# Cable Impedance and Structural Return Loss Measurement Methodologies

Save this PDF as:

Size: px
Start display at page:

## Transcription

1 Cable Impedance and Structural Return Loss Measurement Methodologies Introduction Joe Rowell Joel Dunsmore and Les Brabetz Hewlett Packard Company Santa Rosa, California Two critical electrical specifications for CATV cable are well known: cable impedance and structural return loss. However, as common as these specifications are, the specifications associated with any cable depend upon both the specification definitions and the measurement methodology. This paper introduces exact definitions for cable impedance and structural return loss alongasfg with several measurement methodologies. Several enhancements to traditional measurement methods are introduced which compensate for the largest source of errors. The uncertainties associated with each method are discussed. Definitions of Cable Impedance and Structural Return Loss In the most general terms, cable impedance is the ratio of the voltage to current of a signal traveling in one direction down the cable. In coaxial cable, the value of the impedance will depend upon the ratio of the inner and outer conductor diameters, and the dielectric constant of the material between the inner and outer conductors. The value of the conductivity will affect the impedance to the extent that RF signals do not travel on the surface of the conductor, but propagate into the conductor by what is known as the skin depth. The finite conductivity also causes losses that increase with RF frequency, and this can change the effective cable impedance. Finally, the construction of the cable can change along the length of the cable, with differences in conductor thickness, dielectric material and outer conductor diameter changing due to limitations in manufacturing. Thus the cable impedance can vary along the length of the cable. The extent to which the manufacturing imperfections degrade cable performance is characterized by the specification Structural Return Loss (or SRL). Structural return loss is the ratio of incident signal to reflected signal in a cable. This definition implies a known incident and reflected signal. In practice, the SRL is loosely defined as the reflection coefficient of a cable referenced to the cable's impedance. The reflection seen at the input of a cable, which

2 contributes to SRL, is the sum of all the tiny reflections along the length of the cable. In terms of cable impedance, the SRL can be defined mathematically as: ρ SRL (ω) = Z in (ω) Z cable Z in (ω) + Z cable eq. 1 Z in is the impedance seen at the input of the cable, and Z cable is the nominal cable impedance. Cable impedance is a specification that is defined only at a discrete point along the cable, and at a discrete frequency. However, when commonly referred to, the impedance of the cable is some average of the impedance over the frequency of interest. Structural return loss, on the other hand, is the cumulative result of reflections along a cable as seen from the input of the cable. The above definitions need to be expressed in a more rigorous form in order to apply a measurement methodology. One definition of cable impedance is that impedance which results in minimum measured values for SRL reflections over the frequency of interest. This is equivalent to measuring a cable with a return loss bridge that can vary its reference impedance. The value of reference impedance that results in minimum reflection, where minimum must now be defined in some sense, is the cable impedance. Mathematically, this is equivalent to finding a cable impedance Z cable such that: [ρ(ω, Z cable )] (Z cable ) = 0 eq. 2 where ρ(ω) is some mean reflection coefficient. Thus, cable impedance and SRL are somewhat inter-related; the value of SRL depends upon the cable impedance, and the cable impedance is chosen to give a minimum SRL value. An alternate definition of cable impedance is the average impedance presented at the input of the cable over a desired span. This can be represented as Z avg = F min Fmax Z in (ω)dω 2π(F max F min ) eq. 3 The value found for Z avg would be substituted for Z cable in equation (1) to obtain the structural return loss from the cable impedance measurement. Understanding cable construction, defects and faults. Any discourse on cable measurements should include a discussion of the unique qualities of cables that make measurements so challenging. Because cables are electrically very long, and very low loss, the effect of any periodic defect in the cable will be greatly multiplied. 2

3 Periodic faults and SRL SRL is a reflection of incident energy that is caused by disturbances (bumps) in the cable which are distributed throughout the cable length. These bumps may take the form of a small dent, or a change in diameter of the cable. These bumps are caused by periodic effects on the cable while in the manufacturing process. For example, consider a turn-around wheel with a rough spot on a bearing. The rough spot can cause a slight tug for each rotation of the wheel. As the cable is passed around the wheel, a small imperfection can be created periodically corresponding to the tug from the bad bearing. Each of these small variations within the cable causes a small amount of energy to reflect back to the source due to the non-uniformity of the cable diameter. Each bump reflects so little energy that it is too small to observe with fault location techniques. However, reflections from the individual bumps can sum up and reflect enough energy to be detected as SRL. As the bumps get larger and larger, or more of them are present, the SRL values will also increase. The energy reflected by these bumps can appear in the return loss measurement as a reflection spike at the frequency that corresponds to the spacing of the bumps. The spacing between the bumps is one half the wavelength of the reflection spike and is described by equations. wavelength = c / f c = speed of light eq. 4 f = frequency wavelength / 2 = spacing between the bumps eq. 5 The wavelength/2 spacing corresponds to the frequency at which down and back reflections will add coherently (in- phase). The reflections produce a very narrow response on the analyzer display that is directly related to the spacing of the bumps. We observe the amount of reflected energy as return loss. When this return loss measurement is normalized to the cable impedance, the return loss becomes structural return loss. Figure 1. diagrams reflections from bumps in a cable. We can combine the energy reflected by each bump in a cable and make a few basic assumptions, to mathematically describe SRL by the series shown in equation (6). P ref = [P in LΓL] + [P in L(1 Γ)LΓLL] + [P in L(1 Γ)L(1 Γ)LΓLLL] +... eq. 6 P ref is the reflected Power. P in is the incident Power. L is the cable loss. Γ is the reflection coefficient of the bumps. The bumps are assumed to be uniform in reflection and spaced by a wavelength/2 separation. 3

4 P ref P in λ/2 λ/2 L L L Γ Γ L (1 Γ)L (1 Γ)L Γ Figure 1. Periodic bumps in a cable The series may be reduced to a simple form to leave us with the relationship shown in equation (7). The derivation for the simplified relationship is reproduced in appendix A. The term L is a function of the loss of the cable at a specific frequency and the wavelength at that frequency. The term L 2 /(1 L 2 ) can be thought of as the number of bumps that are contributing to SRL. It represents a balance between the contribution of loss in a single bump and further bumps in the cable for the specified frequency and cable loss. Calculate the distance into the cable by multiplying the term L 2 /(1 L 2 ) by the distance between bumps. eq. 7 Table 1.1 illustrates some calculated values for a typical trunk cable. From the table, bumps spaced 1.5 meters apart out to 307 meters will contribute to SRL. The calculations to obtain the table values are shown in Appendix B. Frequency Spacing (λ/2)m SRL = P ref P in = Γ L2 1 L 2 Table 1.1 SRL Equation Constant Loss /m (db) db/bump bumps Distance (m) L 2 /(1 L 2 ) 100 MHz MHz GHz In actual cables, the reflections from the bumps and the spacing of the bumps may vary widely. The worst case, or best case for a minimum SRL, is that the bumps are totally random and very small. The effects are scattered throughout the frequency range and are small. Real world examples are somewhere in between the uniform bumps and the scattered case. As the 4

5 sizes of the bumps, their spacing and the number of bumps vary within the manufacturing process, varying amounts of SRL are observed. Discrete cable faults and SRL Reflections from faults within the cable will also increase the level of SRL measured. The energy reflected from a fault will sum with the energy reflected from the individual bumps and provide a higher reflection level at the measurement interface. Examining the cable for faults before the SRL measurement is a worthwhile procedure. The time required to perform the measurement is small compared to the time spent in performing an SRL measurement scan. A fault within the cable will provide the same type of effect as a bad connector. If the fault is present within the near end of the cable, the effect will be noticed throughout the entire frequency range. As the fault is located further into the cable, the cable attenuation will reduce the effect at higher frequencies. The reflected energy travels further through the cable at lower frequencies where the cable attenuation is lower. Traditional measurement methods, variable bridge A variable impedance bridge can be used to measure cable impedance and structural return loss. The variable impedance bridge consists of a return loss bridge that can vary its nominal impedance. The nominal impedance is the impedance for which zero reflection occurs, usually from about 70 to 80 ohms. Also, there is compensation for connector mismatch in the form of a variable capacitance term at the bridge test port, with a typical range of ± 1.5 picofarads. In recent years, the variable impedance bridge method has been used with a vector network analyzer taking advantage of the increased dynamic range, synthesized source, and advanced features such as markers, limit lines and vector error correction. Measurement methods have been described in the Society for Cable Television Engineers (SCTE) Interface Practices Subcommittee standards for cable impedance and structural return loss (IPS 406 and 407) To use a variable impedance bridge, a measurement is set up with the source going into power splitter, then into the reference channel of the network analyzer as shown in Figure 2. This is needed for phase-lock of the source to the receiver, but also greatly enhances the source match of the measurement system in ratio measurements, such as the ratio of reflected to incident power. A ohm match pad can be used before the input of the variable impedance bridge, and another on the reflection port output of the bridge to the measurement channel of the network analyzer. The frequency span should be set to the desired range, 5 MHz to 1 GHz for CATV cable. Next, if a vector network analyzer is used, a vector 1-port (open, short, load) calibration can be performed to remove effects such as tracking (also known as frequency response), source match, and directivity. For this, the bridge must be set to exactly 75 ohms, and 0 pf, before the calibration is performed. After the calibration, the bridge will be nearly ideal measuring exactly 75 ohms, and thus impedance measurements will be much better than if no calibration is performed. Even when 5

6 the impedance is changed on the bridge, any residual errors due to changing the reference impedance will be smaller than if no calibration had been performed. To measure cable impedance and structural return loss, the bridge impedance is varied while watching the trace on the screen to get the flattest trace that is lowest in level. Sometimes, there is a trade off between varying the impedance and varying the compensating capacitor. If a cable does not meet specifications, it may be because the connector to the cable has too high of return loss. The connector may be manually "tweaked" by sliding it back and forth slightly, or twisting it, while watching the trace. When the best trace has been achieved, the value of the impedance dial on the variable bridge is considered the cable impedance. This is a physical realization of equation 2, where the operator minimizes cable reflection coefficient by varying Z to get the minimum value. Then, the value of Z is the cable impedance. There are two difficulties here, both related to operator craft and judgment: 1) The value of impedance depends upon the operator's judgment of minimum reflection. Thus, the result can be different for the exact same cable and setup conditions, if a different operator performs the test. 2) The value depends upon the connection, and the manner in which the operator adjusts the connector and adjusts the capacitor compensation. Sometimes, the value for cable impedance is determined by judging only the low frequency trace response where the effect of the connector is not significant. Note that the impedance measurement is a single number, which is obtained by some interpretation of return loss data over frequency, and that the cable impedance is only truly defined at one spot along the cable. Figure 2. SRL test system setup with variable bridge Structural return loss, however, is a measurement made over the entire frequency range of the cable. A typical report shows the worst several frequency points, along with their value for SRL. After finding the cable impedance, the operator should re-measure the cable with a 6

7 finer resolution, because structural return loss can have very narrow, high-q peaks in the response. For a 2500 foot cable, the minimum spacing between points should be less than 200 khz. Fixed bridge method with connector compensation Another technique for measuring structural return loss involves a vector network analyzer, a set of calibration standards, and a technique for emulating a variable bridge with calculations. The calculations give the average impedance of the cable and a C value for connector compensation. The fixed bridge measurement setup is shown in Figure 3. The source output from the network analyzer is connected to the cable being tested. A test lead cable may be used between the analyzer and the end of the cable being tested for convenience. Figure 3. SRL test system setup with fixed bridge Perform a vector 1-port (open, short, load) calibration at the end of the test lead cable to correct for the bridge directivity, the frequency response, the test lead cable, and the measurement system port matches. The standards used for this calibration are an open, a short, a load, and a thru. Each standard is traceable to the National Institute of Standards (NIST). This means that the measurements taken with a calibrated system are traceable to national standards. To measure cable impedance and structural return loss, the frequency span should be set to the desired range, 5 MHz to 1 GHz for CATV cable. Begin by making low resolution measurements with fast update rates to determine the impedance of the cable and verify the setup. The measured return loss data is normalized to the Z-average impedance of the cable 7

8 mathematically by the analyzer. The mathematics to do this calculation emulate the action of balancing the variable bridge Z impedance. A connector compensation can also be mathematically applied which emulates the adjustment of the C value on the variable impedance bridge. The impedance and C compensation values may be entered manually or automatically derived by the analyzer. Once the values for the impedance and the connector compensation are determined, re-measure the cable at higher resolution. The analyzer can scan the cable at resolutions less than 100 khz, then process the measured results and graph the worst responses. The mathematics for the Z-averatge normalization are shown below. The mathematics for the connector compensation are shown in the next section. Z in (ω) = 75 (1 + ρ(ω)) (1 ρ(ω)) eq. 8 Z cable = abs Σ Z in (ω) N eq. 9 ρ SRL (ω) = Z in (ω) Z cable Z in (ω) + Z cable eq. 10 In equation (8), ρ(ω) is the reflection coefficient from the analyzer measured at each frequency and Z in (ω) is the impedance of the cable for that measured reflection coefficient. The calculation of Z cable, described in equation (9) is the Z-average impedance of the cable over the frequency range. Equation (10) is the structural return loss for the cable. This calculation can be done by the analyzer or an external computer. Techniques for removing connector effects Connector effects on SRL When measuring spools of cable, typically two connectors are used: the test lead connector and the termination connector. These connectors provide the cable interface and are measured as part of the cable data. Often, slight changes in the test lead connector can cause significant changes in the values of structural return loss measured at high frequencies. This is because the reflection from a connector increases for high frequencies. In fact, the return loss of a test lead connector can dominate the SRL response at frequencies above 500 MHz. This is where training, good measurement practices, and precision cable connectors are needed, especially for measurements up to 1 GHz. Precision connectors are required to provide repeatability over multiple connections. Slip-on connectors are used to provide rapid connections to the cables, but require careful attention in obtaining good measurement data. Repeatability of measurement data is 8

10 start and stop of gate. In general, the gating is used to measure connector interfaces, where the start is before the connector and the stop is beyond it, somewhere in the cable. With the gating turned on, the time response can be converted back to frequency response with only the connector response shown. In this way, reflections from the cable are removed from the connector measurement. However, if the gate start is set after the connector, and the gate stop is set before the connector, the time response of the connector only will be set to zero, leaving the response of the rest of the cable. This time response can be converted back to the frequency domain. The question is: does this twice converted response represent the structural return loss of the cable, or is some information lost? That is, will time domain gating remove the error caused by a poor connection, but leave the cable response intact. This question arises because the time response of any reflections following a large reflection will be smaller than normal, since the large reflection does not pass all of the "impulse" energy. Thus, the gated result will not have as high a reflection as the actual cable. Also, since the gating function is numerically derived from the limited frequency response data, gating time widths may have unexpected affects in the frequency response. The gated response may still be used to determine cable quality. One technique proposed is to gate in the connector, convert that back the frequency range, and use the resultant reflection as a "correction factor" to be applied to the response with connector gated out. For example, if a connector has a reflection coefficient of 0.05 at 500 MHz, and 0.1 at 1 GHz, then the response of the cable with connector gated out should be divided by 0.95 at 500 MHz and 0.9 at 1 GHz. There are many other factors that must be considered before time domain gating is used, including understanding the affects of frequency sampling points, gating windows, and how the effects of a variable bridge add into the measurement uncertainty. Time domain gating promises to be an effective tool in removing a major cause of repeatability problems, but more work is needed to prove its performance in real world tests. Fixed bridge with connector compensation. Vector error correction is used to provide the most accurate measurements up to the calibration plane defined by the calibration standards. Additional corrections can also be used to minimize the affects of the test lead connector on the measured SRL response. The error corrections done for a fixed bridge can also include connector compensation. The fixed bridge method with connector compensation technique mathematically removes the effects of the test lead connector by compensating the predicted connector response given by a connector model. With a connector model, the adjustment of the C value given in the variable bridge technique can be emulated. This will reduce the requirements on the performance of the connector. One model that can be used for the cable connector is the shunt C connector model. The shunt C connector model assumes the discontinuity at the interface is abrupt and much smaller than a half wavelength of the highest frequency of measurement. With this assumption, the discontinuity can be modeled as a single shunt susceptance, where C= Co + second and third order terms. 10

11 Intuitively this is the right model to choose because the effects of a typical poor connector on structural return loss measurement is an upward sloping response, typically worst at the high frequencies. Using a shunt C to model the connector, a value of the susceptance, -C, may be chosen by the network analyzer to cancel the equivalent C of the connector and mathematically minimize the effect of the connector on the response measurement. The equations for computing structural return loss and the average cable impedance with capacitive compensation are described below. Z in (ω) = Z in (ω) 1 jωc Z in (ω) + 1 jωc eq. 11 Z cable = abs Σ Z in (ω) N eq. 12 ρ SRL (ω) = Z in (ω) Z cable Z in (ω) + Z cable eq. 13 In equation (11), Z in (ω) is calculated from the measured return loss as described in equation (8), previously. The primed values are the new calculation values using the capacitive compensation. With these equations, the network analyzer can compute values for the cable impedance and mathematically compensate for the connector mismatch with a given value of C connector compensation. Deriving the connector C value There are two ways to derive a value for the C connector compensation. The first way to do this is similar to the variable bridge method. Observe the SRL response, then allow the operator to choose a value of C that best flattens the response. The second way to determine the value is to use the computational power of the network analyzer to choose an optimal value of C to achieve the flattest response. The optimization performed is based on the predicted behavior given by the connector model. 11

12 Analyzer based optimization eliminates operator interaction and provides a consistent solution to enable less skilled operators to make SRL measurements. CAE simulations The network analyzer can use modern CAE techniques to compute the optimum values for the connector model. CAE tools can also be used verify the validity of the model. Use CAE to simulate a typical structural return loss measurement by modeling a cable as lengths of transmission line and the cable connector as a connector mismatch. Figure meter cable SRL with fixed bridge and connector compensation Figure 4 displays a simulation (CAE) of a 500 meter length of transmission line with the connector modeled as a shunt capacitance of.5pf. Trace 1 displays the return loss response as would be measured by a fixed bridge with no impedance normalization and no connector compensation. Trace 2 shows the structural return loss measurement now normalized to an average measured impedance of 72.9 ohms with -.5pF capacitive reactance mathematically removed. Notice the 12 db response improvement at high frequency. The response improvement at high frequency is mainly caused by removing the effects of the connector mismatch mathematically. Figure 5 displays a simulation (CAE) of a 1000 meter length of transmission line with a connector mismatch modeled as an 8 mm length of 88 ohm line. Trace 1 displays the uncompensated return loss measurement. Trace 2 displays the response after the network analyzer has mathematically calculated the response including a 0.11 pf shunt capacitance compensation and normalizing to a ohm cable impedance. Notice an 8 db improvement at high frequency but a worse response at lower frequency in this example. 12

13 Figure meter cable with connector mismatch. Figure meter cable with mismatch at 1 meter Adding connector compensation should only correct for connector mismatches and not for cable defects. Figure 6 shows the best compensation for a cable with a cable defect at 1 meters away from the connector. Again the defect is modeled as an 8 mm length of 88 ohm line. The response is improved at some frequencies but made worse at other frequencies. The maximum response is higher than the response of the cable return loss without compensation, for any value of compensation C. The capacitance compensation cannot improve the response of a 13

14 defect greater than a half wavelength down the length of the cable primarily because the summed vector responses from the defects down the length of the cable can not be canceled by a simple reactance. Measurement uncertainty in SRL measurements and cable impedance. In any comparison of cable impedance or structural return loss data, it is important to understand the measurement uncertainty involved in each type of measurement. This is critical for manufacturers, who often use the most sophisticated techniques to reduce manufacturing guard bands. It is also important in field measurements that users choose the proper equipment for their needs, and understand the differences that can occur between manufactures' data and field data. Also, note that measurement uncertainty is usually quoted as the worst case result if the sources of error are at some maximum value. This is not the same as error in the measurement, but rather a way to determine measurement guard band, and to understand how closely to expect measurements to compare on objects measured on different systems. The errors that can occur in a reflection measurement are reflection tracking (or frequency response), T, source match, Γ M, and directivity, D. The total error in a measurement can be shown to be: Γ MEAS = T D + (Γ DUT ) (1 Γ M Γ DUT ) eq. 14 Error correction techniques can effectively remove the effects of tracking. Also, source match effects are small if Γ DUT is small. This leaves directivity as the largest error term in the reflection measurement. The causes and effects of these error terms will be described for each of the measurement methodologies. For variable bridge measurements, the directivity of the bridge is the major error term. One-port vector error correction reduces the effects of tracking and source match, and improves directivity. The directivity after error correction is set by the return loss of the precision load, specified to be better than 49 db at 1 GHz. However, the directivity is only well known at precisely 75 ohms, and the directivity at other impedances should be assumed to be that specified by the manufacturer. For best performance, the bridge should be connected directly to the cable connector, with no intervening cable in between. The directivity of the bridge could be determined at impedances other than 75 ohms, by changing the impedance and measuring the resulting values. With a vector network analyzer, this can be done by changing the reference impedance to the new value, say 76 ohms, changing the bridge to that value, and measuring the impedance on a Smith chart display. The difference from exactly 75 ohms represents the directivity at that impedance. For fixed bridge methods, the reflection port is often connected to the cable connector through a length of test lead. A one-port calibration is performed at the end of the test lead. The directivity will again be set by the load, but any change in return loss of the test lead due to flexing will degrade the directivity of the measurement system. In both fixed and variable 14

16 cable impedance measurements. Also, the loads that are used will invariably be somewhat better than specified, especially over the low frequency range. From this, it is reasonable to assume that the errors in impedance measurements are at least 50% less than listed in the above table. Measurement Uncertainty for Structural Return Loss The same factors that effect cable impedance - directivity, system and test lead stability, and cable connector mismatch - also affect structural return loss. However, since structural return loss is measured at all frequencies, it is much more likely that a worst case condition can occur at any one frequency. For that reason, the measurement uncertainty must include the full effect of the above listed errors. Figure 5 shows a graph of measurement uncertainty curves for a -40 db error term applied to various return loss values. The upper trace is return loss plus error, the bottom term is return loss minus error, and the middle curve, which must be a straight line, is simply the return loss with no error. The x-axis is the value of the return loss, and the y-axis is the value measured, with the error added or subtracted. Notice as the return loss gets larger (closer to 0 db) the effect of the error is smaller. Figure 6 is the same plot, shown for a -49 db error term. One use for this graph is to determine the measurement guard band needed to specify cable performance. If the cable is to meet a -30 db spec., with -49 db system, then the value of -31 db must be measured to guarantee - 30 db. Graphically, if the a spec line is drawn vertically through the x-axis at -30 db, the lower uncertainty trace shows the value on the y-axis that must be measured to guarantee the spec. Conversely, if a cable is shipped with value of -35 db for structural return loss, the value that might be measured by a second system in the field can be determined by drawing a vertical line through the x-axis at -35 db. The upper and lower traces show, on the y-axis, the limits with which the cable SRL can be determined. For this example, the -35 db cable could measure as bad as db, or as good as -37 db. But this is not the whole picture. If the first measurement of the cable was performed on a similar system, it will have a similar uncertainty. For this example, a cable measured as -35 db on the first system could be as bad as db. This could be measured on a second system as good as -35 db (if the directivity error is exactly the same magnitude and phase as the first system), or as bad as (where the directivity magnitude is the same, but the phase is opposite). Fortunately, as with impedance measurements, the directivities are unlikely to be worst case in magnitude at the same frequency and with opposite phase on two different systems. (-db) Return Loss Uncertainty, -40 db system 16

17 (18) (21) (24) (27) (30) (33) (36) (39) (42) (45) (48) Figure 5. Return Loss of Device-Under-Test DUT R.L Measured Minimum Nominal Maximum Table 1.3 Uncertainty "window" for -40 db directivity system Return Loss Uncertainty, -49 db system (-db) 17

18 (20) (22) (24) (26) (28) (30) (32) (34) (36) (38) (40) Figure 6. Return Loss of Device-Under-Test DUT R.L Measured Minimum Nominal Maximum Table 1.4. Uncertainty for -49 db directivity system Appendix A. SRL Equations P ref = Reflected Power 18

19 P in = Incident Power L = Cable Loss Γ = Reflection Coefficient of bump P ref = [P in LΓL] + [P in L(1 Γ)LΓLL] + [P in L(1 Γ)L(1 Γ)LΓLLL] +... P ref = [P in ΓL 2 ] + [P in L 2 (1 Γ)ΓL 2 ] + [P in L 3 (1 Γ) 2 ΓL 3 ] +... P ref = [P in ΓL 2 ] + [P in Γ(1 Γ)L 4 ] + [P in Γ(1 Γ) 2 L 6 ] + [P in Γ(1 Γ) n 1 L 2n ] +... P ref = Σ P in Γ(1 Γ) n 1 L 2n n=1 S n = Σ n P in Γ(1 Γ) n 1 L 2n n=1 S n =P ref = [P in ΓL 2 ] + [P in Γ(1 Γ)L 4 ] + [P in Γ(1 Γ) 2 L 6 ] +... [P in Γ(1 Γ) n 1 L 2n ] (1 Γ)L 2 S n = [P in Γ(1 Γ)L 4 ] + [P in Γ(1 Γ) 2 L 6 ] +... [P in Γ(1 Γ) n 1 L 2n ] + [P in Γ(1 Γ) n L 2(n+1) ] S n (1 Γ)L 2 S n = [P in ΓL 2 ] [P in Γ(1 Γ) n L 2(n+1) ] lim n S n [1 (1 Γ)L 2 ] = [P in ΓL 2 ] 0 L < 1, Γ < 1 19

20 P ref = S [1 (1 Γ)L 2 ] = P in ΓL 2 P ref = P inγl 2 1 (1 Γ)L 2 P ref = P inγl 2 1 (1 Γ)(L 2 ) where Γ << 1 & L << 1 P ref = P inγl 2 1 L 2 ρ = P ref P in ρ SRL = ΓL2 1 L 2 P SRL = Γ individuall 2 1 L 2 L = Loss Wavelength Length 2 20

21 Appendix B. Example calculations of SRL equation constants. TX MHz db/100 m = db/m λ = c f = m = 3m λ 2 = 1.5m L =.0141dB/m 1.5m L =.0212dB/bump 20 log(l) =.0212 L = L 2 = L 2 = L 2 1 L 2 = Bumps TX MHz.0328 db/m λ =.6m L = db = L 2 = L 2 = L 2 1 L 2 = Bumps TX GHz 4.89 db/100m =.0489 db/m λ =.3 m 21

22 λ 2 =.15m L = db = L 2 = L 2 = L 2 1 L 2 = Bumps Distance into cable bumps λ 2 = 88.7 m bumps λ 2 = m bumps 1.5k = 307 m 22

### Reflection Coefficient Applications in Test Measurements Bernhart A. Gebs Senior Product Development Engineer Belden Electronics Division. Rho.

Reflection Coefficient Applications in Test Measurements Bernhart A. Gebs Senior Product Development Engineer Belden Electronics Division Reflection coefficient is the basis of several measurements common

### Cable Analysis and Fault Detection using the Bode 100

Cable Analysis and Fault Detection using the Bode 100 By Stephan Synkule 2014 by OMICRON Lab V1.3 Visit www.omicron-lab.com for more information. Contact support@omicron-lab.com for technical support.

### Agilent De-embedding and Embedding S-Parameter Networks Using a Vector Network Analyzer. Application Note 1364-1

Agilent De-embedding and Embedding S-Parameter Networks Using a Vector Network Analyzer Application Note 1364-1 Introduction Traditionally RF and microwave components have been designed in packages with

### Impedance 50 (75 connectors via adapters)

VECTOR NETWORK ANALYZER PLANAR TR1300/1 DATA SHEET Frequency range: 300 khz to 1.3 GHz Measured parameters: S11, S21 Dynamic range of transmission measurement magnitude: 130 db Measurement time per point:

### Vector Network Analyzer Techniques to Measure WR340 Waveguide Windows

LS-296 Vector Network Analyzer Techniques to Measure WR340 Waveguide Windows T. L. Smith ASD / RF Group Advanced Photon Source Argonne National Laboratory June 26, 2002 Table of Contents 1) Introduction

### A Shunt Fixture for Low Impedance Measurements Sam Wetterlin 4/28/11

A Shunt Fixture for Low Impedance Measurements Sam Wetterlin 4/28/11 Introduction Measurements of impedances in the 0.01-10 ohm range are useful for measuring the resistance of capacitors, inductors and

### 9 Measurements on Transmission Lines

Measurements on Transmission Lines Power and Attenuation Measurements Although a variety of instruments measure power, the most accurate instrument is a power meter and a power sensor. The sensor is an

### Experiment 7: Familiarization with the Network Analyzer

Experiment 7: Familiarization with the Network Analyzer Measurements to characterize networks at high frequencies (RF and microwave frequencies) are usually done in terms of scattering parameters (S parameters).

### MEASUREMENT UNCERTAINTY IN VECTOR NETWORK ANALYZER

MEASUREMENT UNCERTAINTY IN VECTOR NETWORK ANALYZER W. Li, J. Vandewege Department of Information Technology (INTEC) University of Gent, St.Pietersnieuwstaat 41, B-9000, Gent, Belgium Abstract: Precision

### Agilent 8510-13 Measuring Noninsertable Devices

Agilent 8510-13 Measuring Noninsertable Devices Product Note A new technique for measuring components using the 8510C Network Analyzer Introduction The majority of devices used in real-world microwave

### Agilent PN 8753-1 RF Component Measurements: Amplifier Measurements Using the Agilent 8753 Network Analyzer. Product Note

Agilent PN 8753-1 RF Component Measurements: Amplifier Measurements Using the Agilent 8753 Network Analyzer Product Note 2 3 4 4 4 4 6 7 8 8 10 10 11 12 12 12 13 15 15 Introduction Table of contents Introduction

### Coaxial Cable Delay. By: Jacques Audet VE2AZX

Coaxial Cable Delay By: Jacques Audet VE2AZX ve2azx@amsat.org Introduction Last month, I reported the results of measurements on a number of coaxial cables with the VNA (Vector Network Analyzer). (Ref.

### Understanding SWR by Example

Understanding SWR by Example Take the mystery and mystique out of standing wave ratio. Darrin Walraven, K5DVW It sometimes seems that one of the most mysterious creatures in the world of Amateur Radio

### Agilent PNA Microwave Network Analyzers

Agilent PNA Microwave Network Analyzers Data Sheet This document describes the performance and features of the Agilent Technologies PNA microwave network analyzers: E8362B E8363B E8364B E836A MHz to 2

### Calibration Report W7ZOI Power Meter built by W2GNN. Clifton Laboratories Jack Smith K8ZOA 12 March 2010

Calibration Report W7ZOI Power Meter built by W2GNN Clifton Laboratories Jack Smith K8ZOA 12 March 2010 Overview of W7ZOI Power Meter Designed by W7ZOI and published in June 2001 QST. Uses an AD 8307 log

### The Effect of Network Cabling on Bit Error Rate Performance. By Paul Kish NORDX/CDT

The Effect of Network Cabling on Bit Error Rate Performance By Paul Kish NORDX/CDT Table of Contents Introduction... 2 Probability of Causing Errors... 3 Noise Sources Contributing to Errors... 4 Bit Error

### Agilent AN 1316 Optimizing Spectrum Analyzer Amplitude Accuracy

Agilent AN 1316 Optimizing Spectrum Analyzer Amplitude Accuracy Application Note RF & Microwave Spectrum Analyzers Table of Contents 3 3 4 4 5 7 8 8 13 13 14 16 16 Introduction Absolute versus relative

### S-Parameters and Related Quantities Sam Wetterlin 10/20/09

S-Parameters and Related Quantities Sam Wetterlin 10/20/09 Basic Concept of S-Parameters S-Parameters are a type of network parameter, based on the concept of scattering. The more familiar network parameters

### Application Note Noise Frequently Asked Questions

: What is? is a random signal inherent in all physical components. It directly limits the detection and processing of all information. The common form of noise is white Gaussian due to the many random

### Technical Datasheet Scalar Network Analyzer Model 8003-10 MHz to 40 GHz

Technical Datasheet Scalar Network Analyzer Model 8003-10 MHz to 40 GHz The Giga-tronics Model 8003 Precision Scalar Network Analyzer combines a 90 db wide dynamic range with the accuracy and linearity

### One Port Network Analyzer

99 Washington Street Melrose, MA 02176 Phone 781-665-1400 Toll Free 1-800-517-8431 Visit us at www.testequipmentdepot.com One Port Network Analyzer 5.4GHz Impendance : 50Ω(75Ωconnectors via adapters) Test

### Precision Video Coaxial Cables Part 1: Impedance

Precision Video Coaxial Cables Part 1: Impedance Martin J. Van Der Burgt Senior Product Engineering Project Manager Belden Electronics Division Abstract: Over the years, engineers have come to identify

### We may relate the values of voltage and current at z and z+äz by writing KVL and KCL equations for the equivalent circuit.

Transmission Lines The problem of plane waves propagating in air represents an example of unguided wave propagation. Transmission lines and waveguides offer an alternative way of transmitting signals in

### Agilent Time Domain Analysis Using a Network Analyzer

Agilent Time Domain Analysis Using a Network Analyzer Application Note 1287-12 0.0 0.045 0.6 0.035 Cable S(1,1) 0.4 0.2 Cable S(1,1) 0.025 0.015 0.005 0.0 1.0 1.5 2.0 2.5 3.0 3.5 4.0 Frequency (GHz) 0.005

### Engineering Sciences 151. Electromagnetic Communication Laboratory Assignment 3 Fall Term 1998-99

Engineering Sciences 151 Electromagnetic Communication Laboratory Assignment 3 Fall Term 1998-99 WAVE PROPAGATION II: HIGH FREQUENCY SLOTTED LINE AND REFLECTOMETER MEASUREMENTS OBJECTIVES: To build greater

### Solving Difficult Cable Measurements

Slide #1 Solving Difficult Introduce yourself and the topic for the next 40 minutes. Inquire if there are any customers that are currently performing cable testing and determine the level of the audience.

### Shielding Effectiveness Test Method. Harbour s LL, SB, and SS Coaxial Cables. Designs for Improved Shielding Effectiveness

Shielding Effectiveness Test Method Harbour s LL, SB, and SS Coaxial Cables Designs for Improved Shielding Effectiveness Harbour Industries 4744 Shelburne Road Shelburne Vermont 05482 USA 802-985-3311

### 1. The Slotted Line. ECE 584 Microwave Engineering Laboratory Experiments. Introduction:

ECE 584 Microwave Engineering Laboratory Experiments 1. The Slotted Line Introduction: In this experiment we will use a waveguide slotted line to study the basic behavior of standing waves and to measure

### Power Amplifier Gain Compression Measurements

Technical Brief Power Amplifier Gain Compression Measurements GPIB Private Bus Sweep Out Sweep In Pulse In AC Mod Out Blank/Marker Out Blanking In Overview The 1 db gain compression of an amplifier describes

### Crosstalk effects of shielded twisted pairs

This article deals with the modeling and simulation of shielded twisted pairs with CST CABLE STUDIO. The quality of braided shields is investigated with respect to perfect solid shields. Crosstalk effects

### Utilizing Time Domain (TDR) Test Methods For Maximizing Microwave Board Performance

The Performance Leader in Microwave Connectors Utilizing Time Domain (TDR) Test Methods For Maximizing Microwave Board Performance.050 *.040 c S11 Re REF 0.0 Units 10.0 m units/.030.020.010 1.0 -.010 -.020

### Digital Systems Ribbon Cables I CMPE 650. Ribbon Cables A ribbon cable is any cable having multiple conductors bound together in a flat, wide strip.

Ribbon Cables A ribbon cable is any cable having multiple conductors bound together in a flat, wide strip. Each dielectric configuration has different high-frequency characteristics. All configurations

### ECE 435 INTRODUCTION TO THE MICROWAVE NETWORK ANALYZER

ECE 435 INTRODUCTION TO THE MICROWAVE NETWORK ANALYZER Latest revision: October 1999 Introduction A vector network analyzer (VNA) is a device capable of measuring both the magnitude and phase of a sinusoidal

### Understanding Mixers Terms Defined, and Measuring Performance

Understanding Mixers Terms Defined, and Measuring Performance Mixer Terms Defined Statistical Processing Applied to Mixers Today's stringent demands for precise electronic systems place a heavy burden

### 2. The Vector Network Analyzer

ECE 584 Laboratory Experiments 2. The Vector Network Analyzer Introduction: In this experiment we will learn to use a Vector Network Analyzer to measure the magnitude and phase of reflection and transmission

### MANUAL RETURN LOSS MEASUREMENTS Sam Wetterlin 10/5/08

MANUAL RETURN LOSS MEASUREMENTS Sam Wetterlin swetterlin@comcast.net 10/5/08 This paper describes the procedure for using a reflection bridge (a/k/a return loss bridge ) to manually measure return loss

### Using Simple Calibration Load Models to Improve Accuracy of Vector Network Analyzer Measurements

Using Simple Calibration Load Models to Improve Accuracy of Vector Network Analyzer Measurements Nick M. Ridler 1 and Nils Nazoa 2 1 National Physical Laboratory, UK (www.npl.co.uk) 2 LA Techniques Ltd,

### Time and Frequency Domain Analysis for Right Angle Corners on Printed Circuit Board Traces

Time and Frequency Domain Analysis for Right Angle Corners on Printed Circuit Board Traces Mark I. Montrose Montrose Compliance Services 2353 Mission Glen Dr. Santa Clara, CA 95051-1214 Abstract: For years,

### S-Band Low Noise Amplifier Using the ATF-10136. Application Note G004

S-Band Low Noise Amplifier Using the ATF-10136 Application Note G004 Introduction This application note documents the results of using the ATF-10136 in low noise amplifier applications at S band. The ATF-10136

### MEASUREMENT SET-UP FOR TRAPS

Completed on 26th of June, 2012 MEASUREMENT SET-UP FOR TRAPS AUTHOR: IW2FND Attolini Lucio Via XXV Aprile, 52/B 26037 San Giovanni in Croce (CR) - Italy iw2fnd@gmail.com Trappole_01_EN 1 1 DESCRIPTION...3

### Introduction to the Smith Chart for the MSA Sam Wetterlin 10/12/09 Z +

Introduction to the Smith Chart for the MSA Sam Wetterlin 10/12/09 Quick Review of Reflection Coefficient The Smith chart is a method of graphing reflection coefficients and impedance, and is often useful

### RF Network Analyzer Basics

RF Network Analyzer Basics A tutorial, information and overview about the basics of the RF Network Analyzer. What is a Network Analyzer and how to use them, to include the Scalar Network Analyzer (SNA),

### Keysight Technologies Signal Integrity Analysis Series

Keysight Technologies Signal Integrity Analysis Series Part 1: Single-Port TDR, TDR/TDT, and 2-Port TDR Application Note Introduction Interconnect Analysis is Simplified with Physical Layer Test Tools

### Measuring RF Parameters of Networks Bill Leonard NØCU

Measuring RF Parameters of Networks Bill Leonard NØCU NAØTC - 285 TechConnect Radio Club http://www.naøtc.org/ What is a Network? A Network is a group of electrical components connected is a specific way

### Agilent Advanced impedance measurement capability of the RF I-V method compared to the network analysis method. Application Note 1369-2

Agilent Advanced impedance measurement capability of the RF I-V method compared to the network analysis method Application Note 1369-2 1. Introduction Explosive demand for the electronic devices used in

### Selecting Receiving Antennas for Radio Tracking

Selecting Receiving Antennas for Radio Tracking Larry B Kuechle, Advanced Telemetry Systems, Inc. Isanti, Minnesota 55040 lkuechle@atstrack.com The receiving antenna is an integral part of any radio location

### Network Analysis. Specifying calibration standards for the HP 8510 network analyzer Product Note 8510-5A

Network Analysis Specifying calibration standards for the HP 8510 network analyzer Product Note 8510-5A 2 Known devices called calibration standards provide the measurement reference for network analyzer

### A Network Analyzer For Active Components

A Network Analyzer For Active Components EEEfCom 29-30 Juni ULM Marc Vanden Bossche, NMDG Engineering Remi Tuijtelaars, BSW Copyright 2005 NMDG Engineering Version 2 Outline Review of S-parameters Theory

### The Smith chart. Smith chart components. The normalized impedance line CHAPTER

26 CHAPTER The Smith chart The mathematics of transmission lines, and certain other devices, becomes cumbersome at times, especially when dealing with complex impedances and nonstandard situations. In

### ANALYZER BASICS WHAT IS AN FFT SPECTRUM ANALYZER? 2-1

WHAT IS AN FFT SPECTRUM ANALYZER? ANALYZER BASICS The SR760 FFT Spectrum Analyzer takes a time varying input signal, like you would see on an oscilloscope trace, and computes its frequency spectrum. Fourier's

### CRP718 RF and Microwave Measurements Laboratory

CRP718 RF and Microwave Measurements Laboratory Experiment- MW2 Handout. Updated Jan 13, 2011 SPECTRUM ANALYZER BASED MEASUREMENTS (groups of 2 may omit part 2.1, but are advised to study the procedures

### Welcome to this presentation which helps to highlight the capability of Genesys in the design of power amplifiers for LTE applications

Welcome to this presentation which helps to highlight the capability of Genesys in the design of power amplifiers for LTE applications 1 2 3 4 5 Linearity is compromised due to the obvious output waveform.

### Antennas and Propagation. Chapter 3: Antenna Parameters

Antennas and Propagation : Antenna Parameters Introduction Purpose Introduce standard terms and definitions for antennas Need a common language to specify performance Two types of parameters 1. Radiation

### Impedance Matching and Matching Networks. Valentin Todorow, December, 2009

Impedance Matching and Matching Networks Valentin Todorow, December, 2009 RF for Plasma Processing - Definition of RF What is RF? The IEEE Standard Dictionary of Electrical and Electronics Terms defines

### MATRIX TECHNICAL NOTES

200 WOOD AVENUE, MIDDLESEX, NJ 08846 PHONE (732) 469-9510 FAX (732) 469-0418 MATRIX TECHNICAL NOTES MTN-107 TEST SETUP FOR THE MEASUREMENT OF X-MOD, CTB, AND CSO USING A MEAN SQUARE CIRCUIT AS A DETECTOR

### AMERICAN NATIONAL STANDARD

ENGINEERING COMMITTEE Interface Practices Subcommittee AMERICAN NATIONAL STANDARD ANSI/SCTE 05 2014 Test Method for F Connector Return Loss In-Line Pair NOTICE The Society of Cable Telecommunications Engineers

### How to make a Quick Turn PCB that modern RF parts will actually fit on!

How to make a Quick Turn PCB that modern RF parts will actually fit on! By: Steve Hageman www.analoghome.com I like to use those low cost, no frills or Bare Bones [1] type of PCB for prototyping as they

### Optimizing IP3 and ACPR Measurements

Optimizing IP3 and ACPR Measurements Table of Contents 1. Overview... 2 2. Theory of Intermodulation Distortion... 2 3. Optimizing IP3 Measurements... 4 4. Theory of Adjacent Channel Power Ratio... 9 5.

### RF-Microwaves formulas - 1-port systems

RF-Microwaves formulas - -port systems s-parameters: Considering a voltage source feeding into the DUT with a source impedance of. E i E r DUT The voltage into the DUT is composed of 2 parts, an incident

### DDX 7000 & 8003. Digital Partial Discharge Detectors FEATURES APPLICATIONS

DDX 7000 & 8003 Digital Partial Discharge Detectors The HAEFELY HIPOTRONICS DDX Digital Partial Discharge Detector offers the high accuracy and flexibility of digital technology, plus the real-time display

### Impedance Matching of Filters with the MSA Sam Wetterlin 2/11/11

Impedance Matching of Filters with the MSA Sam Wetterlin 2/11/11 Introduction The purpose of this document is to illustrate the process for impedance matching of filters using the MSA software. For example,

### FieldFox Handheld Education Series Part 4: Techniques for Precise Time Domain Measurements in the Field

FieldFox Handheld Education Series Part 4: Techniques for Precise Time Domain Measurements in the Field FieldFox Handheld Education Series Interference Testing Cable and Antenna Measurements Calibration

### Connector interfaces for the small coaxial line sizes

Connector interfaces for the small coaxial line sizes National Metrology Institute of Japan Advanced Industrial Science and Technology Masahiro Horibe 1 Metrological Standards for S- parameters - Standards:

### Comparison of Vector Network Analyzer and TDA Systems IConnect Generated S-Parameters

Comparison of Vector Network Analyzer and TDA Systems IConnect Generated S-Parameters Purpose: This technical note presents single-ended insertion loss ( SE IL) and return loss ( SE RL) data generated

### Transmission Lines. Smith Chart

Smith Chart The Smith chart is one of the most useful graphical tools for high frequency circuit applications. The chart provides a clever way to visualize complex functions and it continues to endure

### Designing the NEWCARD Connector Interface to Extend PCI Express Serial Architecture to the PC Card Modular Form Factor

Designing the NEWCARD Connector Interface to Extend PCI Express Serial Architecture to the PC Card Modular Form Factor Abstract This paper provides information about the NEWCARD connector and board design

### DDX 7000 & 8003. Digital Partial Discharge Detectors FEATURES APPLICATIONS

DDX 7000 & 8003 Digital Partial Discharge Detectors The HAEFELY HIPOTRONICS DDX Digital Partial Discharge Detector offers the high accuracy and flexibility of digital technology, plus the real-time display

### Connectivity in a Wireless World. Cables Connectors 2014. A Special Supplement to

Connectivity in a Wireless World Cables Connectors 204 A Special Supplement to Signal Launch Methods for RF/Microwave PCBs John Coonrod Rogers Corp., Chandler, AZ COAX CABLE MICROSTRIP TRANSMISSION LINE

### Printed Dipole Antenna

Printed Dipole Antenna Reto Zingg Abstract In this project a printed dipole antenna is being designed. Printed dipole antennas are of interest, when an electronic product, which is implemented on a printed

### This article will. A Design Tool for Improving the Input Match of Low Noise Amplifiers LNA INPUT MATCH

From February 2007 High Frequency Electronics Copyright 2007 Summit Technical Media, LLC A Design Tool for Improving the Input Match of Low Noise Amplifiers By Dale D. Henkes Applied Computational Sciences

### Network analyzer and spectrum analyzer two in one

R&S ZVL Vector Network Analyzer Network analyzer and spectrum analyzer two in one The R&S ZVL is the lightest and smallest vector network analyzer in its class. On top of this, it can be used as a full-featured

### Mutual Inductance Matching for Small Loop Antennas

Mutual Inductance Matching for Small Loop Antennas by Alan Bensky Loop antennas are one of the most popular and widespread antenna types for small wireless products, particularly those operating in UHF

### Performing Amplifier Measurements with the Vector Network Analyzer ZVB

Product: Vector Network Analyzer R&S ZVB Performing Amplifier Measurements with the Vector Network Analyzer ZVB Application Note This document describes typical measurements that are required to be made

### EMC STANDARDS STANDARDS AND STANDARD MAKING BODIES. International. International Electrotechnical Commission (IEC) http://www.iec.

EMC STANDARDS The EMC standards that a particular electronic product must meet depend on the product application (commercial or military) and the country in which the product is to be used. These EMC regulatory

Applying Error Correction to Network Analyzer Measurements Application Note 287-3 Table of Contents Page Introduction 2 Sources of Errors and Types of Errors 3 Types of Error Correction 4 One-Port 4 The

### EMC quantities and their uncertainty

EMC quantities and their uncertainty Carlo Carobbi Dipartimento di Elettronica e Telecomunicazioni Università degli Studi di Firenze Politecnico di Torino, I3P - 17 Giugno 2010 1 EMC quantities Frequency

### MILLIMETER WAVE TESTS AND INSTRUMENTATION

MILLIMETER WAVE TESTS AND INSTRUMENTATION Mohamed M. Sayed, Microwave and Millimeter Wave Solutions Santa Rosa, CA 95404 E-mail: mmsayed@sbcglobal.net Phone: (707) 318-5255 Abstract The recent growth in

### Making Spectrum Measurements with Rohde & Schwarz Network Analyzers

Making Spectrum Measurements with Rohde & Schwarz Network Analyzers Application Note Products: R&S ZVA R&S ZVB R&S ZVT R&S ZNB This application note describes how to configure a Rohde & Schwarz Network

### Agilent AN 1315 Optimizing RF and Microwave Spectrum Analyzer Dynamic Range. Application Note

Agilent AN 1315 Optimizing RF and Microwave Spectrum Analyzer Dynamic Range Application Note Table of Contents 3 3 3 4 4 4 5 6 7 7 7 7 9 10 10 11 11 12 12 13 13 14 15 1. Introduction What is dynamic range?

### SIGNAL GENERATORS and OSCILLOSCOPE CALIBRATION

1 SIGNAL GENERATORS and OSCILLOSCOPE CALIBRATION By Lannes S. Purnell FLUKE CORPORATION 2 This paper shows how standard signal generators can be used as leveled sine wave sources for calibrating oscilloscopes.

### Uniprise Solutions COAX 101. White Paper. www.commscope.com

Uniprise Solutions COAX 101 White Paper www.commscope.com Structured cable systems have very thorough standards for fiber optic and twisted pair installations. The cabling components and installed systems

### &"#' ( "#' )*! #+ #,# 1" 1! 2"# ' 6! #* #!"#" +" )\$# # # "#\$#\$ '2 2 ## #1##9 # # ##2 #( # 8##! #9# 9@8!( " " "32#\$#\$2#9 # "#!#1#\$#\$#9 '29# # #9\$

!"## \$#!%!"# &"#' ( "#' )*! #+ #,# "##!\$ -+./0 1" 1! 2"# # -&1!"#" (2345-&1 #\$6.7 -&89\$## ' 6! #* #!"#" +" 1##6\$ "#+# #-& :1# # \$ #\$#;1)+#1#+

### Spectrum Analyzers And Network Analyzers. The Whats, Whys and Hows...

Spectrum Analyzers And Network Analyzers The Whats, Whys and Hows... Bertrand Zauhar, VE2ZAZ ve2zaz@amsat.org June 2010 Today's Program Definitions of Spectrum and Network Analyzers, Differences between

### Agilent PN 89400-13 Extending Vector Signal Analysis to 26.5 GHz with 20 MHz Information Bandwidth

Agilent PN 89400-13 Extending Vector Signal Analysis to 26.5 GHz with 20 MHz Information Bandwidth Product Note The Agilent Technologies 89400 series vector signal analyzers provide unmatched signal analysis

### POWER SENSOR MANUAL. Boonton Electronics A Subsidiary of Noise/Com -- A Wireless Telecom Group Company 25 Eastmans Road Parsippany, NJ

POWER SENSOR MANUAL Revision Date: 10/03 MANUAL P/N 98501900C CD P/N 98501999C DATE 12/02 Boonton Electronics A Subsidiary of Noise/Com -- A Wireless Telecom Group Company 25 Eastmans Road Parsippany,

### Yesterday s discrete. Ordinary Vector Network Analyzers Get Differential Port Measurement Capability DIFFERENTIAL MEASUREMENTS

From November 2003 High Frequency Electronics Copyright 2003 Summit Technical Media, LLC Ordinary Vector Network Analyzers Get Differential Port Measurement Capability By Dale D. Henkes Applied Computational

### Six-Port Reflectometer: an Alternative Network Analyzer for THz Region. Guoguang Wu

Six-Port Reflectometer: an Alternative Network Analyzer for THz Region Guoguang Wu Outline General Background of Network Analyzer Principles of Six-Port Reflectometer WR-15 Six-port Reflectometer Design,

### Techniques for Precise Cable and Antenna Measurements in the Field

Techniques for Precise Cable and Antenna Measurements in the Field Using FieldFox handheld analyzers Application Note This application note introduces the practical aspects of cable and antenna testing,

### SECTION 2 Transmission Line Theory

SEMICONDUCTOR DESIGN GUIDE Transmission Line Theory SECTION 2 Transmission Line Theory Introduction The ECLinPS family has pushed the world of ECL into the realm of picoseconds. When output transitions

### Keysight Technologies Understanding the Fundamental Principles of Vector Network Analysis. Application Note

Keysight Technologies Understanding the Fundamental Principles of Vector Network Analysis Application Note Introduction Network analysis is the process by which designers and manufacturers measure the

### Microwave Circuit Design and Measurements Lab. INTRODUCTION TO MICROWAVE MEASUREMENTS: DETECTION OF RF POWER AND STANDING WAVES Lab #2

EE 458/558 Microwave Circuit Design and Measurements Lab INTRODUCTION TO MICROWAVE MEASUREMENTS: DETECTION OF RF POWER AND STANDING WAVES Lab #2 The purpose of this lab is to gain a basic understanding

### AVX EMI SOLUTIONS Ron Demcko, Fellow of AVX Corporation Chris Mello, Principal Engineer, AVX Corporation Brian Ward, Business Manager, AVX Corporation

AVX EMI SOLUTIONS Ron Demcko, Fellow of AVX Corporation Chris Mello, Principal Engineer, AVX Corporation Brian Ward, Business Manager, AVX Corporation Abstract EMC compatibility is becoming a key design

### DRIVING LOOOONG CABLES

DRIVING LOOOONG CABLES INTRODUCTION Microphone or line level cables may appear to be foolproof compared to loudspeaker cables. However, they are not. In particular you can easily encounter high frequency

### Calibration i of ESD Generators According to IEC Edition 2.0:

Calibration i of ESD Generators According to IEC 61000-4-2 Edition 2.0:2008-12 Presented by : Standards and Calibration Laboratory (SCL) Innovation and Technology Commission The Government of the HKSAR

### A wave lab inside a coaxial cable

INSTITUTE OF PHYSICS PUBLISHING Eur. J. Phys. 25 (2004) 581 591 EUROPEAN JOURNAL OF PHYSICS PII: S0143-0807(04)76273-X A wave lab inside a coaxial cable JoãoMSerra,MiguelCBrito,JMaiaAlves and A M Vallera

### Application Note, AN-7

Application Note, AN-7 Conducted emissions is a measure of the amount of Electromagnetic Interference (EMI) that is conducted from a unit under test (EUT) to the outside world via the power cables. A Line

### Sensor and Simulation Notes. Note 479. October 2003. A Dual-Polarity Impulse Radiating Antenna

Sensor and Simulation Notes Note 479 October 2003 A Dual-Polarity Impulse Radiating Antenna Leland H. Bowen and Everett G. Farr Farr Research, Inc. Dean I. Lawry Air Force Research Laboratory, Directed