ level or 1.5 ' ' req = k k R N s α α ' ' req = k k R N s α α

69 Required Column Length The N req parameter can be used to determine the length of column necessary for a separation. We know that N = L/H; thus ' ' req = k k H R L s α α

70 Minimum Analysis Time The minimum analysis time t min is t min = 16R s H u α α 1 ( ' k ) + 1 ( ' k ) 3

71 The Major Objective in Chromatography The goal in chromatography is the highest possible resolution in the shortest possible time. Optimization techniques aim at choosing conditions that lead to a desired degree of resolution with a minimum expenditure of time

72 Optimization Techniques Minimizing plate height: Reducing d p Reducing column diameter Changing column temperature Reducing the thickness of the liquid film Optimizing the flow rate of the mobile phase

73 Optimization Techniques (cont.) Resolution also improves with L, but it expensive in terms of time of analysis Variation in the selectivity factor: Changing the composition of the mobile phase (in HPLC) Changing the column temperature Changing the stationary phase Using special chemical effects

74

75 Effect of Capacity Factor on Resolution and Elution Time Increases in k enhance resolution but at the expense of elution time The optimum k values are from 1 to 5 The easiest way to improve R s is by optimizing k In GC, k can be improved by temperature programming In LC, k is improved by gradient elution

76

77

78 Advantages of Open Tubular Columns Compared with packed columns, open tubular columns can provide: Higher resolution Shorter analysis time Increased sensitivity Lower sample capacity

79 Advantages of Open Tubular Columns (cont.) Compared with packed columns, open tubular columns allow Increased linear flow rate or a longer column or both Decreased plate height, which means higher resolution

80

81 A Touch of Reality: Asymmetric Bandshapes

82

Gas Chromatography. Let s begin with an example problem: SPME head space analysis of pesticides in tea and follow-up analysis by high speed GC.

Gas Chromatography Let s begin with an example problem: SPME head space analysis of pesticides in tea and follow-up analysis by high speed GC. Samples in 10mL sealed glass vials were placed in the MPS-2

Diffusion and Fluid Flow

Diffusion and Fluid Flow What determines the diffusion coefficient? What determines fluid flow? 1. Diffusion: Diffusion refers to the transport of substance against a concentration gradient. ΔS>0 Mass

H A + B u x. Contributions to Band Broadening as described in Van Deemter Equation

Chem 2001 Summer 2004 Why Do Bands Spread? A band of solute invariably spreads as it travels through the column and emerges at the detector with a standard deviation, σ. Plate height (H) is proportional

Clinical Chemistry (CHE221) Professor Hicks Lecture 12. Chromatography. chromatography. normal phase chromatography like a tug of war

Clinical Chemistry (CHE221) Professor Hicks Lecture 12 Chromatography chromatography technique to separate mixtures each peak a different substance stationery phase = material mixtures comes in contact

Evaluating System Suitability CE, GC, LC and A/D ChemStation Revisions: A.03.0x- A.08.0x

CE, GC, LC and A/D ChemStation Revisions: A.03.0x- A.08.0x This document is believed to be accurate and up-to-date. However, Agilent Technologies, Inc. cannot assume responsibility for the use of this

Chapter 21 Principles of Chromatography and Mass Spectrometry Problems 1,2,3,4,8,9,11,13

Chapter 21 Principles of Chromatography and Mass Spectrometry Problems 1,2,3,4,8,9,11,13 21-1 What is Chromatography Process to separate and purify compounds in a mixture by passing them trough a material

Carrier Gas Selection for Capillary Gas Chromatography

Carrier Gas Selection for Capillary Gas Chromatography Len Sidisky, Greg Baney, Katherine Stenerson, and James L. Desorcie Supelco, Div. of Sigma-Aldrich, Bellefonte, PA 16823 USA T411126 www.sigma-aldrich.com

The Theory of HPLC. Gradient HPLC

The Theory of HPLC Gradient HPLC i Wherever you see this symbol, it is important to access the on-line course as there is interactive material that cannot be fully shown in this reference manual. Aims

Fractional Distillation and Gas Chromatography

Fractional Distillation and Gas Chromatography Background Distillation The previous lab used distillation to separate a mixture of hexane and toluene based on a difference in boiling points. Hexane boils

Gas Chromatography notes

Reminder: These notes are meant to supplement, not replace, the lab manual. Gas Chromatography notes Application: Gas Chromatography (GC) provides a quick, accurate and relatively inexpensive way to analyze

Physics Notes Class 11 CHAPTER 13 KINETIC THEORY

1 P a g e Physics Notes Class 11 CHAPTER 13 KINETIC THEORY Assumptions of Kinetic Theory of Gases 1. Every gas consists of extremely small particles known as molecules. The molecules of a given gas are

The van Deemter Equation:

The van Deemter Equation: A Three Act Play by Christa Colyer Department of Chemistry Wake Forest University Winston-Salem, North Carolina Playbill All too rarely does science meet with society in any kind

Experimental Studies in Temperature Programmed Gas Chromatography. Kari Rojas Urias. Thesis submitted to the Faculty of the

Experimental Studies in Temperature Programmed Gas Chromatography by Kari Rojas Urias Thesis submitted to the Faculty of the Virginia Polytechnic Institute and State University in partial fulfillment of

DETERMINING CAFFEINE CONCENTRATIONS IN SOFT DRINKS (Revised: )

INTRODUCTION DETERMINING CAFFEINE CONCENTRATIONS IN SOFT DRINKS (Revised: 5-18-93) Chromatography is a separation technique that was first used by the Russian botanist Mikhal Tsvet. Around the turn of

Fluid Mechanics: Static s Kinematics Dynamics Fluid

Fluid Mechanics: Fluid mechanics may be defined as that branch of engineering science that deals with the behavior of fluid under the condition of rest and motion Fluid mechanics may be divided into three

Mass Transfer Operations I Prof. Bishnupada Mandal Department of Chemical Engineering Indian Institute of Technology, Guwahati

Mass Transfer Operations I Prof. Bishnupada Mandal Department of Chemical Engineering Indian Institute of Technology, Guwahati Module - 1 Diffusion Mass Transfer Lecture - 6 Diffusion coefficient: Measurement

9.2.3.7 Retention Parameters in Column Chromatography

9.2.3.7 Retention Parameters in Column Chromatography Retention parameters may be measured in terms of chart distances or times, as well as mobile phase volumes; e.g., t R ' (time) is analogous to V R

Energy Transport. Focus on heat transfer. Heat Transfer Mechanisms: Conduction Radiation Convection (mass movement of fluids)

Energy Transport Focus on heat transfer Heat Transfer Mechanisms: Conduction Radiation Convection (mass movement of fluids) Conduction Conduction heat transfer occurs only when there is physical contact

Introduction to HPLC. Basic Module in Bioorganic Chemistry January 16, 2007 Johannes Ranke. Introduction to HPLC p.1/17

Introduction to HPLC Basic Module in Bioorganic Chemistry January 16, 2007 Johannes Ranke Introduction to HPLC p.1/17 Sources R. Kellner, J.-M. Mermet, M. Otto, H. M. Widmer (Eds.), Analytical Chemistry,

Separation of Mixtures by Paper Chromatography

Exercise 14 Page 1 Illinois Central College CHEMISTRY 130 Laboratory Section: Name: Separation of Mixtures by Paper Chromatography Objectives In this experiment we will effect a separation of a mixture

Factors affecting selection of mobile phase in gas Chromatography

etc. (1-3) The opinion of many chromatographers about the role of carrier gas is limited American Journal of Research Communication Factors affecting selection of mobile phase in gas Chromatography A.Y.

Separation Science - Chromatography Unit Thomas Wenzel Department of Chemistry Bates College, Lewiston ME

Separation Science - Chromatography Unit Thomas Wenzel Department of Chemistry Bates College, Lewiston ME 04240 twenzel@bates.edu The following textual material is designed to accompany a series of in-class

Laboratory exercise No. 4 Water vapor and liquid moisture transport

Laboratory exercise No. 4 Water vapor and liquid moisture transport Water vapor transport in porous materials Due to the thermal conductivity of water and other unfavourable properties and effects in porous

EXPERIMENT 7 DISTILLATION PRELAB ASSIGNMENT

EXPERIMENT 7 DISTILLATION PRELAB ASSIGNMENT 1. Prepare a Table of Physical Constants for this experiment, for the compounds, ethyl acetate and 1-butanol. Include boiling point and refractive index. 2.

FLUID DYNAMICS. Intrinsic properties of fluids. Fluids behavior under various conditions

FLUID DYNAMICS Intrinsic properties of fluids Fluids behavior under various conditions Methods by which we can manipulate and utilize the fluids to produce desired results TYPES OF FLUID FLOW Laminar or

Standard Test Method for Analysis of Hydrocarbon Waxes by Gas Chromatography EWF METHOD 001/03

1. Scope Standard Test Method for Analysis of Hydrocarbon Waxes by Gas Chromatography EWF METHOD 001/03 (Version 1 Reviewed 2015) 1. 1 This test method provides for the determination of the carbon number

1. Fluids Mechanics and Fluid Properties. 1.1 Objectives of this section. 1.2 Fluids

1. Fluids Mechanics and Fluid Properties What is fluid mechanics? As its name suggests it is the branch of applied mechanics concerned with the statics and dynamics of fluids - both liquids and gases.

TROUBLESHOOTING GUIDE HPLC

TROUBLESHOOTING GUIDE HPLC Before starting any troubleshooting, whether it is related to instruments or columns, it is essential that safe laboratory practices be observed. The chemical and physical properties

Lecture Chromo-3: Gas Chromatography. CHEM 5181 Fall 2004 Mass Spectrometry & Chromatography. Jessica Gilman and Prof. Jose-Luis Jimenez CU-Boulder

Lecture Chromo-3: Gas Chromatography CHEM 5181 Fall 2004 Mass Spectrometry & Chromatography Jessica Gilman and Prof. Jose-Luis Jimenez CU-Boulder Outline Introduction Instrument overview Carrier gas Sample

At-Column Dilution Application Notes

At-Column Dilution Application Notes 34 Maple Street Milford, MA 01757 71500078010, Revision A NOTICE The information in this document is subject to change without notice and should not be construed as

Background Information

1 Gas Chromatography/Mass Spectroscopy (GC/MS/MS) Background Information Instructions for the Operation of the Varian CP-3800 Gas Chromatograph/ Varian Saturn 2200 GC/MS/MS See the Cary Eclipse Software

PREPARATIVE CHROMATOGRAPHY

1 BOOK 12 Chrom-Ed Book Series Raymond P. W. Scott PREPARATIVE CHROMATOGRAPHY Chrom-Ed Book Series 2 Book 1 Principles and Practice of Chromatography Book 2 Gas Chromatography Book 3 Liquid Chromatography

SIZE OF A MOLECULE FROM A VISCOSITY MEASUREMENT

Experiment 8, page 1 Version of April 25, 216 Experiment 446.8 SIZE OF A MOLECULE FROM A VISCOSITY MEASUREMENT Theory Viscous Flow. Fluids attempt to minimize flow gradients by exerting a frictional force,

Vacuum Technology. Kinetic Theory of Gas. Dr. Philip D. Rack

Kinetic Theory of Gas Assistant Professor Department of Materials Science and Engineering University of Tennessee 603 Dougherty Engineering Building Knoxville, TN 3793-00 Phone: (865) 974-5344 Fax (865)

The term chromatography applies to the separation of chemical constituents

Chem 112: Gas Chromatography Page 103 Determination of Alcohols by Gas Chromatography John Schaumloffel, SUC-Oneonta Chemistry Department Introduction The term chromatography applies to the separation

Lecture 24 - Surface tension, viscous flow, thermodynamics

Lecture 24 - Surface tension, viscous flow, thermodynamics Surface tension, surface energy The atoms at the surface of a solid or liquid are not happy. Their bonding is less ideal than the bonding of atoms

Interpretation of Experimental Data

Lab References When evaluating experimental data it is important to recognize what the data you are collecting is telling you, as well as the strengths and limitations of each method you are using. Additionally,

Kinetic Molecular Theory

Kinetic Molecular Theory Particle volume - The volume of an individual gas particle is small compaired to that of its container. Therefore, gas particles are considered to have mass, but no volume. There

Gases. Gas: fluid, occupies all available volume Liquid: fluid, fixed volume Solid: fixed volume, fixed shape Others?

CHAPTER 5: Gases Chemistry of Gases Pressure and Boyle s Law Temperature and Charles Law The Ideal Gas Law Chemical Calculations of Gases Mixtures of Gases Kinetic Theory of Gases Real Gases Gases The

Separation of Amino Acids by Paper Chromatography

Separation of Amino Acids by Paper Chromatography Chromatography is a common technique for separating chemical substances. The prefix chroma, which suggests color, comes from the fact that some of the

Experiment 4 Analysis by Gas Chromatography

Experiment 4 Analysis by Gas Chromatography In this experiment we will study the method of gas chromatography. Gas chromatography (GC) is one of the most important analytical tools that the chemist has.

Hydrogen Isotope Separation Using Gas Chromatography

1 Hydrogen Isotope Separation Using Gas Chromatography Aaron Jo Victor Senior High School Advisors: Dr. Walter Shmayda, Neil Redden Dec 2012 2 Abstract A gas chromatograph (GC) system was commissioned

1.1 This test method covers the qualitative and quantitative determination of the content of benzene and toluene in hydrocarbon wax.

Standard Method for Analysis of Benzene and Toluene Content in Hydrocarbon Waxes by Headspace Gas Chromatography EWF METHOD 002/03 (Version 1 Reviewed 2015) 1 Scope 1.1 This test method covers the qualitative

Method Development for Size-Exclusion Chromatography of Monoclonal Antibodies and Higher Order Aggregates

Method Development for Size-Exclusion Chromatography of Monoclonal Antibodies and Higher Order Aggregates Paula Hong and Kenneth J. Fountain Waters Corporation, 34 Maple St., Milford, MA, USA APPLICATION

10.7 Kinetic Molecular Theory. 10.7 Kinetic Molecular Theory. Kinetic Molecular Theory. Kinetic Molecular Theory. Kinetic Molecular Theory

The first scheduled quiz will be given next Tuesday during Lecture. It will last 5 minutes. Bring pencil, calculator, and your book. The coverage will be pp 364-44, i.e. Sections 0.0 through.4. 0.7 Theory

The Determination of Benzene and Toluene in Finished Gasolines Containing Ethanol Using the PerkinElmer Clarus 680 GC with Swafer Technology

application Note Gas Chromatography Author A. Tipler PerkinElmer, Inc. Shelton, CT 06484 USA The Determination of and in Finished Gasolines Containing Using the PerkinElmer Clarus 680 GC with Swafer Technology

Chemistry 321, Experiment 8: Quantitation of caffeine from a beverage using gas chromatography

Chemistry 321, Experiment 8: Quantitation of caffeine from a beverage using gas chromatography INTRODUCTION The analysis of soft drinks for caffeine was able to be performed using UV-Vis. The complex sample

Kinetic Theory of Gases. Chapter 33. Kinetic Theory of Gases

Kinetic Theory of Gases Kinetic Theory of Gases Chapter 33 Kinetic theory of gases envisions gases as a collection of atoms or molecules. Atoms or molecules are considered as particles. This is based on

Paper Chromatography: Separation and Identification of Five Metal Cations

Paper Chromatography: Separation and Identification of Five Metal Cations Objectives Known and unknown solutions of the metal ions Ag +, Fe 3+, Co 2+, Cu 2+ and Hg 2+ will be analyzed using paper chromatography.

Chapter 5: Diffusion. 5.1 Steady-State Diffusion

: Diffusion Diffusion: the movement of particles in a solid from an area of high concentration to an area of low concentration, resulting in the uniform distribution of the substance Diffusion is process

Ferda ORHUN Mineral Research and Exploration Institute of Turkey

GAS CHROMATOGRAPHY, EXPERIMENTAL STUDIES ON THE FLOW RATE EFFECTS OF CARRIER GAS AND APPLICATION OF THE METHOD TO HYDROCARBON ANALYSIS OF SOME NATURAL GASES IN TURKEY Ferda ORHUN Mineral Research and Exploration

Guide to Reverse Phase SpinColumns Chromatography for Sample Prep

Guide to Reverse Phase SpinColumns Chromatography for Sample Prep www.harvardapparatus.com Contents Introduction...2-3 Modes of Separation...4-6 Spin Column Efficiency...7-8 Fast Protein Analysis...9 Specifications...10

Fluids and Solids: Fundamentals

Fluids and Solids: Fundamentals We normally recognize three states of matter: solid; liquid and gas. However, liquid and gas are both fluids: in contrast to solids they lack the ability to resist deformation.

THERMAL DESORPTION. Introduction and Principles. Focusing on Volatiles

THERMAL DESORPTION Introduction and Principles Thermal Desorption Thermal desorption is a simple extension of the technique of gas chromatography (GC) It involves the use of heat and a flow of inert gas

Fall 97 Test 1, P. 2

2102 Fall 97 Test 1 Fall 97 Test 1, P. 2 Fall 97 Test 1, P. 3 Fall 97 Test 1, P. 4 Fall 97 Test 1, P. 5 5. (10 points) A spherical rubber balloon has a charge uniformly distributed over is surface. The

15. The Kinetic Theory of Gases

5. The Kinetic Theory of Gases Introduction and Summary Previously the ideal gas law was discussed from an experimental point of view. The relationship between pressure, density, and temperature was later

Basic Equations, Boundary Conditions and Dimensionless Parameters

Chapter 2 Basic Equations, Boundary Conditions and Dimensionless Parameters In the foregoing chapter, many basic concepts related to the present investigation and the associated literature survey were

An HPLC Analysis of Sweeteners in Beverages

CHEM 311L Quantitative Analysis Laboratory Version 1.1 An HPLC Analysis of Sweeteners in Beverages In this laboratory exercise we will perform a separation of the components of diet soft drinks using reversed-phase

Ion Exchange Chromatography

Ion Exchange Chromatography The separation and purification of various elements by ion exchange chromatography takes advantage of the variation of the electrostatic bond energies of ions in solution. Thus,

GC METHODS FOR QUANTITATIVE DETERMINATION OF BENZENE IN GASOLINE

ACTA CHROMATOGRAPHICA, NO. 13, 2003 GC METHODS FOR QUANTITATIVE DETERMINATION OF BENZENE IN GASOLINE A. Pavlova and R. Ivanova Refining and Petrochemistry Institute, Analytical Department, Lukoil-Neftochim-Bourgas

Purification of reaction mixtures using flash chromatography.

Purification of reaction mixtures using flash chromatography. This technical note details the use of ISOLUTE Flash chromatography columns for the purification of reaction mixtures. What is flash chromatography?

Viscoelasticity of Polymer Fluids.

Viscoelasticity of Polymer Fluids. Main Properties of Polymer Fluids. Entangled polymer fluids are polymer melts and concentrated or semidilute (above the concentration c) solutions. In these systems polymer

Natural Convection. Buoyancy force

Natural Convection In natural convection, the fluid motion occurs by natural means such as buoyancy. Since the fluid velocity associated with natural convection is relatively low, the heat transfer coefficient

where h = 6.62 10-34 J s

Electromagnetic Spectrum: Refer to Figure 12.1 Molecular Spectroscopy: Absorption of electromagnetic radiation: The absorptions and emissions of electromagnetic radiation are related molecular-level phenomena

Hydrogen & Nitrogen as Alternative GC Carrier Gases SEPTEMBER/OCTOBER The Technology and Application of Industrial, Specialty and Medical Gases

SEPTEMBER/OCTOBER 2014 VOLUME 8 / ISSUE 5 The Technology and Application of Industrial, Specialty and Medical Gases Hydrogen & Nitrogen as Alternative GC Carrier Gases The Practical Impact of the Use of

Ch. 20 Electric Circuits

Ch. 0 Electric Circuits 0. Electromotive Force Every electronic device depends on circuits. Electrical energy is transferred from a power source, such as a battery, to a device, say a light bulb. Conducting

Alignment and Preprocessing for Data Analysis

Alignment and Preprocessing for Data Analysis Preprocessing tools for chromatography Basics of alignment GC FID (D) data and issues PCA F Ratios GC MS (D) data and issues PCA F Ratios PARAFAC Piecewise

GAS CHROMATOGRAPHY. DETERMINING THE PERCENT COMPOSITION OF A MIXTURE OF VOLATILE LIQUIDS USING A GAS CHROMATOGRAPH. (Revised: 1-12-93)

DETERMINING THE PERCENT COMPOSITION OF A MIXTURE OF VOLATILE LIQUIDS USING A GAS CHROMATOGRAPH. (Revised: 1-12-93) INTRODUCTION Gas chromatography is a technique used to analyze mixtures. The instrument

Chapter 8 Steady Incompressible Flow in Pressure Conduits

Chapter 8 Steady Incompressible Flow in Pressure Conduits Outline 8.1 Laminar Flow and turbulent flow Reynolds Experiment 8.2 Reynolds number 8.3 Hydraulic Radius 8.4 Friction Head Loss in Conduits of

Stability of Evaporating Polymer Films. For: Dr. Roger Bonnecaze Surface Phenomena (ChE 385M)

Stability of Evaporating Polymer Films For: Dr. Roger Bonnecaze Surface Phenomena (ChE 385M) Submitted by: Ted Moore 4 May 2000 Motivation This problem was selected because the writer observed a dependence

Chapter 6 - Solute Transport

SSC 107, Fall 00 Chapter 6 Page 6-1 Chapter 6 - Solute Transport fertilizer movement leaching of salt (reclamation) pollutant movement Transport depends on H O flow velocity Water content Soil characteristics

Detailed simulation of mass spectra for quadrupole mass spectrometer systems

Detailed simulation of mass spectra for quadrupole mass spectrometer systems J. R. Gibson, a) S. Taylor, and J. H. Leck Department of Electrical Engineering and Electronics, The University of Liverpool,

.01 1 10 100 1000 Flow Rate (ml/min)

TECHNICAL NOTE Pressure Drop of alves and Tubes from to 000 ml/min Abstract. The flow resistance of injectors, switching valves, detectors, and connecting tubes can cause problems under some conditions.

General Properties of Gases. Properties of Gases. K is for Kelvin. C is for degrees Celsius. F is for degrees Fahrenheit PROPERTIES OF GASES GAS LAWS

PROPERTIES OF GASES or GAS LAWS 1 General Properties of Gases There is a lot of empty space in a gas. Gases can be expanded infinitely. Gases fill containers uniformly and completely. Gases diffuse and

Table 9.1 Types of intrusions of a fluid into another and the corresponding terminology.

Chapter 9 Turbulent Jets SUMMARY: This chapter is concerned with turbulent jets, namely their overall shape and velocity structure. The first jets being considered are those penetrating in homogeneous

ANALYTICAL METHODS INTERNATIONAL QUALITY SYSTEMS

VALIDATION OF ANALYTICAL METHODS 1 GERT BEUVING INTERNATIONAL PHARMACEUTICAL OPERATIONS TASKS: - Internal auditing - Auditing of suppliers and contract manufacturers - Preparing for and guiding of external

Analysis of Testosterone in Serum using LC/Triple Quadrupole MS/MS

Analysis of Testosterone in Serum using LC/Triple Quadrupole MS/MS Application Note Authors M.P. George Agilent Technologies, Inc. Santa Clara, CA USA Andre Szczesniewski Agilent Technologies, Inc. Schaumburg,

A LAMINAR FLOW ELEMENT WITH A LINEAR PRESSURE DROP VERSUS VOLUMETRIC FLOW. 1998 ASME Fluids Engineering Division Summer Meeting

TELEDYNE HASTINGS TECHNICAL PAPERS INSTRUMENTS A LAMINAR FLOW ELEMENT WITH A LINEAR PRESSURE DROP VERSUS VOLUMETRIC FLOW Proceedings of FEDSM 98: June -5, 998, Washington, DC FEDSM98 49 ABSTRACT The pressure

Ch 2 Properties of Fluids - II. Ideal Fluids. Real Fluids. Viscosity (1) Viscosity (3) Viscosity (2)

Ch 2 Properties of Fluids - II Ideal Fluids 1 Prepared for CEE 3500 CEE Fluid Mechanics by Gilberto E. Urroz, August 2005 2 Ideal fluid: a fluid with no friction Also referred to as an inviscid (zero viscosity)

#13 Chromatography is a Gas: An Inquiry-Based Introduction to Gas Chromatography Cynthia N. Peck, Delta College, University Center, MI 48710

#13 Chromatography is a Gas: An Inquiry-Based Introduction to Gas Chromatography Cynthia N. Peck, Delta College, University Center, MI 48710 INTRODUCTION To close the yellow note, click once to select

Lecture 12. Physical Vapor Deposition: Evaporation and Sputtering Reading: Chapter 12. ECE 6450 - Dr. Alan Doolittle

Lecture 12 Physical Vapor Deposition: Evaporation and Sputtering Reading: Chapter 12 Evaporation and Sputtering (Metalization) Evaporation For all devices, there is a need to go from semiconductor to metal.

Chemistry, The Central Science, 11th edition Theodore L. Brown; H. Eugene LeMay, Jr.; and Bruce E. Bursten. Chapter 10 Gases

Chemistry, The Central Science, 11th edition Theodore L. Brown; H. Eugene LeMay, Jr.; and Bruce E. Bursten Chapter 10 Gases A Gas Has neither a definite volume nor shape. Uniformly fills any container.

Quantitative & Qualitative HPLC

Quantitative & Qualitative HPLC i Wherever you see this symbol, it is important to access the on-line course as there is interactive material that cannot be fully shown in this reference manual. Contents

HWR 431 / 531 HYDROGEOLOGY LAB SECTION LABORATORY 2 DARCY S LAW

HWR 431 / 531 HYDROGEOLOGY LAB SECTION LABORATORY 2 DARCY S LAW Introduction In 1856, Henry Darcy, a French hydraulic engineer, published a report in which he described a series of experiments he had performed

Analysis of Blood Alcohol by Headspace with Simultaneous GC-FID and MS Detection. No. GCMS No. SSI-GCMS-1403

Gas Chromatograph Mass Spectrometer No. GCMS-1403 Analysis of Blood Alcohol by Headspace with Simultaneous GC-FID and MS Detection Introduction Determination of Blood Alcohol Content (BAC) has been a standard

There are certain things that we have to take into account before and after we take an FID (or the spectrum, the FID is not that useful after all).

Data acquisition There are certain things that we have to take into account before and after we take an FID (or the spectrum, the FID is not that useful after all). Some deal with the detection system.

Hydrogen Carrier Gas for Total Petroleum Hydrocarbon. By Ed Connor, GC Product Specialist, Peak Scientific Instruments Ltd

Hydrogen Carrier Gas for Total Petroleum Hydrocarbon By Ed Connor, GC Product Specialist, Peak Scientific Instruments Ltd Many labs are facing helium supply difficulties and rapid price increases as a

Experiment 1: Thin Layer Chromatography

Experiment 1: Thin Layer Part A: understanding R f Part B: R f & solvent polarity Part C: R f & compound functionality Part D: identification of commercial food dye components Reading: MHS Ch. 17 pgs 219-235

The University of Toledo Soil Mechanics Laboratory

The University of Toledo Soil Mechanics Laboratory Permeability Testing - 1 Constant and Falling Head Tests Introduction In 1856 the French engineer Henri D arcy demonstrated by experiment that it is possible

Chemical Composition of Instead Electronic Cigarette Smoke Juice and Vapor

Chemical Composition of Instead Electronic Cigarette Smoke Juice and Vapor Alliance Technologies, LLC 2009 info@alliancetechgroup.com 732.355.1234 (ph) 9 December 2009 Study: Chemical Composition of Instead

ME19b. SOLUTIONS. Feb. 11, 2010. Due Feb. 18

ME19b. SOLTIONS. Feb. 11, 21. Due Feb. 18 PROBLEM B14 Consider the long thin racing boats used in competitive rowing events. Assume that the major component of resistance to motion is the skin friction

Carbon Cable. Sergio Rubio Carles Paul Albert Monte

Carbon Cable Sergio Rubio Carles Paul Albert Monte Carbon, Copper and Manganine PhYsical PropERTieS CARBON PROPERTIES Carbon physical Properties Temperature Coefficient α -0,0005 ºC-1 Density D 2260 kg/m3

IDEAL AND NON-IDEAL GASES

2/2016 ideal gas 1/8 IDEAL AND NON-IDEAL GASES PURPOSE: To measure how the pressure of a low-density gas varies with temperature, to determine the absolute zero of temperature by making a linear fit to

The Viscosity of Fluids

Experiment #11 The Viscosity of Fluids References: 1. Your first year physics textbook. 2. D. Tabor, Gases, Liquids and Solids: and Other States of Matter (Cambridge Press, 1991). 3. J.R. Van Wazer et

Addressing the World Helium Shortage For Gas Chromatography

Addressing the World Helium Shortage For Gas Chromatography Helium Conservation and Converting GC Methods to Nitrogen and Hydrogen Carrier Gas James D. McCurry, Ph.D. Senior Scientist Agilent Technologies

Min-218 Fundamentals of Fluid Flow

Excerpt from "Chap 3: Principles of Airflow," Practical Mine Ventilation Engineerg to be Pubished by Intertec Micromedia Publishing Company, Chicago, IL in March 1999. 1. Definition of A Fluid A fluid

DIFFUSION IN SOLIDS. Materials often heat treated to improve properties. Atomic diffusion occurs during heat treatment

DIFFUSION IN SOLIDS WHY STUDY DIFFUSION? Materials often heat treated to improve properties Atomic diffusion occurs during heat treatment Depending on situation higher or lower diffusion rates desired

4. Introduction to Heat & Mass Transfer

4. Introduction to Heat & Mass Transfer This section will cover the following concepts: A rudimentary introduction to mass transfer. Mass transfer from a molecular point of view. Fundamental similarity