1 Journal of Materials Processing Technology 92±93 (1999) 302±308 Temperature effects in the drilling of human and bovine bone M.T. Hillery a,*, I. Shuaib b a Department of Manufacturing and Operations Engineering, University of Limerick, Limerick, Ireland b Izzak W. Killam Childrens' Orthopaedic Hospital, Halifax, NS, Canada Abstract Bone is an anistropic material, the drilling of which is an essential part of internal xation in orthopaedic and trauma surgery. Some research has been carried out into the drilling of bone, but there are no de nitive recommendations available on the most suitable drill shape or on the optimum drilling speed that should be used in surgical practice. The heat generated during the drilling of bone is proportional to the drilling speed and may be partially dissipated by the presence of blood and tissue uids and also partially carried away by the chips formed. However, some heat may be conducted into the bone which, if exposed to a temperature higher than 558C, may sustain serious damage. This is important in orthopaedics, since most holes drilled are subsequently tapped for retaining screws. If the drill burns the bone to any great depth, the thread will not anchor properly and the xation could work itself loose. This paper describes the drill design for accurate temperature measurement at the drill point and further shows that whilst the temperature may go to much higher than 558C because of the poor conductivity of the bone the depth of damage is not critical insofar as screw xation is concerned. # 1999 Elsevier Science S.A. All rights reserved. Keywords: Drilling; Orthopaedics; Bone; Surgery; Temperature; Thermocouple; Sensor 1. Introduction Bone is a special type of connective tissue consisting of inter-cellular calci ed material. There are two types of bone, the outer hard layer that is called cortical bone and an inner spongy layer called cancellous bone. The outer surface of the bone is covered by a tough layer of osteogenic (boneforming) connective tissue called periosteum. Most of the inside of a bone is hollow and contains bone marrow. The inner surface of the bone is also lined with a similar cell layer with osteogenic properties called endosteum. Both the periosteum and the endosteum contain the bone vascular system which supply it with nutrients and oxygen for bone growth and repair. Bone fracture is a feature of everyday life due to accident or otherwise. When a bone is broken the periosteum and endosteum provide bone-forming cells, which endeavour to bridge the fracture. One of the principal methods of repair and reconstruction of a fracture is achieved by drilling the bone and xing the separate parts together using screws, wires and plates. Fig. 1 shows a typical application of drilling in femoral fracture surgery. Many problems are encountered when drilling bone in orthopaedic and trauma surgery such as hole accuracy, drill *Corresponding author. Tel.: ; fax: address: (M.T. Hillery) wander (walking) and heat generation. Also, other unpredictable situations can occur due to the non-homogeneous structure of the bone material itself. Many different drill-bit designs and geometries have been suggested over the years, [1±3], each with its own claim to success but most of them are based on conventional twistdrill geometry for the drilling of metals. The drills used are: twist drills, guide wires and large diameter drills/reamers Implant and drill materials Implants must be made from corrosion-resistant materials. They mainly consist of 0.3% carbon, 17±19% chromium, 13±15% nickel, 3% molybdenum, 2% manganese and up to 60% iron, with other trace elements. Also, many of the plates and screws used in implant surgery are made from titanium, which, in its pure state, is bio-compatible and corrosion-resistant. There is little or no allergic reaction to it by adjacent tissue, thus allowing implants made from titanium to remain in the human body for long periods of time. There are ISO(5832-1), AISI (316L) and DIN (17-440) standards for implant materials. Drill steels for orthopaedic procedures are not required to be as corrosion-resistant as implant materials, since they are not in contact with body uids or tissue for long periods of time. It is much more important that they retain their cutting /99/$ ± see front matter # 1999 Elsevier Science S.A. All rights reserved. PII: S (99)
2 M.T. Hillery, I. Shuaib / Journal of Materials Processing Technology 92±93 (1999) 302± Fig. 1. Drilling of cancellous bone in femoral fracture surgery. edge and withstand repeated sterilisation cycles in an autoclave at temperatures of up to 1358C. They are manufactured to a standard such as DIN or AISI 440B and have chemical compositions of approximately 0.85% C, 18% Cr, 1% Mo, 1% Mn 1% Si, and trace elements, with the remainder made up of iron Fixation of bone In dynamic bone xation there is some movement at the fracture site so that the load is shared by the implant and the bone, thus allowing the fracture to be subjected to stresses leading to rapid healing. This type of xation allows for early joint mobilisation and reduces or prevents stiffness setting in on other joints. In many orthopaedic procedures holes of 3.2 mm diameter are drilled into the hard cortical bone where f4.5 mm self-tapping surgical screws are used for securing pieces of fractured bone, or implant plates or to anchor wires that are used for binding the pieces of bone together until new growth restores the fractures to their natural state. These screws, plates and/or wires may remain in position for several months. However, it is much more important that the repaired fracture, in the case of a lower limb, is strong enough to support the weight of the patient within hours of the operation. To ensure that this is the case the anchoring of the self-tapping screws has to be as close to perfect as possible. 2. Heat generation when drilling bone Bone is a poor conductor of heat, with the thermal conductivity of fresh cortical bone in the region of 0.38± 2.3 J/msK [4±6]. Heat generation is a common problem during any type of drilling but particularly during the drilling of bone due to the sensitivity of bone to heat damage and the dif culty of conducting heat away from the cutting edge, both as a result of the poor conductivity of the bone and the inability to use a coolant because of the danger of causing infection to the area. However, some of the heat generated during drilling may be partially dissipated by the presence of blood and tissue uids and partially carried away by the chips formed but the temperature rise at the cutting edge in a deep cortical hole could be signi cantly high. Lundskog  states that if bone is exposed for longer than 30 s at 508C cellular necrosis will be induced. Mortiz and Henrique [7,8] established that when epithelial cells are exposed to a temperature of 708C they will be damaged immediately, that when they are exposed to a temperature of 558C for 30 s the result will be the same and at a temperature of 458C harmful effects will occur after 5 h. In general, the literature shows that if the temperature rises above 558C for a period of longer than one-half a minute, serious damage will be done to the bone, which may take several weeks to fully recover. Some previous attempts have been made to measure temperature when drilling bone by inserting thermocouples adjacent to the drill as it progressed into the hole. Krause  attempted to measure temperature when drilling bone by cementing a f1.07 mm copper±constantan thermocouple into a hole previously drilled from the endoesteal surface. The drill under test approached the thermocouple along the same axis from the periosteum surface, stopping just short of it. This gave an indication of the temperature of the cutting edges of the drill as they approached the thermocouple. No irrigation was used during the drilling. As would be expected, indifferent results were obtained using this method. Methews  and Methews and Green  cemented thermocouples into four separate f0.7 mm holes drilled about a centre at radii of 2.5, 3.0, 4.0 and 5.0 mm. They then drilled a f4 mm hole through the centre and measured the temperature from each thermocouple. Their ndings showed that cutting speed had very little effect on temperature change above 558C but that the force applied to the drill caused a much greater temperature increase. Insofar as can be established from the literature, attempts at temperature measurement when drilling cortical bone have been carried out by placing thermocouples into the bone at given distances from the drilling area. Indifferent and sometimes contradictory results have been achieved using these methods. It is the opinion of the present authors that because of the poor thermal conductivity of bone, its structural inconstancy and the great dif culty in modelling for heat-transfer purposes, embedding thermocouples in the bone adjacent to the drilling operation is not a satisfactory method of measuring the temperature effects.
3 304 M.T. Hillery, I. Shuaib / Journal of Materials Processing Technology 92±93 (1999) 302± Aim of the research The aim of the research was to accurately establish the temperature at the cutting edge when drilling both human and animal bone and to examine the effect this increase in temperature would have on the physical aspects of the bone structure with particular emphasis on the ability of the bone to retain a surgical self-tapping screw. This research was carried out as part of an overall investigation into the mechanics of drilling bone . 3. Methodology Research by Hillery and Shuaib  has established that the optimum point angle for drilling bone lies in the range of 70±808 and that a rake angle in the region of 20±308 suf ciently clears the chips and also gives the lowest thrust force without `grabbing' on drill break-through. This drill point geometry was used for the research on temperature measurement at the drill point Bone collection and preparation Drilling tests were carried out on both human and animal bone. A distal ulna, the heads of femurs and many other pieces of cortical and metaphyseal bone were collected from two separate orthopaedic operating theatres. Some of the experiments were performed on fresh bone within 24 h of their removal, the remainder of the bone being immediately frozen to 188C and stored in a refrigerator until required, as recommended by Sedlin and Hirsch . In the case of animal bone the specimens were obtained from a local abattoir and were from 14 months old cattle. The tibia was chosen as a suitable long bone for drilling, the thickness of the cortical bone of the specimens ranging from 7 to 9 mm. A special container that was capable of holding 2 l of normal saline solution was tted with a heating element to heat the uid within the container. A thermostat was also tted which gave a precise temperature control of the uid to 0.5 K. The solution was maintained at 378C to simulate body temperature. During the thawing process the specimen was sealed in a waterproof cover in order to prevent the absorption of uid by the specimen during the process. The specimens were then slowly thawed and when ready they were removed from the container. At this point all the soft tissue and the periosteum was removed from the specimens using a surgical knife. A standard orthopaedic saw was used for cutting the specimens. The bovine tibiae were cut rst at each end at the junction of diaphysis with the metaphyseal bone. The metaphyseal bone was then discarded. The diaphyseal bone was cut longitudinally into halves. These pieces were than cut into lengths of 3 cm with an average width of 2 cm Bone hardness There is a great variation in the hardness of the different bones in a particular body and there can be a great variation in the hardness of a particular bone due to the age, gender, race or diet of the person. This variation will cause a consequent variation in the thrust, torque and power values of a particular drilling operation. Therefore the hardness of all of the bone used was established in order to compare `like with like.' It was dif cult or impossible to use normal hardness testing machines such as the Brinell or Vickers instruments, but a simple hand-held Barber Coleman Impresser model 934-1, which was capable of operating over very small areas, gave accurate and consistent results when compared with results from a Rockwell-H machine on samples large enough to be tested using both machines. The hardness values were taken at short intervals across the thickness of the bone and also values were taken along the bone surface. Bones which had similar hardness characteristics were grouped and used later in drilling tests Design for drilling It was decided that the most ef cient method of measuring the temperature at the cutting edge was to locate a thermocouple as close as possible to the point within the drill-bit itself. The reasoning for this was that stainless steel is a much better conductor of heat than bone, having a thermal conductivity of 14 J/msK, and it was considered that better results could be obtained than those previously available Drill-bit design In order to place the thermocouple in the drill-bit as close as possible to its cutting edge, surgical stainless steel piping of 4.5 mm outside diameter having a bore of approximately 1 mm was selected to make ve separate drill-bits. This steel tube was ground to a diameter of 3.2 mm and uted to a helix angle of 238, as for normal twist drill manufacture. The tip of each tube was closed by welding using a ller rod of the same stainless steel material. Three of the drill-bits were then ground to a 908 point angle and one each to 708 and 808. Previous research  by the authors has shown that these geometries are the most ef cient for the drilling of bone. A K-type insulated thermocouple was placed and `bedded' inside the drill and its leads were then glued along the drill barrel. The wall thickness at the cutting edge was in the region 0.5±0.7 mm. The system was calibrated by placing the drill-bit and a standard thermocouple into a container of water that was heated in steps of 5 K to boiling point. The temperature response time for the drill-bit compared to the thermocouple was, at most, 3 s throughout the range.
4 The drilling machine The machine used for drilling was a modi ed Emco milling machine with an inverter tted to the spindle motor, which gave a speed range from approximately 50±2500 rev/min. The spindle speed reading was detected by a specially tted digital read-out meter that was `switched' by a small proximity magnet attached to the top of the spindle. A small drilling machine chuck was used for holding the drill-bit. The chuck in turn was rotated by the milling machine collet. The chuck also had a hole drilled through its shank to allow for the lead wires from the thermocouple to be threaded through it. These lead wires were then connected to an electrical `connector block,' the other side of which was connected to a pair of brass slip rings tted at the top of the machine spindle. A pair of brushes was then used to make electrical contact with a digital read-out pyrometer capable of reading to 0.1 K. An in nitely variable DC servo motor was connected to the feed lead-screw through a toothed belt and reduction gear for precisely controlling the feed rate from 1 to 200 mm/min The drilling dynamometer All drilling took place by clamping the sample bone onto the work table of a sensitive drilling dynamometer  that was capable of taking pieces of a length of up to 300 mm and height 25 mm. This dynamometer measured both the torque and the thrust forces generated. The active elements of the measuring cantilevers were Kwoya KFD-2-D16-L30-type strain gauges with pre-attached lead wires in a half bridge circuit with separate identical compensation gauges. The dynamometer was calibrated by physically applying different known load and torque values. The general machine arrangement is shown in Fig The drilling procedure M.T. Hillery, I. Shuaib / Journal of Materials Processing Technology 92±93 (1999) 302± The thawed bones were clamped to the work table of the dynamometer and then drilled. At least two holes were drilled for each speed and feed. Occasionally, wedging of the drill-bit occured due to the chips not clearing along the utes. This happened particularly in the high speed drilling of bovine cortical bone. In one case this wedging resulted in the fracture of the drill-bit. Due to this, if there was any major variation from one piece of data to the next, the drill was changed and the experiment was repeated to verify the result. The speeds used were 400, 600, 800, 1000, 1200, 1400, 1600, 1800 and 2000 rev/min. These were chosen in order to go below and above the speeds recommended in the literature for the drilling of bone. The feeds ranged from 40 to 60 mm/min and generally settled at 50 mm/min. These feeds were chosen arbitrarily, since in orthopaedic practice the feed rate varies from surgeon to surgeon and in the case of a particular Fig. 2. Drill and chuck assembly for temperature measurement. surgeon there will also be a variation, since the drill is hand-held. 4. Temperature variation in drilling bone Some holes were drilled using 708, 808 and 908 point angles to ascertain the effect of point geometry on temperature. It was found that temperature variation across the three drills was not signi cant.
5 306 M.T. Hillery, I. Shuaib / Journal of Materials Processing Technology 92±93 (1999) 302±308 Fig. 3. Temperature vs. speed and depth when drilling bovine cortical bone. Fig. 4. Temperature vs. speed and depth when drilling human bone. Since it was easier to manufacture drill-bits with a 908 point angle, this geometry was used to drill over 100 holes, without coolant, to a depth of up to 9 mm in steps of 3 mm for bovine bone and to 6±7 mm for human bone. The drilling was performed at a room temperature of 268C. Fig. 3 shows the temperature variation for bovine cortical bone for a variety of speeds and a variety of depths of hole. The feed was set at 50 mm/min. Considering the rst drilling series to a depth of 3 mm the average temperature variation ranged from 388C at 400 rev/ min down to slightly above room temperature at 1300 rev/ min and rising again to 298C at 2000 rev/min. When the bone was drilled to a depth of 9 mm there was an increase in temperature, which ranged from 102±1178C as the speed ranged from 400 to 2000 rev/min. Fig. 4 shows a similar temperature plot for human bone using the same speeds and feed. The temperature recorded whilst drilling to a depth of 2±3 mm at room temperature of 268C and at a drill speed of 400 rev/min gave a temperature reading of 368C. In the speed range of 600±2000 rev/min the temperature reading did not reach greater than 288C, but as the depth of the hole was increased to 5±6 mm the temperature also increased to 808C at 400 rev/min. As the speed approached 1200 rev/min at this depth, the temperature gradually dropped to 688C and rose again to approximately 758C with an increase in speed to 2000 rev/min. It is obvious that as the drill-bit travels further into the bone there is an increase in temperature due to the inability of the bone to conduct the heat generated away from the point. Earlier research by the authors  has shown that when using the same drill-bit diameter and geometry there is an exponential falling-off of the cutting torque from N m at 400 rev/min to an asymptote of N m at 2000 rev/min. There was a corresponding falling-off of the thrust force from 48 N at 400 rev/min to an asymptote 23 N at 2000 rev/min. These trends are shown in Figs. 5 and 6, respectively. It is suggested that the temperature drop in the cutting range from 400 to 1200 rev/min re ects the reduction in the cutting torque and thrust parameters, which in themselves more than compensate for the increase in heat generation due to the higher speeds. The slight increase in temperature over the speed range of 1200±2000 rev/min probably could be attributed to the drop-off in the rate of torque and thrust change, which were not suf cient to compensate for speed change, since the torque and thrust force at 1200 rev/min was only N m and 25 N, respectively. Fig. 5. Torque±speed relationship for the drilling of bovine cortical bone.
6 M.T. Hillery, I. Shuaib / Journal of Materials Processing Technology 92±93 (1999) 302± insulation characteristics of bone. It also shows that placing thermocouples within the bone, adjacent to the drilling surface, as a means of measuring temperature is not a satisfactory solution to the problem of temperature measurement due to the insulating effect of the bone itself and that if temperatures greater than 458C are reached by this method, then the temperature at the drill-bit cutting edge must be considerably higher with consequent severe damage to the bone surface of the hole Pull test for self-tapping screws Fig. 6. Thrust±speed relationship for the drilling of bovine cortical bone. 5. Examination of the holes For the bovine cortical bone, 10 of the holes that had been drilled to a depth of 9 mm were examined under electron and optical microscopes. The holes produced were cut longitudinally and the surface of each hole was cleaned gently with a cloth and a mild jet of air in order to remove any debris from its surface. These specimens were then left to partially dry at 378C for 3 days. They were then exposed to a vacuum after which they were sputtered with gold and examined on a Joel JSM 840 scanning electron microscope. From this examination it was noted that there were a number of different micro-fractures, which had various shapes made up of: (i) irregular lines; and (ii) dendritic shapes. Circular markings were also noted along the diameter of the hole. These were produced by the rubbing of the drill bit and the trapped chips, which embedded themselves into the hole surface. However, the depth of burning or rubbing damage was comparatively shallow and was only in the region of 0.2± 0.3 mm at the section where the drill recorded a temperature of 1158C. When examined under the optical microscope some of the holes appeared to be slightly burned, but this appearance could also have been due to the rubbing of the drill behind the point as it proceeded into the hole. Whilst drilling there was also a faint smell of burning. A similar examination was carried out on a number of holes drilled in the human bone. In this case, probably due to the lower temperatures, it was not possible to detect any evidence of burning. It is reasonable to assume that this lack of damage to the hole surface at temperatures that would have been previously considered to be very high is due to the heat In order to establish the holding force of the 25 mm long, f4.5 self-tapping screws, 10 f3.2 holes were drilled in a fresh bovine tibia. All of the screws were driven by the same person using a hand-held screw driver, as would be the case in a normal surgical procedure. All of the holes were suf ciently close to each other to give consistent bone surface hardness. The same speeds, ranging from 400 to 2000 rev/min, were selected as already used for the temperature tests and all holes were drilled to a depth of 9 mm. A `pull' test was devised  where a standard sensitive digital read-out `Pull/Push' dynamometer was secured to the chuck of a drilling machine and a special collar was attached to the other end that could t easily under the head of the screw. A tensile load could then be applied to the head of the screw by raising the chuck of the drilling machine. The force required to withdraw the screw was then read off from the dynamometer. Fig. 7 shows the `pull test' arrangement. Fig. 7. Screw pull test.
7 308 M.T. Hillery, I. Shuaib / Journal of Materials Processing Technology 92±93 (1999) 302±308 The pull test for all 10 holes ranged between 350±360 N and it was not possible to detect a trend due to cutting speed or temperature effect. 6. Conclusions There was no signi cant difference noted in the temperatures produced when drilling with tools with point angles of 70, 80 or 908 with a 238 positive rake angle. The temperature increased as the depth of the hole increased. The temperature generated when drilling a bovine bone was higher than that produced when drilling human bone. This is to be expected, since the bovine bone is much harder and the depth of the cortical bone is greater than that of the human bone. The cortical specimen thickness for the bovine bone was 7±9 mm and that of the human ulna 3± 5mm. Speeds of 800±1400 rev/min should be used when drilling with a f3.2 mm drill-bit to provide the best cutting conditions and maintaining temperatures at a manageable level. The temperature developed at the drill cutting edge, whilst high relative to previously suggested limits, does not in fact seriously damage the effective depth of bone thread used in securing surgical self-tapping screws. Acknowledgements The authors wish to thank Mohawk Europe Ltd., manufacturers of specialist twist drills, Shannon, Ireland, for their technical and nancial contribution to the project. References  C.H. Jacobs, A study of the bone machining process, drilling, J. Biomechanics 9 (1976) 343±349.  S. Saha, Surgical drilling and design of an improved drill, J. Biomechanics 104 (1982) 245±252.  M.T. Hillery, I. Shuaib, The drilling of bone using guide wires and twist drills, IMC-13, Ireland, (1996) pp. 33±42.  R. Huskies, Some fundamental aspects of human joint replacement, Acta Orthop. Scand. 185 (1979) 62±63.  A. Lansari, Effects of torque and heat generation during direct cortical trochar tipped pin insertion, Annals of Biomechanical Eng. 19 (1991)  J. Lundskog, Heat and bone tissue, Scand. J. Plastic and Reconstructive Surgery, Sup. 9 (1972).  A.R. Mortiz, F.C. Henrique, Studies of thermal injuries, II, Am. J. Pathol. 23 (1947) 695.  A.R. Mortiz, F.C. Henerique, The relative importance of time and surface temperature in the causation of cutaneous burns, Am. J. Physiol. 23 (1947) 695±719.  W.R. Krause, Temperature elevations in orthopaedic cutting operations, J. Biomechanics 15 (1985) 267±275.  L.S. Methews, Temperature measured in human cortical bone when drilling, J.B.J.S. 54A (1972) 297±308.  L.S. Methews, C.A. Green, The thermal effects of skeletal fixation± Pin insertion in bone, J.B.J.S. 66A (7) (1984).  I. Shuaib, The drilling of bone using guide wires, twist drills and cutters, Ph.D. thesis, University of Limerick,  E.D. Sedlin, C. Hirsch, Factors affecting the deformation of the physical properties of femoral cortical bone, Acta Orthop. Scand. 37 (1966) 29±48.  W.P. Hennessy, M.T. Hillery, The design and manufacture of a drilling dynamometer for drilling Kevlar-49 reinforced composites, Proc. IMC-8, Ireland, (1991) pp. 803±816.  I. Shuaib, M.T. Hillery, Forces generated in guide wires when drilling human bone, Proc. Inst. Mech. Eng., Part H, J. Eng. in Medicine 209(H3) (1995) 157±162.
Think precision, Think HSS REAMING SUMMARY REAMING TOOLS 2 Zoom on a reamer 3 Which HSS for maximum efficiency? 4 Coatings for the best performance 5 Vocabulary 6 Choose the right design 7 Types of bevel
Zimmer Small Fragment Universal Locking System Surgical Technique Zimmer Small Fragment Universal Locking System 1 Zimmer Small Fragment Universal Locking System Surgical Technique Table of Contents Introduction
INTERNATIONAL JOURNAL OF MECHANICAL ENGINEERING AND TECHNOLOGY (IJMET) International Journal of Mechanical Engineering and Technology (IJMET), ISSN 0976 6340(Print), ISSN 0976 6340 (Print) ISSN 0976 6359
Standard Internal Hex Touareg TM -OS Touareg TM -S Swell TM Touareg -S Touareg -OS Swell About ADIN Adin Dental Implant Systems Ltd., designs, manufactures and markets state of the art, technologically
Spark plug thread repair for Ford Triton cylinder heads Step-by-step instructions: Identification Installation Verification Specifically designed and tested for 4.6L, 5.4L, and 6.8L 2 and 4 valve heads,
ORTHODONTIC MINI IMPLANTS Clinical procedure for positioning Orthodontics and Implantology 2 All rights are reserved. Any reproduction of the present publication is prohibited in whole or in part and by
1 von 9 Precision made in Europe. As per DIN 8606. The heart of a system, versatile and expandable. Main switch with auto-start protection and emergency off. Precision lathe chuck as per DIN 6386 (Ø 100mm).
T A B L E T 1 1 of 7 Tests & Inspection Cable PCUT & PCUT-A (Table T1) T E S T S A N D I N S P E C T I O N C A B L E P C U T A N D P C U T - A No. Test Scale MOC Requirements G 20:10:001:01 Defined Test
1 CASAFLEX - 11 10 114,3 x 4,5 3 5 2 4 1 7 6 8 12 9 Do not begin work until you have read the installation instruction carefully. 1 internal backing sleeve 2 graphite packing 3 connecting piece 4 thrust
Screws Screws for definitive use. Made from medical grade 5 ELI Titanium Integrity and soundness guaranteed by stress tests of up to 150% of their nominal torques. Their designs ensure a perfect fit with
Pump Skid Fabrication for Magnetic Coupling Rick Soltis Chief Mechanic City of Bedford Contents Magnetic Couplings What They Are, How They Work, Where They re Used Fabrication and Manufacturing of Pump
RESEARCH ARTICLE ISSN: 2321-7758 DESIGN AND DEVELOPMENT OF A DYNAMOMETER FOR MEASURING THRUST AND TORQUE IN DRILLING APPLICATION SREEJITH C 1,MANU RAJ K R 2 1 PG Scholar, M.Tech Machine Design, Nehru College
Installation Manual Elo Touch Solutions Wall-mounting Kit for the 5501L IDS Touchmonitors SW602206 Rev B Table of Contents Chapter 1: Safety Warning... 3 Chapter 2: Kit Contents... 4 Included in Kit...
Aesculap Veterinary Orthopaedics Targon VET Interlocking Nail Flexibility Stability Dynamics The Targon VET is based on new, technology which reverses the known principles of interlocking nails. The rotation-stabilising
Guidelines for Earthquake Bracing of Residential Water Heaters Department of General Services Division of the State Architect 1102 Q Street, Suite 5100 Sacramento, CA 95814 Phone: (916) 324-7099 Fax: (916)
Femur Shaft Fractures (Broken Thighbone) Page ( 1 ) Your thighbone (femur) is the longest and strongest bone in your body. Because the femur is so strong, it usually takes a lot of force to break it. Car
Resharpening Companion 10950 Correct Angles, Pictures, and Step-By-Step Instructions The Resharpening Companion is meant to be a guide and quick reference to help you resharpen. It is not meant to replace
Lathe Milling Attachment By L C. MASON BY CLEVERLY stacking cold-rolled flat stock together, T-slots and slide for this lathe milling attachment are made without costly machinery. In fact, only two tools,
Rebuild Instructions for 70001 and 70010 Transmission Brinn, Incorporated 1615 Tech Drive Bay City, MI 48706 Telephone 989.686.8920 Fax 989.686.6520 www.brinninc.com Notice Read all instructions before
w Technical Product Information Subminiature Load Cell 1. Introduction... 2 2. Preparing for use... 2 2.1 Unpacking... 2 2.2 Using the instrument for the first time... 2 2.3 Grounding and potential connection...
Technique Guide 3.5 mm LCP Clavicle Hook Plates. Part of the Synthes locking compression plate (LCP) system. Table of Contents Introduction 3.5 mm LCP Clavicle Hook Plates 2 AO Principles 4 Indications
Orthopedic Foot Instruments. Dedicated instruments for reconstructive foot surgery. Surgical Technique This publication is not intended for distribution in the USA. Instruments and implants approved by
Operating Instructions Drill rig DRU160 Index 000 / 001 Original operating instructions 10988825 en / 20.10.2009 Congratulations! With a Hydrostress unit from TYROLIT you have chosen a tried and tested
Technique Guide Norian SRS Fast Set Putty. Calcium phosphate bone void filler. TableofContents Introduction Norian SRS Fast Set Putty 2 Indications and Contraindications 3 Basic Science 4 Surgical Technique
Trillium 40 Axis Spring Tensioner Wire Replacement Instructions 1 Overview The objective is to replace the broken axis spring tensioner wire. This requires the following tasks: 1. Remove the seismometer
Victorian Certificate of Education 2013 SUPERVISOR TO ATTACH PROCESSING LABEL HERE STUDENT NUMBER Letter Figures Words VCE VET ENGINEERING STUDIES Written examination Section Wednesday 20 November 2013
TECHNICAL INFORMATION Models No. Description HP2070, HP2070F, HP2071, HP2071F 20mm (3/4") 2-Speed Hammer Drill CONCEPT AND MAIN APPLICATIONS L W NEW TOOL P 1 /13 W L These hammer drills have been developed
HOW TO MAKE A MOUNTING BRACKET FOR AN ELECTRIC MOTOR Often, brackets supplied with purchased items (e.g. electric motors or servos) are not ideally suited for fitting into our models or they are not supplied
machine tools for railways RAFAMET S.A. Comprehensive solutions, advanced technologies and cost efficient productivity now much more than ever before are obvious requirements that the right equipment supplier
Tool Holder Catalog INDEX ABC VDI 25 LL3602.30051-10.04.2014 0414/Rö Disclaimer The illustrations in this documents may deviate from the delivered product. We reserve the right to make changes due to errors
Technology Design Task -Artefact Production- Design Task Artefact A number of factors can be considered in influencing the extent to which the completed artefact fully satisfies the chosen design brief
2100 AD 015 0009 Mirror Elevator Ball Nut Replacement Procedure Derek Guenther 1/28/2015 Rev. Purpose The purpose of this document is to describe the procedure necessary to replace one of the ball nuts
Home"" """"> ar.cn.de.en.es.fr.id.it.ph.po.ru.sw Milling of Grooves, Elongated Slots and Break-throughs - Course: Techniques for machining of material. Trainees' handbook of lessons (Institut fr Berufliche
Implant materials How to use this handout? The left column is the information as given during the lecture. The column at the right gives you space to make personal notes. Learning outcomes At the end of
Computer Aided Manufacturing (CAM) CNC HARDWARE & TOOLING BASICS Assoc. Prof. Dr. Tamer S. Mahmoud 1. Parts of CNC Machine Tools Any CNC machine tool essentially consists of the following parts: Part Program,
Overview Milling Machine Fundamentals Wayne Staats, UW-Madison FSAE Safety Shop Etiquette Before Machining Indicating Calculating Feeds and Speeds Machining Maintenance Safety Respect the machines Common
Mounting Instructions for SP4 Power Modules Pierre-Laurent Doumergue R&D Engineer Microsemi Power Module Products 26 rue de Campilleau 33 520 Bruges, France Introduction: This application note gives the
Laboratory 4 Torsion Testing Objectives Students are required to understand the principles of torsion testing, practice their testing skills and interpreting the experimental results of the provided materials
The information contained in this document is intended for healthcare professionals only. Dall-Miles Cabling System Dall-Miles Recon and Trauma Cable System Trochanteric Reattachment Using the Trochanteric
APPLICATIONS Process Industry Weak Acids Alkalies Compressed Air Pulp & Paper MODELS ASM - Flanged Connection OPTIONS Control Rods Oil & Gas Water & Waste Pump suction & discharge Sea water Chemical lines
Mayo Clinic CoNGRUENT RADIAL HEAD PLATE Since 1988 Acumed has been designing solutions to the demanding situations facing orthopedic surgeons, hospitals and their patients. Our strategy has been to know
The "DAVID" Steam Engine By Alan Marconett Hobbit Engineering HTTP://WWW.HobbitEngineering.com Alan@HobbitEngineering.com (c) 2/20/03 The David steam engine is a simple oscillating steam engine, many plans
This specification covers the requirements for application of ELCON Drawer Series Connectors: True Hot Plug, Blind Mating Mixed Signal and Power Connectors. These connectors are designed for use in pluggable
Tridelta Magnetsysteme A Tridelta Group Company Permanent Magnetic Couplings and Brakes for Drive Technology Raw Materials Magnets Systems and Components Magnet N1 Soft iron N2 Resin Introduction and principals
POWERSANDING WOOD BOWLS by John Franklin We have all been taught to consider sanding a less than desirable function when turning wood. Although skill with your tools will normally reduce the need to do
CHAPTER 10 RESTS AND DEFINITIONS A REST is any rigid part of an RPD framework which contacts a properly prepared surface of a tooth. A REST PREPARATION or REST SEAT is any portion of a tooth or restoration
Guidelines for Earthquake Bracing Residential Water Heaters Department of General Services Division of the State Architect In accordance with the Health and Safety Code Section 19215, the Division of the
Use of Strain Gauge Rosette to Investigate Stress concentration in Isotropic and Orthotropic Plate with Circular Hole Mr.V.G.Aradhye 1, Prof.S.S.Kulkarni 2 1 PG Scholar, Mechanical department, SKN Sinhgad
Surgical technique End Cap for TEN. For axial stabilization and simultaneous protection of soft tissue. Table of contents Indications and contraindications 3 Implants 4 Instruments 4 Preoperative planning
Modular Sternal Cable System. Flexibility and strength in sternal closure and repair. Multiple closure options Stable fixation in all configurations Improved strength and flexibility compared to wire Modular
THE INFLUENCE OF STEEL GRADE AND STEEL HARDNESS ON TOOL LIFE WHEN MILLING IN HARDENED TOOL STEEL S. Gunnarsson, B. Högman and L. G. Nordh Uddeholm Tooling AB Research and Development 683 85 Hagfors Sweden
The Do s and Don ts of Pressure Transducers ABSTRACT When specifying a pressure transducer for a process measurement, a number of items have to be considered. Some of the more important ones are discussed
PERIPROSTHETIC IMPLANTS PRODUCT OVERVIEW CLINICAL SOLUTIONS Periprosthetic fractures present unique challenges, such as how to gain fixation when the medullary canal is occupied. Special techniques and
tapping Technical Information Troubleshooting Guide 115 TAP DOES NOT START Program depth: Tap drill size: Tap sharpness: Compression stroke may use up the entire program depth. Check for tap drill size.
Mr. Will Rood B & B Precise Products 25 Neck Road Benton, ME 04901 Date June 3, 2008 Quote # 06032008 Valid for 30 Days Quoted by Norman Signature Norman Crepeau Special Condition Subject to prior sale
Posterior Referencing Surgical Technique Posterior Referencing Surgical Technique INTRO Introduction Instrumentation Successful total knee arthroplasty depends in part on re-establishment of normal lower
MACHINING OPERATIONS AND MACHINE TOOLS 1. Turning and Related Operations 2. Drilling and Related Operations 3. Milling 4. Machining & Turning Centers 5. Other Machining Operations 6. Shape, Tolerance and
IMPLANTS CATALOG ABOUT US Edison Medical The Complete and Comprehensive solution for implants! Edison Medical Ltd. is engaged in the import, export, marketing and distribution of dental Implants and other
Super X3 Mill Dismantling and Reassembly Guide A picture story book to help you dismantle and reassemble your Sieg Super X3 Mill Arc Euro Trade Ltd. 10 Archdale Street, Syston, Leicester, LE7 1NA. Web:
Technical Service Bulletin: Triac Printed Circuit Board Replacement TRONIC 5000C Pro Models: WH17, WH27, WH36 Introduction Fig. 1 ELECTRICITY IS EXTREMELY DANGEROUS. TAKE EXTRA PRECAUTIONS AND ENSURE ALL
Product Facts Pins and sockets can be intermixed in the same housing Positive polarization Rear cavity identification completely enclosed in housings Positive locking housings Insulation capability to.00
Installation, operation and maintenance manual TX 35A Rev.10 may 2012 Page 1 of 18 1.0.0 Table of contents INSTALLATION, OPERATION AND MAINTENANCE MANUAL...1 1.0.0 TABLE OF CONTENTS...2 2.0.0 ILLUSTRATIONS...2
Solstice/Sky Water Pump Replacement The water pump on the Solstice/Sky is starting to need replacement on some vehicles. This guide will help in replacing the water pump while the engine is still in the
Technique Guide 3.5 mm and 4.5 mm Curved Locking Compression Plates (LCP). For minimally invasive osteosynthesis. Table of Contents Introduction 3.5 mm and 4.5 mm Curved Locking Compression 2 Plates (LCP)
1 Unit 6: EXTRUSION Introduction: Extrusion is a metal working process in which cross section of metal is reduced by forcing the metal through a die orifice under high pressure. It is used to produce cylindrical
Understanding Dental Implants Comfort and Confidence Again A new smile It s no fun when you re missing teeth. You may not feel comfortable eating or speaking. You might even avoid smiling in public. Fortunately,