ABB Flowmeters Review of industrial processing flowmeters

Save this PDF as:

Size: px
Start display at page:

Transcription

1 White paper ABB Flowmeters Review of industrial processing flowmeters Understanding flowmeters and their pros and cons By Ron DiGiacomo, ABB Measurement Products One of the most critical measurements in the processing industries is rate of flow. In 2010, the ARC Advisory Group estimated the total instrumentation market (not including process analytical devices) to be \$13.4 billon. The chemical industry accounted for \$2.5 billion of this figure. Flowmeters account for nearly 30% of the amount spent on these measurement products. Flow metering technologies fall into four classifications: velocity, inferential, positive displacement, and mass. This article summarizes the considerations in selecting and applying these flowmeters, and provides examples of the kinds of flowmeters in each category. Velocity meters Many kinds of flowmeters on the market sense a fluid's average velocity through a pipe. Multiplying the measured average velocity by the cross-sectional area of the meter or pipe results in volumetric flow rates. For example, if the average fluid velocity is 2.5 ft/sec and the inside diameter of the pipe or flowmeter is 12 inches (0.79 sq ft area), the volumetric flow rate equals (2.5 x 0.79) or 1.98 cu ft/sec or about 14.8 gal/sec. When specifying velocity meters, chemical engineers Figure 1. Fluid velocity profiles for turbulent and laminar flow. must be concerned with the fluid's velocity profile in the pipe, which depends on piping geometry and Reynolds number. Assuming sufficient straight piping runs, the cross-sectional view shown in Figure 1 illustrates two flow profile situations: turbulent (red) and laminar (blue). With relatively small piping friction loss and low fluid viscosity, the flow profile of velocities is uniform across the entire cross-section of the pipe--called fully developed turbulent flow. In this case, the fluid velocity at the pipe walls closely matches the fluid velocity at the center and at all points in-between. The velocity at any point is the average velocity. This condition results when the Reynolds number is 10,000 or above. Calculating volumetric flow rates is relatively easy, as noted above. Figure 2. Pipe fittings such as elbows, tees, and valves will distort the flow profile. But depending on the pipe diameter and the fluid's density, viscosity, and momentum (variables affecting the Reynolds number), the flow velocity within a pipe can vary significantly between the pipe wall and its center. The average velocity through a pipe becomes increasingly difficult to measure precisely when Reynolds numbers are low. For long, straight pipe runs and low Reynolds numbers, the flow velocity would be highest at the pipe's center, and trail off in symmetrical fashion toward the pipe wall; such conditions are typical of laminar flow profiles. With an insertion probe flowmeter, engineers can establish the fluid velocities across the diameter of a pipe at multiple cross-sectional locations. By determining the 1 ABB

2 A sampling of velocity flowmeters and their principle of operation Electromagnetic flowmeters subject conductive liquids to alternating or pulsating DC magnetic fields. Electrodes on either side of the pipe wall pick up the induced voltage following Faraday's Law, which is proportional to fluid velocity. Swirl meters are similar to vortex meters, except vanes at the inlet swirl the flow, creating the pressure variations. Straightening vanes at the outlet de-swirl the flow. Ultrasonic meters come in two types. The Doppler flowmeter sends an ultrasonic beam into the flow and measures the frequency shift of reflections from discontinuities in the flow. Transit-time flowmeters have an ultrasonic transmitter and receiver separated by a known distance. The difference in transit time for a signal aided by the flow versus the signal moving against the flow is a function of fluid velocity. Vortex meters place a bluff obstacle in the flow stream, which creates vortices or eddies whose frequency is proportional to flow velocity. Sensors detect and count the pressure variations produces over a fixed time. Turbine meters contain a turbine. The flow against the turbine's vanes causes the turbine to rotate at a rate proportional to flow velocity. A sensor detects the rotational rate. pipe s actual flow profile, and integrating the data to determine the mean flow velocity, they can check on how accurately a flowmeter measures the true flow rate. Manufacturers will specify the length of straight pipe upstream and downstream of a velocity flowmeter for achieving high accuracies. But often plant piping geometries in a chemical plant will be such that sufficiently long straight pipe runs are not feasible. The flowmeter may have to be located near an elbow, tee, valve, or change in pipe diameter. In this case the flow will not be fully developed and result in the distorted profile one example shown in Figure 2. Various flow straightening devices (engineered, bundled tubes) installed upstream of the flowmeter can help correct these distortions by creating uniform flow profiles, permit ting average velocity to be inferred. The most practical liquid pipeline flow rates in the chemical and processing industries range from 0.5 to 12 ft/sec, providing a range (turndown) of 24:1. Lower rates can be difficult to measure accurately and higher rates result in higher pressure drops, pumping energy costs, and erosion (if abrasive solids are present). In the case of pipelines carrying gases, the practical flow velocities range from 15 to 200 ft/sec, a range of about 13:1. Many actual applications have flow rate ranges well within these extremes. Velocity flowmeters include electromagnetic, vortex, Swirl, turbine, and ultrasonic types, as indicated in the diagrams at right and on the next page. Inferential flowmeters An inferential flowmeter calculates flow rates based on a 2 ABB

3 Orifice plates--these are the most common DP element in the processing industries. Their flow characteristics are well documented in the literature. They're inexpensive and available in a variety of materials. The rangeability, however, is less than 5:1, and accuracy is moderate -- 2 to 4 percent of full scale. Maintenance of good accuracy requires a sharp edge to the upstream side, which degrades with wear. Pressure loss is high relative to other DP elements. Nozzles--These elements mimic the properties of the Venturi. They come in three standard, documented types: ISA 1932 nozzle, common outside of the U.S.; the long radius nozzle; and Venturi nozzle, which combines aspects of the other two Flow tubes--the ASME defines flow tubes as any DP element whose design differs from the classic Venturi, which includes short-form Venturies, nozzles, and wedges. Flow tubes come in several proprietary shapes, but all tend to be more compact than the classic and short-form Venturies. Being proprietary, flow tubes vary in configuration, tap locations, differential pressure, and pressure loss for a given flow. Their manufacturers must supply test data for flow tube. 3 ABB Flowmeters based on differential pressure measurements Venturi meter-- Characterized by a gradual tapered restriction on inlet and outlet, this element has high discharge coefficients near the ideal of 1. Pressure loss is minimal. It finds use primarily in water and wastewater applications and has limited acceptance in the processing industries. Rangeability of about 6:1 is better than orifice plates. Performance characteristics are well documented. Wedge--This element consists of a V- shaped restriction molded into the top of the meter body. This basic meter has been on the market for more than 40 years, demonstrating its ability to handle tough, dirty fluids. The slanted faces of the wedge provide self-scouring action and minimize damage from impact with secondary phases. Wedge meter rangeability of 8:1 is relatively high for a DP element. Accuracies are possible to ± 0.5 percent of full scale. Pitot tubes-- These are lowcost DP elements used to measure fluid flow, especially air flow in ventilation and HVAC systems. They work by converting the kinetic energy of the flow velocity into potential energy (pressure). Engineers can easily insert pitot tubes into existing piping, minimizing installation costs. One type makes a measurement at a point within the pipeline or ductwork, requiring knowledge of the flow profile. Another, photo, contains multiple orifices, providing an averaging effect.

4 non-flow measurement that has widely accepted correlations to rate of flow. Differential pressure--most of these flow measurement devices depend on three principles. First, despite the restriction in a pipe, the overall flowrate remains the same, which pertains to the continuity equation. Second, Bernoulli's Law says the fluid flow velocity (kinetic energy) through the restriction must increase. Third, the law of conservation of energy says the increased kinetic energy comes at the expense of fluid pressure (potential energy). The unrecoverable pressure drop across the restriction is a function of the fluid velocity, which can be calculated. Variables in the calculation of flow rate for differential flowmeters include: - the square root of the measured differential pressure - the fluid density, - pipe cross sectional area - area through the restriction - a coefficient that's specific to the device. When a fluid passes through a restriction in a pipe it does not follow the contour of restriction perfectly. It produces a jet stream that's narrower than the restrictive bore. The smaller jet diameter results in a faster stream velocity through the restriction, resulting in higher pressure loss than if the fluid perfectly followed the contour of the bore. Calculated flow rates from measured pressure drop and a known restriction bore diameter would tend to overstate the fluid flow rate. So the rate must be corrected downward from the ideal discharge coefficient of 1. The overall flow coefficient applied to the basic equation is often specific to both the device and the application, and depends on additional factors involving gas expansion and velocity of approach. This coefficient (K factor) can range from 0.6 to 0.98 for DP flowmeters. Flowmeters based on differential pressure represent a popular choice in the processing industries, constituting nearly 30 percent of installations. They have good application flexibility since they can measure liquid, gas, and steam flows and are suitable for extreme temperatures and pressures with moderate pressure losses. These losses depend on restriction size and type (orifice; wedge; pitot; Venturi, etc) and can be quite high and permanent given a low enough Beta ratio. (Beta ratio is the diameter of the restrictive orifice divided by the pipe diameter). Accuracy ranges from 1 percent to 5 percent. Compensation techniques can improve accuracy to 0.5 percent to 1.5 percent. On the other hand restrictive flowmeter piping elements are relatively expensive to install. Their dependence on the square root of differential pressure can severely diminish rangeability. Additionally they require an instrument to measure differential pressure and compute a standard flow signal. Changes in temperature, pressure, and viscosity can significantly affect accuracy of DP flowmeters. And while they have no moving parts, maintenance can be intensive. Variable area meters Often called rotameters, these are another kind of inferential flowmeter. Simple and inexpensive, these devices provide practical flow measurement solutions for many applications. They basically consist of two components: a tapered metering tube and a float that rides within the tube. The float position, a balance of upward flow and float weight, is a linear function of flow rate. Operators can take direct readings based on the float position with transparent glass and plastic tubes. Rotameters having metal tubes include a magnetically coupled pointer to indicate float position. Rotameters are easy to install and maintain, but must be mounted perfectly vertical. Accuracy (± 2 percent of full scale) is relatively low and depends on precise knowledge of the fluid and process. They're also susceptible to vibration and plugging by solids. They apply primarily to flow rates below 200 gpm and pipe sizes less than 3 inches. Target meters These flowmeters insert a physical target within the fluid flow. The moving fluid deflects a force bar attached to the target. The deflection depends on the target area, as well as the fluid density and velocity. Target meters measure flows in line sizes above 0.5 inches. By changing the target size and material, engineers can adapt them to different fluids and flow rate ranges. In most cases their calibration must be verified in the field. Target meters are relatively uncommon. They're found 4 ABB

5 primarily on water, steam applications and on wet gases. Positive displacement flowmeters These are true volumetric flow devices, measuring the actual fluid volume that passes through a meter body with no concern for velocity. Accordingly, fluid velocity, pipe internal diameters, and flow profiles are not a concern. Volume flow rate is not calculated but rather measured directly. These flowmeters capture a specific volume of fluid and pass it to the outlet. The fluid pressure moves the mechanism that empties one chamber as another fills. Residential gas meters are a common example. Counting the cycles of rotational or linear motion provides a measure of the displaced fluid. A transmitter converts the counts to true volumetric flow rate. Some examples include: - Single or multiple reciprocating piston meters. - Oval-gear meters with synchronized, close fitting teeth. - Movable nutating disks mounted on a concentric sphere located in spherical side-walled chambers. - Rotary vanes creating two or more compartments and sealed against the meter's housing. Engineers can apply these flowmeters to a wide range of non-abrasive fluids, including high-viscosity fluids. Examples include heating oils, lubrication oils, polymer additives, and ink. Accuracy may be up to +/- 0.1 percent of full scale with a rangeability of 70:1 or better. They require no power and can handle high pressures. Positive displacement flowmeters cannot handle solids, entrapped air in liquids, or entrained liquids in gases. They are expensive to install and maintain, having many moving parts. Pressure drop across the meters is high. Direct measurement of mass flow rates In the processing industries two kinds of flowmeters directly measure mass rates of fluid flow: Coriolis flowmeters for liquids and gases and thermal flowmeters for gases. Chemical processes are generally concerned more with rates of mass flow because reactions within the plant depend on mass rather than volume ratios. Coriolis flowmeters--in the early 1800s Gustave-Gaspard Coriolis, a French engineer and mathematician, discovered and described Coriolis forces. These forces come into play on rotating (or oscillating) bodies. Since the earth is a rotating body, Coriolis forces Figure 3. With flow, Coriolis forces twist the oscillating tube. The amount of twist is proportional to the mass flow. affect the weather, ballistics, and oceanography. To get an understanding of this principle, suppose you stood very near the north pole of the earth. The rotational distance you would travel over 24 hours would be relatively small. But as you walked toward the equator, you would gain rotational speed. At the equator your rotational distance traveled in 24 hours would be about 25,000 miles, amounting to a rotational speed of more than 1000 miles per hour. Obviously, while walking away from the rotational axis, you would be experiencing acceleration. Any acceleration requires a force--in this case the Coriolis force. Physics tells us that force equals mass times acceleration. So the force developed is proportional to mass. Commercial Coriolis flowmeters are a relatively recent innovation, having emerged in the mid to late 70s. Steady technical improvements since then have greatly increased their acceptance in the process industries. No other flow device is more versatile and capable. Aside from measuring mass flow rates, Coriolis flowmeters can provide simultaneous outputs for volumetric flow rate, total flow, density, temperature, and percent concentration. These meters are unaffected by flow profiles or viscosity, so they don't require long runs of straight pipe upstream and downstream. The fluid flow can be turbulent, laminar, or anything in between. The fluid can be viscous or freely flowing. Additionally, mass is not affected by changes in temperature or pressure. Accuracies can be as high as +/- 0.05% of rate. Purchase prices are relatively high but falling as these meters become more popular. Pressure drop through the meters is relatively high because of circuitous tube geometries, and, typically, its separation into two tubes. Available sizes are up to 12 inch. Entrained gases can be problematical, so control valves should be 5 ABB

6 downstream to keep pressure on the meter to prevent emergence of gas bubbles. Coriolis flowmeters are somewhat sensitive to vibration, but this can often by overcome by harmonic studies and sophisticated signal processing. Since rotating flow tubes are impractical, Coriolis flowmeters resort to oscillation. Usually a single tube or dual tubes oscillating 180 degrees out of phase take the fluid away from the axis of oscillation and back again. The Coriolis forces Figure 4. Coriolis flow tubes come in a variety of shapes, depending on the manufacturer. Above is the S-shape of ABB's Coriolis flowmeter. developed within the fluid push against the elastic tubes, twisting them. Strategically mounted magnetic pickup coils measure the degree of tube twist or distortion, which corresponds to the mass flow rate. At zero flow rate no Coriolis forces are developed, so the tubes retain their normal shape. With flow, signals from the pickoff coils experience a difference in phase that's proportional to the mass flow rate. In short, a Coriolis flowmeter comprises the following parts: - flow tube or tubes that take the fluid away from and back toward the axis of oscillation - a flow splitter to divert the fluid into two flow tubes - a drive coil to oscillate the flow tubes at their natural (resonant) frequency - pickoff coils that measure the distortion of the tubes - a resistance thermometer (RTD) to measure tube temperature, which can affect the elasticity of the tubes and thus their degree of twisting. The resonant frequency developed by the drive coil depends on the mass that's oscillating. Since the tube mass and volume are constant, this frequency is also a measure of the fluid's density. Measurement of fluid density is a bonus provided by Coriolis flowmeters. Applications in the chemical industry for Coriolis flowmeters include: - custody transfer - critical process control - filling and dosing - reactor charging - blending - loading and unloading. Their accuracy makes Coriolis flowmeters obvious candidates for custody transfer. The capability for a single flowmeter to measure a variety of different fluids suggests applications such as batch operations and tanker truck loading/unloading. For anything sold by weight, they can replace load cells for filling and dosing. They're good candidates for material balances and blending by weight. Thermal mass flow meters-- Introducing heat into fluids (mostly gases) offers a way to measure their mass flow rates. The heat dissipated by the flow stream, using temperature sensors, is a measure the mass flow rate. Thermal mass flowmeters have no moving parts, are easy to install, and provide a relatively unobstructed flow path, Since they're measuring mass, temperature and pressure corrections are unnecessary. They are accurate over a wide range of gaseous flow rates. But because they essentially measure flow at a point within the gas stream, they require some flow conditioning or knowledge of the flow profile. Figure 5. Thermal mass flowmeters introduce heat into the flow and measure its dissipation. They measure at a point within the gas stream. Manufacturers of thermal mass flowmeters use two different methods to measure heat dissipation. Both depend on the principle that higher mass flow rates have a greater cooling effect on the sensors. - Constant temperature differential--this technique uses a heated sensor (generally a resistance temperature detector) upstream from another RTD that measures gas temperature. The electrical power needed to maintain the same temperature difference between the heated and unheated sensor is a function of the mass flow. 6 ABB

7 - Constant current--here the electrical current to heat the upstream sensor is kept constant. The downstream sensor measures the heat gain. In this case mass flow is a function of the temperature difference between the two sensors. Applications include boiler control, biogas measurements, compressed air accounting, pharmaceuticals, pneumatics, and the food and beverage industries. RONALD W. DiGIACOMO manages business development for flow technologies in North America for ABB Inc. (125 E. County Line Rd., Warminster, PA 18974; Phone: (215) ; He has 25 years of experience in process instrumentation and control, primarily in flow measurement. WP_FLOWMETER_SELECTION_ Contact: ABB Instrumentation 125 East County Line Road Warminster, PA USA Tel: Fax: Copyright ABB All rights reserved 7 ABB

Fluid Flow Measurement Selection and Sizing (ENGINEERING DESIGN GUIDELINE)

KLM Technology Group Sdn. Bhd. Guidelines for Processing Plant Page : 1 of 80 (ENGINEERING DESIGN GUIDELINE) Author: A L Ling Checked by: Karl Kolmetz TABLE OF CONTENT INTRODUCTION 5 Scope 5 Flow Measurement

Michael Montgomery Marketing Product Manager Rosemount Inc. Russ Evans Manager of Engineering and Design Rosemount Inc.

ASGMT / Averaging Pitot Tube Flow Measurement Michael Montgomery Marketing Product Manager Rosemount Inc. Russ Evans Manager of Engineering and Design Rosemount Inc. Averaging Pitot Tube Meters Introduction

Chapter 10. Flow Rate. Flow Rate. Flow Measurements. The velocity of the flow is described at any

Chapter 10 Flow Measurements Material from Theory and Design for Mechanical Measurements; Figliola, Third Edition Flow Rate Flow rate can be expressed in terms of volume flow rate (volume/time) or mass

Energy and Flow Measurement for Hydronic Systems

Energy and Flow Measurement for Hydronic Systems Presented By: George Szyszko Applications Consultant MCR for ONICON Incorporated Why Measure Thermal Energy? To control something, you must first measure

Grant Agreement No. 228296 SFERA. Solar Facilities for the European Research Area SEVENTH FRAMEWORK PROGRAMME. Capacities Specific Programme

Grant Agreement No. 228296 SFERA Solar Facilities for the European Research Area SEVENTH FRAMEWORK PROGRAMME Capacities Specific Programme Research Infrastructures Integrating Activity - Combination of

CHE 253M Experiment No. 3 LIQUID FLOW MEASUREMENT

Rev 8/15 AW/GW CHE 253M Experiment No. 3 LIQUID FLOW MEASUREMENT The objective of this experiment is to familiarize the student with several types of flow-measuring devices commonly used in the laboratory

Laminar and Turbulent flow. Flow Sensors. Reynolds Number. Thermal flow Sensor. Flow and Flow rate. R = Mass Flow controllers

Flow and Flow rate. Laminar and Turbulent flow Laminar flow: smooth, orderly and regular Mechanical sensors have inertia, which can integrate out small variations due to turbulence Turbulent flow: chaotic

Using meters to measure steam flow

Page 1 of 6 April 1998 Plant Engineering SELECTING NEW TECHNOLOGY: PROBLEM SOLVING Using meters to measure steam flow Jesse L. Yoder, Ph.D., Senior Analyst, Automation Research Corp., Dedham, MA Key Concepts

Flow Measurement in Pipes and Ducts. Flow Measurement in Pipes and Ducts, Course #503. Presented by:

Flow Measurement in Pipes and Ducts, Course #503 Presented by: PDH Enterprises, LLC PO Box 942 Morrisville, NC 27560 www.pdhsite.com This course is about measurement of the flow rate of a fluid flowing

Flow Measurement Options for Pipeline and Open Channel Flow

Flow Measurement Options for Pipeline and Open Channel Flow October 2013 Presented by Molly Skorpik - 2013 Montana Association of Dam and Canal Systems Conference Irrigation Training and Research Center

Outdated Publication, for historical use. CAUTION: Recommendations in this publication may be obsolete.

IRRIGATION MANAGEMENT S E R I E S Irrigation Water Measurement Danny H. Rogers Extension Agricultural Engineer Richard D. Black Extension Agricultural Engineers Cooperative Extension Service Kansas State

FLUID FLOW Introduction General Description

FLUID FLOW Introduction Fluid flow is an important part of many processes, including transporting materials from one point to another, mixing of materials, and chemical reactions. In this experiment, you

Technical Publication

Technical Publication Flow Meter Selection for Improved Gas Flow Measurements: Comparison of Differential Pressure and Thermal Dispersion Technologies Art Womack, Sr. Applications Engineer Fluid Components

Aeration Air & Digester Gas Flow Metering Using Thermal Mass Technology. HWEA 2011 Conference Craig S. Johnson

Aeration Air & Digester Gas Flow Metering Using Thermal Mass Technology HWEA 2011 Conference Craig S. Johnson Presentation Overview Introduction Aeration Air & Digester gas challenges Gas flow metering

Liquid Flowmeters A Guide for Selecting a Flowmeter for Pressurized Systems

Liquid Flowmeters A Guide for Selecting a Flowmeter for Pressurized Systems Selecting a meter requires a first-rate technical assessment of the pressurized system and a thorough evaluation of the capabilities

The Unique Accelabar Flow Meter

The Unique Accelabar Flow Meter The Accelabar is a new and unique flow meter that combines two differential pressure technologies to produce operating ranges never before attainable in a single flow meter.

191: Calibration and Standards in Flow Measurement. Richard Paton National Engineering Laboratory, Scotland, UK 1 GENERAL PRINCIPLES

191: Calibration and Standards in Measurement Richard Paton National Engineering Laboratory, Scotland, UK 1 General Principles 1 2 Gravimetric Calibration of Liquid meters 2 3 Volumetric Calibration of

Vortex Shedding Meters Class # Curtis Gulaga Business Development Manager CB Engineering Ltd. #20, Street S.E. Calgary, Alberta, Canada

Vortex Shedding Meters Class # 8150 Curtis Gulaga Business Development Manager CB Engineering Ltd. #20, 5920 11 Street S.E. Calgary, Alberta, Canada 1.0 Introduction Vortex meters have proven to be repeatable,

FLOW METER SPECIFICATION AND APPLICATION ANALYSIS Class #41

FLOW METER SPECIFICATION AND APPLICATION ANALYSIS Class #41 Curtis Gulaga Business Development Manager CB Engineering Ltd. #20, 5920 11 th Street SE Calgary, AB T2H 2M4 1.0 Introduction Today s working

THE THERMAL FLOW METER, A GAS METER FOR ENERGY MEASUREMENT

THE THERMAL FLOW METER, A GAS METER FOR ENERGY MEASUREMENT Kazuto Otakane, Tokyo Gas Co., Ltd Katsuhito Sakai, Tokyo Gas Co., Ltd Minoru Seto, Tokyo Gas Co., Ltd 1. INTRODUCTION Tokyo Gas s new gas meter,

Thermal Dispersion Mass Flow

Thermatel Thermal Dispersion Mass Flow Measurement Handbook Table of Contents Introduction...1 What is Mass Flow Measurement?...1 Types of Flow Transmitters...4 Differential Pressure...4 Orifice...4 Venturi...6

UNIVERSITY OF MINNESOTA DULUTH DEPARTMENT OF CHEMICAL ENGINEERING ChE MEASUREMENTS WITH FLOW METERS

UNIVERSITY OF MINNESOTA DULUTH DEPARTMENT OF CHEMICAL ENGINEERING ChE 3211-4211 MEASUREMENTS WITH FLOW METERS OBJECTIVE The purpose of this experiment is to calculate the coefficient of discharge from

Flow Sensors. - mass flow rate - volume flow rate - velocity. - stream line parabolic velocity profile - turbulent vortices. Methods of measurement

Flow Sensors Flow - mass flow rate - volume flow rate - velocity Types of flow - stream line parabolic velocity profile - turbulent vortices Methods of measurement - direct: positive displacement (batch

COMPACT CORIOLIS MASS FLOW METER/CONTROLLER FOR LOW FLOWS

COMPACT CORIOLIS MASS FLOW METER/CONTROLLER FOR LOW FLOWS W. Jouwsma Bronkhorst Cori-Tech B.V. Nijverheidsstraat -, 71 AK Ruurlo, The Netherlands Phone: +31 573 59; Fax: +31 9375; Internet: www.bronkhorst-cori-tech.com

A LAMINAR FLOW ELEMENT WITH A LINEAR PRESSURE DROP VERSUS VOLUMETRIC FLOW. 1998 ASME Fluids Engineering Division Summer Meeting

TELEDYNE HASTINGS TECHNICAL PAPERS INSTRUMENTS A LAMINAR FLOW ELEMENT WITH A LINEAR PRESSURE DROP VERSUS VOLUMETRIC FLOW Proceedings of FEDSM 98: June -5, 998, Washington, DC FEDSM98 49 ABSTRACT The pressure

measurement, but almost any pipe elbow can be calibrated Elbow meters are not as potentially accurate as venturi,

Lecture 14 Flow Measurement in Pipes I. Elbow Meters An elbow in a pipe can be used as a flow measuring device much in the same way as a venturi or orifice plate The head differential across the elbow

The Versatile Differential Pressure Transmitter. By David Gunn Honeywell Process Solutions

The Versatile Differential Pressure Transmitter By David Gunn Honeywell Process Solutions The Versatile Differential Pressure Transmitter 2 Table of Contents Abstract... 3 Pressure Fundamentals... 3 Applications...

Advanced Differential Pressure Flowmeter Technology V-CONE FLOW METER TECHNICAL BRIEF

Advanced Differential Pressure Flowmeter Technology V-CONE FLOW METER TECHNICAL BRIEF Table of Contents Section 1 - General Introduction 1.1 1 Principles Of Operation 1.2 1 Reshaping The Velocity Profile

Energy Systems Engineering Technology

College of Technology Instrumentation and Control Module # 4 Flow Measurement Document Intent: The intent of this document is to provide an example of how a subject matter expert might teach Flow Measurement.

df dt df dt df ds df ds

Principles of Fluid Mechanics Fluid: Fluid is a substance that deforms or flows continuously under the action of a shear stress (force per unit area). In fluid mechanics the term fluid can refer to any

Fluid Flow Instrumentation

Fluid Flow Instrumentation In the physical world, mechanical engineers are frequently required to monitor or control the flow of various fluids through pipes, ducts and assorted vessels. This fluid can

ME 120 Experimental Methods. Flow Measurement. BJ Furman 02MAY06

ME 120 Experimental Methods Flow Measurement 02MAY06 Fluid Flow Measurement Classification of flow rate measurements Volume flow Industrial process control Heat transfer and cooling Mass flow Industrial

A basic introduction to steam

A basic introduction to steam FOR HOT, COLD, MOIST AND DRY, FOUR CHAMPIONS FIERCE. STRIVE HERE FOR MASTERY Milton 1666 Steam Wonderful Steam Very high heat content Recyclable Clean, non toxic Biodegradable

COMBIMASS. Technical Data COMBIMASS eco-bio +

COMBIMASS Technical Data THE SYSTEM COMBIMASS The field transmitters of the COMBIMASS eco series are suitable for gas flow measurement and cover a wide range of different applications. The instruments

Selection Guide for Mechanical Differential Pressure Gauges

Selection Guide for Mechanical Differential Pressure Gauges A common specialized pressure gauge that uses the input from two pressure sources is the differential pressure gauge. A differential pressure

F-1100 Series Single Turbine Meters Suitable for pipe sizes 1 ¼ and larger.

ONICON Insertion Turbine Flow Meters Provide an accurate, cost effective option for flow measurement of relatively clean liquids such as those found in closed loop HVAC systems. Each meter is built to

PARTIALLY CLOSED VALVE EFFECTS ON THE V-CONE FLOWMETER

PARTIALLY CLOSED VALVE EFFECTS ON THE V-CONE FLOWMETER Stephen A. Ifft (McCrometer Inc., Hemet, CA 9545, USA) ABSTRACT Research conducted from 199 through 1996 indicates that the V-Cone flowmeter is less

Flow System Calibration in Medical and Bio-medical Production Processes.

Flow System Calibration in Medical and Bio-medical Production Processes. Speaker / Author: Mark Finch Fluke Calibration Norfolk, Norwich, NR6 6JB, UK e-mail: mark.finch@fluke.com Phone: + 44 (0)1908 678

Head Loss in Pipe Flow ME 123: Mechanical Engineering Laboratory II: Fluids

Head Loss in Pipe Flow ME 123: Mechanical Engineering Laboratory II: Fluids Dr. J. M. Meyers Dr. D. G. Fletcher Dr. Y. Dubief 1. Introduction Last lab you investigated flow loss in a pipe due to the roughness

Applied Fluid Mechanics

Applied Fluid Mechanics Sixth Edition Robert L. Mott University of Dayton PEARSON Prentkv Pearson Education International CHAPTER 1 THE NATURE OF FLUIDS AND THE STUDY OF FLUID MECHANICS 1.1 The Big Picture

GOOD PRACTICE GUIDE AN INTRODUCTION TO DIFFERENTIAL-PRESSURE FLOW METERS

GOOD PRACTICE GUIDE AN INTRODUCTION TO DIFFERENTIAL-PRESSURE FLOW METERS www.tuvnel.com Introductory guide to differential-pressure flow meters This introductory guide to differential-pressure (Δp) flow

Effects of Beta Ratio and Reynold s Number on Coefficient of Discharge of Orifice Meter

J Agric Rural Dev 7(&2), 5-56, June 29 ISSN 8-86 K wl I cj x Dbœqb zj Available online at http://www.banglajol.info/index.php/jard JARD Journal of Agriculture & Rural Development Effects of Beta Ratio

F ox W hi t e Paper. Reduce Energy Costs and Enhance Emissions Monitoring Systems

F ox W hi t e Paper Reduce Energy Costs and Enhance Emissions Monitoring Systems A Technical White Paper from Fox Thermal Instruments Rich Cada, VP Sales & Marketing, Fox Thermal Instruments, Inc. 399

Advances in Thermal Dispersion Mass Flow Meter Accuracy

Advances in Thermal Dispersion Mass Flow Meter Accuracy By Dan McQueen President Fluid Components International (FCI) Advances in Thermal Dispersion Mass Flow Meter Accuracy By Dan McQueen President Fluid

Fundamentals of Mass Flow Control

Fundamentals of Mass Flow Control Critical Terminology and Operation Principles for Gas and Liquid MFCs A mass flow controller (MFC) is a closed-loop device that sets, measures, and controls the flow of

Overview of the state of the art heat measurement technologies for larger solar thermal systems

Overview of the state of the art heat measurement technologies for larger solar thermal systems Overview of the state of the art heat measurement technologies for larger solar thermal systems... 1 Introduction...

Annex A (informative) Specification forms tables of contents. Table A.1 Operating parameter device categories and form titles

17 ISA-TR20.00.01-2007 Annex A (informative) Specification forms tables of contents Table A.1 Operating parameter device categories and form titles Form Number Form Title Development Status, ISA20 20A1001

Experiment # 3: Pipe Flow

ME 05 Mechanical Engineering Lab Page ME 05 Mechanical Engineering Laboratory Spring Quarter 00 Experiment # 3: Pipe Flow Objectives: a) Calibrate a pressure transducer and two different flowmeters (paddlewheel

FUNDAMENTALS OF GAS TURBINE METERS John Gorham Sensus Metering Systems 805 Liberty Blvd. DuBois, PA 15801

FUNDAMENTALS OF GAS TURBINE METERS John Gorham Sensus Metering Systems 805 Liberty Blvd. DuBois, PA 15801 INTRODUCTION The majority of all gas measurement used in the world today is performed by two basic

Testing Protocol for Differential Pressure Measurement Devices API MPMS Chapter 22.2

Testing Protocol for Differential Pressure Measurement Devices API MPMS Chapter 22.2 Steve Baldwin Chevron Energy Technology Company Houston, Texas USA Casey Hodges CEESI Measurement Solutions. Nunn, Colorado

HEAVY OIL FLOW MEASUREMENT CHALLENGES

HEAVY OIL FLOW MEASUREMENT CHALLENGES 1 INTRODUCTION The vast majority of the world s remaining oil reserves are categorised as heavy / unconventional oils (high viscosity). Due to diminishing conventional

Tadahiro Yasuda. Introduction. Overview of Criterion D200. Feature Article

F e a t u r e A r t i c l e Feature Article Development of a High Accuracy, Fast Response Mass Flow Module Utilizing Pressure Measurement with a Laminar Flow Element (Resistive Element) Criterion D200

UNIT Engineering: Measurement Technology Flow (SCQF level 6)

National Unit Specification: general information CODE H0W6 12 SUMMARY This Unit can be delivered as part of a National Qualification Group Award but can also be taken as a free-standing Unit. This Unit

Custody Transfer Measurement. with the V-Cone Flowmeter

Custody Transfer Measurement with the V-Cone Flowmeter Stephen A. Ifft McCrometer Inc. Hemet, California, USA Abstract This paper will discuss the approval of the McCrometer V-Cone flowmeter for custody

Industry faces unrelenting pressure to increase production

Reprinted with permission from CEP (Chemical Engineering Progress), March 2013. Copyright 2013 American Institute of Chemical Engineers (AIChE). Back to Basics Coriolis: The Direct Approach to Mass Flow

Conditioning Orifice Plate Technology; Taking the Standard to a New Level of Capability

Technical Note Conditioning Orifice Plate Conditioning Orifice Plate Technology; Taking the Standard to a New Level of Capability Conditioning Orifice Plates, the 1595 Conditioning Orifice Plate and the

Flue gas measurement and control Partial-insertion pitot tubes in CEMS applications

Article AT/FLOW/002 EN Flue gas measurement and control Partial-insertion pitot tubes in CEMS applications Measurement made easy Specifying the most appropriate pitot flowmeter design in stack applications

KNOWLEDGE. Understanding and Selecting Coriolis Technology for Drilling Fluid Monitoring MICRO MOTION WHITE PAPER. Abstract. Applying the Technology

MICRO MOTION WHITE PAPER BY DENVER SMART, CHRIS RUSSELL, MARK SIMONS MICRO MOTION, INC. Understanding and Selecting Coriolis Technology for Drilling Fluid Monitoring KNOWLEDGE Abstract Continuous, accurate

Gas Custody Transfer Calibration

Gas Custody Transfer Calibration Using multi variable temperature / pressure calibrators for flowmeter calibration 2013 Introduction Gas custody transfer flow computers require special calibration to perform

FIGURE P8 50E FIGURE P8 62. Minor Losses

8 48 Glycerin at 40 C with r 1252 kg/m 3 and m 0.27 kg/m s is flowing through a 4-cm-diameter horizontal smooth pipe with an average velocity of 3.5 m/s. Determine the pressure drop per 10 m of the pipe.

The Buyer s Guide to Ultrasonic Flowmeters

Spire Metering Technology The Buyer s Guide to Ultrasonic Flowmeters Selecting the Right Flowmeter for Your Liquid Application Introduction Today s advanced ultrasonic flow meters enable the accurate measurement

Introduction Process Piping and Instrumentation Diagram (P&ID)

Session 2313 Use of Lab Experiments to Build Transport Concepts Abstract Introduction Process Piping and Instrumentation Diagram (P&ID) Viscosity Measurement Absolute Viscosity vs. Sucrose Conce ntration

ngenieros Ingenieros Asociados De Control S.L.

ngenieros sociados de ontrol 1 Company founded in 1985 to serve commercial and technical assistance in the fields of instrumentation, control and installation for all types of industries. We offer top

CASE STUDY. Accurate Flow Meter Helps Sugar Mill Measure Steam Consumption of Distillery Plant Producing Bio-Ethanol

CASE STUDY Accurate Flow Meter Helps Sugar Mill Measure Steam Consumption of Distillery Plant Producing Bio-Ethanol Page 1 Accurate Flow Meter Helps Sugar Mill Measure Steam Consumption of Distillery Plant

THEORETICAL UNCERTAINTY OF ORIFICE FLOW MEASUREMENT page 1 THEORETICAL UNCERTAINTY OF ORIFICE FLOW MEASUREMENT INTRODUCTION COEFFICIENT OF DISCHARGE

THEORETICAL UNCERTAINTY OF ORIFICE FLOW MEASUREMENT page 1 DANIEL MEASUREMENT AND CONTROL WHITE PAPERS THEORETICAL UNCERTAINTY OF ORIFICE FLOW MEASUREMENT www.daniel.com INTRODUCTION Orifice meters are

Minor losses include head losses through/past hydrants, couplers, valves,

Lecture 10 Minor Losses & Pressure Requirements I. Minor Losses Minor (or fitting, or local ) hydraulic losses along pipes can often be estimated as a function of the velocity head of the water within

V-Cone, An Alternative to Orifice Meter in Wet Gas Applications

V-Cone, An Alternative to Orifice Meter in Wet Gas Applications Stephen A. Ifft McCrometer Inc. Hemet, California, USA Abstract This paper will discuss the use of the V-Cone differential pressure flowmeter

COUNTERBALANCE VALVES

COUNTERBALANCE VALVES Introduction They are modulating valves which allow free flow into the actuator and then block the reverse flow until they feel a pilot pressure inversely proportional to the load

Applied Fluid Mechanics

Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and

Contents. Microfluidics - Jens Ducrée Physics: Fluid Dynamics 1

Contents 1. Introduction 2. Fluids 3. Physics of Microfluidic Systems 4. Microfabrication Technologies 5. Flow Control 6. Micropumps 7. Sensors 8. Ink-Jet Technology 9. Liquid Handling 10.Microarrays 11.Microreactors

CENTRIFUGAL PUMP OVERVIEW Presented by Matt Prosoli Of Pumps Plus Inc.

CENTRIFUGAL PUMP OVERVIEW Presented by Matt Prosoli Of Pumps Plus Inc. 1 Centrifugal Pump- Definition Centrifugal Pump can be defined as a mechanical device used to transfer liquid of various types. As

Accurate Air Flow Measurement in Electronics Cooling

Accurate Air Flow Measurement in Electronics Cooling Joachim Preiss, Raouf Ismail Cambridge AccuSense, Inc. E-mail: info@degreec.com Air is the most commonly used medium to remove heat from electronics

ENERGY TRANSFER SYSTEMS AND THEIR DYNAMIC ANALYSIS

ENERGY TRANSFER SYSTEMS AND THEIR DYNAMIC ANALYSIS Many mechanical energy systems are devoted to transfer of energy between two points: the source or prime mover (input) and the load (output). For chemical

Sanitary Type Electronic Flow Switches

Industries Pharmaceutical Sanitary Type Electronic Flow Switches Food and Beverage Biotechnology Semiconductor Water Treatment S Series Ameritrol, Inc. Instruments and Controls Features No Moving Parts

DIRECT STEAM INJECTION HOT WATER SYSTEMS FOR JACKETED HEATING

By Philip Sutter Pick Heaters, Inc. DIRECT STEAM INJECTION HOT WATER SYSTEMS FOR JACKETED HEATING INTRODUCTION Many process plants currently use steam or hot water to heat jacketed devices such as tanks,

Testing methods applicable to refrigeration components and systems

Testing methods applicable to refrigeration components and systems Sylvain Quoilin (1)*, Cristian Cuevas (2), Vladut Teodorese (1), Vincent Lemort (1), Jules Hannay (1) and Jean Lebrun (1) (1) University

Theory overview of flow measurement using differential pressure devices based on ISO-5167 standard.

Theory overview of flow measurement using differential pressure devices based on ISO-567 standard. rian FL40 flow computer description. Flow Cad software users manual. Technical note, Differential pressure

THEORY AND APPLICATION OF KURZ THERMAL CONVECTION MASS FLOW METERS INSTRUMENTS

THEORY AND APPLICATION OF KURZ THERMAL CONVECTION MASS FLOW METERS INTRODUCTION Many versions of thermal convection mass flow meters have been made, starting in the early 1900 s. Originally labeled, hot

A Review of the Revisions to API 14.3 / AGA 3 Part 2

A Review of the Revisions to API 14.3 / AGA 3 Part 2 Tom Cathey Wedge Measurement & Control, Inc. 1415 Louisiana Suite 1500, Houston Texas 77056 Introduction In April of 2000, revisions to the specification

Class A Foam Mixing and Application Equipment

Class A Foam Mixing and Application Equipment Adding Phos-Chek Class A foam concentrate to water: There are many methods of adding Phos-Chek Class A foam concentrate to water. The different methods have

DEVELOPMENT OF HIGH SPEED RESPONSE LAMINAR FLOW METER FOR AIR CONDITIONING

DEVELOPMENT OF HIGH SPEED RESPONSE LAMINAR FLOW METER FOR AIR CONDITIONING Toshiharu Kagawa 1, Yukako Saisu 2, Riki Nishimura 3 and Chongho Youn 4 ABSTRACT In this paper, we developed a new laminar flow

INDIAN STANDARDS FOR NATURAL GAS PIPELINE SYSTEM

INDIAN STANDARDS FOR NATURAL GAS PIPELINE SYSTEM 1. IS 15663(Part 1):2006 This code covers requirements and recommendations for the design, materials, construction and testing of pipelines made of steel

Chapter 5 MASS, BERNOULLI AND ENERGY EQUATIONS

Fluid Mechanics: Fundamentals and Applications, 2nd Edition Yunus A. Cengel, John M. Cimbala McGraw-Hill, 2010 Chapter 5 MASS, BERNOULLI AND ENERGY EQUATIONS Lecture slides by Hasan Hacışevki Copyright

Take total control of your HVAC system with cost-saving flow and thermal energy meters

Take total control of your HVAC system with cost-saving flow and thermal energy meters Because you can t manage what you don t measure usa.siemens.com/pi Answers for Industry. Contents Introduction 03

Control ball valves for severe services. Author: Michele Ferrante, PARCOL S.p.A., Italy

Control ball valves for severe services Author: Michele Ferrante, PARCOL S.p.A., Italy Control valves are primarily classified according to the type of their obturator motion which can be linear or rotary.

Direct Mounting Allows Differential Pressure (DP) Based Flow Measurement Optimization

Direct Mounting Allows Differential Pressure (DP) Based Flow Measurement Optimization Gregory Robert Strom, Marketing Product Manager Gregory Livelli, Manager of Worldwide Pressure Marketing Rosemount,

Application of the Orifice Meter for Accurate Gas Flow Measurement page 1. Application of the Orifice Meter for Accurate Gas Flow Measurement.

Application of the Orifice Meter for Accurate Gas Flow Measurement page 1 DANIEL MEASUREMENT AND CONTROL WHITE PAPER Application of the Orifice Meter for Accurate Gas Flow Measurement www.daniel.com Summary

IFC 070 Technical Datasheet

IFC 070 Technical Datasheet Electromagnetic signal converter Battery driven with low energy consumption Quick and easy to install and operate External data logger and GSM module for remote data transfer

Flow. 1.1 Introduction. 1.2 Units of Flow. 1.3 Measurement of Flow

1 Flow 1.1 Introduction Industrial flow measurements include measuring of flow rate of solids, liquids and gases. There are two basic ways of measuring flow ; one on volumetric basis and the other on weight

EXPERIMENT NUMBER 7 PERFORMANCE TEST OF A CENTRIFUGAL PUMP

EXPERIMENT NUMBER 7 PERFORMANCE TEST OF A CENTRIFUGAL PUMP OBJECTIVE The primary objectives of this experiment is to measure the performance of a centrifugal pump and compare the results to the manufacturers

CO 2 41.2 MPa (abs) 20 C

comp_02 A CO 2 cartridge is used to propel a small rocket cart. Compressed CO 2, stored at a pressure of 41.2 MPa (abs) and a temperature of 20 C, is expanded through a smoothly contoured converging nozzle

INLINE flowmeter for continuous flow measurement

INLINE flowmeter for continuous flow measurement Economic integration in pipe systems without any additional piping 3-wire frequency pulse version to directly interface with PLC s (both PNP and NPN) Connection

Industrial Instrumentation Course Catalog

Industrial Instrumentation Course Catalog 603-01 Instrumentation and Control Overview When you complete this lesson, you will be able to describe the basic operating principles behind the instrumentation

Heating Water by Direct Steam Injection

Heating Water by Direct Steam Injection Producing hot water by direct steam injection provides a solution where large volumes of hot water at precise temperatures are required, and where energy and space

T U R B I N E G A S M E T E R

TURBINE GAS METER TURBINE GAS METER CGT 1 2 3 4 5 6 7 Design and function page 2 General technical data page 3 Measurement outputs page 4 Dimensions and weights page 5 Performance page 7 Pressure loss

AP2 Fluids. Kinetic Energy (A) stays the same stays the same (B) increases increases (C) stays the same increases (D) increases stays the same

A cart full of water travels horizontally on a frictionless track with initial velocity v. As shown in the diagram, in the back wall of the cart there is a small opening near the bottom of the wall that

Applied Fluid Mechanics

Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and

4.What is the appropriate dimensionless parameter to use in comparing flow types? YOUR ANSWER: The Reynolds Number, Re.

CHAPTER 08 1. What is most likely to be the main driving force in pipe flow? A. Gravity B. A pressure gradient C. Vacuum 2.What is a general description of the flow rate in laminar flow? A. Small B. Large