1 KleenSpec 790 Series Cordless Illumination System for the 590 Series Vaginal Speculum REF Directions for Use
2 ii 2012 Welch Allyn. All rights are reserved. To support the intended use of the product described in this publication, the purchaser of the product is permitted to copy this publication, for internal distribution only,from the media provided by Welch Allyn. Caution: Federal US law restricts sale of the device identified in this manual to, or on the order of, a licensed physician. Welch Allyn assumes no responsibility for any injury, or for any illegal or improper use of the product, that may result from failure to use this product in accordance with the instructions, cautions, warnings, or indications for use published in this manual. Welch Allyn and KleenSpec are registered trademarks of Welch Allyn. CaviWipe is a registered trademark of Metrex Research Corporation. Caution: Changes or modifications not expressly approved by Welch Allyn could void the purchaser's authority to operate the equipment. Patents pending. Mat. Number: , Ver: A Welch Allyn 4341 State Street Road, PO Box 220 Skaneateles Falls, NY Printed in USA
3 iii Contents Introduction...1 Intended use...1 Contraindications...1 User profile...2 Symbols...2 Using the system safely...4 Utilisation sécuritaire du système...7 Charging the illuminator...10 Using the illuminator for a pelvic exam...11 Using the illuminator as light source...11 Maintenance...12 Cleaning and disinfecting Inspecting the system...13 Recycling the system components...13 Troubleshooting...14 Contacts...14 Appendix Specifications Part numbers...17 Limited warranty...18 Service policy EMC guidance and manufacturer s declarations...20
5 Directions for use Cordless illumination system 1 Introduction The Welch Allyn KleenSpec 790 series cordless illumination system can be used with or without the KleenSpec 590 series disposable vaginal speculum. The cordless illumination system consists of the following items. Cordless illuminator Intended use Charging station (base and power cord) When used with the vaginal speculum, the cordless illuminator provides illumination during pelvic examinations and other gynecological procedures, such as pap smears, dilation and curettage (D&C) biopsy, and electrosurgery. The cordless illuminator can be used independent of the speculum as a general-purpose light source. Contraindications Auxiliary power cord (optional) The 790 series illuminator (either by itself or in conjunction with a 590 series speculum) is not intended to be used for eye examinations. The 790 series illuminator (either by itself or in conjunction with a 590 series speculum) is not intended to be used for diagnosis.
6 2 Welch Allyn User profile Symbols The intended users of the illuminator are medical professionals who are qualified to perform a pelvic examination. The symbols illustrated on the following pages may appear in this manual or on the illuminator, charging station, auxiliary power cord, packaging, or shipping container. Table 1. Documentation symbols WARNING Caution In this manual, this symbol and WARNING indicate conditions or practices that could lead to illness, injury, or death. In this manual, this symbol and Caution indicate conditions or practices that could damage the equipment or other property. On the product, this symbol means consult operating instructions. Table 2. Shipping, storing, and environment symbols This end up Keep dry Fragile 95% Relative humidity limit 15% +49 C Temperature limits -20 C
7 Directions for use Cordless illumination system 3 Table 3. Certification symbols Meets essential requirements of European Medical Device Directive 93/42/EEC. EC REP Regulatory Affairs Representative Welch Allyn Limited Navan Business Park Dublin Road, Navan County Meath, Republic of Ireland CERTIFIED TO: CAN/CSA STD C22.2 NO CONFORMS TO: IEC , IEC , UL N344 EMC framework of Australia Table 4. Operation symbols Caution, hot surface The speculum is a Type BF applied part Alternating current Direct current Double insulation IPXØ Not protected from the ingress of liquid. Recycle the product separately from other disposables. See Recycling the system components on page 13. Power plug
8 4 Welch Allyn Using the system safely Before using the cordless illumination system, you must read and understand the following safety-related information. General warnings The following warning statements apply to use of the cordless illumination system in general. Warning statements that apply specifically to particular procedures, such as cleaning, appear in the corresponding sections of the manual. Warning statements indicate conditions or practices that could lead to illness, injury, or death. WARNING To avoid the risk of electric shock, this equipment must only be connected to a supply mains with protective earth. WARNING No modification to this equipment is allowed. WARNING The illuminator may generate heat after prolonged use. Operation time limits: Illuminator with speculum 10 minutes on, 5 minutes off between uses. Illuminator alone 2 minutes on, 5 minutes off between uses. Illuminator could be harmful to skin if touched after being on for extended times. To avoid fire, do not place illuminator onto a flammable surface if LED is on. Illuminator is not suitable for use in the presence of a flammable anesthetic mixture with air or oxygen or nitrous oxide. An explosion may result.
9 Directions for use Cordless illumination system 5 General cautions WARNING To prevent shock hazards, the power outlet must be properly grounded. Grounding reliability is achieved only when connected to hospital-use or hospital-grade receptacles. WARNING The illuminator is not intended to be used for eye examinations. WARNING The illuminator is not intended to be used for diagnosis. WARNING The illuminator is intended to be used by trained medical professionals. WARNING This product comes with a power supply and/or power cord that is intended for use only with this product. The power supply and/or power cord have not been tested and approved for use with other products that may have the same power connectors. If you cannot locate the original power supply and/or power cord, please contact Welch Allyn to obtain replacement parts. The following caution statements apply to use of the cordless illumination system in general. Caution statements that apply specifically to particular procedures, such as pelvic examinations, appear in the corresponding sections of the manual. Caution statements indicate conditions or practices that could damage the equipment or other property. Caution Disconnect the AC power cord to remove power to the device. Caution The cordless illuminator could be a biohazard after use. Follow good hospital practices. See Cleaning and disinfecting on page 12. Caution Cordless illuminator, charging station, and auxiliary power cord are not protected from the ingress of liquid. Caution There are no user-serviceable parts inside the device. In case of a malfunction, call Welch Allyn and precisely describe the problem.
10 6 Welch Allyn Caution This device complies with current required standards for electromagnetic interference and should not present problems to other equipment or be affected by other devices. As a precaution, avoid using this device in close proximity to other equipment. Caution Use only Welch Allyn-approved parts. See Part numbers on page 17.
11 Directions for use Cordless illumination system 7 Utilisation sécuritaire du système Avant d'utiliser le système d éclairage sans fil, vous devez bien lire et comprendre les renseignements de sécurité suivants. Avertissements généraux Les avertissements suivants s appliquent à l'utilisation du système d éclairage sans fil en général. Les avertissements qui s'appliquent à des procédures particulières, comme le nettoyage, sont abordés dans les sections correspondantes de ce manuel. Ces avertissements indiquent des conditions ou des pratiques qui pourraient causer des maladies, des blessures ou des décès. AVERTISSEMENT Pour éviter les risques de choc électrique, cet équipement doit seulement être branché sur une alimentation secteur dotée d'une mise à la masse. AVERTISSEMENT Aucune modification de cet équipement n'est autorisée. AVERTISSEMENT Le système d éclairage peut dégager de la chaleur après une utilisation prolongée. Les durées d'opération maximales : Illuminateur avec spéculum 10 minutes de marche, 5 minutes d arrêt entre les utilisations. Illuminateur seul 2 minutes de marche, 5 minutes d arrêt entre les utilisations. Le système d éclairage peut causer des brûlures cutanées si vous y touchez après une utilisation prolongée. Pour éviter les risques d'incendie, ne déposez pas le système d éclairage sur une surface inflammable lorsque la DEL est allumée. Le système d éclairage ne devrait pas être utilisé en présence d'un mélange d'anesthésique inflammable et de l air ou encore d'oxygène ou d'oxyde nitreux. Cela pourrait causer une explosion.
12 8 Welch Allyn AVERTISSEMENT La prise électrique doit être correctement mise à la terre afin d éviter les chocs électriques. Vous ne pouvez être certain de la mise à la terre que si l'appareil est branché dans une prise utilisée par les hôpitaux ou de qualité hospitalière. AVERTISSEMENT Le système d éclairage ne devrait pas être utilisé pour les examens des yeux. AVERTISSEMENT Le système d éclairage ne devrait pas être utilisé pour effectuer un diagnostic. AVERTISSEMENT Le système d éclairage est conçu pour être utilisé par des professionnels médicaux formés. AVERTISSEMENT Un bloc d'alimentation et/ou un cordon d'alimentation, destinés à une utilisation avec ce produit seulement, sont fournis avec le produit. Le bloc d'alimentation et/ou le cordon d'alimentation n'ont pas été testés et approuvés pour une utilisation avec d'autres produits qui pourraient être munis des mêmes connecteurs d'alimentation. Si vous n'avez plus le bloc d'alimentation et/ou le cordon d'alimentation originaux, veuillez communiquer avec Welch Allyn pour obtenir des pièces de remplacement.
13 Directions for use Cordless illumination system 9 Mises en garde générales Les mises ne garde suivantes s appliquent à l'utilisation du système d éclairage sans fil en général. Les énoncés de mise en garde qui s'appliquent à des procédures particulières, comme les examens pelviens, sont abordées dans les sections correspondantes de ce manuel. Les énoncés de mise en garde indiquent des conditions ou des pratiques qui pourraient endommager les équipements ou la propriété. Mise en garde Débrancher le cordon d'alimentation c.a. pour mettre l'appareil hors tension. Mise en garde Le système d éclairage sans fil pourrait être considéré à risque biologique après utilisation. Conformezvous aux bonnes pratiques hospitalières. Voir Cleaning and disinfecting à la page 12. Mise en garde Le système d éclairage sans fil, la borne de recharge et le cordon d'alimentation auxiliaire ne sont pas protégés contre les infiltrations de liquide. Mise en garde Cet appareil ne contient aucune pièce à entretenir. En cas de mauvais fonctionnement, appelez Welch Allyn et décrivez le problème de manière précise. Mise en garde Cet appareil est conforme aux normes actuelles obligatoires sur les interférences électromagnétiques. Il ne devrait présenter aucun problème pour les autres équipements et ne devrait pas être affecté par d'autres appareils. Par mesure de précaution, évitez d'utiliser cet appareil à proximité d'autres équipements. Mise en garde N'utilisez que des pièces approuvées par Welch Allyn. Voir Part numbers à la page 17.
14 10 Welch Allyn Charging the illuminator Charge the illuminator before using it the first time, or whenever its low-charge indicator turns on. LED Power switch Low-charge indicator: On = battery low To charge the illuminator 1. If LED is lit, turn power switch off. 2. Connect charging base to AC power. 3. Place the illuminator (either direction) into the charging station. Full charging takes 6 to 8 hours. Status indicator: Off = no AC power Green = AC power Amber (illuminator inserted) = charging. Full charging takes 6 to 8 hours. 4. Remove the charged illuminator when you are ready to use it. It is safe to leave the illuminator in the charging station after it is charged.
15 Directions for use Cordless illumination system 11 Using the illuminator for a pelvic exam 1. Fully insert the illuminator into a 590 series speculum (a). The LED automatically turns on. 2. Conduct the examination. 3. If the LED turns off, do not remove the speculum. Do either of these things: Leave the illuminator in place, and use an external light source. a or Connect the auxiliary power cord to AC power. Hold the speculum in place, and remove the illuminator. Connect the auxiliary power cord to illuminator (b). Reinsert the illuminator. b 4. When finished, remove the speculum, and then remove the illuminator. The LED automatically turns off. Using the illuminator as light source To use the illuminator as a light source, slide the power switch to turn it on (all the way down) and off (all the way up). Note For operation time limits, see Intermittent operation on page 16. Power switch
16 12 Welch Allyn Maintenance Cleaning and disinfecting Use CaviWipe wipes for both cleaning and disinfecting.* *EPA-approved intermediate-level disinfectant appropriately labeled as effective against TB mycobacterium tuberculosis (per AAMI TIR-12, 2004, section 5.2.3) Caution Use only the cleaning and disinfection methods described in this manual. Clean the illuminator prior to disinfection. To clean and disinfect the illuminator 1. After using the illuminator, unplug any power cords. 2. Clean the illuminator. a. Following the wipe manufacturer's instructions, wipe the entire unit to completely remove any gross debris. b. Gently wipe the gold contacts on the bottom with 70% alcohol. If debris builds up on these contacts, the charging or auxiliary power functions might be affected. 3. Disinfect the illuminator. a. Following the wipe manufacturer's instructions, thoroughly wet the surface of the entire illuminator. b. Using sterile gauze soaked with 70% isopropyl alcohol, thoroughly wet the surface of the entire illuminator. The treated surface must remain wet for five minutes. If needed, reapply the solution to keep the surface continuously wet during this time. c. Allow the surface to dry. To clean the charging base or auxiliary power cord 1. After using the charging base or auxiliary power cord, unplug the power cord. 2. Clean the component. Following the wipe manufacturer's instructions, wipe the surface to completely remove any gross debris.
17 Directions for use Cordless illumination system 13 Inspecting the system Examine all components of the cordless illumination system monthly. Components include the illuminator, charging station, and auxiliary power cord. If any component is worn or damaged, replace it with the Welch Allyn-approved part. For ordering information, see Part numbers on page 17. Recycling the system components Within the EU Do not dispose of this product as unsorted municipal waste. Prepare it for reuse or separate collection as specified by Directive 2002/96/EC of the European Parliament and the Council of the European Union on Waste Electronic and Electrical Equipment (WEEE). If this product is contaminated, this directive does not apply. For more specific disposal information, see or contact Welch Allyn. Outside the EU When the product reaches its end of life, recycle the components according to national, state, and local regulations, or return the product to Welch Allyn.
18 14 Welch Allyn Troubleshooting If you try these suggestions and still have problems, contact Welch Allyn. Problem LED does not light. Probable causes & solutions Battery is low. Charge the illuminator. Illuminator is not seated correctly in speculum. Insert illuminator fully into speculum. LED does not light when the auxiliary power cord is attached. LED does not automatically turn off when removing the illuminator from the speculum. Illuminator does not charge in the charging base. Auxiliary power cord is not plugged in. Plug the auxiliary power cord into an AC outlet. Auxiliary power cord is not seated properly. Plug the cord into the illuminator properly. Power switch is damaged or blocked. Clean the switch area. If the problem continues, replace the illuminator. Charging base is not plugged in. Connect charging base to AC power. Contacts Illuminator is not seated properly in charging base. Reinsert the illuminator into the charging base, and verify that the status indicator glows amber. For information about any Welch Allyn product, call the nearest Welch Allyn representative: USA Australia Canada China European Call Center France Germany Japan Latin America Netherlands Singapore South Africa United Kingdom Sweden
19 Directions for use Cordless illumination system 15 Appendix Specifications Item Auxiliary power cord transformer Classification Input Output Category Charging station transformer Classification Input Output Category Power cord length Auxiliary power cord Charging station Weight Illuminator Charging station (base and power cord) Auxiliary power cord Battery cell Capacity Voltage Chemistry Protection against ingress of water, per IEC (illuminator, auxiliary power cord, charging station) Operating conditions Temperature Relative humidity Altitude Specification U.S. & Canada Class II and internally powered, type BF applied part 120V~60Hz 5.5 VDC, 300 ma Not AP/APG equipment U.S. & Canada Class I and internally powered 120V~60Hz 9.8 VDC, 150 ma Not AP/APG equipment 305 cm (120 in) 92 cm (36 in) 48 g (0.1 lb) 142 g (0.31 lb) 290 g (0.64 lb) 420 mah 3.7 V Li-ion (single lithium ion) IPXØ +10 C to +35 C (+50 F to +95 F) 30% to 75% noncondensing 700 to 1060 hpa International Class I and internally powered, type BF applied part VAC ma 5.5 VDC, 300 ma Not AP/APG equipment International Class I, Class II and internally powered 230 V (240 V in Australia) ~ 50 Hz VDC, 200 ma Not AP/APG equipment
20 16 Welch Allyn Item (continued) Shipping and storage conditions Temperature Relative humidity Altitude Intermittent operation Illuminator with speculum Illuminator alone Safety, EMC, and regulatory compliance LED radiation output Specification (continued) -20 C to +49 C (-4 F to +120 F) 15% to 95% noncondensing 500 hpa to 1060 hpa 10 minutes on, 5 minutes off between uses 2 minutes on, 5 minutes off between uses CAN/CSA C22.2 No M90, including amendments 1 & 2 EN/IEC EN/IEC EN/IEC UL AS/NZS CISPR11 < 6.67 mw at nm wavelengths
21 Directions for use Cordless illumination system 17 Part numbers Item Cordless illuminator system with charging station Order Number REF U.S. & Canada: International: 79915* Cordless illuminator system with charging station and auxiliary power cord U.S. & Canada: International: 79925* Cordless illuminator U.S. & Canada: Charging station (base and power cord) U.S. & Canada: International: 73915* Auxiliary power cord U.S. & Canada: International: 75915* * Specify country and voltage.
22 18 Welch Allyn Limited warranty Welch Allyn warrants that the KleenSpec vaginal speculum cordless illumination system products meet the labeled specifications of the Products and will be free from defects in materials and workmanship that occur within one year after the date of purchase. The date of purchase is the date specified in our records, if you purchased the Product directly from us or 30 days after the date on which the Product was sold to the dealer from whom you bought the Product, as documented in our records. This warranty does not cover damage caused by: 1) handling during shipping, 2) use or maintenance contrary to labeled instructions, 3) alteration or repair by anyone not authorized by Welch Allyn, and 4) accidents. If a Product or accessory covered by this warranty is determined to be defective because of defective materials, components, or workmanship, and the warranty claim is made within the warranty period described above, Welch Allyn will, at its discretion, repair or replace the defective Product or accessory free of charge. If your Product requires repairs covered by this warranty, upon your request Welch Allyn will loan to you, at no cost, a substitute Product for use until your repaired Product is returned. You must obtain a return authorization from Welch Allyn to return your Product before you send it to Welch Allyn s designated service center for repair. Contact Welch Allyn Technical Support. THIS WARRANTY IS IN LIEU OF ALL OTHER WARRANTIES, EXPRESS OR IMPLIED, INCLUDING BUT NOT LIMITED TO THE IMPLIED WARRANTIES OF MERCHANTABILTY AND FITNESS FOR A PARTICULAR PURPOSE. WELCH ALLYN'S OBLIGATION UNDER THIS WARRANTY IS LIMITED TO REPAIR OR REPLACEMENT OF PRODUCTS CONTAINING A DEFECT. WELCH ALLYN IS NOT RESPONSIBLE FOR ANY INDIRECTOR CONSEQUENTIAL DAMAGES RESULTING FROM A PRODUCT DEFECT COVERED BY THE WARRANTY.
23 Directions for use Cordless illumination system 19 Service policy All repairs on products under warranty must be performed or approved by Welch Allyn. Unauthorized repairs will void the warranty. In addition, whether or not covered under warranty, any product repair shall exclusively be performed by Welch Allyn certified service personnel. If the product fails to function properly or if you need assistance, service, or spare parts contact Welch Allyn. Before contacting Welch Allyn, try to duplicate the problem, and check all accessories to ensure that they are not causing the problem. When calling, please be prepared to provide: Product name and model number and complete description of the problem. Serial number of your product (if applicable). Complete name, address and phone number of your facility. For out-of-warranty repairs or spare parts orders, a purchase order (or credit card) number. For parts orders, the required spare or replacement part numbers. If your product requires warranty, extended warranty, or non-warranty repair service, please call Welch Allyn. A representative will assist you troubleshooting the problem and will make every effort to solve it over the phone, avoiding potential unnecessary returns. In case a return cannot be avoided, the representative will record all necessary information and will provide a Return Material Authorization (RMA) number, as well as the appropriate return address. An RMA number must be obtained prior to any return. If you have to return goods for service, follow these recommended packing instructions: Remove all hoses, cables, sensors, power cords, and ancillary products (as appropriate) before packing, unless you suspect they are associated with the problem. Wherever possible use the original shipping carton and packing materials. Include a packing list and the Welch Allyn Return Material Authorization (RMA) number. It is recommended that all returned goods be insured. Claims for loss or damage to the product must be initiated by the sender.
24 20 Welch Allyn EMC guidance and manufacturer s declarations Table 1. Emissions for all EQUIPMENT and SYSTEMS The 790 series cordless illumination system is intended for use in the electromagnetic environment specified below. The customer or user of the 790 series cordless illumination system should assure that it is used in such an environment. Emissions test Compliance Electromagnetic environment - guidance RF emissions CISPR 11 RF emissions CISPR 11 Harmonic emissions IEC Voltage fluctuations/ flicker emissions IEC Group 1 Class B Class A Complies The 790 series cordless illumination system uses RF energy only for its internal function. Therefore, its RF emissions are very low and are not likely to cause any interference in nearby electronic equipment. The 790 series cordless illumination system is suitable for use in all establishments including domestic and those directly connected to the public low-voltage power supply network that supplies buildings used for domestic purposes.
25 Directions for use Cordless illumination system 21 Table 2. Immunity for all EQUIPMENT and SYSTEMS The 790 series cordless illumination system is intended for use in the electromagnetic environment specified below. The customer or user of the 790 series cordless illumination system should assure that it is used in such an environment. Immunity test IEC test level Electrostatic discharge (ESD) IEC Electrical fast transient (EFT)/ burst IEC Surge IEC Voltage dips, short interruptions, and voltage variations on power supply input lines. IEC Power frequency (50/60Hz) magnetic field IEC ± 6 kv contact ±8kVair ±2 kv mains ±1 kv I/O ±1 kv differential ±2 kv common >95% dip in 0.5 cycle 60% dip in 5 cycles 30% dip for 25 cycles >95% dip in 5 seconds Compliance level ± 6 kv contact ±8kVair ±2 kv mains Not applicable ±1 kv differential ±2 kv common >95% dip in 0.5 cycle 60% dip in 5 cycles 30% dip for 25 cycles >95% dip in 5 seconds Electromagnetic environment - guidance Floors should be wood, concrete, or ceramic tile. If floors are covered with synthetic material, the relative humidity should be at least 30%. Mains power quality should be that of a typical commercial or hospital environment. Mains power quality should be that of a typical commercial or hospital environment. Mains power quality should be that of a typical commercial or hospital environment. If the user of the 790 series cordless illumination system requires continued operation during power mains interruptions, it is recommended that the 790 series cordless illumination system be powered from an uninterruptible power supply or battery. 3 A/m 3 A/m Power frequency magnetic fields should be at levels characteristic of a typical location in a typical commercial or hospital environment.
26 22 Welch Allyn Table 3. Electromagnetic immunity The 790 series cordless illumination system is intended for use in the electromagnetic environment specified below. The customer or user of the 790 series cordless illumination system should assure that it is used in such an environment. Immunity test IEC test level Compliance level Electromagnetic environment - guidance Portable and mobile RF communications equipment should be used no closer to any part of the 790 series cordless illumination system, including cables, than the recommended separation distance calculated from the equation applicable to the frequency of the transmitter. Recommended separation distance Conducted RF, IEC Radiated RF, IEC Vrms 150 khz to 80 MHz 3 V/m 80 MHz to 2.5 GHz 3 Vrms d = (1.17) 3 V/m d = (1.17) P 80 MHz to 800 MHz d = (2.33) P P 800 MHz to 2.5 GHz where P is the maximum output power rating of the transmitter in watts according to the transmitter manufacturer and d is the recommended separation distance in meters. Field strengths from fixed RF transmitters, as determined by an electromagnetic site survey, a should be less than the compliance level in each frequency range. b Interference may occur in the vicinity of equipment marked with the following symbol:
27 Directions for use Cordless illumination system 23 Table 3. Electromagnetic immunity (continued) Note 1: At 80 MHz and 800 MHz, the higher frequency range applies. Note 2: These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection from structures, objects, and people. a Field strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and land mobile radios, amateur radio, AM and FM radio broadcast and TV broadcast cannot be predicted theoretically with accuracy. To assess the electromagnetic environment due to fixed RF transmitters, an electromagnetic site survey should be considered. If the measured field strength in the location in which the 790 series cordless illumination system is used exceeds the applicable RF compliance level above, the 790 series cordless illumination system should be observed to verify normal operation. If abnormal performance is observed, additional measures may be necessary, such as reorienting or relocating the 790 series cordless illumination system. b Over the frequency range 150 khz to 80 MHz, field strengths should be less than 3 V/m.
28 24 Welch Allyn Table 4. Recommended separation distances between portable and mobile RF communications equipment and the 790 series cordless illumination system The 790 series cordless illumination system is intended for use in an electromagnetic environment in which radiated RF disturbances are controlled. The customer or user of the 790 series cordless illumination system can help prevent electromagnetic interference by maintaining a minimum distance between portable and mobile RF communications equipment (transmitters) and the 790 series cordless illumination system as recommended below, according to the maximum output power of the communications equipment. Rated max. output power of transmitter (W) Separation distance according to frequency of transmitter (m) 150 khz to 80 MHz 80 MHz to 800 MHz 800 MHz to 2.5 GHz d = (1.17) P d = (1.17) P d = (2.33) P For transmitters rated at a maximum output power not listed above, the recommended separation distance d in meters (m) can be estimated using the equation applicable to the frequency of the transmitter, where P is the maximum output power rating of the transmitter in watts (W) according to the transmitter manufacturer. Note 1: At 80 MHz and 800 MHz, the separation distance for the higher frequency range applies. Note 2: These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection from structures, objects, and people.
5392 Dual Chamber Temporary External Pacemaker Electromagnetic Compatibility Declaration Caution: Federal law (USA) restricts this device to sale by or on the order of a physician. The following list includes
Regulatory Compliance Statement 0560 EU Declaration of Conformity The declaration of conformity may be consulted at www.kobo.com/userguides SAR Limits The exposure standard for wireless devices employs
1053621 EN, DE DSF 4/16/08 EverFlo EverFlo Q English Table of Contents Symbol Key... 2 Abbreviations... 2 Chapter 1: Introduction... 3 Intended Use... 3 About Your EverFlo / EverFlo Q...3 Parts of Your
Operation Instructions PowerCool Series TOP VIEW BOTTOM VIEW PDCOOL-05RA PDCOOL-00RK PDCOOL-5R PDCOOL-0R ALSO AVAILABLE Thank you for purchasing the PowerCool Series component cooler and power distribution.
767 Diagnostic System 3.5v 767 Wall Transformer Patent Pending Listed to: UL 60601-1 CSA C22.2 No. 601 Thank you for purchasing the Welch Allyn 3.5v 767 Wall Transformer. This manual is meant to provide
i800 AC Current Clamp Instruction Sheet Introduction The i800 AC Current Clamp, the Clamp, has been designed for use with multimeters, recorders, power analyzers, safety testers, etc., for accurate non-intrusive
Revision A Direct AC Wiring Option Installation Guide The direct AC wiring option enables you to wire AC power directly in to the ADP InTouch. The option kit contains the following parts. 25mm 12mm Caution:
Installation Instructions Compact I/O End Caps/Terminators Catalog Numbers 1769-ECL, 1769-ECR Topic Page Important User Information 2 Environment and Enclosure 3 North American Hazardous Location Approval
Gateway Port Replicator User Guide Using the Port Replicator Identifying features Connecting and disconnecting the port replicator 1 Using the Port Replicator Top Power button Component Icon Description
INSTALL SHEET BDA Broadband Telecommunications Drop Amplifier Introduction The Broadband Telecommunications Drop Amplifier (BDA) is a two-way, 1 GHz amplifier designed for customer-premise amplification
System One Heated Humidifier USER MANUAL 2009 Respironics Inc. and its affiliates. All rights reserved. Table of Contents Intended Use... 2 Warnings... 2 Cautions... 2 Symbol Key... 2 System Features and
User Guide Nokia Portable Wireless Charging Plate DC-50 Issue 1.1 EN User Guide Nokia Portable Wireless Charging Plate DC-50 Contents For your safety 3 About your accessory 4 Keys and parts 5 Top up your
aseries A13B Mini Bullet Camera User Manual Thank you for purchasing our product. If there are any questions, or requests, please do not hesitate to contact the dealer. This manual applies to the MicroView
Welch Allyn ProXenon Headlight and Fiberoptic Cable Directions for Use ii Copyright Information Welch Allyn ProXenon Headlight and Fiberoptic Cable Copyright 2008 Welch Allyn Welch Allyn is a registered
StarLight 3000 OPERATING INSTRUCTIONS h h 2 4 1 1. Quick Start Power Button 2. SOS/Flashlight Power Switch 3. USB Out/In 4. Fuel Gauge (Battery Status Indicator) 3 Figure 1 A B Included items: - 1 HALO
BroadBand PowerShield User Manual 990-0375G 12/2006 Chapter 1 General Information The PowerShield provides a power source for broadband telephony and other DC applications. Safety This Safety Guide contains
VistaMAX Series Safety Precautions Read this document before using this product. Download and read the full user manual for complete information (see the VistaMAX Support File Access document for download
Heated HumidifieR M Series USER Manual 1031849 HM 1/6/06 Table of Contents Introduction...3 In This Manual...3 Intended Use...3 Warnings and Cautions...3 Warnings...3 Cautions...4 Symbol Key...5 System
WorkHorse SERIES 1 FERRORESONANT BATTERY CHARGERS Installation and Operating Instructions Applied Energy Solutions, LLC. One Technology Place Caledonia, New York 14423 (585) 538-4421 www.appliedenergysol.com
UR Series LED Upgrade Kit Includes: 48" Linear Option IMPORTANT SAFEGUARDS When using electrical equipment, basic safety precautions should always be followed including the following: READ AND FOLLOW ALL
AD511 Active Iridium Antenna User Manual Mar 12 V4.0 AD511 Active Iridium Transmitter/Receiver Antenna with up to 160 metres of coaxial down-lead and DC Power Break-In Box for Iridium Satellite Systems
Laptop Power Adapter for home/office User Manual Introduction Congratulations and thank you for purchasing the Laptop Power Adapter for home/office from Belkin. This new Adapter provides a single source
Rechargeable Power Pack User s Manual 2013 VTech Printed in China 91-009656-000 US INTRODUCTION The Rechargeable Power Pack makes it easier than ever to keep the InnoTab 3 or InnoTab 3S charged and ready
Cisco TelePresence MCU 4500 Series safety and compliance information On this page: Safety information symbols Operating guidelines Safety warnings Technical specifications Compliance information WEEE information
Technical Support Thank you for selecting NETGEAR products. After installing your device, locate the serial number on the label of your product and use it to register your product at www.netgear.com/register.
Transcend Heated Humidifier User Guide Page 1 Transcend Heated Humidifier TM User guide Notices Revised Notice Trademark Copyright Transcend Heated Humidifier User Guide 103404 Rev A Published April 15,
Home Care Provider Setup Instructions Always use these instructions along with the User Manual when assembling or adjusting this equipment. For clinical systems, refer to the setup guide entitled Respironics
Mini Speaker Printed in China Part No. 480-919-P Version D CWA BT300 Product Information Guide IMPORTANT To get the full capability of your new Cobra Airwave Mini, please read this manual. It's more than
INSTALLATION GUIDE AXIS T8133 Midspan 30W ENGLISH Liability Every care has been taken in the preparation of this document. Please inform your local Axis office of any inaccuracies or omissions. Axis Communications
FortiFone QuickStart Guide for FON-670i and FON-675i FortiFone QuickStart Guide for FON-670i and FON-675i Revision 2 September 25, 2015 Copyright 2015 Fortinet, Inc. All rights reserved. Fortinet, FortiGate,
User Manual CDM250 Digital Multimeter 070-6736-03 Copyright Tektronix, Inc. 1987. All rights reserved. Tektronix products are covered by U.S. and foreign patents, issued and pending. Information in this
HP SuperSpeed USB 3.0 PCIe x1 Card User Guide Copyright 2010 Hewlett-Packard Development Company, L.P. The information contained herein is subject to change without notice. Microsoft, Windows, and Windows
Entire document: 7 pages THIS DOCUMENT CONTAINS GENERAL INSTRUCTION FOR SERVICING STENNER PRODUCTS. PLEASE BE SURE TO READ ALL SAFETY INFORMATION. NOTICE: Indicates special instructions or general mandatory
Quick-Start Guide Arrival Sensor Meet your Arrival Sensor The Arrival Sensor lets you stay connected to people, pets, and cars by sending you a push notification when they arrive and leave home. How to
Backup Scanner Module Catalog Number 1747-BSN Installation Instructions 2 Backup Scanner Module Important User Information Because of the variety of uses for the products described in this publication,
Power Supply Guide Version 1.0 for D-Show Digidesign 2001 Junipero Serra Boulevard Daly City, CA 94014-3886 USA tel: 650 731 6300 fax: 650 731 6399 Technical Support (USA) tel: 650 731 6100 fax: 650 731
Electronic Stethoscope MODEL 3000 with Ambient Noise Reduction Made in Denmark for 3M Health Care, St. Paul, MN 55144 (U.S.A.) 1 800 228-3957 Fax 651 736-2803 Visit our web site: http://www.3m.com/littmann
IE-MiniMc LFPT and Telco/LFPT Operation Manual FCC Radio Frequency Interference Statement Class A (using DC terminal power or Power Over Ethernet (POE)) This equipment has been tested and found to comply
v. 02.november.2010 owner s manual 1. IMPORTANT SAFETY INSTRUCTIONS 1) Read these instructions; 2) Keep these instructions; 3) Heed all warnings; 4) Follow all instructions; 5) Do not use this apparatus
FLUORESCENT UV- RING LIGHT OPERATING INSTRUCTION Caution! UV-radiation of this device is in the range of UV-A (320-400 nm). Direct exposure to eyes shall therefore be avoided. UV protection glasses shall
www.iconnectivity.com USER GUIDE WARRANTY INFORMATION iconnectivity warrants to the original purchaser that this unit is free of defects in materials and workmanship under normal use and maintenance for
Installation Instructions USB-to-Data Highway Plus Cable Catalog Number 1784-U2DHP Topic Page Important User Information 2 Environment and Enclosure 3 Install the Cable 5 Configure the DH+ Node Address
USER S GUIDE User s Guide Aaron 940 1 TABLE OF CONTENTS Introduction...3 Precautions...3 Unit Operation...6 Setup Procedures...7 Maintenance...9 Sterilization...9 Accessories...9 Technical Description...9
Introduction This guide is to assist the reader with the most basic form of installation and connection to switches in this series. As there is more than one switch in this series, the diagrams might slightly
INSTALLATION GUIDE AXIS T81B22 DC 30W Midspan ENGLISH About this Document This document includes instructions for installing AXIS T81B22 on your network. Previous experience of networking will be beneficial
SECTION 7 - REPAIR/REPLACEMENT PROCEDURES INTRODUCTION Although the modular power system is designed to give long, troublefree service, some components may need to be replaced periodically. This section
DC50Xu Digital Transport Adapter Quick Start Guide BRINGING TECHNOLOGY HOME www.pace.com IMPORTANT SAFETY INSTRUCTIONS Before you install or use the apparatus, you must read and understand these Important
Operating Instructions Vision 5L Digital Holter Recorder Document Number: 010-1646-00 Revision: H Bothell, Washington 98021-8969 (608)-764-1919 Caution: U.S. Federal law restricts this device to sale by
USER GUIDE TURBOCORD TM PORTABLE CHARGER 120V/240V: DUAL VOLTAGE AeroVironment EV Solutions 2013 AeroVironment, Inc. All rights reserved. AeroVironment, EV Solutions, and the AeroVironment logo are trademarks
Agilent 87421A/87422A Power Supply Technical Overview Designed specifically for Agilent Technologies microwave system amplifiers Bias cable permits remote placement Compact size for easy system integration
FortiFone QuickStart Guide for FON-370i FortiFone QuickStart Guide for FON-370i Revision 2 August 17, 2015 Copyright 2015 Fortinet, Inc. All rights reserved. Fortinet, FortiGate, FortiCare and FortiGuard,
user s manual model #: SPB3200 Charger for Samsung Galaxy S 4 What s Included Unpack the battery case and make sure all accessories are put aside so they will not be lost. hello. USB to Micro USB Cable
PO Box 8680 Madison, WI 53708 Owner s Manual Phoenix CAM Installation, Operation & Service Instructions Read and Save These Instructions Low amp draw, balanced airflow characteristics and maximum velocity
SMS SERIES MIXER SYSTEM INSTALLATION AND MAINTENANCE MANUAL PERISTALTIC METERING PUMPS SINCE 1957 TABLE OF CONTENTS WARRANTY AND SERVICE POLICY... 3 SAFETY INFORMATION... 4-5, 8-12 MIXER SYSTEM IDENTIFICATION...
Instruction Manual ET50 ENGLISH AC / DC VOLTAGE TO 600V POSITIVE AND NEGATIVE DC POLARITY LIGHT AND VIBRATION INDICATORS CAT IV 600V SAFETY RATING MODERN SOLID-STATE DESIGN C 1 www.kleintools.com For Professionals...Since
GSM-EXT Cable Assembly Installation Guide For Documentation and Online Support: http://www.security.honeywell.com/hsc/resources/mywebtech General Information The GSM-EXT cable assembly is used to connect
Wireless Keyboard and Optical Mouse Bundle Enjoy the freedom of wireless optical precision User Manual F8E832-BNDL TABLE OF CONTENTS Introduction....................................1 Compatible Equipment............................2
CDR USB Remote HS User Guide B1051063 Sirona Dental, Inc. 30-30 47 th Avenue Long Island City, NY 11101 USA (718) 937-5765 (718) 937-5962 (fax) Sirona Dental GmbH Sirona Straße 1 5071 Wals bei Salzburg
PROGRAMMER INSTALLATION INSTRUCTIONS TABLE OF CONTENTS Description 4 Installation 4 Operation 5 Safety Precautions 6 Regulatory & Service Information 6 Important Safety and Usage Information 7 Regulatory
Owner s Manual 2600mAh Mobile Power Bank Model: UPB-02K6-1U PROTECT YOUR INVESTMENT! Register your product for quicker service and ultimate peace of mind. You could also win an ISOBAR6ULTRA surge protector
EPM3 Phase Sequence and Motor Rotation Tester Users Manual 1 L1 L2 L3 2 3 A B C CAT 600V 3-PHASE TESTER 5 TEST M1 M2 BATT M3 EPM3 MOTOR ROTATION TESTER EPM3 MOTOR ROTATION DETERMINED WHILE FACING MOTOR
Messager USB Installation & User Manual Note: All units support MP3 playback of up to 128Kbps @ 48KHz. Units with serial numbers starting with 15 or above will support the addition of WAV playback done
ReachDSL ALM Hotwire ReachDSL Modem, Model 6390 with Inline Phone Filter Installation Instructions Document Number 6390-A2-GN10-00 May 2002 Package Checklist Verify that your package contains the following:
CONTENTS Thank you for purchasing the Welch Allyn Fiber Optic Headlight. Follow the guidelines provided in this manual, and you will enjoy years of dependable, trouble-free service from your new headlight.
AMBR AMATIS Border Router User Guide TABLE OF CONTENTS Version 4.0 October 2015 1.0 INTRODUCTION 1.1 BEFORE YOU START 1.2 NETWORK FIREWALL REQUIREMENTS 2.0 INSTALLING AMBR 2.1 TROUBLESHOOTING AMBR CONNECTIVITY
Safety, Installation and Operating Instructions Instructions importantes concernant la sécurité Manual for the Following Battery Charger Models i2412, i3612, i4809, i2425, i3625, and i4818 081808 IMPORTANT
User s Manual TM Fire Command Rescue Center 2014 VTech Printed in China 91-009683-001 US INTRODUCTION Thank you for purchasing the VTech Go! Go! Smart Wheels Fire Command Rescue Center playset learning
Operating Guide Manuel de l utilisateur Model RF-110 Modèle RF-110 English Thank you for purchasing this Ringmaster; please read this operating guide carefully before use and keep it for future reference.
IMPORTANT SAFETY INSTRUCTIONS Before you install or use the apparatus, you must read and understand these Important Safety Instructions. At all times when using the apparatus you must follow these Important
Welch Allyn Connex VM data management system Directions for use 2011 Welch Allyn. All rights are reserved. To support the intended use of the product described in this publication, the purchaser of the