1 Liquid Flowmeters A Guide for Selecting a Flowmeter for Pressurized Systems Selecting a meter requires a first-rate technical assessment of the pressurized system and a thorough evaluation of the capabilities and needs of the operator. Technical knowledge required for metering ranges from very basic ability to read a meter and write down the rates of diversion and total use to a high level of technical ability for automated systems that enable a user to operate/control a system from almost any location (see discussion on Supervisory Control and Data Acquisition, also called SCADA). Selecting a meter for pressurized pipe systems Experts claim that over 75 percent of the flow meters installed are not performing satisfactorily, and improper selection accounts for 90 percent of problems users have with meters. Selecting a meter for pressurized systems can be confusing because of the different types and styles available on the market. A meter must be matched to the system, and the water user has to tell the meter dealer the following basic information to make sure a meter matches the system. If not, readings and recordings will be inaccurate, and inaccuracies will not likely favor the water right holder. Know the range of flow under which the system operates. Range of flow is the lowest rate of pumping to the highest rate of pumping of a system needed for the operator. For instance, a system may need to pump as little as 20 gallons per minute for the smallest set of water to as much as 120 gallons per minute for the largest set. The lower rate may eliminate some types of meters because the low flow may be outside the range the meter can measure. In some cases, a meter that will measure the lowest rate for a system may not be able to measure the highest rate used in that system. A meter must be able to measure the instantaneous quantity (Qi) and record the total quantity of water used over time (Qa). Qi is the rate at which water is pumped, such as 20 gallons per minute, 550 gallons per minute, and so forth. The total quantity pumped over time may be recorded as gallons, tens of gallons, hundreds of gallons, or thousands of gallons. Totalizing use in gallons is standard for meters. If recording Qa in acre-feet is desired, the user must order that requirement. For most uses in agriculture, recording Qa in acre-feet is best. If totalizing Qa in acre-feet is not available, then order a meter that records in thousands of gallons, or 10 thousand gallons, if possible. Once installed, be sure the dealer explains how to read the meter and record the readings. The size of the pipe is very important. Some systems have pipe installed where the diameter of the pipe is larger than necessary. In those cases, it is not likely that a meter of reasonable cost for the system will be available. It is best to place a smaller diameter pipe in the system and install a meter in the smaller diameter section that will measure small flows. Be sure that there is enough straight pipe before and after the meter. Read the instructions for installation. Not all meters have the same requirements. The required lengths are a function of the diameter of the pipe. If, for example, a meter requires 5 times the diameter of straight pipe before the meter and 2 times the diameter after the meter, a 6-inch pipe would need 30 inches of straight pipe before the meter and 12 inches of straight pipe after the meter. If the required length of straight pipe is not available, particularly before the meter, then straightening vanes would be required or the system redesigned near the outlet of the pump to install a meter properly. Be sure the accuracy of the meter is within prescribed limits, which are plus or minus 5% of the amount actually being pumped. Most, if not all, meters are calibrated at the factory and operate well within ±5%. All meters should be inspected periodically as part of a scheduled maintenance program, especially meters that have moving parts. Generally, meters are checked for two reasons: to confirm that it s still operating or to assess its accuracy.
2 What type of meter to install It is just as important to know what a flowmeter cannot do as well as what it can do when selecting a meter. Each type and style has advantages and disadvantages, and the degree of satisfactory performance is directly related to how well the capabilities and shortcomings of an instrument is matched to the requirements of the application. Users often expect performance from a meter that is not consistent with what the supplier has provided. Most suppliers want to make sure the right flowmeter is installed and used for a particular system. Many provide questionnaires, checklists, and specification sheets designed to obtain the critical information necessary to match the correct flowmeter to the job. Contact your local conservation district administering cost-share funding to see if they have a checklist to use. Technological improvements of flowmeters must also be considered. A common mistake is to select a design that was most popular for a given application some years ago and to assume that it is still the best instrument for the job. Many changes and innovations may have occurred in recent years in the development of flowmeters for that particular application, making the choice much broader. A recent development is the availability of computer programs to perform the tedious calculations often necessary for selecting flowmeters. Calculations that used to take an hour can be performed in a matter of seconds. Considerations of Cost There are a wide range of prices for flowmeters. Prices range from approximately $50 to over $10,000. Generally, meters for large systems will cost more than those for small systems. Specific designs for an individual system can be even higher. However, total system costs must always be considered when selecting flowmeters. For example, a meter for a 6" pipeline may cost around $1200 plus installation. Adding a data logger may cost an additional $200 to $400, and telemetry and installation would add to the cost. Users should balance added cost with the efficiencies of having automatic recordings of date, telemetry, and so forth with labor savings over time. As with many products, users generally get what they pay for when purchasing a flowmeter. The satisfaction received with the product will depend on the care used in selecting and installing the instrument. That, again, involves knowing the process, the products, and the flow-metering requirements. "Overbuying" is not uncommon. Users need not buy a flowmeter more capable or complicated than they need, but neither should a meter be purchased that will not or can not satisfy future needs, such as adding telemetry. WORKING WITH FLOWMETERS Although suppliers are ready to install meters, around 75 percent of the users install their own equipment. Mistakes when installing are not uncommon. One of the most common is not allowing sufficient upstream and downstream straight-run piping for the flowmeter. Every design has some tolerance to variable velocity conditions in the pipe, but all units require a certain amount of straight pipe to provide a normal flow pattern for the device. Without it, accuracy and performance are will not be correct. With electrical components, safety is an important consideration in hazardous areas. Stray magnetic fields may exist, particularly when other electrical equipment is operating nearby. Power lines, relays, solenoids, transformers, motors, and generators all contribute their share of interference. Be sure that the flowmeter selected is immune to such interference. Problems occur primarily with the electronic components in secondary elements, which must be protected. Most flowmeter suppliers offer designs for such uses, and strict adherence to the manufacturers recommended installation practices will usually prevent such problems. Reading a meter One of the biggest sources of confusion with metering is knowing how to read a meter to determine what the meter is recording. Figure 1 is an example of what some readouts look like. All meters read and record the same designated units, but the displays may be different, as Figure 1
3 shows. Readouts usually are in gallons, 10s of gallons, 100s of gallons, 1,000s of gallons, 10,000s of gallons, and acre-feet. Always select a meter that displays in as large a number as possible, and preferably in acre-feet if available. What does the readout on a meter head look like? Rate of withdrawal or diversion in gallons per minute or cubic feet per second, the Qi Figure 1. An example of the readings on an electronic readout meter head Total amount of water used, the totalizer or Qa
4 Figure 2. What a mechanical meter head might look like and what readings are provided. For agricultural applications, totalizing in acre-feet is best. Calibration All flowmeters require an initial calibration. Most manufacturers calibrate a meter for the specified service conditions. If qualified personnel are available, the user can either perform their own calibrations or hire an outside consultant. If telemetry equipment is in place and a user uses wireless internet, for example, a user can often contact the manufacturer or other qualified people to conduct a diagnostic on the equipment over the internet and corrections made that way. The need to recalibrate usually depends on how well the meter fits the application. Some liquids passing through flowmeters tend to be abrasive, erosive, or corrosive. In time, portions of the device will deteriorate sufficiently to affect performance. Some designs are more susceptible to damage than others. For example, wear of individual turbine blades will cause performance changes. In other cases, recalibration may not be necessary for years because nothing that could change the meter's performance has occurred. Most manufacturers will recalibrate a meter in their plant or in the user's facility. Maintenance A number of factors influence maintenance requirements and the life expectancy of flowmeters. The major factor, of course, is matching the right instrument to the particular application. Poorly selected devices invariably will cause problems and/or fail at an early date. While flowmeters with no moving parts usually will require less attention than units with moving parts, all flowmeters eventually require some kind of maintenance. Flowmeters with moving parts require periodic internal inspection for wear and tear, especially if water contains sediments. Installing filters ahead of such units will help minimize fouling and wear. Meters with no moving parts, such as ultrasonic or electromagnetic meters, may develop problems with their secondary element's electronic components. Pressure sensors associated with secondary elements should be periodically removed and inspected. Applications where coatings such as calcium or iron may occur are potential problems. If the coating is insulating, the operation of magnetic flowmeters will be impaired if the coating insulates the electrodes from the flow of water. Periodic cleaning would be necessary to remove any buildup of coatings on the electrodes. With ultrasonic flowmeters, refraction angles may change and the sonic energy absorbed by the coating will cause the meter to become inoperative. In that case the inside of the pipe wall would have to be cleaned. The chart below is a general guide only. There are several other types of meters available; the chart discusses meters most commonly used in agriculture for large to small irrigation systems. General Guide for Selecting Among the More Common Types of Pressurized Pipe Flow Meters Used in Agricultural Applications
5 Type of Flowmeter Electromagnetic full pipe meter Electromagnetic insertion meter Ultrasonic (time-of-travel) Propeller Positive displacement Vortex Conditions of water suited Few limitations Few limitations Clean water with some particles or air bubbles Clean, relatively free of sediments preferred Pipe sizes 4, 6, 8, larger sizes available by order Pressure loss None Typical accuracy in percent ±0.5 % of rate 3 to 48 None ±1 % of full scale 1 to 60 or None ±2 % of full more scale 2 to 96 Low ±2 % of reading Required length of straight pipe upstream of meter 5 or less. Consult dealer and specification sheet Required length of straight pipe downstream of meter 2 or less. Consult dealer and specification sheet 10 times the diameter of the pipe 5 times the diameter of the pipe 10 times the inside diameter of 5 times the inside the pipe diameter of the pipe Cost Depends on brand and size of pipe (increases with pipe size) Low to medium Medium to high, depending on configuration needed and diameter of pipe 5 to 10 diameters 1 to 2 diameters Low to high, depending on pipe size Clean 5/8 to 2 High ±1.5 to 2 % Minimal Minimal Low Clean to dirty 2 to 20 Low to medium ±1 % of full scale 10 times the diameter of the pipe 5 times the diameter of the pipe Low to medium With most pressurized pipe flow meters, the flow rate (instantaneous quantity, or Qi) is determined by measuring the velocity of the liquid flowing in the pipe. Velocity depends on the pressure forcing the liquid through a pipe or conduit. Because the pipe's cross-sectional area is known and remains constant, the average velocity is an indication of the flow rate. Factors that affect the rate of flow include the liquid's viscosity and density, and the friction of the liquid in contact with the pipe. Flow through a pipe is described generally in Figure 1. A characteristic of laminar flow is the parabolic shape of its velocity profile, Fig. 1. Uniform laminar flow provides the most accurate measurement. Turbulent flow is caused by bends, valves, restrictions, and so forth in the pipe and provides the least accurate measurement. Each style of meter will require a certain amount of straight pipe before and after the meter to ensure laminar flow. In some applications, the piping system may have to be redesigned to make sure that both laminar flows are ensured by having the recommended length of straight pipe before and after a meter to reduce if not eliminate turbulent and non-laminar flow.
6 Non-uniform and turbulent flow can be reduced or eliminated by locating the meter where there is plenty of straight pipe, by redesigning the configuration of the pipeline where the meter is to be located, or by using straightening vanes. Figure 3. Laminar and turbulent flow. Straightening Vanes A meter should not be placed in pipe where turbulence exists because turbulence causes the meter to read the amount of water incorrectly. The error likely will not be in favor of the water right holder. Meters should be placed where turbulence is reduced or eliminated. Some meters require more length of straight pipe than others. Look at the directions for installing the meter and make sure that the minimum length of straight pipe, both before and after the meter, is available to ensure accuracy (see the table above). The length of straight pipe needed ahead of the meter is more than that required the meter. If the required amount of straight pipe is not available, straightening vanes are an option to reduce turbulence and improve accuracy for the end user. Straightening vanes need to meet requirements and standards of the American Petroleum Institute. Straightening vanes may be necessary for any meter, depending on the length of straight pipe ahead of a meter. Look at the instructions for the meter, and ask the meter installer about the need for straightening vanes. Direction of flow Figure 4. Meter with straightening vanes ahead of the meter. Installing a Meter on the Pipe There are a number of ways to install a meter on a pipe. The usual methods are:
7 Threaded pipe fitting: involves cutting a hole in the pipe, welding on a threaded pipe fitting, and screwing the meter into the pipe fitting. This type of installation is usually for insertion-type meters where the meter is inserted into the pipe (insertion means that a part of the metering system has an element of the meter inserted into the pipe flow), such as with paddle wheel meters, vortex meters, and some types of magnetic meters. Saddle mount: involves cutting a hole in the pipe and installing a saddle mounted meter. This type of installation can be used for propeller meters, paddle wheel meters, vortex meters, and some types of magnetic meters. Figure 5. Typical saddle mounts for installing a meter in a pipe Figure 6. Example of a propeller meter on a saddle mount Flanged meter: involves cutting a section of the existing pipe out of the system, installing flanges on both ends, and placing the flanged meter in place of the cut out section. Often used when straightening vanes are required. Figure 7. Example of a typical flanged meter with straightening vanes.
8 TYPES OF PRESSURIZED PIPE FLOW METERS Many types of flowmeters are available for pressurized-pipe systems. In general, meters for most water systems are positive displacement or velocity meters. Positive displacement meters include piston, oval-gear, nutating-disk, and rotary-vane types. Velocity meters consist of turbine, vortex shedding, electromagnetic, and sonic designs. Mechanical Flow Meters Mechanical meters are probably the most used type of meter for pressurized systems. There are several types of mechanical flow meters available over a range of costs. The following is not to be considered an all-inclusive list of mechanical flow meters. There are others; this discussion only provides a brief overview of meters commonly used in many applications. Propeller meters Propeller meters are the most common type of meter in use for measuring water at the source in pressurized water delivery systems. The basic unit consists of a rotating propeller mounted in the pipe with a meter for measuring the rate of pumping and the volume of water pumped. The revolutions of the propeller are counted and the rate and volume measured. The rotational speed is a direct function of flow rate and can be sensed by magnetic pick-up, photoelectric cell, or gears. Propeller meters are relatively accurate and meet the needs of many users. The readout on the meter provides Qa and Qi, and does not require power to operate unless equipped with an electronic readout for Qi and Qa. Major concerns with propeller meters are wear on the bearings and turbidity in water causing excessive wear on the propellers or plugging of bearings with sediment. Propeller meters are not recommended when water is dirty and contains abrasive material such as sand. A rule of thumb is that if the impellers on a pump have to be replaced often (for example, every two years or so), then propeller meters are probably not the best choice due to abrasive wear on the meter propeller. Propeller meters can be equipped with electronics for SCADA (see SCADA webpage). Figure 8. Flanged propeller meter with straightening vanes Figure 9. Saddle mount propeller meter
9 Positive-Displacement Meters are mechanical meters that separate liquids into accurately measured increments representing a discrete volume. Each segment is counted by a connecting register. Positive-displacement units work well for small pipe systems 2 or smaller and where a simple mechanical meter system satisfies requirements. With all positive-displacement meters, variations in viscosity below a given threshold will affect measuring accuracies (viscosity is the resistance to flow, honey is more viscous than water). Many sizes and capacities of positive-displacement meters are available. Figure 10. Example of a positive displacement meter. Vortex meters make use of a natural phenomenon that occurs when a liquid flows around a bluff object. Eddies or vortices are shed alternately downstream of the object, much like a flag flapping in the breeze. The faster the wind blows, the more the flag flaps. The flapping in liquid flows is called shedding. The frequency of the shedding is directly proportional to the velocity of the liquid flowing through the meter. The more frequent the shedding, the higher the rate of flow of water. Figure 11. Example of vortex meters Bluff body The three major components of the flowmeter are a bluff body strut-mounted across the bore of the flowmeter, a sensor to detect of the vortex and generate an electrical impulse, and signal amplification and conditioning transmitter whose output is proportional to the rate of flow. The meter is equally suitable for rate of flow or for totalizing flow measurements. Use for slurries or high viscosity liquids is not recommended. Electromagnetic meters can handle most liquids and slurries, providing that the material being metered is can conduct electricity. Major components are the flow tube (primary element). The flow tube mounts directly in the pipe. Pressure drop across the meter is the same as it is
10 through an equivalent length of pipe because there are no moving parts or obstructions to the flow. The voltmeter can be attached directly to the flow tube or can be mounted remotely and connected to it by a shielded cable. Figure 12. Examples of full pipe size mag meters with readout/datalogger capabilities Electromagnetic flowmeters operate on Faraday's law of electromagnetic induction that states that a voltage will be induced when a conductor moves through a magnetic field. The liquid serves as the conductor; the magnetic field is created by energized coils outside the flow tube. The amount of voltage produced is directly proportional to the rate of flow. Two electrodes mounted in the pipe wall detect the voltage, which is measured by the secondary element. Electromagnetic flowmeters have major advantages: they can measure difficult and corrosive liquids and slurries, and they can measure forward as well as reverse flow with equal accuracy. They also have no moving parts to wear. Electromagnetic flowmeters are also available as insertion meters. Magnetic insertion meters use a probe inserted into the pipe, usually through a hot tap arrangement. Insertion meters have to be installed carefully as the diameter of the pipe affects the depth the probe has to extend into the pipe. Insertion meters work well when expensive major modifications would be required to install a full pipe meter. Figure 13. Examples of two types of magnetic insertion meters Ultrasonic flowmeters are Doppler meters and time-of-travel (or transit) meters. Doppler meters measure the frequency shifts caused by liquid flow. Two transducers are mounted in a case attached to one side of the pipe. A signal of known frequency is sent into the liquid to be measured. Solids, bubbles, or any discontinuity in the liquid, cause the pulse to be reflected to the receiver element. Because the liquid causing the reflection is moving, the frequency of the returned pulse is shifted. The frequency shift is proportional to the liquid's velocity. A portable Doppler meter capable of being operated on AC power or from a rechargeable power pack is available. The sensing heads are clamped to the outside of the pipe, and the instrument is ready to be used. A set of 4 to 20 millampere output terminals permits the unit to be connected to a strip chart recorder or other remote device.
11 Figure 14. Example of ultrasonic meters installed on a pipe. Time-of-travel meters have transducers mounted on each side of the pipe. The configuration is set so that the sound waves traveling between the devices are at a 45 deg. angle to the direction of liquid flow. The speed of the signal traveling between the transducers increases or decreases with the direction of transmission and the velocity of the liquid being measured. A time-differential relationship proportional to the flow can be obtained by transmitting the signal alternately in both directions. A limitation of time-of-travel meters is that the liquids being measured must be relatively free of entrained gas or solids to minimize signal scattering and absorption.
Flow Measurement Options for Pipeline and Open Channel Flow October 2013 Presented by Molly Skorpik - 2013 Montana Association of Dam and Canal Systems Conference Irrigation Training and Research Center
Chapter 10 Flow Measurements Material from Theory and Design for Mechanical Measurements; Figliola, Third Edition Flow Rate Flow rate can be expressed in terms of volume flow rate (volume/time) or mass
Energy and Flow Measurement for Hydronic Systems Presented By: George Szyszko Applications Consultant MCR for ONICON Incorporated Why Measure Thermal Energy? To control something, you must first measure
Overview of the state of the art heat measurement technologies for larger solar thermal systems Overview of the state of the art heat measurement technologies for larger solar thermal systems... 1 Introduction...
Grant Agreement No. 228296 SFERA Solar Facilities for the European Research Area SEVENTH FRAMEWORK PROGRAMME Capacities Specific Programme Research Infrastructures Integrating Activity - Combination of
Flow and Flow rate. Laminar and Turbulent flow Laminar flow: smooth, orderly and regular Mechanical sensors have inertia, which can integrate out small variations due to turbulence Turbulent flow: chaotic
STATIC WATER LEVEL MEASUREMENT FLOW METER USER MANUAL www.enoscientific.com Eno Scientific 1606 Faucette Mill Rd Hillsborough, NC 27278 USA www.enoscientific.com 919-778-2660 Copyright Notice Copyright
White paper ABB Flowmeters Review of industrial processing flowmeters Understanding flowmeters and their pros and cons By Ron DiGiacomo, ABB Measurement Products One of the most critical measurements in
FLOW METER SPECIFICATION AND APPLICATION ANALYSIS Class #41 Curtis Gulaga Business Development Manager CB Engineering Ltd. #20, 5920 11 th Street SE Calgary, AB T2H 2M4 1.0 Introduction Today s working
ME 120 Experimental Methods Flow Measurement 02MAY06 Fluid Flow Measurement Classification of flow rate measurements Volume flow Industrial process control Heat transfer and cooling Mass flow Industrial
Magnetic Flowmeter Fundamentals TDS 3032A00 English September 1995 Magnetic Flowmeter Fundamentals INTRODUCTION TO MAGNETIC FLOWMETERS The magnetic flowmeter is one of the most flexible and universally
Measuring Irrigation Flow Ron E. Sheffield, Biological and Agricultural Engineering, LSU AgCenter Christopher G. Henry, Biological and Agricultural Engineering, University of Arkansas David Bankston, Food
Fluid Flow Instrumentation In the physical world, mechanical engineers are frequently required to monitor or control the flow of various fluids through pipes, ducts and assorted vessels. This fluid can
INLINE flowmeter for continuous flow measurement Economic integration in pipe systems without any additional piping 3-wire frequency pulse version to directly interface with PLC s (both PNP and NPN) Connection
Flow Measurement in Pipes and Ducts, Course #503 Presented by: PDH Enterprises, LLC PO Box 942 Morrisville, NC 27560 www.pdhsite.com This course is about measurement of the flow rate of a fluid flowing
ABE18 Selection and Use of Water Meters for Irrigation Water Measurement 1 Melissa C. Baum, Michael D. Dukes, and Dorota Z. Haman 2 The demand for water by agriculture, industry, urban users, and recreation
WaterMAG Electromagnetic Flowmeter for Water Applications OVERVIEW The WaterMAG is an electromagnetic flowmeter for water measurement applications based on field proven Azbil Corporation's technologies.
instrumentation for fluids Electromagnetic sensor Series FLOMID FX Instructions Manual! Conforms with the Pressure Equipment Directive 97/23/EC. 0830 This equipment is considered as being a pressure accessory
IFC 070 Technical Datasheet Electromagnetic signal converter Battery driven with low energy consumption Quick and easy to install and operate External data logger and GSM module for remote data transfer
Rev 8/15 AW/GW CHE 253M Experiment No. 3 LIQUID FLOW MEASUREMENT The objective of this experiment is to familiarize the student with several types of flow-measuring devices commonly used in the laboratory
GE Measurement & Control Solutions Sentinel LCT Panametrics Ultrasonic Flowmeter for Liquid Custody Transfer Measurement Benefits Extremely reliable and highly accurate flowmeter, designed specifically
Electromagnetic Flow Meter Description The EL 2200 series of electromagnetic sensors represent the state of the art for the accurate measurement for water cycle and process applications. Flow meters are
Know the Flow Flowmetering Training Manual Acknowledgments ANCID and its member organisations National Program for Irrigation Research and Development Goulburn Murray Water Training Centre Manly Hydraulics
BULLETIN BULLETIN EM20607 VORTEX IN-LINE FLOW METERS Design Features For Multi-Parameter Meters see m Principles of Operation n No moving parts to wear or fail. n Electronics can be remotely mounted up
Page 1 of 6 April 1998 Plant Engineering SELECTING NEW TECHNOLOGY: PROBLEM SOLVING Using meters to measure steam flow Jesse L. Yoder, Ph.D., Senior Analyst, Automation Research Corp., Dedham, MA Key Concepts
Non-invasive CLAMP-ON ultrasonic flowmeter for liquids Model CU100: non-invasive installation flowmeter, with transducers installed outside the pipe Installation in horizontal or vertical pipe Easy and
Meters Frederick C. Schwarz Badger / Instromet, LLC. 3731 Briar Park Suite. 100 Houston, TX 77042 Introduction All rotary piston meters, commonly known as rotary meters, utilize the fundamental 1846 lobed
Flow Assurance & Operability Erosion due to sand production Date Business Name Overview 1. What causes erosion? 2. Sand production and transport 3. Sand management 4. Instrumentation / monitoring of sand
Thermal Mass meter and Temperature Transmitter Description The MTI10 insertion thermal mass flowmeter and the MTL10 in-line thermal mass flowmeter provide accurate mass flow measurement of clean, dry gases
Chapter2. Installation Power Supply Requirements The ULM-1000C operates from a DC supply of 20 30V and will typically draw less than 0.08A. All electronic products are susceptible to electrostatic shock,
Technical Publication Flow Meter Selection for Improved Gas Flow Measurements: Comparison of Differential Pressure and Thermal Dispersion Technologies Art Womack, Sr. Applications Engineer Fluid Components
Transit Time Flow Meters TT-TDS-100H-HS ECEFast offers amazing prices on Transit Time Flow Meter portable kits. We are buying from the manufacturer who supplies 60% of the Chinese market and the majority
LLB-OCMFM 2000 PORTABLE DOPPLER AREA-VELOCITY FLOWMETER System for flow measurement, data acquisition and transmission for wastewater with long battery life With Long Life Battery Open Channel Flow Meter
College of Technology Instrumentation and Control Module # 4 Flow Measurement Document Intent: The intent of this document is to provide an example of how a subject matter expert might teach Flow Measurement.
INDIAN STANDARDS FOR NATURAL GAS PIPELINE SYSTEM 1. IS 15663(Part 1):2006 This code covers requirements and recommendations for the design, materials, construction and testing of pipelines made of steel
Thermatel Thermal Dispersion Mass Flow Measurement Handbook Table of Contents Introduction...1 What is Mass Flow Measurement?...1 Types of Flow Transmitters...4 Differential Pressure...4 Orifice...4 Venturi...6
FUNDAMENTALS OF GAS TURBINE METERS John Gorham Sensus Metering Systems 805 Liberty Blvd. DuBois, PA 15801 INTRODUCTION The majority of all gas measurement used in the world today is performed by two basic
Open Channel & Partially Filled Pipe Ultrasonic Flow Meters DYNAMETERS series DMDF-OP has three types as below: DMDF-OP-A for partially filled pipe application; DMDF-OP-B for partially filled pipe, channel
KFF/KFG Low-Volume Flow Meter User Instructions KFG_KFF_manual_0304 1. General Information Please read this service manual carefully before unpacking and readying the unit for operation, and follow the
THE THERMAL FLOW METER, A GAS METER FOR ENERGY MEASUREMENT Kazuto Otakane, Tokyo Gas Co., Ltd Katsuhito Sakai, Tokyo Gas Co., Ltd Minoru Seto, Tokyo Gas Co., Ltd 1. INTRODUCTION Tokyo Gas s new gas meter,
Turbine Flowmeters for Gas Applications Description Flow Technology s FT Series turbine flowmeters utilize a proven flow measurement technology to provide exceptionally reliable digital outputs. Because
Understanding the Flo-Dar Flow Measuring System Independent tests verify non-contact flowmeter is highly accurate under both open channel and surcharge conditions TM Flo-Dar is the only non-contact open
Accurate Air Flow Measurement in Electronics Cooling Joachim Preiss, Raouf Ismail Cambridge AccuSense, Inc. E-mail: email@example.com Air is the most commonly used medium to remove heat from electronics
Digital inductive conductivity transmitter Optimal solution for conductivity measurements in difficult fluids (polluted, dirty,...) PEEK/PPA version for CIP applications Large range of process connections
FAQ s on Siemens Magmeters 1. The cables for the coils and electrodes for the magmeters look the same. Does it matter which cable is used for coils and which one is used for the electrodes? Both cables
The Do s and Don ts of Pressure Transducers ABSTRACT When specifying a pressure transducer for a process measurement, a number of items have to be considered. Some of the more important ones are discussed
A Practical Overview of Level Measurement Technologies Martin Bahner Director of Marketing Drexelbrook Engineering Co. 205 Keith Valley Rd. Horsham, PA 19044 Abstract: There are multiple technologies available
FLR 1000 & 1000-BR Series Flow Sensors & Meters For Gases READ THIS MANUAL COMPLETELY BEFORE ATTEMPTING TO CONNECT OR OPERATE YOUR FLOW SENSOR. FAILURE TO DO SO MAY RESULT IN INJURY TO YOU OR DAMAGE TO
w Technical Product Information Subminiature Load Cell 1. Introduction... 2 2. Preparing for use... 2 2.1 Unpacking... 2 2.2 Using the instrument for the first time... 2 2.3 Grounding and potential connection...
MODEL 03002 WIND SENTRY FEBRUARY 1999 MANUAL PN 03002-90 INTRODUCTION MODEL 03002 The Wind Sentry Anemometer and Vane measure horizontal wind speed and wind direction. The small size, simplicity, and corrosion
Description Air Eliminators and Combination Air Eliminator Strainers are designed to provide separation, elimination and prevention of air in piping systems for a variety of installations and conditions.
CEE 410: Hydraulic Engineering Lab 2 Flow Measurement Test at Unit Well 30 November 26 th 2014 Tyler Gates Bill Hussey Megan Long Senn Mettel Introduction This laboratory experiment involved the collecting
Building Intelligence into Water Metering WPW-WAT00-051101 White Paper NEW TECHNOLOGIES CAN REDUCE ENERGY CONSUMPTION AND OPERATING COSTS FOR WATER UTILITIES, WHILE IMPROVING REVENUE AND PERFORMANCE. Water
Steven R. McManus Environmental Product Support Teledyne Isco Flow Measurement Applications and Technologies. Methods of flow measurement Technologies of flow measurement Flow Rate Applications Channel
Gas Fluxi 2000/TZ Turbine Gas Meter Turbine gas meters are flow meters. The flow of gas turns a turbine wheel, and thus, the rotating speed of the turbine is proportional to the linear speed of the gas.
EXPERIMENT NUMBER 7 PERFORMANCE TEST OF A CENTRIFUGAL PUMP OBJECTIVE The primary objectives of this experiment is to measure the performance of a centrifugal pump and compare the results to the manufacturers
APPENDIX A (IFGS) SIZING AND CAPACITIES OF GAS PIPING (This appendix is informative and is not part of the code.) A.1 General. To determine the size of piping used in a gas piping system, the following
6. FLOW MEASUREMENT Contents Page A. Evaluation of Permittee's Objectives and Requirements... 6-1 Evaluation of Facility-Installed Flow Devices and Data... 6-1 Evaluation of Permittee Data Handling and
Streaming Potential System (SPT1000, SPD1000) User's Manual Sentrol Systems, Inc. 3949 Cotswold Dr. SW, Lilburn, GA 30047-2371 Tel: 770-564-1541, Fax: 770-564-8605 www.sentrolsystems.com Since the streaming
ABSTRACT: PIPELINE INSPECTION UTILIZING ULTRASOUND TECHNOLOGY: ON THE ISSUE OF RESOLUTION By, M. Beller, NDT Systems & Services AG, Stutensee, Germany Today, in-line inspection tools are used routinely
Catalog CG/FLOW/001 EN Rev. B catalog Flow products Whatever your need our team will keep your measurement and plant performing Measurement made easy Introduction When it comes to ensuring your flowmeters
Applied Fluid Mechanics Sixth Edition Robert L. Mott University of Dayton PEARSON Prentkv Pearson Education International CHAPTER 1 THE NATURE OF FLUIDS AND THE STUDY OF FLUID MECHANICS 1.1 The Big Picture
3-06 INTRODUCTION INDUSTRIAL PROCESS AIR HANDLING & MEASUREMENT Air Velocity/Volume Measurement The movement of air is a factor in almost all modern industrial facilities. Industrial air movement is usually
WATER METER CALIBRATION, REPAIR, AND REPLACEMENT PROGRAM EPD Guidance Document AUGUST 2007 TABLE OF CONTENTS Part 1: OVERVIEW OF WATER METERING 3 SUMMARY 3 NECESSITY OF METERS 3 TYPES OF WATER METERS 4
White Paper FSG-WP-0012 February 2014 Fiscal Measurement Natural Gas Fiscal Measurement How much? Fiscal Measurement must not be confused with Custody Transfer; in fact, fiscal measurement is a more general
Fiberstars Lighted Laminar Flow Fountain Installation Manual 79-15037-00 REV Xx http://www.fiberstars.com Page 1 of 8 SAVE THESE DIRECTIONS! These directions are provided to ensure the proper installation
Connection Technology Center, Inc. 590 Fishers Station Drive Victor, New York 14564 Toll Free: (800) 999-5290 Phone: (585) 924-5900 Fax: (585) 924-4680 Industrial Vibration Analysis for Predictive Maintenance
SECTION 15318 - CHLORINATION SYSTEM PART 1 - GENERAL 1.1. SCOPE OF WORK A. Under this item the Contractor shall furnish and install a sodium hypochlorite solution metering system complete with all accessories
TM M711 Rev. B Mark V Target Strain Gage Flowmeter Installation, Operation and Maintenance Manual Table of Contents Installation General... 2 Retractable... 3 Preparation of Pipeline for Probe Style Meters...
Data Sheet Electro-Magnetic Flowmeters COPA-XT 2-Wire DT43 2-Wire Operation Significantly reduces installation costs when compared to a traditional 4-wire magmeter. ±0.5% of rate system accuracy. Patented
1 Abstract The goal of this project was to create precise optical flow meters for the oil & gas industry optimized for measuring flare gas flow in explosive atmospheres. The flow meters had to incorporate
STANDARD SPECIFICATIONS SECTION 02512 CLEANING AND LINING WATER MAINS PART 1 GENERAL 1.1 DESCRIPTION A. Section includes requirements for cleaning and lining existing cast iron and ductile iron water mains
DIRECT CURRENT GENERATORS Revision 12:50 14 Nov 05 INTRODUCTION A generator is a machine that converts mechanical energy into electrical energy by using the principle of magnetic induction. This principle
Introduction To SCADA and Telemetry Joe Mullaney Senior I&C Engineer MSE Technology Applications, Inc. Tetragenics Division firstname.lastname@example.org Overview Definitions What is SCADA? What is Telemetry?
Flowmeter for Gases and Installation in Existing Pipes Type can be combined with... Depth scale for accurate installation Usable in pipes from 1/2 " up to 12" (DN300) Easy installation under pressure Analogue
KEY FACTORS AFFECTING WATER METER ACCURACY Francisco Arregui*, Enrique Cabrera Jr., Ricardo Cobacho, Jorge García-Serra * email@example.com. Institute for water technology. Universidad Politecnica de