1 Modbus Plus Adapter EIG Model MP1, a Modbus Plus adapter for DMMS 300+ and Futura+ Monitors User Manual and Reference Guide Version 1.6
3 Modbus Plus Adapter User Manual and Reference Guide Version 1.6 Pub lished by: Elec tro In dus tries/gaugetech 1800 Shames Drive West bury, NY All rights re served. No part of this pub li ca tion may be re pro duced or trans mit ted in any form or by any means, electronic or me chani cal, in clud ing pho to copy ing, re cord ing, or in for ma tion stor age or re trieval sys tems or any fu ture forms of du pli - ca tion, for any pur pose other than the pur chas er s use, with out the ex pressed writ ten per mis sion of Elec tro Industries/GaugeTech Elec tro In dus tries/gaugetech Released November 1998 Printed in the United States of America Modbus and Modbus Plus are registered trademarks of Schneider Automation, Inc.
5 Cus tomer Serv ice and Sup port Cus tomer sup port is avail able 9:00 a.m. to 4:30 p.m., east ern stan dard time, Mon day through Fri day. Please have the model, se rial number and a de tailed prob lem de scrip tion avail able. If the prob lem con cerns a par ticu lar read ing, please have all me ter read ings avail - able. When re turn ing any mer chan dise to EIG, a re turn authori za tion number is re quired. For cus tomer or tech ni cal as sis tance, re pair or cali bra tion, phone or fax Prod uct Warranty Elec tro In dus tries/gaugetech war rants all prod ucts to be free of de fects in ma te rial and work man ship for one year from the date of ship ment. Dur ing the war ranty pe riod, we will, at our op tion, re pair or re place any prod uct that proves to be de fec tive. To ex er cise this war ranty, fax or call our customer- service de part ment. You will re ceive prompt as sis tance. Send the in stru ment, trans por ta tion pre paid, to Elec tro In dus tries/ GaugeTech at 1800 Shames Drive, West bury, NY Re pairs will be made and the instrument will be re turned. Limitation of Warranty This war ranty does not ap ply to de fects re sult ing from un au thor ized modi fi ca tion, mis use, or use for any rea son other than elec tri cal power moni tor ing. OUR PRODUCTS ARE NOT TO BE USED FOR PRIMARY OVER-CURRENT PRO- TECTION. ANY PROTECTION FEATURES IN OUR PRODUCTS ARE TO BE USED FOR ALARM OR SECONDARY PROTECTION ONLY. THIS WARRANTY IS IN LIEU OF ALL OTHER WARRANTIES, EXPRESSED OR IMPLIED, INCLUDING ANY IMPLIED WARRANTY OF MERCHANTABILITY OR FITNESS FOR A PARTICULAR PURPOSE. ELECTRO INDUSTRIES/GAUGETECH SHALL NOT BE LIABLE FOR ANY INDIRECT, SPECIAL OR CONSEQUENTIAL DAMAGES ARISING FROM ANY AUTHORIZED OR UNAUTHORIZED USE OF ANY ELECTRO INDUSTRIES/GAUGETECH PRODUCT. LIABILITY SHALL BE LIMITED TO THE ORIGINAL COST OF THE PRODUCT. State ment of Cali bration Our in stru ments are in spected and tested in ac cor dance with speci fi ca tions pub lished by Electro In dus tries/gaugetech. The ac cu racy and cali bra tion of our in stru ments are trace able to the Na tional Bu reau of Stan dards through equip ment that is cali brated at planned in ter vals by com pari son to cer ti fied stan dards. Disclaimer The in for ma tion pre sented in this pub li ca tion has been care fully checked for re li abil ity; how - ever, no re spon si bil ity is as sumed for in ac cu ra cies. The in for ma tion con tained in this docu - ment is sub ject to change with out no tice. Elec tro In dus tries/gaugetech i.
6 ii. Elec tro In dus tries/gaugetech
7 Electro Industries/ GaugeTech Elec tro In dus tries/gaugetech was founded in 1973 by en gi neer and in ven tor Sam uel Ka gan. Dr. Ka gan s first in no va tion, which revo lu tion ized the power- monitoring field, was the de vel op ment of an af ford able, easy- to- use AC power me ter. In the 1980s, Dr. Ka gan and his team at EIG de vel oped a digi tal mul ti func tion moni tor. This moni tor, with its abil ity to meas ure every as pect of power, trans formed AC power me ter ing and power dis tri bu tion. Un der Dr. Ka gan s lead er ship, EIG again de vel oped a prod uct that sur passed everything else on the mar ket: the Fu tura+ de vice. It sup plied all the func tion al ity of a fault re corder, an event re corder and a data log ger in the con figu ra tion of a sin gle me ter. To day, as a leader in the de vel op ment and pro duc tion of power- monitoring prod ucts, EIG aspires to at tain zero- defect manu fac tur ing. Elec tro In dus tries/gaugetech iii.
8 Products All of EIG s prod ucts are de signed, manu fac tured, tested and cali brated at our fa cil ity in West bury, New York. EIG manu fac tures the most so phis ti cated digi tal power moni tors avail able. Our prod ucts han dle such things as: Mul ti func tion power moni tor ing Power- quality moni tor ing On board data log ging for trending power usage and quality Dis tur bance analy sis EIG manu fac tures both sin gle and mul ti func tion digi tal power moni tors. These utility- grade de vices are highly re li able and so phis ti cated. Futura+ Se ries As the ul ti mate power- quality moni tor, the Fu tura+ is widely used at auto mated sub sta tions. In addition to having nearly all of the capabilities of DM meters, it also handles: Power- quality moni tor ing High- accuracy AC me ter ing On board data log ging On board fault and voltage re cord ing Ten channels of analog outputs DM Se ries DM meters are the sub sta tion stan dard for many utili ties and large in dus tri al companies. These three-phase multifunction monitors meas ure every as pect of power. Watt age, voltage, amperage, var, VA, power factor, frequency and harmonics (%THD) P rotocols: Mod bus, Modbus Plus, DNP 3.0 and Ethernet A na log outputs (0-1 and 4-20mA) iv. Elec tro In dus tries/gaugetech
9 Single- Function Me ters AC volt age and am per age DC volt age and am per age AC watt age Single- phase moni tor ing with maxi mum and mini mum de mands Trans ducer read outs Portable Analyzers Power-quality analysis Energy analysis Elec tro In dus tries/gaugetech v.
10 vi. Elec tro In dus tries/gaugetech
11 Table of Contents Overview Introduction Installation MP1 Installation Meter Installation Operation Operating Modes Normal Mode Program Mode Programming the MP Program-Mode Commands Baud Rate Boot Software Version Echo Global Registers Meter Address Product Code Resetting the MP Appendices Appendix 1: Command Summary Appendix 2: Specifications Appendix 3: Mounting Information Appendix 4: Programming Connections Appendix 5: Standard Connections Elec tro In dus tries/gaugetech vii.
12 viii. Elec tro In dus tries/gaugetech
13 Overview Introduction The Modbus Plus Adapter (MP1) is handy gadget that allows a Futura+ or DMMS 300+ monitor to communicate over a Modbus Plus network. The monitor becomes just another node on the network: It can take full advantage of the speed of the Modbus Plus architecture. A monitor equipped with an MP1 can be polled by other nodes on the same network, or by nodes on adjacent bridged networks. The MP1 can be configured to globally broadcast any 32 registers of information. This rugged device is opto isolated and features an upgradable flash RAM. Use the MP1 to: Con vert Mod bus to Mod bus Plus. Poll me ters for Mod bus Plus global vari ables and pres ent them to a Mod bus Plus network. Program: The meter address. The meter baud rate. The global variables to poll. Read the meter s Modbus Plus address (which is set with its top switches). To program the MP1, all you need is the MP Cable and an RS485 two-wire terminal interface operating at 19.2 kilobaud. Just connect the MP1 between the monitor and the network, and you re off and running. Elec tro In dus tries/gaugetech 1.
14 Installation The Modbus Plus Adapter supports a single meter in register polling and global variables. Use a terminal interface to program the global variables, meter address and baud rate. MP1 Installation Set the MP1 s address to 1 (1,0). Slide the MP1 s PROG/NORM switch to NORM. Plug the 5-pin meter connector cable (MP Cable) into both the meter and the MP1. Attach the DB9-female connector to the Modbus Plus network. Meter Installation Most meters can be used right out of the box, because the required settings are the default values. If necessary: Set the meter address to 1. Set the baud rate to Set the protocol to Modbus RTU. 2. Elec tro In dus tries/gaugetech
15 Operation Operating Modes Normal Mode In Normal mode, the MP1 transfers data between its meter side and its Modbus Plus side. No commands can be executed in Normal mode. Program Mode Program mode is used with a computer to read, set and erase parameters, or check the MP1 s status or boot software version. The functions available in this mode are described below. (See Appendix 1: Command Summary.) In Program mode, you can read, program or erase: The meter s baud rate. The meter s address. The global registers. In Program mode, you can read: The MP1 s product code. The boot software version. Elec tro In dus tries/gaugetech 3.
16 Programming the MP1 To program the MP1: Use the MP Cable to connect the computer s RS232 port to the Modbus Plus Adapter, via a Unicom 2500 (or another RS485-to-RS232 converter). Set the DCE/DTE switch on the Unicom 2500 (or equivalent) to DCE. Turn the RS232 BAUD dial on the MP1 (or Unicom) to Connect the 2-wire RS485 cable from the MP1 (or Unicom) and external power to the 5-wire connector (MP Cable) on the Modbus Plus Adapter, as shown in Appendix 4: Programming Connections. Set the baud rate in the terminal program to 19.2 kilobaud. To enter Program mode: Slide the NORM/PROG switch to PROG. Unplug the 9V AC adapter and wait five seconds. Plug in the 9V AC adapter. Use Windows 95 HyperTerminal or another communications program to select local mode and a baud rate of 19.2 kilobaud. Once you ve finished programming, to reenter Normal mode: Slide the NORM/PROG switch to NORM. Replace the MP Cable (see Appendix 4: Programming Connections) with the 5-wire cable (see Appendix 5: Standard Connections). 4. Elec tro In dus tries/gaugetech
17 Program-Mode Commands Baud Rate Use the following commands to read, program or erase the baud rate in the meter s flash. Erase Baud Rate Type: BE[Enter] B0 will be returned, indicating that the baud rate has been erased. Program Baud Rate Type: BPrate[Enter], where rate is the baud rate (of up to five digits). B0<CR> will be returned, indicating that the baud rate has been programmed. Read Baud Rate Type: BR[Enter] BRrate<CR> will be returned if a baud rate is in the flash, where rate is the baud rate (of up to five digits). NA<CR> will be returned if the flash was not programmed with a baud rate. Boot Software Version Type: Q[Enter] Qn<CR> will be returned, where n is the three-digit version number of the boot software. Echo The echo command is used to determine whether the Modbus Plus Adapter is responding correctly. Type: A[Enter] A<CR> will be returned if the Modbus Plus Adapter is responding correctly. Elec tro In dus tries/gaugetech 5.
18 Global Registers Use the following commands to read, program or erase the global registers in the meter s flash. Erase Global Registers Type: GE[Enter] G0<CR> will be returned, indicating that the data has been erased. Program Global Registers Type: GP02data[Enter], where data is a maximum of 162 characters, delimited by commas in groups of four. For example: GP02,1111,2222[Enter]. G0<CR> will be returned, indicating that the data has been programmed. Read Global Registers Type: GR[Enter] GRrate<CR>data<CR> will be returned if flash has been programmed with global registers, where rate is the baud rate and data is the programmed data, delimited by commas in groups of four. For example: GR02<CR>1111,2222<CR>. NA<CR> will be returned if the flash was not programmed with data. Meter Address Use the following commands to read, program or erase the meter address in the meter s flash. Erase Meter Address Type: ME[Enter] M0<CR> will be returned, indicating that the meter address has been erased. Program Meter Address Type: MPaddress[Enter], where address is a maximum of three digits. M0<CR> will be returned, indicating that the meter address has been programmed. 6. Elec tro In dus tries/gaugetech
19 Read Meter Address Type: MR[Enter] MRaddress<CR>will be returned if a meter address is in the flash, where address is the meter address (of up to three digits). NA<CR> will be returned if the flash was not programmed with a meter address. Product Code This command returns the product code assigned to the Modbus Plus Adapter. Type: E[Enter] Product code<cr> will be returned, where product code is the Modbus Plus Adapter s product code, Modbus Plus Adapter. Resetting the MP1 The Modbus Plus Adapter will automatically reset after five to ten minutes if no serial interrupt has occurred. To manually reset the MP1: Type: K[Enter] K<CR> will be returned, indicating that the Modbus Plus Adapter has been reset. Elec tro In dus tries/gaugetech 7.
20 8. Elec tro In dus tries/gaugetech
21 Appendix 1: Command Summary Description Input Output Baud BE[Enter] B0<CR> Rate: verifies that the baud rate has Erase been erased Baud BPrate[Enter] B0<CR> Rate: rate= the baud rate verifies that the baud rate has Program (of up to five digits) been programmed Baud BR[Enter] BRrate<CR> Rate: rate=the programmed baud rate Read (of up to five digits) NA<CR> indicates no baud rate was programmed Boot Q[Enter] Qn<CR> Software n=the three-digit version number Version of the MP1 s boot software Echo A[Enter] A<CR> verifies that the MP1 is responding properly Global GE[Enter] G0<CR> Registers: verifies that the data has been erased Erase Global GP02data[Enter] G0<CR> Registers: data=up to 162 verifies that the data has been Program characters of data programmed Global GR[Enter] GRrate<CR>data<CR> Registers: rate=the programmed baud rate Read data=the programmed data NA<CR> indicates no data was programmed Elec tro In dus tries/gaugetech 9.
22 Description Input Output Meter ME[Enter] M0<CR> Address: verifies that the meter address has Erase been erased Meter MPaddress[Enter] M0<CR> Address: address=meter address verifies that the meter address has Program (of up to three digits) been programmed Meter MR[Enter] MR[address]<CR> Address: address=the programmed meter address Read (of up to three digits) NA<CR> indicates no meter address was programmed Product E[Enter] product code<cr> Code: product code=the MP1 s factory- Read assigned product code Reset K[Enter] K<CR> verifies that the MP1 has been reset 10. Elec tro In dus tries/gaugetech
23 Appendix 2: Specifications Power Voltage Requirements Maximum Power Consumption Isolation Voltage Input Data Rate LED Indicators Switches Ports and Connectors Enclosure Dimensions Mounting Operating Temperature None (supplied by meter) 3VA 2500V RMS (RS485 to Modbus Plus) ,600 baud POWER ERR (error) TX (transmit) CNTRL (control) RX (receive) ACTIVE (Modbus Plus) Modbus Plus Address (two dials) PROG/NORM (program/normal selection) Modbus Plus (DB9 female) Power (RS485, 2-wire detachable terminal block) Rugged aluminum, resistant to electromagnetic interference Length: (169mm) Width: (117mm) Height: 1 (27mm) Stand alone or wall mount (mounting plate supplied) 20 to +70 C Elec tro In dus tries/gaugetech 11.
24 Appendix 3: Mounting Information 12. Elec tro In dus tries/gaugetech
25 Appendix 4: Programming Connections MP1 Modbus Plus Adapter 9V AC RS485- to-rs232 Converter Power 9V AC DCE DB9 male serial cable DTE DB9 female 9V AC Computer MP Cables Elec tro In dus tries/gaugetech 13.
26 Appendix 5: Standard Connections To Modbus Plus Network MP1 Modbus Plus Adapter 5-Wire Connection (end view of MP1 and monitor) Wire Cable 1 5 Modbus Plus Interface Futura+ or DMMS Elec tro In dus tries/gaugetech
User Manual UNICOM 2500 Installation Operation Programming Universal Communication Converter UNICOM 2500 Electro Industries/GaugeTech 1800 Shames Drive Westbury, New York 11590 Tel 516.334.0870 Fax 516.338.4741
Vissersdijk 4 4301 ND Zierikzee The Netherlands Tel. +31 111 413656 Fax. +31 111 416919 www.deltapowersupplies.com DELTA ELEKTRONIKA B.V. SM 3300 - Interface modules Mod els Description INT MOD M/S Master/Slave
Vissersdijk 4 4301 ND Zierikzee The Netherlands Tel. +31 111 413656 Fax. +31 111 416919 www.deltapowersupplies.com DELTA ELEKTRONIKA B.V. SM 3300 Series 3300 W DC POWER SUP PLIES One or Three phase input
User Manual IC-485AI Note: This equipment has been tested and found to comply ith the limits for a Class A digital device pursuant to Part 15 of the FCC Rules. These limits are designed to provide reasonable
Freeze Protection Thermostats MODELS SST 2 Part Number 22744 MANUAL We Manage Heat Installation and Operation Manual Environmental Technology, Inc. 1850 N Sheridan Street South Bend, Indiana 46628 (574)
Freeze Protection Thermostats MODELS SST 2 Part Number 22744 MANUAL We Manage Heat Installation and Operation Manual Environmental Technology, Inc. 1850 N Sheridan Street South Bend, Indiana 46628 (574)
Restoration Certificate No. 201501 19th Century Alabaster Clock adorned with Cupid, French 8-day Japy Movement with Hourly and Half-Hourly Strike, Alabaster and Gilt Dial Issued by Dr N Bjergstrom, All
Vissersdijk 4 4301 ND Zierikzee The Netherlands Tel. +31 111 413656 Fax. +31 111 416919 www.deltapowersupplies.com DELTA ELEKTRONIKA B.V. ES 150 - Series 150 W DC POWER SUP PLIES Mod els Volt age range
2 To 1 Serial (RS-232) AUTO SWITCH AS-251S User Manual Read this manual thoroughly and follow the installation procedures carefully to prevent any damage to the AS-251S and/or the devices it connects to.
Power-monitoring unit PowerLogic System Technical data sheet 2003 Power Meter Series 800 Function and characteristics PB100313 The PowerLogic Power Meter Series 800 offers all the high-performance measurement
Aquadyne TechTip TITLE: TROUBLESHOOTING PC COM PORT AND MODEM COMMUNICATIONS WITH AN OCTOPUS. Article # 072297-1 Last reviewed: 03/25/98 Keywords: Serial Port, Modem, Polling, AquaWeb, Node Not Responding
P.O. BOX 27 4300 AA ZIERIKZEE NETHERLANDS TEL. +31 111 413656 FAX +31 111 416919 www.deltapowersupplies.com ES150 - se ries 150 watts DC POWER SUP PLIES ES 015-10 0-15 V 0-10 A ES 030-5 0-30 V 0-5 A ES
TCP/IP MODULE CA-ETHR-A INSTALLATION MANUAL w w w. c d v g r o u p. c o m CA-ETHR-A: TCP/IP Module Installation Manual Page Table of Contents Introduction...5 Hardware Components... 6 Technical Specifications...
IntesisBox Server SAMSUNG Air Conditioners Gateway for monitoring and control of Samsung Air Conditioning Systems from any master device TCP or RTU (BMS, PLC, SCADA, HMI, TouchPanel ) Master TCP Master
User Manual Humidity-Temperature Chart Recorder Model RH520 Introduction Congratulations on your purchase of the Extech RH520 Temperature + Humidity Chart Recorder. The RH520 measures and displays Temperature,
SW485 485C RS-232 to RS-485/422 Converter Introduction ntroduction: The SW485C is an industrial grade bi-directional port-powered RS232 to RS485 or RS422 converter in a 9 pin format. It can convert any
Vissersdijk 4 4301 ND Zierikzee The Netherlands Tel. +31 111 413656 Fax. +31 111 416919 www.deltapowersupplies.com DELTA ELEKTRONIKA B.V. SM 6000 - Se ries 6000 W DC POWER SUP PLIES Three phase input Mod
OWNERS MANUAL Status Monitor for Windows 95, 98, ME, NT 4, 2000 & XP SIGNALCRAFTERS TECH, INC. www.signalcrafters.com 57 Eagle Rock Avenue, East Hanover, NJ 07936 Tel: 973-781-0880 or 800-523-5815 Fax:
DX4010V2 EN Installation Instructions RS-232/USB Serial Interface Module DX4010V2 Installation Instructions Trademarks Trademarks BlackBox is either a registered trademark or a trademark of BlackBox Corporation
F2400 FOM II Series Fiber Optic Modem Technical Manual T1 Revision B Copyright April 2003 VERSITRON, Inc. 83 Albe Drive / Suite C Newark, DE 19702 www.versitron.com A030430283T PROPRIETARY DATA All data
IntesisBox ModBus Server - KNX Gateway for integration of KNX equipment into Modbus (RTU and TCP) control systems. Integrate KNX based lighting control into your SCADA, BMS, PLC "talking" Modbus. Master
IntesisBox Server Fidelio IP Gateway for integration of Micros Fidelio Ho management system into (RTU and TCP) enabled control systems. Integrate your Micros Fidelio Ho management system into your master
IPG/7700 Hardware Manual SYSTECH C O R P O R A T I O N Document number 80-001099-7 Revision A Created 2010, and Protected Under the U.S. Copyright Act of 1976. Copyright 2010, SYSTECH Corporation All Rights
APPLICATION Note TELECOM SOLUTIONS FOR THE 21ST CENTURY Using the PB-100 for All-Call Dialing March 22, 2002 Features Compatible with Windows 95, 98, ME, NT 4.0, 2000 Male sub-d 9 pin cable provided 20
FOSTCDR pn 8684R1 FOSTCDR-0812ds page 1/5 Industrial Serial to Multimode Fiber Optic Converter Data Rates up to 115.2 kbps 2.5 Mile (4 km) Range 10 to 30 VDC Input Voltage Wide Operating Temperature 2000V
Data Bulletin 1210DB0002R3/05 03/2005 LaVergne, TN, USA Communications Wiring for POWERLINK G3 Systems Class 1210 Retain for future use. ABOUT THIS BULLETIN This data bulletin describes the proper wiring
08/17/12 Rev2 LTR Series General Specification 700 VA to 2.1 KVA 1.0 General LTR Series Uninterruptible Power Systems 700 VA - 2.1 KVA General Specification This specification describes the features and
Installation Instructions Universal Serial Bus (USB) to DH-485 Interface Converter Catalog Number 1747-UIC Contents Overview..................................................3 Computer and Operating System
Programming Flash Microcontrollers through the Controller Area Network (CAN) Interface Application te Programming Flash Microcontrollers through the Controller Area Network (CAN) Interface Abstract This
12843 Foothill Blvd., Suite D Sylmar, CA 91342 818 898 3380 voice 818 898 3360 fax www.dnfcontrolscom The Analyst RS422/RS232 Tester With VTR, Monitor, and Data Logging Option (LOG2) User Manual Manual
a g l Educa t Le H LSNJ io...justice for all n s a n d b o ok Copyright 2010 Legal Services of New Jersey All rights reserved. No part of this handbook may be reproduced without permission. For further
Keep it Simple Timing Support... 1 Introduction... 2 Turn On and Go... 3 Start Clock for Orienteering... 3 Pre Start Clock for Orienteering... 3 Real Time / Finish Clock... 3 Timer Clock... 4 Configuring
PowerLogic power-monitoring units Power Meter Series 700 Technical data sheet 2007 Functions and characteristics E90463 The PowerLogic Power Meter Series 700 offers all the measurement capabilities required
TM New MINIMAG Magnetic Stripe Reader USB/CDC Interface Quickstart Manual ID TECH 10721 Walker Street Cypress, California 90630 (714) 761-6368 www.idtechproducts.com 80066509-001 Rev. C R04/06 #419 DEFAULT
User's Guide 99 Washington Street Melrose, MA 02176 Phone 781-665-1400 Toll Free 1-800-517-8431 Visit us at www.testequipmentdepot.com Back to the Extech 407780 Product Page Integrating Sound Level Datalogger
to RS422/RS485 Isolated Universal Converter 1. Introduction The converts serial data from the standard to either RS422 or RS485. Features: Universal converter can be used with both and other manufacturer
MPBB1 Uploader user manual 1997-2000 Martin Professional A/S, Denmark. All rights reserved. No part of this manual may be reproduced, in any form or by any means, without permission in writing from Martin
Industrial RS-232/ RS-422/ RS-485 over Ethernet Media Converter Serial Interface One RS-232 port and one RS-422/485 port to 10/Base- TX Media Conversion Cost effective solutions of RS-232/ RS-422/ RS-485
POS 208 / 808 DISTRIBUTED BY INTELLIGENT CASH DRAWER WEDGES MADE IN CANADA USER S MANUAL 2 Table of Contents Introduction............................................... 3 POS 208 Installation........................................
Eaton Innovative Technology PEM PC Software Installation and Operation Instruction Manual: IM01005006E Rev. 2 March 2007 Modbus RTU (RS-485) TCP/IP (Ethernet) Effective Date 3/2007 For more information
IntesisBox KNX TCP master Gateway for integration of TCP slave devices into KNX control systems. Integrate any TCP slave device into KNX. KNX TCP slave EIB Bus IntesisBox Ethernet slave LinkBoxEIB Configuration
Smarthome SELECT Bluetooth Wireless Stereo Audio Receiver and Amplifier INTRODUCTION The Smarthome SELECT Bluetooth Wireless Stereo Audio Receiver and Amplifier is a multi-functional compact device. It
BIT COMMANDER Serial RS232 / RS485 to Ethernet Converter (Part US2000A) Copyrights U.S. Converters 1 Contents Overview and Features... 3 Functions..5 TCP Server Mode... 5 Httpd Client Mode.5 TCP Auto mode....6
MAKING MODERN LIVING POSSIBLE AK-SC255 On-Site Installation Guide DANFOSS ELECTRONIC CONTROLS & SENSORS How to Use This Guide Read this Guide completely as you install and start up your new AK-SC 255 controller.
RSC232 Communication Converter Installation and Operation Manual LIMITED WARRANTY The manufacturer offers the customer a 24-month functional warranty on the instrument for faulty workmanship or parts from
IntesisBox KNX Panasonic Air Conditioners Gateway for Panasonic ECOi and PACi system integration into KNX networks Up to 30 Outdoor units Up to 64 Indoor units PA-AC-KNX-64/128 EIA485 U1U2 line U1U2 line
Data Sheet Process Alarm Monitor Reliable Supervision and Control 10 inputs with LED indications Supports both NO/NC input contacts 10 open collector outputs Built-in siren relay Text label for alarm descriptions
Weatherford s automated systems accessory components are used to expand input/ output (I/O) capacity, meet power requirements, provide communications connectivity, and supply the instrumentation interface
Programming and Using the Courier V.Everything Modem for Remote Operation of DDF6000 1.0 Introduction A Technical Application Note from Doppler System July 5, 1999 Version 3.x of the DDF6000, running version
SE2000 (Part No. 2005457) Serial to Ethernet Converter with VoIP User s Guide 255373A January 2008 i TABLE OF CONTENTS 1 GENERAL DESCRIPTION...1 1.1 INTRODUCTION...1 1.2 FEATURES SUMMARY...1 2 SPECIFICATIONS...2
(19) United States US 20030035525A1 (12) Patent Application Publication (10) Pub. No.: US 2003/0035525 A1 Wu et al. (43) Pub. Date: (54) (76) (21) (22) SYSTEM AND METHOD FOR DISTINGUISHING TELEPHONE NUMBER
Sense Box VOICE RECORDING SYSTEM ISDN PRI / BRI / Analog SUB-SYSTEM ISDN2, ISDN30 Sense Box USER MANUAL TABLE OF CONTENTS 1. STATUTORY INFORMATION... 3 1.1 SAFETY WARNING... 3 1.2 SAFETY EXTRA LOW VOLTAGE
RS-232 to RS-422/485 Converters User Guide Models : SER-COMi-M SER-COMi-SI-M Web: www.titan.tw Support: email@example.com Document Part Number TN-068 Version 1.00 Issue Date: 2011-09-11 The computer programs
Monitoring For Business-Critical Continuity Liebert IntelliSlot Web Cards Firmware Upgrade Manual Liebert IntelliSlot Web Card, Liebert IntelliSlot Web Card-LB, Liebert IntelliSlot Web Card-LBDS, Liebert
WEB log LIGHT+ 20 BASIC 100 PRO unlimited Device connection plans Version 20151210* Copyright Copyright for this manual remains with the manufacturer. No part of this manual may be reproduced or edited,
Vissersdijk 4 4301 ND Zierikzee The Netherlands Tel. +31 111 413656 Fax. +31 111 416919 www.deltapowersupplies.com DELTA ELEKTRONIKA B.V. SM 3000 - Se ries 3000 W DC POWER SUP PLIES Mod els Volt age range
Up dated 12/7/1999 General Ledger User s Guide : 1999 Print ing Copy right 1994 Boy Scouts of Amer ica Irving, Texas No. 80-173 Contents: Contents Con tents......................... 1 In tro duc tion.......................
Technical Manual For use with Caller ID signaling types: Belcore 202, British Telecom, & ETSI Caller ID.com WHOZZ CALLING? POS 2 Caller ID Monitoring Unit Technical Manual For use with Caller ID signaling
CoBox WSE CoBox Master/Slave Terminal Server Setup NSCC 2.0/NSM 2.1 Field Configuration and Installation Guide 47102 Mission Falls Court Fremont, CA 94539-7818 an ISO 9000 certified company PHONE (510)
ICS-2 S15 Industrial RS-232/ RS-422/ RS-485 over Ethernet Media Converter Cost Effective Solution for RS-232 / RS-422 / RS-485 to Industrial Ethernet Application PLANET ICS-210x series Smart Media Converter
DMX USB PRO User Manual www.enttec.com Firmware V1.43 February 2007 Package Contents Your DMX USB PRO package should contain these items: DMX USB PRO (Part No. 70304) Driver for Windows software on the
RS-232 to Converter Click 304 The Click 304 converts half-duplex RS-232 communication to communication and vice versa. It also auto-detects baud rates and features LEDs that light up when data is successfully
IntesisBox KNX SAMSUNG Air Conditioning Gateway for integration of Samsung air conditioners into KNX control systems. Integrate Samsung Air Conditioners into KNX. KNX EIB Bus IntesisBox RS485 MIM-B13x/B04A
IntesisBox Server - Mitsubishi Electric G50 Gateway for monitor and control Mitsubishi Electric City Multi Air Conditioning systems from any master device TCP or RTU (BMS, PLC, SCADA, HMI, TouchPanel )
Table of Contents 1 Introduction...3 Unpacking...4 Warranty Instructions...4 Installation...7 Operation...9 Remote Control Buttons...11 Place a Call...11 End a Call...11 Adjust the Volume in the Room...11
ENET-710 ENET-710 - Ethernet Module JAN / 06 ENET-710 FOUNDATION E N E T 7 1 0 ME smar www.smar.com Specifications and information are subject to change without notice. Up-to-date address information is
Installation Guide Solar Connect-11 Version 1.1 Contents Important Product Information 3 System Registration Form 4 Solar Connect-11 Overview 5 Internet & Power Connections 6 Single Phase CT Connections
Bluetooth to serial HC-06 wireless module Product Description: 1,Mainstream CSR Bluetooth chip, Bluetooth V2.0 protocol standards 2,serial module operating voltage 3.3V. 3,the baud rate for 1200, 2400,
Application : The RISH CON - Hz transducer is used for frequency measurement. The output signal is proportional to measured frequency and is either load independent DC Current or load independent DC Voltage.
PI 1115A 703 5863 Parallel to Serial and Serial to Parallel Converter Product Code PI 1115A Table of Contents Chapter 1: Introduction...3 Packing List...3 Specifications...3 Chapter 2: Parallel to Serial
TM Mico-Switch/8 8 Port 10/100Base-TX Switch (Auto MDI/MDIX) FEP-32008T-3 U SER S M ANUAL (Auto MDI/MDIX) FEP-32008T-3 TM TM Package Contents Package contents include the following: Micro-Switch/8; 8 Port
TECHNICAL DATASHEET #TDAX100200 BLDC Motor Drive Drives a 12V, 24V or 48V BLDC motor Bidirectional, up to 25A Smooth speed control using Hall Sensors CAN (SAE J1939) with Electronic Assistant Features:
CONSOLE CONSOLE REMOTE I /O AC 9V AC 9V REMOTE I /O User Manual CE-220 Read this guide thoroughly and follow the installation and operation procedures carefully in order to prevent any damage to the units
USB to RS-422/485 Serial Adapter (ID-SC0911-S1/SC0A11-S1) User s Manual 1 Copyright Statement No part of this publication may be reproduced in any form by any means without the prior written permission.
Universal I/O EDI Distributor Control Interface Wiring and Setup Instructions EDI UNIVERSAL I/O INTERFACE MODULE The only interface needed for EDI-V5 controls Network compatible with all older EDI controls
E90463 PB101556 The EGX100 serves as an Ethernet gateway for PowerLogic System devices and for any other communicating devices utilizing the Modbus protocol. The EGX100 gateway offers complete access to
Real Life Online Seminars The HEART of Accounting TM A 4-hour course for non-accountants! John W. Day, MBA Real Life Online Sem i nars Pres ents The HEART of Account ing A 4-Hour Step-By-Step Course That