1 Correlation between OATS, Fully Anechoic Room and GTEM Radiated Emissions Stephen Clay Introduction: Just a few words about myself. My name is Steve Clay and I work for Nokia Mobile Phones, and it is my job to measure our phones which are designed in England. 1
2 Radiated Emissions for AMPS Mobile Phone In this presentation, I shall briefly describe the experiments that I've done with a mobile phone which will show how the different radiated emissions measurements work in practice. From an EMC point of view, an AMPS Analogue phone has advantage of continuous RF signal with some interesting radiation patterns - easy to measure because it is very small and can be measured with no connecting leads which can upset repeatability: This has enabled me to avoid the usual statistical comparison techniques which are very unconvincing.
3 SPECIFICATIONS FCC CFR47 Part 22 Transmitter Power (at 850 MHz, +28dBm) Transmitter Harmonics (<-13 dbm) Receive Band (-80 dbm) ANSI C63.4 I work in a very fast moving business, and for AMPS phones, we need to FCC approval. As we cannot afford to delay a product launch, it is my jo to ensure that we do not fail by checking the performance during development in our lab. Just a quick look at one of the specifications we have to meet. The specifications usually use a 3metre OATS
4 RADIATION PATTERNS Half Wave Resonant Dipole Top View Side View First let us look at some of the patterns we could be measuring. Here is the familiar picture of the radiation pattern of a vertical half wave dipole; it is omnidirectional from above, and doughnut shaped from the side - approximately. Elemental dipole gives similar shapes
5 RADIATION PATTERNS Fundamental 2 x frequency 3 x frequency However, if we increase the frequency, we get an increased number of lobes, something like this. For the case of a mobile phone, non linearities in the Power Amplifier generate 3rd harmonic signals which leak through the duplex filter and can give a similar pattern to that shown. This assumes a perfect dipole and no radiation from the case.
6 RADIATION PATTERNS Mobile Phone fundamental Top View Side View 23 dbm 26 dbm 20 dbm 23 dbm Another cause of directivity was found due to the relative positioning of the antenna and the phone case, and the p.c.b. layout. This gave greater radiation away from the antenna corner, and also the radiation was also slightly tilted downwards. I should just strees that this is not a paper about mobile phones, but I do need to explain these patterns to show how the results vary if we choose different methods, and this will now be looked at.
7 FCC TEST HOUSE 3 m OATS TEST SITES CAMBERLEY Fully Anechoic Room Small GTEM Large GTEM I should just explain that this work came about because of differences that we got between our FCC Test House which uses a 3m OATS, and our own facilities which due to lack of space could only house a couple of GTEMs, and an Screened Room which could only be used Fully Anechoic; it wasn't big enough to be Semi Anechoic.
8 SIMPLIFIED GTEM THEORY Before describing this, I would just like to describe the simplified method that I have used to derive the GTEM equations. I should say that anybody who likes pages of equations had better turn away now!
9 Centre Line h E GTEM Theory V E = V / h 20log E = 20log V - 20log h AF(gtem) = V(dBV) - E(dBV/m) = 20 log h I think it's fair to say that this can be done on the back of a cigarette packet: For an EUT placed on the centre line, a GTEM is sensitive to vertically polarized radiation in the direction of the apex as shown. Mathematically, the field at any point is determined simply by the ratio of the voltage and septum height. If we express this in dbs as an Antenna Factor, the difference between terminal voltage (in dbuv) and field (in dbuv/m)is 20logh.
10 h Substitution Method E Vout Vin Vin - Vout = 20log h log f Now, mobile phones are usually specified in terms of achieving the power that would be fed to a half wave dipole substituted in place of the EUT. My computer used to have the full IEC equations to give the field and then worked backwards to give the dipole power, although I couldn't understand the theory at all. I had one of those bathtub thoughts where I wondered "What happens if you just add the GTEM and dipole Antenna Factors together?" Answer: Exactly the same as for the complicated method!! Not even a need for any correction factor, and it was simply verified using a Signal Generator and Spectrum Analyser.
11 Free Space at 'd' metres E(at 0) Vout E(at d) = Vout + 20log h - ( log d - 20 log f) In a similar simple way, it is possible to extrapolate to the equivalent field at say 10 metres by correcting the GTEM Antenna Factor by the Free Space Normalised Site Transmission Loss. All of these assume a far field situation of course. From this we can imagine a GTEM as the limit case of an EUT placed opposite to a Horn antenna with the distance 'd' =0.
12 IEC Correlation Algorithm Free Space Electric Field Resultant Voltage Vres = (Vx² + Vy² + Vz²) Ground Reflection Simulation Approx. +5dB For interest, should you wish the full IEC equations, they can be obtained by using the resultant voltage for 3 orthogonal measurements in the previous equation, and adding a composite V & H ground reflection simulation. As we shall see, I don't use these equations because they make assumptions that there is no directionality in the EUT, and my mobile phone does not meet this criteria.
13 OATS Consider a radiation source with directional characteristics on an OATS.
14 3 m Site (V) 4 m direct path phone +45 reflected path 1 m turntable Picture shows a 3m OATS measuring a 3rd harmonic signal, for example. Surely there is an ambiguity here. 1. Are we measuring the radiation pattern on a sphere of up to +45? 2. Or are we optimizing the direct and reflected rays? It is actually a mixture. We could look at the lobe coming out at +45 as antenna moved up and down. Or the lobe coming out horizontally, as antenna moved up and down. I challenge anybody to work out exactly what we will get.
15 10 m Site (V) 4 m direct path phone +17 reflected path 1 m Now not only is the OATS measurement ambiguous, it is also inconsistent between sites - Here's a 10m site Again, looking at the 3rd harmonic. It is 1. Distance dependent - a 10m site measures up to +17degrees instead. 2. Antenna directivity dependent - the amount of signal for the direct and reflected paths may not be equal etc. 3. It is very tilt dependent - tilting to an angle half way between the 2 may not give the same gain on the 2 paths and nulls could affect the results. It is a fundamental assumption that there is no directionality using the OATS, and this si not necessarily be the case.
16 Universal Specification "The maximum field strength allowed at a specified distance from the EUT, at a specified frequency, at all angles in an azimuthal plane for both horizontal and vertical polarization" In an attempt to define what is the purpose of the OATS where the omnidirectional criteria is not met, I am suggesting this as a definition Reading the specifications, it seems that the idea behind the OATS is to measure the radiation from an EUT in the horizontal plane. Could argue this is undertesting. What about radiation vertically upwards etc. Or could say overtesting. Shouldn't we only measure a few faces so we don't miss any time varying signals.
17 Fully Anechoic Room Here is a photograph of our Fully Anechoic Room used to simulate free space.
18 phone d turntable This is actually a simple case because there is no ground plane, and consequently no height search - the main disadvantage is cost, as it's a case of "The bigger the better" with the best absorber to minimise reflections. Here, the distance is not critical. In the 3rd harmonic case shown, there is no ambiguity; we are simply measuring the horizontal lobes and ignoring the angled lobes.
19 Small GTEM And now for the GTEMs. Here's a rather old picture of our Small GTEM and its equipment rack.
20 Phone mounted inside And another picture of a phone mounted on the turntable/ manipulator. The leads and charger are removed for the measurements. The turntable need to be improved to make it more electromagnetically transparent.
21 3 Axis Method ( IEC ) Ey Fundamental Emax Ex Emin Ez If we look at the specified IEC method with 3 orthogonal measurements, at the fundamental frequency for a phone. We see that in the horizontal plane, only if E max aligns with Ez, the direction of maximum sensitivity of the GTEM towards the APEX, will we measure the maximum. As the positioning in any plane is arbitrary, Emin could be in the direction of maximum sensitivity instead giving a result which is too low The other axes do not couple significantly to the GTEM, and so make little difference to the result.
22 3 Axis Method (IEC ) Ey 3 x frequency Ex Ez With the 3rd harmonic, it is also possible to measure a maximum lobe which is not in the horizontal plane when the EUT is oriented with the horizontal plane in the Ex or Ey planes instead of Ez. This can give a result which is too high. As with OATS, the standard IEC makes assumptions about lack of EUT directivity which is not satisfactory Defies common sense to measure in more than one direction at once.
23 4 Faces Method (V) phone turntable Or Full Rotation For this work, I then tried measuring 4 faces for vertical polarization and plotting the maximum for all frequencies. Basically, this method can be considered to be similar to the anechoic room method with no distance between the EUT and the antenna, except that the turntable was rotated by 90degrees to give horizontal polarization instead of the GTEM. A full rotation was not possible because of the absence of an an automated turntable -the manipulator was just moved by hand. Such a method is dependent on the radiation not varying significantly over a 90degree arc.
24 EXPERIMENTAL RESULTS USING A MOBILE PHONE Now to my actual results.
25 (a) OATS (b) FAR (c) GTEM 5305 (d) GTEM 5305 (d) GTEM 5311 AMPS Ch799 S/N 059 V&H max Vert Horiz max min Vert Horiz Vert Horiz TX fund 849MHz ± ± TX 2nd harm 1.7GHz ±3-39 ± TX 3rd harm 2.55GHz ±5-31 ± TX 4th harm 3.4GHz ±2-42 ±3-41 < < TX 5th harm 4.25GHz ±4-33 ± TX 6th harm 5.1GHz ±4-23 ± TX 7th harm 5.95GHz ±5-22 ± RX Band 878MHz RX Band 887MHz FAR: This method is closest to the universal specification. OATS Surprisingly, the OATS results are fairly comparable with the other methods. This is partly due to the lack of ground reflections at higher frequencies, and no antenna tilting was employed. It is noticeable that the transmitter power wasn't approximately doubled as expected, and the large 3rd harmonic angled lobe was not measured. Hence, the results were comparable because the full OATS method had not or could not be followed. GTEM 3 axis algorithm, for the transmitter, we get an arbitrary range of results, even getting 17dBm instead of the 25dBm of the other methods which is not acceptable. GTEM 4faces, although not fill rotation, reasonably comparable. With more work, I hope to improve the agreement particularly at the higher frequencies.
26 CONCLUSIONS Both OATS and GTEM 3 Axis Correlation Algorithm Methods have directional ambiguities. For consistent results, the directions to be measured must be specified. The first obvious point is that the OATS and GTEM 3 axis and other methods are unsuitable for EUTs with directional characteristics, and the users of these methods should prove that their EUT is not directional before accepting any results. It is particularly disturbing when methods which give arbitrary or inconsistent are mandatory. Secondly, what we have managed to do is to specify different directions for OATS, for Antioch Room, and for GTEMs. I find it almost embarrassing to stand here and say something so obvious, but is important to measure the specified and consistent directions.
27 FURTHER WORK Effect of EUT size Effect of leads Lower frequency and Higher frequency limitations One of the things I hope this paper will do is to stimulate people to look at the consequences of this work, and some examples are shown on the slide. The first 2 depend on antenna beamwidth. How far down in frequency can you go before worrying about near field effects / ground reflections? Improved tilted and electromagnetically transparent turntables for GTEMs would be useful. Images in GTEMs Field direction in GTEM off centre line is not towards the apex.
28 SPECIFICATIONS Remove ambiguities - cannot assume no directionality Reciprocity between Emissions and Immunity A final shot if I may: 1. Specifications need some overhaul. In particular, the OATS method has problems below 1GHz. 2. Relationship between an EUT and its environment must be reciprocal. Going from immunity to emissions just just involve removing an amplifier and connecting a spectrum analyzer. This applies to: 4 face FARs for immunity vs full rotation OATS for emissions. RF Common Mode CDNs vs Conducted Emissions LISNs etc.
Antenna Properties and their impact on Wireless System Performance Dr. Steven R. Best Cushcraft Corporation 48 Perimeter Road Manchester, NH 03013 Phone (603) 627-7877 FAX: (603) 627-1764 Email: firstname.lastname@example.org
EMC STANDARDS The EMC standards that a particular electronic product must meet depend on the product application (commercial or military) and the country in which the product is to be used. These EMC regulatory
Evaluation of Anechoic Chambers For EMC Measurements Using A CNE Overview Anechoic chambers are used for EMC testing primarily for radiated emissions (RE) and radiated immunity (RI) in the frequency range
Antenna Trainer EAN Technical Teaching Equipment Products Products range Units 3.-Communications INTRODUCTION Antennas are the main element of aerial communications. They are the transition between a transmission
Avaya WLAN 9100 External Antennas for use with the WAO-9122 Access Point Overview To optimize the overall performance of a WLAN in an outdoor deployment it is important to understand how to maximize coverage
Antennas for EMC Testing Zhong Chen ETS-Lindgren 1301 Arrow Point Drive Cedar Park, TX 78613 Zhong.Chen@ets-lindgren.com Outline EMC Terms and Definitions Typical EMC Antennas Calibration of EMC Antennas
Predicting radiated emissions from cables in the RE02/RE102/DO- 160/SAE J113-41 test set up, using measured current in NEC and simple TX equations. D. A. Weston RE02Tx.rep 14-6-2004 NARTE Certified EMC
PreCompliance Test Method for Radiated Emissions of Automotive Components Using Scattering Parameter Transfer Functions D. Schneider 1*, S. Tenbohlen 1, W. Köhler 1 1 Institute of Power Transmission and
AN1200.04 Application Note: FCC Regulations for ISM Band Devices: Copyright Semtech 2006 1 of 15 www.semtech.com 1 Table of Contents 1 Table of Contents...2 1.1 Index of Figures...2 1.2 Index of Tables...2
Application Note: Spread Spectrum Oscillators Reduce EMI for High Speed Digital Systems Introduction to Electro-magnetic Interference Design engineers seek to minimize harmful interference between components,
Omni Antenna vs. Directional Antenna Document ID: 82068 Contents Introduction Prerequisites Requirements Components Used Conventions Basic Definitions and Antenna Concepts Indoor Effects Omni Antenna Pros
Antenna Measurement 1 Antenna Ranges An antenna range is a facility where antenna radiation characteristics are measured. An antenna range includes the following typical components: 1. A substantial space
SUSCEPTIBILITY MODELING ANALYSIS of PARASITICS IN UNSHIELDED PRODUCTS with LONG WIRES ATTACHED Bruce Archambeault SETH Corporation Johnstown, PA H. Stephen Berger Siemens ROLM Communications Inc. Austin,
Antenna Selection for Optimum Wireless LAN Performance Dr. Steven R. Best Cushcraft Corporation 48 Perimeter Road Manchester, NH 03108 (603) 627-7877 PURPOSE Overview of antenna properties and performance
Electromagnetic Compatibility (EMC) IEEE Definition Origin, control, and measurement of electromagnetic effects on electronic and biologic systems. IEEE EMC Society Areas of Interest EMC Standards: Standards
Time and Frequency Domain Analysis for Right Angle Corners on Printed Circuit Board Traces Mark I. Montrose Montrose Compliance Services 2353 Mission Glen Dr. Santa Clara, CA 95051-1214 Abstract: For years,
Rec. ITU-R F.1336 1 RECOMMENDATION ITU-R F.1336* Rec. ITU-R F.1336 REFERENCE RADIATION PATTERNS OF OMNIDIRECTIONAL AND OTHER ANTENNAS IN POINT-TO-MULTIPOINT SYSTEMS FOR USE IN SHARING STUDIES (Question
Report No.: AGC01039140401FE09 Page 1 of 54 FCC Part 97 Test Report Report No.: AGC01039140401FE09 FCC ID : PODTH-9800 PRODUCT DESIGNATION : Mobile radio BRAND NAME : TYT MODEL NAME : TH-9800,TH-9800D,TH-7800,TH-7900
Design and Certification of ASH Radio Systems for Japan RFM s second-generation ASH radio hybrids are being used in a wide variety of applications in Japan, operating under the Japanese BIJAKU radio regulations.
Draft EN 301 783-1 V1.2.1 (2009-07) European Standard (Telecommunications series) Electromagnetic compatibility and Radio spectrum Matters (ERM); Land Mobile Service; Commercially available amateur radio
TECHNOTE No. 1 Joe Carr's Radio Tech-Notes Directional or Omnidirectional Antenna? Joseph J. Carr Universal Radio Research 6830 Americana Parkway Reynoldsburg, Ohio 43068 1 Directional or Omnidirectional
Coaxial Cable Feeder Influence on Yagi Antenna Dragoslav Dobričić, YU1AW email@example.com Introduction o far, in several previous articles [1, 2, 3], we investigated how boom radius and its S distance
九 州 工 業 大 学 学 術 機 関 リポジトリ Title FDTD analysis of site free space VS disturbance measurement above 1 GHz Author(s) Kuwabara N; Midori M; Kawabata M Issue Date 9-8 URL http://hdl.handle.net/8/368 9 IEEE.
ANTENNA An antenna is a device to transmit and/or receive electromagnetic waves. Electromagnetic waves are often referred to as radio waves. Most antennas are resonant devices, which operate efficiently
Test Report Prepared for: Technologic Systems, Inc. Model: TS-8820 Description: Single Board Computer with Analog and Digital Inputs and Outputs, Relays, and RS-232/485 Ports To IEC 61000-6-1 (2005-03)
CABLES CABLES radiating Link Budget 3. 1. LINK BUDGET The basic elements to calculate a link budget can be illustrated by considering the example shown in Figure 4. It involves a GSM 900 radio coverage
Electromagnetic radiation exposure: assessment against ACA mandated limits General radio services (operating above 0 MHz) (Edition May 0) Disclaimer Unless otherwise specified, the information contained
TESTING CERT #803.01, 803.02, 803.05, 803.06 MYRICOM, INC. TEST REPORT FOR THE SERVER RACK, 10G-PCIE2-8B2-2S & 10G-PCIE2-8B2-2QP FCC PART 15 SUBPART B SECTIONS 15.107 AND 15.109 CLASS A TESTING DATE OF
Rec. ITU-R F.699-5 1 RECOMMENATION ITU-R F.699-5 * REFERENCE RAIATION PATTERNS FOR LINE-OF-SIGHT RAIO-RELAY SYSTEM ANTENNAS FOR USE IN COORINATION STUIES AN INTERFERENCE ASSESSMENT IN THE FREQUENCY RANGE
APPLICATION NOTE - 016 Testing RFI Line Filters Frequency Response Analysis Testing RFI line filters Radio frequency interference (RFI) is unwanted electromagnetic noise within a radio communications frequency
HUMAN EXPOSURE TO EMR: ASSESSMENT OF AMATEUR RADIO STATIONS FOR COMPLIANCE WITH ACA REQUIREMENTS May 2005 Version 2.0 PO Box 78 BELCONNEN ACT 2616 Telephone (02) 6219 5555 Facsimile (02) 6219 5353 www.aca.gov.au
EN 301 489-17 v1.2.1 TEST REPORT FOR 802.11ag/Draft 802.11n WLAN PCI-E Mini Card MODEL NUMBER: BCM94322MC REPORT NUMBER: 07U11529-5 ISSUE DATE: JANUARY 29, 2008 Prepared for BROADCOM CORPORATION 190 MATHILDA
14713221 001 Seite 2 von 37 Page 2 of 37 Model List: No Model Rated Voltage(V) 1. EG-LED0827-01 Rated Power(W) 8W 2. EG-LED0840-01 8W 3. EG-LED1027-01 10W 4. EG-LED1040-01 AC 100-240V, 10W 5. EG-LED1027-02
P a g e 1 EE 172 Extra Credit Project 2.4 GHz Yagi-Uda Antenna Created by Mario Delgadillo Maringan Pardamean Panggabean P a g e 2 Abstract This report will define antenna theory and design as it relates
RALI : LM 8 DATE OF EFFECT : 23/09/2015 Radiocommunications Assignment and Licensing Instruction FREQUENCY ASSIGNMENT REQUIREMENTS FOR THE LAND MOBILE SERVICE AUSTRALIAN COMMUNICATIONS AND MEDIA AUTHORITY
Radio Physics for Wireless Devices and Networking By Ron Vigneri The Radio Physics of WiFi The standard for wireless LANs (WLANs) was completed in 1997 with the release of the IEEE 802.11 specification
Report ITU-R M.44 (11/011) Isolation between antennas of IMT base stations in the land mobile service M Series Mobile, radiodetermination, amateur and related satellite services ii Rep. ITU-R M.44 Foreword
Test & Measurement Data Sheet 02.00 R&S R-Line Compact Test Chamber Precise measurements of radiated emissions of wireless terminals 800 MHz to 18 GHz R&S R-Line Compact Test Chamber At a glance The R-Line
UNIVERSITY OF TORONTO FACULTY OF APPLIED SCIENCE AND ENGINEERING The Edward S. Rogers Sr. Department of Electrical and Computer Engineering ECE422H1S: RADIO AND MICROWAVE WIRELESS SYSTEMS EXPERIMENT 3:
Page: 1 of 35 R&TTE (99/5/EC ) Directive EN 300 440-2 V 1.4.1: 2010 TEST REPORT FOR Product Name: Brand Name: Model No.: Model Difference: Report No.: TomTom Cardio GPS Watch TomTom 8RA0 N/A ER/2014/10096
Troubleshooting Range: Understanding the Cell Size Jerome Henry, Senior Member, IEEE, TME @ Cisco Systems How Large is My Cell? Long ago, vendors used to be able to provide this Most don t provide cell
Automotive RF immunity test set-up analysis Why test results can t compare Mart Coenen, EMCMCC, Eindhoven, the Netherlands Hugo Pues, Melexis Technologies NV, Tessenderlo, Belgium Thierry Bousquet, Continental,
Electromagnetic radiation exposure: assessment against ACA mandated limits Paging services (Edition May 2002) Disclaimer Unless otherwise specified, the information contained in these guidelines is intended
TX SAW MID 5V TRANSMITTER SAW Transmitter module with external antenna, for utilisations with ON-OFF modulation of a RF carrier with digital data. Buffer stage for enhanced power and low harmonics on output
Uncertainties of immunity measurements DTI-NMSPU project R2.2b1 Details of chamber performance (radiated immunity) Details of chamber performance (radiated immunity) Details of chamber performance This
Antennas A good antenna works A bad antenna is a waste of time & money Antenna systems can be very inexpensive and simple They can also be very, very expensive Antenna Considerations The space available
Cellular Wireless Antennas A Technical Brief GarrettCom Inc., November 2010 Overview The Cellular Wireless Antenna Technical brief is provided to assist with the design and deployment of the DX940 Cellular
1 Embedded FM/TV Antenna System Final Report Prepared for By January 21, 2011 2 Table of Contents 1 Introduction... 5 2 Technical Specification... 6 3 Prototype Antenna... 7 4 FASTROAD Active module fabrication...
RX-AM4SF Receiver The super-heterodyne receiver RX-AM4SF can provide a RSSI output indicating the amplitude of the received signal: this output can be used to create a field-strength meter capable to indicate
AUTOMATED THREE-ANTENNA POLARIZATION MEASUREMENTS USING DIGITAL SIGNAL PROCESSING John R. Jones and Doren W. Hess Abstract In this paper we present a three-antenna measurement procedure which yields the
EMC Basics Speaker : lain Lafuente firstname.lastname@example.org WHT IS EMC? 2 CE Marking With the formation of the single European market, standardization was required to remove technical barriers to trade.
Antenna Basics Introduction Antennas are a very important component of communication systems. By definition, an antenna is a device used to transform an RF signal, traveling on a conductor, into an electromagnetic
Rec. ITU-R F.1113 1 RECOMMENDATION ITU-R F.1113 RADIO SYSTEMS EMPLOYING METEOR-BURST PROPAGATION (Question ITU-R 157/9) (1994) Rec. ITU-R F.1113 The ITU Radiocommunication Assembly, considering a) that
Selecting Receiving Antennas for Radio Tracking Larry B Kuechle, Advanced Telemetry Systems, Inc. Isanti, Minnesota 55040 email@example.com The receiving antenna is an integral part of any radio location
BACKGROUND INFORMATION Infrared Spectroscopy Before introducing the subject of IR spectroscopy, we must first review some aspects of the electromagnetic spectrum. The electromagnetic spectrum is composed
HALF-WAVE PARABOLIC REFLECTOR ANTENNA OPTIMIZATION Parker Singletary, Carson Smith Advisor: Dr. Gregory J. Mazzaro Department of Electrical & Computer Engineering The Citadel, The Military College of South
Progress In Electromagnetics esearch C, Vol. 42, 191 203, 2013 CONDUCTED EMISSION MEASUEMENT OF A CELL PHONE POCESSO MODULE Fayu Wan *, Junxiang Ge, and Mengxiang Qu Nanjing University of Information Science
1. Introduction HB Series of microwave motion sensor module are X-Band Mono-static DRO Doppler transceiver front-end module. These modules are designed for movement detection, like intruder alarms, occupancy
Electronic Communications Committee (ECC) within the European Conference of Postal and Telecommunications Administrations (CEPT) COMPATIBILITY STUDY FOR UMTS OPERATING WITHIN THE GSM 900 AND GSM 1800 FREQUENCY
RF Basics - Part 1 This is the first article in the multi-part series on RF Basics. We start the series by reviewing some basic RF concepts: Decibels (db), Antenna Gain, Free-space RF Propagation, RF Attenuation,
538 Progress In Electromagnetics Research Symposium 2006, Cambridge, USA, March 26-29 A Study of RF Absorber for Anechoic Chambers Used in the Frequency Range for Power Line Communication System K. Shimada
Evaluating Cell Phone and Personal Communications Equipment and their EMC Effects on Automotive Audio and In-Cabin Modules Craig W. Fanning Elite Electronic Engineering, Inc. 1516 Centre Circle Downers
Application Note #0430 September 2012 Revised: Volts/m For EMC and Antenna Measurement Application Note Image courtesy of Wikipedia Commons This note describes the concept of electric field strength in
Radio Frequency Exposure EN 62311 January 2008 Assessment of electronic and electrical equipment related to human exposure restrictions for electromagnetic fields (0Hz 300GHz) (IEC 62311:2007, modified)
FCC Test Report FCC Part 22, 24 / RSS 132, 133 FOR: GSM Cellular Telephone with Bluetooth and WiFi Model #: A1203 Apple Inc. 1 Infinite Loop Mail Stop26A Cupertino, California 95014 U.S.A FCC ID: BCGA1203
SAR ASSOCIATED WITH THE USE OF HANDS-FREE MOBILE TELEPHONES S.J. Porter, M.H. Capstick, F. Faraci, I.D. Flintoft and A.C. Marvin Department of Electronics, University of York Heslington, York, YO10 5DD,
Sound Power Measurement A sound source will radiate different sound powers in different environments, especially at low frequencies when the wavelength is comparable to the size of the room 1. Fortunately
How to make a Quick Turn PCB that modern RF parts will actually fit on! By: Steve Hageman www.analoghome.com I like to use those low cost, no frills or Bare Bones  type of PCB for prototyping as they
RF Path Loss & Transmission Distance Calculations By Walter Debus Director of Engineering Axonn, LLC Technical Memorandum August 4, 2006 INTRODUCTION DOC# 8545-0003-01 For radio transmission systems that
Radiated Emission and Susceptibility Tzong-Lin Wu, Ph.D. EMC Lab Department of Electrical Engineering National Taiwan University Differential-Mode v.s. Common-mode Currents 1 Differential-Mode v.s. Common-mode
MANUAL FOR RX700 LR and NR 2013, November 11 Revision/ updates Date, updates, and person Revision 1.2 03-12-2013, By Patrick M Affected pages, ETC ALL Content Revision/ updates... 1 Preface... 2 Technical
FCC PART 15 SUBPART C EMI MEASUREMENT AND TEST REPORT For AeroComm, Inc. 10981 Eicher Drive Lenexa, KS 66219 2003-05-22 This Report Concerns: Original Report Equipment Type: 900MHz Transceiver Test Engineer:
An Algorithm for Automatic Base Station Placement in Cellular Network Deployment István Törős and Péter Fazekas High Speed Networks Laboratory Dept. of Telecommunications, Budapest University of Technology
Krešimir Malarić EMI Protection for Communication Systems CD-ROM INCLUDED Contents Preface xiii CHAPTER 1 Communications Systems 1 1.1 Components of Communications Systems 1 1.2 Transmitter Systems 2 1.2.1
November 8, 2007 Charlie Hughes Using All Pass Filters to Improve Directivity Response All pass filters are perhaps one of the most overlooked and poorly understood tools for the general audio practitioner.
AN INTRODUCTION TO TELEMETRY PART 1: TELEMETRY BASICS Telemetry is defined as the sensing and measuring of information at some remote location and then transmitting that information to a central or host
Your consent to our cookies if you continue to use this website.