1 I/Q Modulation Generator Supplement The following description relates to the Operating Manuals, version 03 of R&S AFQ100A, and version 01 of R&S AFQ100B. It encloses the following topics: LXI features, see Chapter 1, "LXI (R&S AFQ100B only)", on page 2. Note to the triggering when setting up several connected R&S AFQs, see Chapter 3, "Setting Up Several Connected R&S AFQs", on page 19 instead of see Chapter 2.1, "Remote Access via on page 12 Standard password for remote desktop connection, see "Remote Access Password" on page 12 TRIGGER KEY functions, see Chapter 2.3, "Activating Remote Desktop with the Trigger Key", on page 18 Using R&S VISA for remote control over LAN, see Chapter 2.2, "Remote Control over LAN using VXI-11 Protocol", on page 13 (;ÙÓÜ2) Test & Measurement Supplement
2 Contents LXI Configuration Contents 1 LXI (R&S AFQ100B only) Remote Control Setting Up Several Connected R&S AFQs LXI (R&S AFQ100B only) Refers to "Configuring the R&S AFQ for Network Operation" in the Operating Manual. 1.1 LXI Configuration LAN extensions for Instrumentation (LXI) is an instrumentation plattform for measuring instruments and test systems that is based on standard Ethernet technology. LXI is intended to be the LAN-based successor to GPIB, combining the advantages of Ethernet with the simplicity and familiarity of GPIB. Like GPIB, LXI determines and standardizes the way the instrument behaves in a LAN. The LXI implementation by the Rohde & Schwarz Signal Generators allows you to change certain LAN settings, to reset the LAN connection as well as to easy identify the instrument. For information about the LXI standard refer to the LXI website at See also "News from Rohde & Schwarz, article 2006/II - 190". The R&S AFQ complies with LXI and provides an integrated "LXI Status" dialog for LXI status indication and reset of the LAN configuration (LAN Configuration Initialize, LCI). To access the LXI Status dialog, press the SETUP key and select "Remote > LXI Status". For further information, see Chapter 1.2, "LXI Status", on page 10. Firmware update After a firmware update, shut-down and re-start the instrument in order to enable the full LXI functionality. Default state of the network settings According to the LXI standard, an LCI must set the following parameters to a default state. 2
3 LXI (R&S AFQ100B only) LXI Configuration Parameter TCP/IP Mode Dynamic DNS ICMP Ping Password for LAN configuration Value DHCP + Auto IP Address Enabled Enabled LxiWebIfc The LCI for the R&S AFQ also resets the following parameters: Parameter Hostname Description Negotiation VXI-11 Discovery Value <Instrument-specific host name> Vector Signal Generator Auto Detect Enabled The LAN settings are configured using the instrument's LXI Browser Settings The instrument's LXI browser interface works correctly with all W3C compliant browsers. Typing the instrument's host name (LAN device name) or IP address in the address field of the browser on your PC opens the "Instrument Home Page" (welcome page). To access the instrument via the web browser: Type in the instrument's host name or IP address in the address field of the browser on your PC, for example Tip: The specific IP address (host address / instrument name) is used by the controller programs to identify and control the instrument. It is automatically assigned by the DHCP server the first time the instrument is registered on the network. The host name is usually composed of the instrument name and the serial number: <instrument name><serial number>, e.g. RSAFQ100B Note: Do not add the missing zeros in the IP address, while opening the instrument home page. The instrument home page (welcome page) opens. 3
4 LXI (R&S AFQ100B only) LXI Configuration The "Instrument Home Page" displays the device information required by the LXI standard including the VISA resource string in read-only format. The navigation pane of the browser interface contains the following elements: "LXI" "Home" opens the instrument home page. The home page displays the device information required by the LXI standard, including the VISA resource string in read-only format. "Device Indicator" activates or deactivates the LXI status indication. When activated, the LXI LEDs flash, both in the browser dialog and in the LXI dialog of the connected instrument. A green LXI status symbol indicates that a LAN connection has been established; a red symbol indicates that no LAN cable is connected. "Lan Configuration" allows you to configure LAN parameters and to initiate a ping, see Chapter , "Ping Parameter", on page 7. "Status" displays information about the LXI status of the instrument. "Utilities" provides access to the LXI event log functionality required by the LXI standard. "Instrument Control" "Web Control" provides remote access to the instrument, see Chapter , "Web Control", on page 8. "Diagnostics" "SCPI Remote Trace" records messages exchanged via the remote control interface, see Chapter , "SCPI Remote Trace", on page 9. "Help" "Glossary" explains terms related to the LXI standard. opens the Rohde & Schwarz home page. 4
5 LXI (R&S AFQ100B only) LXI Configuration Press the "Inactive (press to toggle)" button to activate the connection. A green flashing status LED indicates the active connection. If the LAN connection fails, the LED turns red. The status of the LAN connection is also indicated by the "LAN Status indicator" in the "LXI Status" dialog. The "Device Indicator" setting is not password-protected. For further information, see section Chapter 1.2, "LXI Status", on page 10. The navigation pane of the browser interface contains the following control elements: "LAN Configuration" opens the LAN Configuration page. "LXI Glossary" opens a document with a glossary of terms related to the LXI standard. "Status" displays information about the LXI status of the instrument LAN Configuration The "LAN Configuration" web page provides access to the parameters required for identifying the R&S AFQ in the network, and allows modification. Password protection Changing the LAN configuration is password-protected. The default password is Lxi- WebIfc (notice upper and lower case characters). This password cannot be changed in the current firmware version. It comprises the following navigation entries. IP Configuration... 5 Advanced LAN Configuration...6 Ping Parameter... 7 Web Control... 8 SCPI Remote Trace IP Configuration The "IP Configuration" page displays all mandatory LAN parameters. 5
6 LXI (R&S AFQ100B only) LXI Configuration The "IP Address Mode" selects a configuration mode for the IP address of the R&S AFQ. With static configuration, the entered IP address, subnet mask, and default gateway are used. With dynamic configuration, DHCP or dynamic link local addressing (automatic IP) are used to obtain the instrument IP address Advanced LAN Configuration The "Advanced LAN Configuration" menu provides LAN settings that are not declared mandatory by the LXI standard. 6
7 LXI (R&S AFQ100B only) LXI Configuration The following advanced parameters are available: "mdns and DNS-SD": The additional protocols "multicast DNS" and "DNS service discovery" are used for device communication in zero configuration networks, working without DNS and DHCP. "ICMP Ping enabled": Must be enabled to use the ping utility. If you disable this setting, the instrument does not answer ping requests. The setting does not affect the LXI ping client. You can ping other hosts from the instrument, even if the setting is disabled. "VXI-11 Discovery": Must be enabled to detect the instrument in the LAN. If you disable this setting, the instrument cannot be detected by the VXI-11 discovery protocol mechanism. The setting does not affect other detection mechanisms. Setting up a VXI-11 connection via the IP address or the host name is independent of this setting Ping Parameter The "Ping Client" page provides the ping utility to verify the connection between the LXI-compliant instrument and another device. The ping is initiated from the instrument. Using the ICMP echo request and echo reply packets, the function checks whether the communication with a device via LAN works. Ping is useful for the diagnosis of IP network or router failures. The ping utility is not password-protected. To initiate a ping from the instrument to the device: 1. Enable "ICMP Ping" on the "Advanced LAN Configuration" page (enabled after an LCI). 2. Select the "Ping Client" page. 3. In the "Destination Address" field, enter the IP address of the device to be pinged (without the ping command and without any further parameters), e.g Select "Submit". 7
8 LXI (R&S AFQ100B only) LXI Configuration Web Control The web control functionality provides remote operation via VNC using a Web browser (with HTML5). This mode does not require additional installation or activation. The VNC protocol allows simultaneous operation from several remote devices. The instrument remains locally operable. The GUI of the R&S AFQ is visible. To perform the settings, you can operate the instrument as with the manual control. The instrument controls are available via the front panel simulation. File upload and download between the instrument and the remote PC is also available. Starting a Remote Control via the LXI web browser This section assumes that the instrument and the controller PC are connected in the LAN. 1. Start a web browser that supports html5 (W3C compliant). 2. Enter the IP address of the R&S AFQ in the browser's address bar. The R&S AFQ's welcome page is displayed. 3. In the navigation pane, select "Instrument Control" > "Web Control". Remote access to the instrument requires the password. The default password is instrument. 4. Enter the password and confirm with the ENTER key. After the connection is established, the current screen of the R&S AFQ is displayed in the browser window. 8
9 LXI (R&S AFQ100B only) LXI Configuration 5. Use the mouse cursor and keyboard to access the functionality of the instrument as you would directly perform the settings on the instruments front panel SCPI Remote Trace The remote trace functionality allows you to trace input and output strings at the remote control interface of the R&S AFQ. A recorded trace (message log) can be evaluated directly in the dialog. Use the highlighting and navigation functions provided in the lower toolbar to locate error messages and messages containing arbitrary search strings. You can also export the message log to a *.csv file and evaluate the file using a suitable program. To trace and display messages: 1. In the navigation pane, select "Diagnostics" > "SCPI Remote Trace". 2. In the toolbar bar of the "SCPI Remote Trace" page, select "live mode" > "on" and "logging" > "on". "live mode > on" displays all commands and responses, and "logging > on" also traces messages. 3. If you now control the instument with SCPI commands, using an appropriate tool, the SCPI remote trace records the information sent and received. The function records all sent commands, received responses and messages, and stores them in an internal database. If "live mode" is disabled, you can display the recent traces upon request, using the "refresh" button. You can also store the log in a file. Note: The diagnostics functionality will be extended in later releases, e.g. to download or upload SCPI command files from / to the instrument. 9
10 LXI (R&S AFQ100B only) LXI Status Toolbars The toolbar at the top of the dialog provides basic settings and functions. "Live mode" / "logging": If logging is switched on, messages are traced. They are stored in an internal database and can be displayed upon request, using the refresh button (live mode off) or they can be displayed automatically (live mode on). "Filter": applies a filter to columns and/or rows when working (live mode off) "Refresh": reads the message log from the internal database and displays it "Download": stores the SCPI trace log to a *.csv file "Clear": deletes all message log entries in the database and on the screen "Details": displays details of the selected message, for example an SCPI command in hex format (also possible by double\-clicking a message) Columns The following columns are available if no column filter is applied: "Rec": record number of the message within the message log I: number of the subinstrument "MT": indicates the type of the message. Possible values and related message contents are: > = incoming command < = outgoing response to a query E = error message, highlighted by red color T = execution time, i.e. time required by the instrument to process the command internally "message": indicates the type of the message. Possible values and related message contents are: > = incoming command < = outgoing response to a query E = error message, denoted in red T = execution time, i.e. time required by the instrument to process the command internally 1.2 LXI Status Refers to "General Instrument Settings > Setup Key" in the Operating Manual. The "LXI - LAN extensions for Instruments - Status..." dialog displays the settings and status of the LAN and allows to reset the LAN connection. For more information about LXI, seechapter 1.1, "LXI Configuration", on page 2. 10
11 Remote Control LAN Status The LED indicates the LXI status. The LAN Status is also indicated with the LED "LAN Status" on the front panel of the instrument. "green" normal operation "green (flashing)" device identification "red" LAN fault LAN Reset Initiates the network configuration reset mechanism for the instrument and resets the hostname, MAC address, and IP address. Alternatively, the LAN RESET button on the front panel of the instrument can be used. To initiate the network configuration reset, press the button longer than 4 sec and wait for the signal tone. See "Default state of the network settings" on page 2 for information on the reset parameters. To open the "Instrument Home Page" (welcome page), type the instrument's computer name (host name) or IP address in the address field of the browser on your PC, e.g. Note: A LAN reset may require an instrument reboot. 2 Remote Control Refers to "Putting into Operation > Manual Remote Control via an External Controller" in the Operating Manual. 11
12 Remote Control Remote Access via As an alternative to remote control using programmed commands, the R&S AFQ can also be manually controlled from a remote location via remote access, or remote desktop. These operating modes are fundamentally different, even though they are often seen as interchangeable: Remote Desktop In this mode, a Windows desktop connection from the remote computer (external controller) and the R&S AFQ is established and you can set your application as with manual operation. Remote Access In this mode, the instrument is accessed from an external controller via a LAN connection, using a standard Web browser and the common cross-plattform technology VNC (Virtual Network Computing). While the instrument is in operation, the user interface of the instrument is displayed on the remote computer, and VNC provides access to all of the applications files and network resources of the instrument. The remote computer requires the VNC client, either as a stand-alone program or as a plug-in for a browser. Remote Control In this mode, all settings are configured from a remote PC using remote control commands. The GUI of the instrument is not displayed. Remote control commands can be executed either individually, or you can also create programs for repeating applications, that are performed faster than with manual operation and are always reproducible. 2.1 Remote Access via Refers to "Configuration for Manual Remote Control" in the Operating Manual. Instead of the remote access via the R&S AFQ allows you to remotely control the instrument via the program an open source desktop software application. It uses extended VNC RFB protocol, and additionally features e.g. listening mode or password protection. The software and additional information is available at the website. For the setting up and operating, you can proceed similarly as described to in the manual. Remote Access Password The default password for remotely accessing the R&S AFQ is instrument. 1. To change the password, click the VNC icon in the windows task bar and select "Properties". 12
13 Remote Control Remote Control over LAN using VXI-11 Protocol In the properties dialog, you can set your own password. 2. Enter the new password and confirm with "OK" 2.2 Remote Control over LAN using VXI-11 Protocol Refers to "Remote Control Basics" > "Switchover to Remote Control " > "Remote Control over LAN Interface". "Remote Control over LAN using VXI-11 Protocol"substitutes the description of "Setting up Control of the R&S AFQ in the 'Measurement & Automation Control' Program" in the user manual. Rohde & Schwarz provides the standardized I/O software library R&S VISA for communication via TCP/IP (LAN: HiSlip, VXI-11 and raw socket) or USB (USBTMC) interfaces. R&S VISA is available for download at the Rohde & Schwarz website In this example, the R&S VISA software library is used to set up a LAN remote control link and remotely control the R&S AFQ. R&S VISA is running on a controller PC with Windows operating system. When the connection is set up you can send commands to the instrument, and receive the responses. The remote control connection requires a VISA installation but no additional hardware on the controller PC. The LAN I/O channel is selected at initialization time using the VISA resource string (also referred to as "address string"). A VISA alias (short name) is used to replace the complete resource string. The host address is either the R&S AFQ's hostname or IP address. See "Remote Control Basics" > "Switchover to Remote Control " > "Remote Control via LAN Interface" in the user manual of the R&S AFQ. In this example, it is assumed that: A LAN remote control link between the controller and the R&S AFQ is already set up. The R&S VISA program is installed on the remote PC, see " > RS VISA Release Notes". Configuring the controller To remote control the R&S AFQ, we use the R&S VISA Tester application. 13
14 Remote Control Remote Control over LAN using VXI-11 Protocol The instrument is preconfigured for networks using DHCP (dynamic host configuration protocol). If this configuration is used, enter the computer name in the position of the IP address. To enable the external controller to communicate with the R&S AFQ via TCP/IP protocol, set up a remote control link as follows: 1. Make sure that the controller and the instrument are connected in the network (network cable) and switched on. 2. On the controller, start "R&S VISA > Tester 32bit" or "R&S VISA > Tester 64bit", respectively. 3. In the menu bar, select "Choose VISA Implementation > Rohde & Schwarz Visa". 4. Select "Rohde & Schwarz Visa" and confirm with "OK". 5. In the menu bar, select "Find Resource" to search for the instrument in the LAN. 14
15 Remote Control Remote Control over LAN using VXI-11 Protocol 6. Select "VXI-11" and "Find Resources". R&S VISA scans the network for connected instruments and lists all detected instuments in the "Resource" list. Note: The search may take some time, particularly in large networks. 7. Select the required instrument and confirm with "Select". The "Find and select resource" dialog closes and R&S VISA indicates the instruments IP address in the "Resource" field of the main application window. 8. As an alternative to the IP address, you can assign an alias name to the R&S AFQ: a) In the menu bar, select "RsVisaConfig". b) In the toolbar, select "+" to access the "VISA Resource String Composer". 15
16 Remote Control Remote Control over LAN using VXI-11 Protocol c) Fill in the "Alias" name, the "VISA Resource String" and the "Device IP Address or host name" as shown in the figure, and confirm with "OK". The "Alias" name is assigned to the instrument. d) Close the dialog. The R&S AFQ is now registered in the program and can be addressed via the resource string or alias name. 9. In the main window, select "Connect". R&S VISA establishes the connection to the R&S AFQ. Now you can send settings to configure the instrument and receive its responses. Note: If the connection cannot be set up, R&S VISA displays an error in the log view. Check whether the controller and the instrument are connected to the network (network cable) and switched on. Correct spelling of the IP address or the computer name can also be checked. For further error location, inform the network administrator. In large networks, specification of additional addresses may be required for link setup, e.g. gateway and subnet mask, which are known to the network administrator. For further information on the functions to read and write to an open session, as well as the utility applications the software provides, see the R&S VISA User Manual. 16
17 Remote Control Remote Control over LAN using VXI-11 Protocol Starting a remote control over LAN (using VXI-11) To set the instrument to remote control, you can use the addressed command >R, or send any command from the controller. 1. Start the R&S VISA Tester and establish the connection to the R&S AFQ, see "Configuring the controller" on page In the R&S VISA "Basics" tab, enter a SCPI command, e.g. "*IDN?" and confirm with "Query". The instrument is switched to remote control when it receives a command from the controller. Tip: Switching from manual operation to remote control and vice versa does not affect the other instrument settings. 3. Select "Read" to obtain the instrument response. Tip: If the "Show Log" checkbox is checked R&S VISA displays each VISA function call in the log-view on the left. If you check the "Write Log" checkbox the log-view entry is written to the log file as well. You can operate the log-view in two modes: the "Live Mode" shows only the most recent messages whereas the "View Mode" allows you to scroll the history. 4. To set, e.g. the frequency, enter SOURCE:STAT 1 and select "Write". To check the performed setting, select SOURCE:STAT?"Read". The instrument response is 1, i.e. the ARB generator is switched on. While remote control is active, the REMOTE status LED on the front panel indicates that the instrument is in remote control mode. The operation via the front panel or via mouse and keyboard are locked, allowing a remote control program to be performed without interruption. 17
18 Remote Control Activating Remote Desktop with the Trigger Key 5. Although operation via front panel, mouse and keyboard is disabled, the dialog boxes can still be opened, for example to verify settings. The buttons and setting fields are grayed out and cannot be activated. Additionally, you can disable the access to the dialogs with the command SYST:KLOC ON to protect the instrument against unauthorized readings. 6. To prevent unintentional return to manual operation, use the command &LLO (see section "VXI-11 Interface Messages" in the operating manual). The instrument switches to "REM-LLO" state. The automatic transition from local state to remote state by a subsequent remote command, and the command *GTL are disabled. To return to manual mode is only possible via remote control. 7. To return to manual operation send the interface command >L. Note: Before returning to manual control, command processing must be completed. Otherwise, the instrument switches back to remote control immediately. Use the >R to disable the "REM-LLO" state. 2.3 Activating Remote Desktop with the Trigger Key Refers to "Trigger Key" in the Operating Manual. Apart from triggering and LAN reset, the trigger key also enables you to activate remote desktop operation: Pressing short triggers manually Pressing long executes an LXI (LAN) reset Pressing the trigger key during startup (while the LEDs are flashing) enables you to modify the firewall settings, and temporarily enables access to the instrument via remote desktop. The start screen of the instrument software displays a dialog which allows you to enable remote desktop operation either permanently or for the current session only. Refer also to chapter Putting into Operation > Booting the R&S AFQ in the Operating Manual. 18
19 Setting Up Several Connected R&S AFQs 3 Setting Up Several Connected R&S AFQs Refers to "Setting Up Several Connected R&S AFQs for Precise Simultaneous Signal Output" in the Operating Manual. When setting up several connected instruments, the master R&S AFQ must be triggered externally. 19
I/Q Modulation Generator Supplement The following description relates to the Operating Manuals, version 03 of R&S AFQ100A, and version 01 of R&S AFQ100B. It encloses the following topics: LXI features,
I/Q Modulation Generator Supplement The following description relates to the Operating Manuals, version 03 of R&S AFQ100A, and version 01 of R&S AFQ100B. It encloses the following topics: LXI features,
PURPOSE Verify that communication is established for the following products programming option (488.2 compliant, SCPI only): DCS - M9C & DCS M130, DLM M9E & DLM-M9G & DLM M130, DHP - M9D, P series, SG,
Rohde & Schwarz Products: R&S FSV3, R&S FSV7, R&S FSV13, R&S FSV30 Remote Monitoring and Control of the R&S FSV with a Web Browser Application Note This application note describes remote operation or monitoring
KRAMER ELECTRONICS LTD. USER GUIDE Ethernet Configuration Guide (Lantronix) P/N: 2900-300321 Rev 6 Contents 1 Connecting to the Kramer Device via the Ethernet Port 1 1.1 Connecting the Ethernet Port Directly
Remote Monitoring and Control of the R&S RTO with a Web Browser Application Note Products: R&S RTO1012 R&S RTO1014 R&S RTO1022 R&S RTO1024 This application note describes remote operation or monitoring
VPN Hello and welcome. In the VPN section we will cover the steps for enabling the VPN feature on the Allworx server and how to set up a VPN connection to the Allworx System from your PC. Page 1 VPN The
Agilent Technologies USB/LAN/GPIB Interfaces Connectivity Guide Agilent Technologies Notices Agilent Technologies, Inc. 2003, 2004 No part of this manual may be reproduced in any form or by any means (including
AXIS 205 Quick Installation Guide Page 1 of 6 AXIS 205 Network Camera Quick Installation Guide Focus Ring Status Indicator (surrounding the focus ring) Serial number Network Indicator Control Button Power
Networking Guide Redwood Manager 3.0 August 2013 Table of Contents 1 Introduction... 3 1.1 IP Addresses... 3 1.1.1 Static vs. DHCP... 3 1.2 Required Ports... 4 2 Adding the Redwood Engine to the Network...
126.96.36.199 Networking for Homes and Small Businesses Student Packet Tracer Lab Manual This document is exclusive property of Cisco Systems, Inc. Permission is granted to print and copy this document for non-commercial
L force Controls Ä.GdWä LDCDP 11999.GdW Software Manual Industrial PC WindowsR CE Thin Client Operating system l Please read these instructions before you start working! Follow the enclosed safety instructions.
Chapter 10 Troubleshooting This chapter provides troubleshooting tips and information for your ProSafe Dual WAN Gigabit Firewall with SSL & IPsec VPN. After each problem description, instructions are provided
SNMP Web Management User s Manual For SNMP Web Card/Box Management Software for Off-Grid Inverter Version: 1.2 Table of Contents 1. Overview... 1 1.1 Introduction... 1 1.2 Features... 1 1.3 Overlook...
Please Install Your EASE Scan Tool DVD Before Installing Any Other Software That Came With Your Package. SECTION I - Installation ATTENTION: Do NOT connect an EASE Vehicle Interface Device to your Computer
ABB solar inverters User s manual ABB Remote monitoring portal List of related manuals Title ABB Remote monitoring portal User s manual NETA-01 Ethernet adapter module User s manual Code (English) 3AUA0000098904
IX System IP network-compatible intercom IX Support Tool Setting Manual Software version 188.8.131.52 or later Before configuring and using the system, read Setting Manual and Operation Manual (PDF) carefully.
Device Installer User Guide Part Number 900-325 Revision B 12/08 Table Of Contents 1. Overview... 1 2. Devices... 2 Choose the Network Adapter for Communication... 2 Search for All Devices on the Network...
the following parts are needed to test the unit: Arduino UNO R3 Arduino Wifi shield And reciever 5V adapter Connecting wifi module on shield: Make sure the wifi unit is connected the following way on the
Version 2.0 About This Guide 2 About This Guide This user guide is intended for Photovoltaic (PV) system owners, installers, technicians, maintainers, administrators, and integrators who are authorized
University of Jordan Faculty of Engineering & Technology Computer Engineering Department Computer Networks Laboratory 907528 Lab. 2 Network Devices & Packet Tracer Objectives 1. To become familiar with
Ethernet Radio Configuration Guide for Gateway, Endpoint, and Repeater Radio Units April 20, 2015 Customer Service 1-866-294-5847 Baseline Inc. www.baselinesystems.com Phone 208-323-1634 FAX 208-323-1834
About This Guide 3 About This Guide This user guide is intended for Photovoltaic (PV) system owners, installers, technicians, maintainers, administrators and integrators who are authorized to configure
PowerLogic ION Enterprise 5.6 Power Management Software User Guide April 2007 Notices This section describes the symbols used in this guide. Danger This alerts you to things that may cause serious injury
0215 IS SERIES IP Network Direct System QuikStart Programming Guide 1 4 GHI 2 ABC 5 JKL 3 DEF 6 MNO PRIVACY TRANSFER LIST SETTING 7 PQRS 8 TUV 9 WXYZ ZOOM 0 # WIDE Programming Videos ATTENTION: This is
Ross Video Limited DashBoard Server and User Rights Management User Manual DashBoard Server and User Rights Management User Manual Ross Part Number: 8351DR-004A-01 Release Date: March 22, 2011. Printed
Table of Contents Page 1: Motion Detection 1.0 Configuring Motion Detection for LX Apollo / LX Zeus DVRs 2 1.1 Motion Playback on LX Apollo / LX Zeus DVRs 3 1.2 Scheduling Motion and Continuous Recording
Amcrest 960H DVR Quick Start Guide Welcome Thank you for purchasing our Amcrest 960H DVR! This quick start guide will help you become familiar with our DVR in a very short time. Before installation and
SOHO 6 Wireless Installation Procedure Windows 95/98/ME with Internet Explorer 5.x & 6.0 Before You Begin Before you install the SOHO 6 Wireless, you must have: A computer with a 10/100BaseT Ethernet card
USER MANUAL GUIMGR Graphical User Interface Manager for FRM301/FRM401 Media Racks CTC Union Technologies Co., Ltd. Far Eastern Vienna Technology Center (Neihu Technology Park) 8F, No. 60 Zhouzi St. Neihu,
Andover Continuum Remote Communication Configuration Guide 2010, Schneider Electric All Rights Reserved No part of this publication may be reproduced, read or stored in a retrieval system, or transmitted,
BASIC ANALYSIS OF TCP/IP NETWORKS INTRODUCTION Communication analysis provides powerful tool for maintenance, performance monitoring, attack detection, and problems fixing in computer networks. Today networks
Objectives University of Jordan Faculty of Engineering & Technology Computer Engineering Department Computer Networks Laboratory 907528 Lab.4 Basic Network Operation and Troubleshooting 1. To become familiar
LifeSize Networker Installation Guide November 2008 Copyright Notice 2006-2008 LifeSize Communications Inc, and its licensors. All rights reserved. LifeSize Communications has made every effort to ensure
genie app and genie mobile app User Manual 350 East Plumeria Drive San Jose, CA 95134 USA June 2012 202-10933-02 v1.0 2012 NETGEAR, Inc. All rights reserved No part of this publication may be reproduced,
020-101186-01 020-101186-01 이 기기는 업무용 급 으로 전자파적합등록을 한 기기이오니 판매자 또는 사용자는 이점을 주의하시기 바라며 가정 외의 지역에서 사용하는 것을 목적으로 합니다 Table of Contents About this Document... 1 Document Conventions... 1 Audience... 1 Related
IDDERO HOME SERVER Quick start guide www.iddero.com 150302-02 TABLE OF CONTENTS 1 INTRODUCTION... 3 2 IDDERO CONFIGURATION SOFTWARE... 3 2.1 Installation and First Steps...3 3 IDDERO HOME SERVER CONFIGURATION...
ADSL Router Quick Installation Guide Revised, edited and illustrated by Neo A typical set up for a router PCs can be connected to the router via USB or Ethernet. If you wish to use a telephone with the
This guide will step you through the process of configuring your laptop and HiQnet TM London Architect TM to communicate with a network of Soundweb London devices. 1 Install London Architect Install the
CPEi 800/825 Series User Manual * Please see the Introduction Section Contents Introduction...iii Chapter 1: CPEi 800/825 User Guide Overview... 1-1 Powerful Features in a Single Unit... 1-2 Front of the
SNMP Web card User s Manual Management Software for Uninterruptible Power Supply Systems Table of Contents 1. Overview... 3 1.1 Introduction... 3 1.2 Features... 3 1.3 Overlook... 3 1.4 Installation and
ProSAFE 8-Port and 16-Port Gigabit Click Switch Model GSS108E and GSS116E User Manual March 2015 202-11520-01 350 East Plumeria Drive San Jose, CA 95134 USA Support Thank you for selecting NETGEAR products.
User Guide for Windows 10 System requirements E10684 First Edition July 2015 To facilitate a smoother transition from your previous operating system, read the system requirements below before upgrading
AMC-IP-Config_EN Table Of Contents 1. IP Configuration of the AMC... 1 Using AmcIPConfig... 1 Starting the AmcIpConfig application... 1 Menu and tool bar... 2 File > Options... 3 Scanning... 4 Scanning
SD7000 Digital Microphone Monitor Software manual Table of Contents 1. Installing The Monitor Software 1.1 Setting Up Receivers For Monitoring 1.2 Running The Application 1.3 Shutdown 2. The Detail Monitoring
Avaya Wireless AP Device Manager User Guide February 2003 Avaya Wireless AP Device Manager User Guide Copyright Avaya Inc. 2003 ALL RIGHTS RESERVED The products, specifications, and other technical information
Quick Note 32 Using Digi RealPort with a Digi TransPort Router UK Support September 2012 1 Contents 1 Introduction... 3 1.1 Outline... 3 1.2 Assumptions... 3 1.3 Version... 3 2 Configuration & scenario...
your Gateway Windows network installationguide 802.11b wireless series Router model WBR-100 Installing Configuring Contents 1 Introduction...................................................... 1 Features...........................................................
14.1 bs^ir^qfkd=obcib`qflk= Ñçê=emI=rkfuI=~åÇ=léÉåsjp=eçëíë bî~äì~íáåö=oéñäéåíáçå=ñçê=emi=rkfui=~åç=lééåsjp=eçëíë This guide walks you quickly through key Reflection features. It covers: Getting Connected
Legal Notes Unauthorized reproduction of all or part of this guide is prohibited. The information in this guide is subject to change without notice. We cannot be held liable for any problems arising from
Intel Active Management Technology with System Defense Feature Quick Start Guide Introduction...3 Basic Functions... 3 System Requirements... 3 Configuring the Client System...4 Intel Management Engine
Panduit Corp. 2010 TDP43ME NetPS PA26306A01 Rev. 01 11-2010 Network Printer Server Control Center for Ethernet Module NOTE: In the interest of higher quality and value, Panduit products are continually
1.1 Overview Technical Note Cameleon requires that you have a static IP address defined for the server PC the Cameleon server application runs on. Even if the server PC has a static IP address, you may
Vodafone PC SMS 2010 (Software version 4.7.1) User Manual July 19, 2010 Table of contents 1. Introduction...4 1.1 System Requirements... 4 1.2 Reply-to-Inbox... 4 1.3 What s new?... 4 2. Installation...6
BIT COMMANDER Serial RS232 / RS485 to Ethernet Converter (Part US2000A) Copyrights U.S. Converters 1 Contents Overview and Features... 3 Functions..5 TCP Server Mode... 5 Httpd Client Mode.5 TCP Auto mode....6
EnGenius ERB9250 300M Range Extender Quick Start Guide Package Contents One ERB9250 300M Range Extender One 12V/1A power adapter Two 2dBi 2.4GHz SMA antennas One Ethernet cable One CD-ROM with user s manual
Movie Cube User s Guide to Wireless Function Table of Contents 1. WLAN USB Adapter Connection...3 2. Wireless Setup...4 2.1 Infrastructure (AP)...5 2.2 Peer to Peer (Ad Hoc)...7 2.3 Settings for PC...8
Quick Start Guide NVR DS-7104NI-SL/W NVR NOTE: For more detailed information, refer to the User s Manual on the CD-ROM. You must use your PC or MAC to access the files. www.hikvision.com Quick Start 1.
Management Software AT-S106 Web Browser User s Guide For the AT-GS950/48 Gigabit Ethernet Smart Switch Version 1.0.0 613-001339 Rev. A Copyright 2010 Allied Telesis, Inc. All rights reserved. No part of
Contents of Package: DG-102S VoIP Station Gateway Power adapter CD-ROM, including User s Manual Quick Install Guide Requirements: RS-232 Console Cable Two RJ-45 CAT-5 Straight-Through Cables For more information
Debugging Network Communications Situation: you have a computer and your NetBurner device on a network, but you cannot communicate between the two. This application note provides a set of debugging steps
Difficulty printing from your PC can occur for various reasons. The most common reason a networked Brother machine may stop printing, is because the connection between the computer and the Brother machine
Chapter 7 Troubleshooting This chapter provides troubleshooting tips and information for your ProSafe VPN Firewall 200. After each problem description, instructions are provided to help you diagnose and
Allworx OfficeSafe Operations Guide Release 6.0 No part of this publication may be reproduced, stored in a retrieval system, or transmitted, in any form or by any means, electronic, mechanical, photocopy,
1.0 Key Features Addonics T E C H N O L O G I E S NAS Adapter Model: NASU2 User Manual Convert any USB 2.0 / 1.1 mass storage device into a Network Attached Storage device Great for adding Addonics Storage
Operating Instructions (For Setting Up) Digital Imaging Systems Installation Overview Model No. DP-8060 / 8045 / 8035 Table of Contents Setting Your Machine Installation Overview General Installation Overview...
Lab 8.3.2 Conducting a Network Capture with Wireshark Objectives Perform a network traffic capture with Wireshark to become familiar with the Wireshark interface and environment. Analyze traffic to a web
Setup Manual and Programming Reference Stanford Research Systems Revision 1.05 (11/2010) Certification Stanford Research Systems certifies that this product met its published specifications at the time
Back-up Server DOC-OEMSPP-S/2014-BUS-EN-10/12/13 The information contained in this guide is not of a contractual nature and may be subject to change without prior notice. The software described in this
Quick Start Guide DVR DS-7300HWI-SH Series DVR DS-7200HWI-SH Series DVR NOTE: For more detailed information, refer to the User s Manual on the CD-ROM. You must use your PC or MAC to access the files. www.hikvision.com
AXIS Camera Station Quick Installation Guide Copyright Axis Communications AB April 2005 Rev. 3.5 Part Number 23997 1 Table of Contents Regulatory Information.................................. 3 AXIS Camera
Computer Networks I Laboratory Exercise 1 The lab is divided into two parts where the first part is a basic PC network TCP/IP configuration and connection to the Internet. The second part is building a