Lecture 14. Point Spread Function (PSF)


 Nancy Jemima Cannon
 2 years ago
 Views:
Transcription
1 Lecture 14 Point Spread Function (PSF), Modulation Transfer Function (MTF), Signaltonoise Ratio (SNR), Contrasttonoise Ratio (CNR), and Receiver Operating Curves (ROC) Point Spread Function (PSF) Recollect that Image restoration refers to the removal or minimization of known degradations in an image This includes deblurring of images degraded by the limitations of the sensor or its environment, noise filtering, and correction of geometric distortions or nonlinearities due to sensors.
2 Point Spread Function (PSF) The figure below shows a typical situation in an imaging system: Point Spread Function (PSF) The image of a point source is blurred and degraded due to noise by an imaging system. If the imaging system is linear, the image of an object can be expressed as: where (x,y) is the additive noise function, f() is the object, g(x,y) is the image, and h(x,y;) is the Point Spread Function (PSF). The ; is used to distinguish the input and output pairs of coordinates in this case.
3 Point Spread Function (PSF) A typical image restoration problem, give the formulation in the previous slide is of the form: Find an estimate of f(a,b) given The Point Spread Function, The blurred image, and, The statistical properties of the noise and/or the factors affecting the noise contribution. PSF The point spread function (PSF) describes the imaging system response to a point input, and is analogous to the impulse response. A point input, represented as a single pixel in the ideal image, will be reproduced as something other than a single pixel in the real image.
4 PSF A Point Source PSF The PSF need not be isotropic (radially symmetric). In ultrasound, Xray CT, and radionuclide tomography it is typical to have a nonisotropic PSF for various physical reasons. In MRI it is possible to have either isotropic or nonisotropic PSF depending on the type of acquisition since spatial frequency coverage can be different for the two inplane directions.
5 PSF The output image may then be regarded as a twodimensional convolution of the ideal image with the PSF: g 2 =g 1 h where h is the impulse response, or PSF. NOTE: Both * and ** are used to represent convolution In some medical imaging systems (e.g. planar X ray) the PSF can vary gradually over the field of view. In this case it is convenient to define a zone of constant PSF (isoplanatic region) to allow use of convolutional forms and the transform domain.
6 Isotropic PSFs Anisotropic PSFs PSF Optical (e.g. microscopy) also manifest asymmetric point spread functions due to lens imperfections (both material and geometry). It is typical for the PSF to degrade as distance from the center of the FOV is increased.
7 PSF: Examples From Fundamentals of Digital Image Processing by A.K. Jain Modulation Transfer Function (MTF) Another measure of system performance is the modulation transfer function (MTF). This is analogous to the frequency response typically used for onedimensional applications. In the case of a complex transfer function, MTF is usually expressed as the magnitude portion of the function. MTF allows for simplified descriptions of an imaging system s spatial resolution capabilities.
8 MTF PSF expresses system performance in the spatial domain, while MTF expresses system performance in the frequency domain. The two parameters are related by the Fourier transform: PSF MTF PSF 1 MTF MTF and OTF For spatially invariant imaging systems, the Optical Transfer Function (OTF) is defined as the normalized frequency response i.e., OTF = H( 1, 2 )/H(0,0) The Modulation Transfer Function (MTF) is defined as the magnitude of the OTF, i.e., MTF = OTF = H( 1, 2 ) / H(0,0)
9 MTF: Examples MTFs of the PSFs displayed earlier From Fundamentals of Digital Image Processing by A.K. Jain MTF Consider the discrete convolutional representation of a blurring function: a b n, m sn a, mbh a b x, Where x is the blurry image, s is the ideal image, and h is the point spread function.
10 MTF The DFT of this PSF is given by: n0 m0, v hn, H u N N u 1,, 2 2 M M v 1,, 2 2 m e un um 2i N M MTF and Frequency Response The coefficients of H(u,v) are those for plane waves of various frequencies and orientations. These are the spatial frequency components that exactly represent the PSF (blurring function). H(u,v) is the transfer function, also referred to as the frequency response. Examination of the magnitude H(u,v) allows for determination of limiting spatial resolution.
11 A Numerical Example Consider this example: a 3x3 blurring kernel typical of what has been seen before: A Numerical Example The transform is given by: u 2v 2u 2v, v 34cos 4cos 4cos cos H u N M N M N=M=33
12 2D Plots of the example Obtaining the MTF from a kernel PSF To obtain the MTF of a PSF represented as a kernel: Apply a scaling factor (= (array elements)) if you want a normalized (range 01) MTF plot. Embed the kernel in the center of an array of zeros equal to the size of the image to which it is to be applied. fftshift(abs(fft2( ))) Result will be 2D array representing the spatial frequency version of the blurring/filter function.
13 MTF As with systems seen in onedimensional applications, the MTF of imaging systems typically demonstrate a rolloff (like a lowpass filter) with higher spatial frequencies. An MTF curve is typically used to express the spatial frequency response of a system, and expresses normalized contrast as a function of spatial frequency (expressed in units of inverse length, e.g. mm 1 ). Signaltonoise Ratio (SNR) Signaltonoise ratio is an engineering term for the power ratio between a signal (meaningful information) and the background noise: Because many signals have a very wide dynamic range, SNRs are usually expressed in terms of the logarithmic decibel scale.
14 SNR In decibels, the SNR is 20 times the base10 logarithm of the amplitude ratio, or 10 times the logarithm of the power ratio: where P is average power and A is RMS amplitude. Both signal and noise power are measured within the system bandwidth. SNR SNR is usually taken to indicate an average signal to noise ratio, as it is possible that (near) instantaneous signal to noise ratios will be considerably different. In general, higher signal to noise is better. (i.e. cleaner.) In image processing, the SNR of an image is usually defined as the ratio of the mean pixel value to the standard deviation of the pixel values. Related measures are the "contrast ratio" and the "contrast to noise ratio".
15 SNR In the case of MRI, the noise is distributed uniformly throughout the image. The SNR can be measured by computing the mean signal intensity over a certain region of interest (ROI) and dividing this by the standard deviation of the signal from a region outside the image. In other modalities, this is not true; the noise is not uniformly distributed over the image. As a consequence, other methods must be used to estimate the SNR. Contrasttonoise Ratio (CNR) CNR is a measure for assessing the ability of an imaging procedure to generate clinically useful image contrast. The image contrast itself is not precise enough to qualify an image, because in a noisy image it is unclear where the contrast originates. It may be due to true tissue contrast, or it may be due to noise fluctuations. The human ability to distinguish between objects is proportional to contrast, and it decreases linearly with noise.
16 CNR These laws of perception are taken into account by the definition of the contrast to noise ratio CNR = contrast/noise. Therefore the CNR gives an objective measure of useful contrast. For instance, if an acquisition technique generates images with twice the contrast of those produced by another technique, the noise must increase less than twice in order to provide clinically better images. CNR A practical problem of CNR definition is that it relies on the measurement of the photon flux; this depends upon the display system and is difficult to perform. An equivalent yet much more feasible approach is to use the signal difference in the original data instead of assessing the contrast of the displayed image.
17 CNR Even if the image has a high signaltonoise ratio, it is not useful unless there is a high enough CNR to be able to distinguish among different tissues and tissue types, and in particular between healthy and pathological tissue. Various definitions of image contrast exist, but the most common is: C AB = S A S B where C AB is the contrast between tissues A and B and S A and S B are the signals from tissues A and B. CNR The CNR between two tissues is defined in terms of their respective signal noisetoratios of the two tissues: CNR AB = C AB / N = S A S B / N = SNR A SNR B where N is the standard deviation of the noise. * Note that this is analogous to the Signal difference to noise ratio
18 Receiver Operating Curve (ROC) in imagingbased diagnoses There are four possibilities for a practitioner making a diagnosis: a true positive (where true refers to a correct diagnosis and positive refers to say a tumor being present), a true negative, a false positive, and, a false negative. ROC plot for tumor diagnosis ROC Figure (Left) A table showing the four possible outcomes of a tumor diagnosis. (Right) The ROC represented by the dashed line represents a random diagnosis. The upper curve represents an improved diagnosis. The better the diagnosis, the larger is the integrated area under the ROC.
19 ROC The ROC plots the fraction of true positives versus the fraction of false positives for a series of images acquired under different conditions, or with a different value of some parameter, or with different SNRs, or with different practitioners, for example. The area under the ROC is a measure of the effectiveness of the imaging system and/or the practitioner. The greater the area under the curve the more effective is the diagnosis. ROC There are three measures commonly used in ROC analysis: 1. The accuracy is the correct number of diagnoses divided by the total number of diagnoses 2. The sensitivity is the number of true positives divided by the sum of the true positives and false negatives 3. The specificity is the number of true negatives divided by the number of true negatives and false positives
The Fundamentals of MTF, Wiener Spectra, and DQE. Motivation
The Fundamentals of MTF, Wiener Spectra, and DQE Robert M Nishikawa Kurt Rossmann Laboratories for Radiologic Image Research Department of Radiology, The University of Chicago Motivation Goal of radiology:
More informationImage Enhancement: Frequency domain methods
Image Enhancement: Frequency domain methods The concept of filtering is easier to visualize in the frequency domain. Therefore, enhancement of image f ( m, n) can be done in the frequency domain, based
More informationLectures 6&7: Image Enhancement
Lectures 6&7: Image Enhancement Leena Ikonen Pattern Recognition (MVPR) Lappeenranta University of Technology (LUT) leena.ikonen@lut.fi http://www.it.lut.fi/ip/research/mvpr/ 1 Content Background Spatial
More informationSignal to Noise Instrumental Excel Assignment
Signal to Noise Instrumental Excel Assignment Instrumental methods, as all techniques involved in physical measurements, are limited by both the precision and accuracy. The precision and accuracy of a
More informationPersonal Identity Verification (PIV) IMAGE QUALITY SPECIFICATIONS FOR SINGLE FINGER CAPTURE DEVICES
Personal Identity Verification (PIV) IMAGE QUALITY SPECIFICATIONS FOR SINGLE FINGER CAPTURE DEVICES 1.0 SCOPE AND PURPOSE These specifications apply to fingerprint capture devices which scan and capture
More informationImage Enhancement in the Frequency Domain
Image Enhancement in the Frequency Domain Jesus J. Caban Outline! Assignment #! Paper Presentation & Schedule! Frequency Domain! Mathematical Morphology %& Assignment #! Questions?! How s OpenCV?! You
More informationConvolution, Noise and Filters
T H E U N I V E R S I T Y of T E X A S Convolution, Noise and Filters Philip Baldwin, Ph.D. Department of Biochemistry Response to an Entire Signal The response of a system with impulse response h(t) to
More informationCHAPTER 6 Frequency Response, Bode Plots, and Resonance
ELECTRICAL CHAPTER 6 Frequency Response, Bode Plots, and Resonance 1. State the fundamental concepts of Fourier analysis. 2. Determine the output of a filter for a given input consisting of sinusoidal
More informationCharacterization of the Xray eye
Characterization of the Xray eye Report by Willem Rischau Friedrich Schiller University Jena, Germany DESY Summer School 2009 21.07.200910.09.2009 1 Introduction Detectors for Xray imaging can be realised
More informationSignal Amplitude and Head Geometry Measurement Accuracy Delivered by Digital and Analog Methods
Signal Amplitude and Head Geometry Measurement Accuracy Delivered by Digital and Analog Methods 1. Introduction... 2 2. Influence of Noise on Signal Amplitude Measurements... 4 2.1. Measurement Setup...
More informationMICROPHONE SPECIFICATIONS EXPLAINED
Application Note AN1112 MICROPHONE SPECIFICATIONS EXPLAINED INTRODUCTION A MEMS microphone IC is unique among InvenSense, Inc., products in that its input is an acoustic pressure wave. For this reason,
More informationCHAPTER 3: DIGITAL IMAGING IN DIAGNOSTIC RADIOLOGY. 3.1 Basic Concepts of Digital Imaging
Physics of Medical XRay Imaging (1) Chapter 3 CHAPTER 3: DIGITAL IMAGING IN DIAGNOSTIC RADIOLOGY 3.1 Basic Concepts of Digital Imaging Unlike conventional radiography that generates images on film through
More informationModule 4. Contents. Digital Filters  Implementation and Design. Signal Flow Graphs. Digital Filter Structures. FIR and IIR Filter Design Techniques
Module 4 Digital Filters  Implementation and Design Digital Signal Processing. Slide 4.1 Contents Signal Flow Graphs Basic filtering operations Digital Filter Structures Direct form FIR and IIR filters
More informationImage Processing with. ImageJ. Biology. Imaging
Image Processing with ImageJ 1. Spatial filters Outlines background correction image denoising edges detection 2. Fourier domain filtering correction of periodic artefacts 3. Binary operations masks morphological
More informationSGN1158 Introduction to Signal Processing Test. Solutions
SGN1158 Introduction to Signal Processing Test. Solutions 1. Convolve the function ( ) with itself and show that the Fourier transform of the result is the square of the Fourier transform of ( ). (Hints:
More information5.4 CT Number Accuracy and Noise
186 Chapter 5 5.4 CT Number Accuracy and Noise In many clinical practices, radiologists rely on the value of measured CT numbers to differentiate healthy tissue from disease pathology. Although this practice
More informationImage Processing Introduction and Overview
Image Processing Introduction and Overview Prof. Eric Miller elmiller@ece.tufts.edu Fall 2007 EN 74ECE Image Processing Lecture 11 Today s Lecture All the dull administrative stuff first Introduction
More informationDigital Filter Design (FIR) Using Frequency Sampling Method
Digital Filter Design (FIR) Using Frequency Sampling Method Amer Ali Ammar Dr. Mohamed. K. Julboub Dr.Ahmed. A. Elmghairbi Electric and Electronic Engineering Department, Faculty of Engineering Zawia University
More informationUSE OF SYNCHRONOUS NOISE AS TEST SIGNAL FOR EVALUATION OF STRONG MOTION DATA PROCESSING SCHEMES
USE OF SYNCHRONOUS NOISE AS TEST SIGNAL FOR EVALUATION OF STRONG MOTION DATA PROCESSING SCHEMES Ashok KUMAR, Susanta BASU And Brijesh CHANDRA 3 SUMMARY A synchronous noise is a broad band time series having
More informationCSE168 Computer Graphics II, Rendering. Spring 2006 Matthias Zwicker
CSE168 Computer Graphics II, Rendering Spring 2006 Matthias Zwicker Last time Sampling and aliasing Aliasing Moire patterns Aliasing Sufficiently sampled Insufficiently sampled [R. Cook ] Fourier analysis
More informationComputer Vision & Digital Image Processing
Intensity Transformations and Spatial Filtering Basics Computer Vision & Digital Image Processing Intensity Transformations and Spatial Filtering Operations take place in the spatial domain Operate directly
More informationFILTER CIRCUITS. A filter is a circuit whose transfer function, that is the ratio of its output to its input, depends upon frequency.
FILTER CIRCUITS Introduction Circuits with a response that depends upon the frequency of the input voltage are known as filters. Filter circuits can be used to perform a number of important functions in
More informationMoving Average Filters
CHAPTER 15 Moving Average Filters The moving average is the most common filter in DSP, mainly because it is the easiest digital filter to understand and use. In spite of its simplicity, the moving average
More informationAliasing, Image Sampling and Reconstruction
Aliasing, Image Sampling and Reconstruction Recall: a pixel is a point It is NOT a box, disc or teeny wee light It has no dimension It occupies no area It can have a coordinate More than a point, it is
More informationDigital Camera Imaging Evaluation
Digital Camera Imaging Evaluation Presenter/Author J Mazzetta, Electro Optical Industries Coauthors Dennis Caudle, Electro Optical Industries Bob Wageneck, Electro Optical Industries Contact Information
More informationAdvantages of High Resolution in High Bandwidth Digitizers
Advantages of High Resolution in High Bandwidth Digitizers Two of the key specifications of digitizers are bandwidth and amplitude resolution. These specifications are not independent  with increasing
More informationToday. next two weeks
Today Temporal and spatial coherence Spatially incoherent imaging The incoherent PSF The Optical Transfer Function (OTF) and Modulation Transfer Function (MTF) MTF and contrast comparison of spatially
More informationTCOM 370 NOTES 994 BANDWIDTH, FREQUENCY RESPONSE, AND CAPACITY OF COMMUNICATION LINKS
TCOM 370 NOTES 994 BANDWIDTH, FREQUENCY RESPONSE, AND CAPACITY OF COMMUNICATION LINKS 1. Bandwidth: The bandwidth of a communication link, or in general any system, was loosely defined as the width of
More informationOptical Fibres. Introduction. Safety precautions. For your safety. For the safety of the apparatus
Please do not remove this manual from from the lab. It is available at www.cm.ph.bham.ac.uk/y2lab Optics Introduction Optical fibres are widely used for transmitting data at high speeds. In this experiment,
More informationSignal Processing First Lab 08: Frequency Response: Bandpass and Nulling Filters
Signal Processing First Lab 08: Frequency Response: Bandpass and Nulling Filters PreLab and WarmUp: You should read at least the PreLab and Warmup sections of this lab assignment and go over all exercises
More informationUntangling the megapixel lens myth! Which is the best lens to buy? And how to make that decision!
Untangling the megapixel lens myth! Which is the best lens to buy? And how to make that decision! 1 In this presentation We are going to go over lens basics Explain figures of merit of lenses Show how
More informationVCO Phase noise. Characterizing Phase Noise
VCO Phase noise Characterizing Phase Noise The term phase noise is widely used for describing short term random frequency fluctuations of a signal. Frequency stability is a measure of the degree to which
More informationT = 1 f. Phase. Measure of relative position in time within a single period of a signal For a periodic signal f(t), phase is fractional part t p
Data Transmission Concepts and terminology Transmission terminology Transmission from transmitter to receiver goes over some transmission medium using electromagnetic waves Guided media. Waves are guided
More informationIntroduction to Digital Filters
CHAPTER 14 Introduction to Digital Filters Digital filters are used for two general purposes: (1) separation of signals that have been combined, and (2) restoration of signals that have been distorted
More informationInterferometric Dispersion Measurements
Application Note 2004022A Interferometric Dispersion Measurements Overview This application note describes how the dbm 2355 Dispersion Measurement Module operates. Summary There are two primary methods
More informationDADiSP / Octave. 1/N Octave Analysis Module. DSP Development Corporation KEY FEATURES. Toll Free: 18004243131
DADiSP / Octave 1/N Octave Analysis Module DADiSP/Octave is a menu driven module designed to simplify the task of octave, 1/3 octave and 1/N octave analysis used to characterize acoustic energy in a wide
More informationPCM Encoding and Decoding:
PCM Encoding and Decoding: Aim: Introduction to PCM encoding and decoding. Introduction: PCM Encoding: The input to the PCM ENCODER module is an analog message. This must be constrained to a defined bandwidth
More informationGregory Hollows Director, Machine Vision Solutions Edmund Optics
Imaging Optics Fundamentals Gregory Hollows Director, Machine Vision Solutions Edmund Optics Topics for Discussion Fundamental Parameters of your system Field of View Working Distance Sensor Sizes Understanding
More informationComputer Vision: Filtering
Computer Vision: Filtering Raquel Urtasun TTI Chicago Jan 10, 2013 Raquel Urtasun (TTIC) Computer Vision Jan 10, 2013 1 / 82 Today s lecture... Image formation Image Filtering Raquel Urtasun (TTIC) Computer
More informationIIR Filter design (cf. Shenoi, 2006)
IIR Filter design (cf. Shenoi, 2006) The transfer function of the IIR filter is given by Its frequency responses are (where w is the normalized frequency ranging in [ π, π]. When a and b are real, the
More informationMODULATION TRANSFER FUNCTION MEASUREMENT METHOD AND RESULTS FOR THE ORBVIEW3 HIGH RESOLUTION IMAGING SATELLITE
MODULATION TRANSFER FUNCTION MEASUREMENT METHOD AND RESULTS FOR THE ORBVIEW3 HIGH RESOLUTION IMAGING SATELLITE K. Kohm ORBIMAGE, 1835 Lackland Hill Parkway, St. Louis, MO 63146, USA kohm.kevin@orbimage.com
More informationIntroduction to Digital Audio
Introduction to Digital Audio Before the development of highspeed, lowcost digital computers and analogtodigital conversion circuits, all recording and manipulation of sound was done using analog techniques.
More informationUnderstand the principles of operation and characterization of digital filters
Digital Filters 1.0 Aim Understand the principles of operation and characterization of digital filters 2.0 Learning Outcomes You will be able to: Implement a digital filter in MATLAB. Investigate how commercial
More informationRightMark Audio Analyzer
RightMark Audio Analyzer Version 2.5 2001 http://audio.rightmark.org Tests description 1 Contents FREQUENCY RESPONSE TEST... 2 NOISE LEVEL TEST... 3 DYNAMIC RANGE TEST... 5 TOTAL HARMONIC DISTORTION TEST...
More information7. Lecture. Image restoration: Spatial domain
7. Lecture Image restoration: Spatial domain 1 Example: Movie restoration ² Very popular  digital remastering of old movies ² e.g. Limelight from Joanneum Research 2 Example: Scan from old film 3 Example:
More informationSpectral Imaging by Upconversion
35 Spectral Imaging by Upconversion Jeppe Seidelin Dam 1, Christian Pedersen 1 and Peter TidemandLichtenberg 1, 1 DTU Fotonik, Deparment of Photonics Engineering, Frederiksborgvej 399, 4000 Roskilde,
More informationFrequency domain filtering fundamentals
Frequency domain filtering fundamentals by Gleb V. Tcheslavski: gleb@ee.lamar.edu http://ee.lamar.edu/gleb/dip/index.htm Spring 2008 ELEN 4304/5365 DIP 1 Preliminaries For a digital image f(x,y) the basic
More informationSIGNAL PROCESSING & SIMULATION NEWSLETTER
1 of 10 1/25/2008 3:38 AM SIGNAL PROCESSING & SIMULATION NEWSLETTER Note: This is not a particularly interesting topic for anyone other than those who ar e involved in simulation. So if you have difficulty
More informationRobot Perception Continued
Robot Perception Continued 1 Visual Perception Visual Odometry Reconstruction Recognition CS 685 11 Range Sensing strategies Active range sensors Ultrasound Laser range sensor Slides adopted from Siegwart
More informationJitter Measurements in Serial Data Signals
Jitter Measurements in Serial Data Signals Michael Schnecker, Product Manager LeCroy Corporation Introduction The increasing speed of serial data transmission systems places greater importance on measuring
More informationComputer Graphics. Course SS 2007 Antialiasing. computer graphics & visualization
Computer Graphics Course SS 2007 Antialiasing How to avoid spatial aliasing caused by an undersampling of the signal, i.e. the sampling frequency is not high enough to cover all details Supersampling 
More informationDepartment of Electronics and Communication Engineering 1
DHANALAKSHMI COLLEGE OF ENGINEERING, CHENNAI DEPARTMENT OF ELECTRONICS AND COMMUNICATION ENGINEERING III Year ECE / V Semester EC 6502 PRINCIPLES OF DIGITAL SIGNAL PROCESSING QUESTION BANK Department of
More informationProcess Control and Instrumentation Prof. D. Sarkar Department of Chemical Engineering Indian Institute of Technology, Kharagpur
Process Control and Instrumentation Prof. D. Sarkar Department of Chemical Engineering Indian Institute of Technology, Kharagpur Lecture  38 Instrumentation: General Principle of Measurement Systems (Contd.)
More informationBlind Deconvolution of Corrupted Barcode Signals
Blind Deconvolution of Corrupted Barcode Signals Everardo Uribe and Yifan Zhang Advisors: Ernie Esser and Yifei Lou Interdisciplinary Computational and Applied Mathematics Program University of California,
More informationModule 6: Pinhole camera model Lecture 30: Intrinsic camera parameters, Perspective projection using homogeneous coordinates
The Lecture Contains: Pinhole camera model 6.1 Intrinsic camera parameters A. Perspective projection using homogeneous coordinates B. Principalpoint offset C. Imagesensor characteristics file:///d /...0(Ganesh%20Rana)/MY%20COURSE_Ganesh%20Rana/Prof.%20Sumana%20Gupta/FINAL%20DVSP/lecture%2030/30_1.htm[12/31/2015
More informationDiffusione e perfusione in risonanza magnetica. E. Pagani, M. Filippi
Diffusione e perfusione in risonanza magnetica E. Pagani, M. Filippi DWMRI DIFFUSIONWEIGHTED MRI Principles Diffusion results from a microspic random motion known as Brownian motion THE RANDOM WALK How
More informationDigital Imaging and Multimedia. Filters. Ahmed Elgammal Dept. of Computer Science Rutgers University
Digital Imaging and Multimedia Filters Ahmed Elgammal Dept. of Computer Science Rutgers University Outlines What are Filters Linear Filters Convolution operation Properties of Linear Filters Application
More informationSignal Processing for Intelligent Sensor Systems with MATLAB 9
Signal Processing for Intelligent Sensor Systems with MATLAB 9 Second Edition David С Swanson @ CRC Press Taylor & Francis Gro jroup Boca Raton London New York CRC Press is an imprint of the Taylor & Francis
More informationTTT4120 Digital Signal Processing Suggested Solution to Exam Fall 2008
Norwegian University of Science and Technology Department of Electronics and Telecommunications TTT40 Digital Signal Processing Suggested Solution to Exam Fall 008 Problem (a) The input and the inputoutput
More informationHigh School Algebra 1 Common Core Standards & Learning Targets
High School Algebra 1 Common Core Standards & Learning Targets Unit 1: Relationships between Quantities and Reasoning with Equations CCS Standards: Quantities NQ.1. Use units as a way to understand problems
More informationElectromagnetic Spectrum
Introduction 1 Electromagnetic Spectrum The electromagnetic spectrum is the distribution of electromagnetic radiation according to energy, frequency, or wavelength. The electromagnetic radiation can be
More informationOptical Versus Video Magnification Primer
Optical Versus Video Magnification Primer Customers have requested additional clarification of true magnifications for inspection needs and quality procedures. This is certainly a valid requirement as
More informationPerformance of an IF sampling ADC in receiver applications
Performance of an IF sampling ADC in receiver applications David Buchanan Staff Applications Engineer Analog Devices, Inc. Introduction The concept of direct intermediate frequency (IF) sampling is not
More informationAnalog Representations of Sound
Analog Representations of Sound Magnified phonograph grooves, viewed from above: The shape of the grooves encodes the continuously varying audio signal. Analog to Digital Recording Chain ADC Microphone
More informationFilter Comparison. Match #1: Analog vs. Digital Filters
CHAPTER 21 Filter Comparison Decisions, decisions, decisions! With all these filters to choose from, how do you know which to use? This chapter is a headtohead competition between filters; we'll select
More informationSynthetic Sensing: Proximity / Distance Sensors
Synthetic Sensing: Proximity / Distance Sensors MediaRobotics Lab, February 2010 Proximity detection is dependent on the object of interest. One size does not fit all For noncontact distance measurement,
More informationPowerTeaching i3: Algebra I Mathematics
PowerTeaching i3: Algebra I Mathematics Alignment to the Common Core State Standards for Mathematics Standards for Mathematical Practice and Standards for Mathematical Content for Algebra I Key Ideas and
More informationMPEG, the MP3 Standard, and Audio Compression
MPEG, the MP3 Standard, and Audio Compression Mark ilgore and Jamie Wu Mathematics of the Information Age September 16, 23 Audio Compression Basic Audio Coding. Why beneficial to compress? Lossless versus
More informationThis unit will lay the groundwork for later units where the students will extend this knowledge to quadratic and exponential functions.
Algebra I Overview View unit yearlong overview here Many of the concepts presented in Algebra I are progressions of concepts that were introduced in grades 6 through 8. The content presented in this course
More informationModule 13 : Measurements on Fiber Optic Systems
Module 13 : Measurements on Fiber Optic Systems Lecture : Measurements on Fiber Optic Systems Objectives In this lecture you will learn the following Measurements on Fiber Optic Systems Attenuation (Loss)
More informationECGAmplifier. MB Jass 2009 Daniel Paulus / Thomas Meier. Operation amplifier (opamp)
ECGAmplifier MB Jass 2009 Daniel Paulus / Thomas Meier Operation amplifier (opamp) Properties DCcoupled High gain electronic ec c voltage amplifier Inverting / noninverting input and single output
More informationFourier Transform and Image Filtering. CS/BIOEN 6640 Lecture Marcel Prastawa Fall 2010
Fourier Transform and Image Filtering CS/BIOEN 6640 Lecture Marcel Prastawa Fall 2010 The Fourier Transform Fourier Transform Forward, mapping to frequency domain: Backward, inverse mapping to time domain:
More informationMATLAB for Audio Signal Processing. MATLAB for Audio Signal Processing. Aliasing. Getting data into the computer
MATLAB for Audio Signal Processing P. Professorson UT Arlington Night School MATLAB for Audio Signal Processing Getting real world data into your computer Analysis based on frequency content Fourier analysis
More informationLecture 3: Quantization Effects
Lecture 3: Quantization Effects Reading: 6.76.8. We have so far discussed the design of discretetime filters, not digital filters. To understand the characteristics of digital filters, we need first
More informationUnderstanding astigmatism Spring 2003
MAS450/854 Understanding astigmatism Spring 2003 March 9th 2003 Introduction Spherical lens with no astigmatism Crossed cylindrical lenses with astigmatism Horizontal focus Vertical focus Plane of sharpest
More informationMatrix, Gradients & Signals Image formation & kpace
Outline 6.1 MR Image Formation, s & Signals Image formation & kpace Imaging s & Signals, and Amplitude selection encoding encoding Carolyn Kaut Roth, RT (R)(MR)(CT)(M)(CV) FSMRT CEO Imaging Education
More informationAVR127: Understanding ADC Parameters. Introduction. Features. Atmel 8bit and 32bit Microcontrollers APPLICATION NOTE
Atmel 8bit and 32bit Microcontrollers AVR127: Understanding ADC Parameters APPLICATION NOTE Introduction This application note explains the basic concepts of analogtodigital converter (ADC) and the
More informationL9: Cepstral analysis
L9: Cepstral analysis The cepstrum Homomorphic filtering The cepstrum and voicing/pitch detection Linear prediction cepstral coefficients Mel frequency cepstral coefficients This lecture is based on [Taylor,
More informationThis is the 23 rd lecture on DSP and our topic for today and a few more lectures to come will be analog filter design. (Refer Slide Time: 01:1301:14)
Digital Signal Processing Prof. S.C. Dutta Roy Department of Electrical Electronics Indian Institute of Technology, Delhi Lecture  23 Analog Filter Design This is the 23 rd lecture on DSP and our topic
More informationIntroduction to Inverse Problems (2 lectures)
Introduction to Inverse Problems (2 lectures) Summary Direct and inverse problems Examples of direct (forward) problems Deterministic and statistical points of view Illposed and illconditioned problems
More informationRoundoff Noise in IIR Digital Filters
Chapter 16 Roundoff Noise in IIR Digital Filters It will not be possible in this brief chapter to discuss all forms of IIR (infinite impulse response) digital filters and how quantization takes place in
More informationDigital vs. Analog Transmission Nyquist and Shannon Laws
1 Digital vs. Analog Transmission Nyquist and Shannon Laws Required reading: Garcia 3.1 to 3.5 CSE 3213, Fall 2010 Instructor: N. Vlajic Transmission Impairments 2 Transmission / Signal Impairments caused
More informationDesign of High Pixel Mobile Phone Camera Lens
Research Journal of Applied Sciences, Engineering and Technology 6(7): 11601165, 2013 ISSN: 20407459; eissn: 20407467 Maxwell Scientific Organization, 2013 Submitted: January 19, 2013 Accepted: February
More informationTechnical Application Note
Technical Application Note Technical Application Note TAN2010002 Revised August 19, 2015 1.1 Subject Technical Application Note (TAN2010002): 1.2 Applicable Product(s) All imaging cameras 1.3 Application
More informationOptimizing IP3 and ACPR Measurements
Optimizing IP3 and ACPR Measurements Table of Contents 1. Overview... 2 2. Theory of Intermodulation Distortion... 2 3. Optimizing IP3 Measurements... 4 4. Theory of Adjacent Channel Power Ratio... 9 5.
More informationHuman visual perception  topics. Anatomy of the human eye
Human visual perception  topics Visual acuity WeberFechner Law Lateral inhibition and excitation Transfer functions of the color channels Spatial and temporal masking Eye movements Bernd Girod: EE368b
More informationCS263: Wireless Communications and Sensor Networks
CS263: Wireless Communications and Sensor Networks Matt Welsh Lecture 2: RF Basics and Signal Encoding September 22, 2005 2005 Matt Welsh Harvard University 1 Today's Lecture Basics of wireless communications
More informationCBS RECORDS PROFESSIONAL SERIES CBS RECORDS CD1 STANDARD TEST DISC
CBS RECORDS PROFESSIONAL SERIES CBS RECORDS CD1 STANDARD TEST DISC 1. INTRODUCTION The CBS Records CD1 Test Disc is a highly accurate signal source specifically designed for those interested in making
More informationFIR Filter Design. FIR Filters and the zdomain. The zdomain model of a general FIR filter is shown in Figure 1. Figure 1
FIR Filters and the Domain FIR Filter Design The domain model of a general FIR filter is shown in Figure. Figure Each  box indicates a further delay of one sampling period. For example, the input to
More informationPassive Optical Resonators
Passive Optical Resonators Optical Cavities and Feedback Back mirror I 0 I 1 Laser medium with gain, G Output mirror I 3 R = 100% R < 100% I 2 Cavities are essential components of the lasers. They provide
More informationThe Effective Number of Bits (ENOB) of my R&S Digital Oscilloscope Technical Paper
The Effective Number of Bits (ENOB) of my R&S Digital Oscilloscope Technical Paper Products: R&S RTO1012 R&S RTO1014 R&S RTO1022 R&S RTO1024 This technical paper provides an introduction to the signal
More informationImplementation of Digital Signal Processing: Some Background on GFSK Modulation
Implementation of Digital Signal Processing: Some Background on GFSK Modulation Sabih H. Gerez University of Twente, Department of Electrical Engineering s.h.gerez@utwente.nl Version 4 (February 7, 2013)
More informationChapter 1: Machine Vision Systems & Image Processing
Chapter 1: Machine Vision Systems & Image Processing 1.0 Introduction While other sensors, such as proximity, touch, and force sensing play a significant role in the improvement of intelligent systems,
More informationTheory From the diffraction pattern to the distribution size
Theory From the diffraction pattern to the distribution size 1 Principle This method is based on diffraction and diffusion phenomenon. To obtain the particle size Fraunhofer and Mie theory are used. When
More informationResolution Enhancement of Photogrammetric Digital Images
DICTA2002: Digital Image Computing Techniques and Applications, 2122 January 2002, Melbourne, Australia 1 Resolution Enhancement of Photogrammetric Digital Images John G. FRYER and Gabriele SCARMANA
More informationFast Fourier Transforms and Power Spectra in LabVIEW
Application Note 4 Introduction Fast Fourier Transforms and Power Spectra in LabVIEW K. Fahy, E. Pérez Ph.D. The Fourier transform is one of the most powerful signal analysis tools, applicable to a wide
More informationIntroduction to IQdemodulation of RFdata
Introduction to IQdemodulation of RFdata by Johan Kirkhorn, IFBT, NTNU September 15, 1999 Table of Contents 1 INTRODUCTION...3 1.1 Abstract...3 1.2 Definitions/Abbreviations/Nomenclature...3 1.3 Referenced
More informationOptics for Machine Vision
The Old Barn Grange Court Tongham Surrey GU10 1DW Optics for Machine Vision Photonex 2002 Simon Hickman Firstsight Vision Ltd The purpose of a camera Lens To transmit light onto the camera sensor in a
More informationIntroduction to Medical Imaging  Medical Student Clerkship Lecture
PATIENT CARE IN RADIOLOGY Introduction to Medical Imaging CLINICAL Kalpana M. Kanal, PhD, DABR Assistant Professor, Diagnostic Physics Dept. of Radiology UW Medicine a copy of this lecture may be found
More informationImplementing and Using the EMVA1288 Standard
Implementing and Using the EMVA1288 Standard A. Darmont *, J. Chahiba, J.F. Lemaitre, M. Pirson, D. Dethier Aphesa, Rue de Lorcé, 39, 4920 Harzé, Belgium ABSTRACT The European Machine Vision Association
More information