1 QuickVet Analyzer Diagnostic System Package insert for QuickVet /RapidVet DEA 1.1 Blood Typing Test Version: February 7, 2012 Description and intended use The QuickVet /RapidVet Canine DEA 1.1 Test Cartridge is used in conjunction with the QuickVet Analyzer for in vitro blood group determination testing. The QuickVet /RapidVet Canine DEA 1.1 cartridge is intended for use in performing a rapid blood group determination for Canine DEA 1.1 from a drop of canine blood. The practice of veterinary transfusion medicine has undergone tremendous growth in recent years, and as a result the understanding of the importance of identifying blood types has increased [1 4]. While it is broadly true that dogs do not possess alloantibodies to incompatible blood groups and thus will generally tolerate well an initial incompatible transfusion, sound practice of veterinary medicine dictates that such transfusions be avoided. The half life of the transfused incompatible cells will be quite short and, thus, the intended therapeutic result may not even be attained. Also, the potential future needs of the canine patient must be considered. Antibodies resulting from a transfusion of incompatible blood may form in only 5 to 7 days and will have long-term viability [5 9]. This eliminates the option of using incompatible blood in a future emergency situation.
2 In addition, antibodies developed in bitches by sensitization resulting from transfusion of incompatible blood groups must be of special concern to breeders. Since antibodies are present in the colostrum, bitches with alloantibodies to a given blood type should not be bred to a sire possessing that blood group if they are expected to nurse the resulting puppies . The nursing puppies will develop isoerythrolysis and may be susceptible to disease or even die due to hemolytic anemia [6, 10 12]. Eight specific antigens have been identified on the surface of the canine erythrocytes [1, 5]. The internationally accepted canine blood group system, the Dog Erythrocyte Antigen (DEA), is based on these antigens. It currently characterizes eight common blood groups, the antigens DEA 1.1, 1.2, 3, 4, 5, 6, 7, and 8. As occurred in human medicine, new antigens are now being discovered on the red cells of dogs. One, DAL, has been reported and is the subject of journal articles. These are often breed specific and thus can be considered rare . The DEA 1.1 blood group is the most significant blood factor in the dog. The DEA 1.1 blood group is highly antigenic and is the primary lytic factor in canine transfusion medicine [1, 6, 14 21]. Although all of the blood group antigens are capable of stimulating formation of alloantibodies, DEA 1.1 has the greatest stimulation potential. Thus most reactions resulting from the transfusion of incompatible cells occur when DEA 1.1 positive blood is given to a DEA 1.1 negative recipient [3, 4, 6]. Clinically significant reactions to DEA 1.2 may occur but are less severe. DEA 7 may be a factor in transfusion reactions, but since it is a cold agglutinin and a naturally occurring isoantibody, it is considered to have very low clinical significance. The remaining antigens are considered to cause clinically insignificant transfusion problems . Ideally, all transfused blood would be DEA 1.1 negative. However, until the concept of using blood rapidly available from commercial sources is more widely accepted, transfusion from dogs that are present in the area at the
3 time of need will remain the norm. It is estimated that 40 % of all dogs are DEA 1.1 positive . Because a number of dogs auto-agglutinate and because a very anemic dog may give equivocal results, typing prior to an urgent need for the information is indicated. Identifying a particular dog as DEA 1.1 positive or negative at birth greatly simplifies future decision making. A DEA 1.1 positive dog can receive both DEA 1.1 positive and negative blood. A dog that is DEA 1.1 negative should not receive DEA 1.1 positive blood. The QuickVet /RapidVet Canine DEA 1.1 test cartridge used with the QuickVet Analyzer is intended for use to classify dogs as DEA 1.1 positive or negative. Principle and explanation of the assay The QuickVet /RapidVet Canine DEA 1.1 assay is based on the agglutination reaction that occurs when an erythrocyte which contains a DEA 1.1 antigen on its surface membrane interacts with a murine monoclonal antibody proven specific to DEA 1.1. The test cartridge has two fluidic channels. The monoclonal antibody is deposited in one of the fluidic channels and a control reagent in the other. The monoclonal antibody is reconstituted during the test. The sample moves through the cartridge by capillary action and the QuickVet Analyzer reads the status of the blood/antibody interaction or lack thereof. All DEA 1.1 positive erythrocytes react with the reconstituted antibody within the cartridge causing agglutination which is automatically examined by the QuickVet Analyzer. The control channel is used to verify that the agglutination is DEA 1.1 specific and not caused by auto-agglutination. No visual reading is needed or possible. The QuickVet Analyzer determines if agglutination has occurred and displays the result. The antibodies are completely nonreactive with all DEA 1.1 negative erythrocytes.
4 Reagents and materials Each test is individually packaged in a sealed pouch with a desiccant bag, and contains the following 1 QuickVet /RapidVet Canine DEA 1.1 test cartridge. 1 plastic pipette tip. Provided separately in the kit for use with the test packages are the following items 1 dropper bottle prefilled with diluent. The plastic bottle contains phosphate buffered saline (PBS) at ph 7.4. The dropper tip dispenses approximately 40 µl of fluid. The bottle contains enough fluid to perform all the tests in the kit. 1 tube for each cartridge for sample preparation. Required but not provided with the test kit are the following items QuickVet Analyzer. The QuickVet Analyzer is designed for use for other hematology determinations in addition to this assay. Refer to the Operator s Manual for complete information concerning the Analyzer. A 100 µl fixed volume pipette. Storage and stability All components of the test kit are stable at refrigerated temperatures or room temperature (2 C to 25 C or 36 F to 77 F) for 24 months. It is recommended that storage of any kit components for more than a week should be at refrigerated temperatures (2 C to 8 C or 36 F to 46 F). The stability of the test components has been determined through the use of
5 a routine accelerated aging protocol. Kits and components should not be exposed to direct sunlight. Each QuickVet /RapidVet Test Cartridge contains a monoclonal antibody specific to the DEA 1.1 antigen on a red blood cell. Each test cartridge can only be used once. Each cartridge is individually packaged with a disposable plastic pipette tip in a sealed pouch containing a desiccant bag to keep moisture out. A torn or otherwise damaged pouch may allow moisture to reach the test cartridge and adversely affect performance. Do not use a test cartridge from a damaged pouch. Open the pouch just prior to use. Once the pouch has been opened, the test cartridge should be used within one day. The test cartridge is stable for 24 months from the date of manufacture if stored according to the instructions above. The expiration date of each cartridge is printed on the pouch label. Do not use the cartridge after the expiration date. The diluent is stable for 24 months from the date of manufacture. Each bottle is labeled with an expiration date. Operating instructions Before performing any tests make sure that the analyzer software has been upgraded to a version capable of performing the QuickVet /RapidVet DEA 1.1 test according to the instructions in the QuickVet Analyzer Operator s Guide. The time it takes to perform a blood type determination with this system depends on factors such as type of test and test environment. At room temperature it takes about 90 seconds for the instrument to warm up the test cartridge to 37 C (98.6 F). Lower and higher room temperatures may change the warm up time. Once the procedure of the assay begins, an answer should be available within 5 minutes.
6 To complete a blood typing test follow the steps below Step 1 Step 2 Step 3 Step 4 Step 5 Step 6 Step 7 Step 8 Step 9 Draw blood from the patient into a syringe or into a tube coated with or containing EDTA as an anticoagulant. The assay requires only 100 µl whole blood but the syringe or tube should be full to ensure the correct concentration of EDTA. If the blood type is not to be determined immediately, nutrients such as CPDA should not be added. Determine the hematocrit of the sample. Remove the cartridge and pipette tip from the foil package and the diluent bottle from the kit box. Touch the Analyze button on the Analyzer s main screen and follow the on-screen instructions. Insert the test cartridge when the message Please insert new cartridge is displayed on the screen. The insertion of a new cartridge into the slot on the front of the QuickVet Analyzer will trigger the warming up of the cartridge to 37 C (98.6 F). When prompted to do so, enter the 7-digit cartridge code located on the pouch label and touch the Next button. When prompted to do so, enter Patient ID and Sample ID (optional) and touch the Next button. When prompted to do so, enter the hematocrit of the sample. Based on the specified hematocrit the QuickVet Analyzer will calculate the required number of diluent drops needed for correct dilution of the sample. Dilute the sample following the on-screen instructions. First dispense the number of drops of diluent stated on the screen into one of the supplied tubes. Then mix 100 µl of blood into the diluent using a fixed volume pipette and the supplied pipette tip. Ensure proper mixing by aspirating a couple of times using the pipette.
7 Step 10 When the message Add 100 µl sample and wait is displayed on the screen, use the pipette to dispense 100 µl diluted blood into the sample well on the cartridge. Avoid bubbles. The determination will start automatically once the blood is detected by the Analyzer. Step 11 The QuickVet Analyzer will determine whether or not agglutination has occurred and the test result will be displayed on the screen. If a printer is connected, the test result can be printed out by pressing the Print button. To obtain suggested advice about safe transfusion/donation with the tested sample touch the Advice button. To print a blood type certificate touch the Certificate button. Step 12 Return to the Analyzer s main screen by touching the Done button and, when the message Please remove cartridge is displayed on the screen, remove the used test cartridge and dispose of it properly in accordance to policies and regulations in practice at the place of operation. Important Make sure that the test sample has been collected and prepared according to the guidelines in this package insert. Limitations of the procedure The accuracy of the test results is dependent on the quality of the blood sample. The quality of the blood sample is dependent upon the blood sample collection, the proper blood to EDTA ratio, and the proper introduction of the diluted sample into the sample well. Please observe all precautions cited in the QuickVet Analyzer Operator s manual and use good blood sample collection techniques at all times.
8 A very limited number of canine patients exhibit auto-agglutination of varying degrees due to serum factors that cause agglutination of the patient s own red cells. The test cartridge has two channels a reference channel and a test channel. An auto-agglutinating patient should receive only DEA 1.1 negative blood. Donor blood exhibiting auto-agglutination should never be used for transfusion. Performance characteristics A total of ninety-four canine erythrocyte samples from ninety-three dogs arriving at an independent animal clinic and an independent university laboratory during the period of the trial were tested utilizing the DiaMed DEA 1.1 Tube Test and the QuickVet /RapidVet Canine DEA 1.1 Blood Typing Test. Eight samples were from healty blood donors, and eighty-six samples were submitted routinely to the laboratory, including blood from dogs with various known diseases, including immunemediated anemia, blood loss anemia, neoplasia, diarrhea, liver disease, severe vasculitis, meningitis and inflammatory bowel disease. All samples were also run in parallel on the DiaMed system. Fifty-four (57 %) samples were DEA 1.1 positive and thirty-five (37 %) were DEA 1.1 negative. Six positive samples retested positive and six negative samples retested negative. No negative samples retested positive and no positive samples retested negative. No samples that were initially positive or negative were inconclusive on retest. Five samples were initially inconclusive. When retested, three were consistently inconclusive, one was negative in accordance with the DiaMed result, and one yielded false positive compared to the DiaMed system. Definitive QuickVet results exhibited an accuracy of 97 % with a sensitivity of 100 % and specificity of 92 % compared to DiaMed results.
9 Trouble shooting guide In most circumstances the instrument will interrupt the test and report an advisory message if a problem occurs during the sample analysis. If this happens, make a note of the advisory number and follow the instructions given in the advisory message. In most cases the problem can be resolved by obtaining a fresh blood sample and repeating the test. Make sure to follow the operating instructions given on-screen and in this package insert. Contact customer support if the problem persists and include the advisory number from the original advisory message in your inquiry. Special attention should be paid to the potentional issues with the collected blood samples listed in Table 1. Table 1: Potential sample quality issues and suggested corrective actions. Problem Agglutinates in the sample tube Hematocrit below 10 % Hemolysis Corrective action Only use fresh blood properly collected in an EDTA tube. If a non-agglutinating sample cannot be obtained the blood should not be used for transfusion and the animal should only receive blood from DEA 1.1 negative donors. Centrifuge the sample and remove plasma to obtain a hematocrit of 15 %. Only use fresh blood properly collected in an EDTA tube. The degree of hemolysis can be estimated by visual inspection of the blood plasma when measuring the hematocrit of the sample. If the plasma is clear and more yellow than red the sample is suitable for use with the QuickVet /RapidVet Canine DEA 1.1 Blood Typing Test.
10 Quality control All materials incorporated into the test package have been quality controlled by standard testing procedures using a routine quality control program during manufacture. Disposal Dispose of all biological materials, pipettes and cartridges in a biohazard container. Manufacturer information The Monoclonal Antibody is licensed from Kansas State University. The QuickVet /RapidVet Canine DEA 1.1 Test is a joint development of dmslaboratories, inc. and Scandinavian Micro Biodevices ApS and is manufactured by Scandinavian Micro Biodevices ApS. RapidVet is a registered trademark of dmslaboratories, inc. QuickVet is a registered trademark of Scandinavian Micro Biodevices ApS. Scandinavian Micro Biodevices ApS Gammelgaardsvej 87C 3520 Farum Denmark dmslaboratories, inc. 2 Darts Mill Road Flemington, NJ USA (908) (800) (908)
11 References  A. Kristensen and B. Feldman, The veterinary clinics of North America (USA); Small animal practice (1995).  H. Vriesendorp, E. Albert, J. Templeton, S. Belotsky, B. Taylor, D. Blumenstock, R. Bull, F. Cannon, R. Epstein, J. Ferrebee, et al., in Transplantation proceedings (1976), vol. 8, p. 289, ISSN  U. Giger and J. Bonagura, Kirk s current veterinary therapy XIII: small animal practice. pp (2000).  U. Giger and M. Blais, Proc Am Coll Vet Intern Med 721, 723 (2005).  M. Pichler and G. Turnwald, Physiology, collection, storage, and indications for whole blood therapy. Comp Cont Ed Pract Vet 7, 64 (1985).  G. Turnwald and M. Pichler, The Compendium on continuing education for the practicing veterinarian (USA) (1985).  H. Ejima, K. Kurokawa, and S. Ikemoto, Bulletin of the Nippon Veterinary and Zootechnical College (Japan) (1980).  L. Young, D. Ervin, and C. Yuile, Blood 4, 1218 (1949), ISSN  L. Young, W. O Brien, G. Miller, S. Swisher, D. Ervin, R. Christian, and C. Yuile, Transactions of the New York Academy of Sciences 13, 209 (1951), ISSN  L. Young, R. Christian, D. Ervin, R. Davis, W. O Brien, S. Swisher, C. Yuile, M. Izzo, and J. Peters, Blood 6, 291 (1951), ISSN  L. Young, D. Ervin, R. Christian, and R. Davis, Science 109, 630 (1949), ISSN  R. Christian, D. Ervin, S. Swisher, W. O Brien, and L. Young, Canadian Journal of Comparative Medicine and Veterinary Science 14, 125 (1950).  M. Blais, L. Berman, D. Oakley, and U. Giger, Journal of veterinary internal medicine 21, 281 (2007).  S. Swisher and L. Young, Physiological Reviews 41, 495 (1961), ISSN  S. Swisher, L. Young, and N. Trabold, Annals of the New York Academy of Sciences 97, 15 (1962), ISSN  G. Wit, N. Goenegracht, P. Poll, and J. Linde, Journal of Small Animal Practice 8, 285 (1967), ISSN  L. Young, W. O Brien, S. Swisher, G. Miller, and C. Yuile, American journal of veterinary research 13, 207 (1952), ISSN  B. Kohn, S. Reitemeyer, and U. Giger, Kleintierpraxis 43, 77 (1998).  C. Knottenbelt and A. Mackin, In Practice 20, 110 (1998), ISSN  C. Knottenbelt and A. Mackin, In Practice 20, 191 (1998), ISSN  G. Lubas, Obiettivi e documenti veterinari 18, 15 (1997), ISSN
442880JVDXXX10.1177/1040638712442880Kohn, Classe, WeingartCanine DEA 1.1 blood typing with QuickVet /RapidVet Clinical evaluation of the QuickVet / RapidVet canine dog erythrocyte antigen 1.1 blood-typing
TESTING BLOOD PRE-TRANSFUSION COMPATIBILITY IN A GROUP OF CATS Meda Moldovan, L. Ognean, Alexandra Arion, I. Morar UNIVERSITY OF AGRICULTURAL SCIENCE AND VETERINARY MEDICINE, FACULTY OF VETERINARY MEDICINE,
Objectives: Cold Agglutination Titer detecting Cold Reacting Antibodies 1. Perform a serial dilution to determine the amount of cold reacting antibody present in a patient specimen with the results obtained
Undergraduate Course in Veterinary Clinical PathologySocrates Programme Haematology LECTURE 10. BLOOD GROUPS AND TRANSFUSIONS 10-1 OVERVIEW 1. Blood Group Systems 2. Blood group testing and cross-matching
Blood Grouping Reagent A/B/D Monoclonal and Reverse Grouping Card MTS080515 Intended Use For the performance of a forward and reverse ABO and D antigen grouping on a single Gel Card For use with the ID-Micro
ARE YOU MY TYPE AND WHAT TO DO WHEN INCOMPATIBLE? Urs Giger, PD Dr. med. vet. Dipl. ACVIM & ECVIM-CA, Dipl. ECVCP Transfusion Center and Penn Animal Blood Bank, Section of Medical Genetics School of Veterinary
Exercise 8 Exercise 9 Direct Antiglobulin Test (DAT) Elution Study Task Aim Introduction To perform the DAT and elution procedure with correct interpretation of results. To perform with 100% accuracy the
IgM ELISA For the quantitative determination of IgM in human serum and plasma For Research Use Only. Not For Use In Diagnostic Procedures. Please read carefully due to Critical Changes, e.g., Calibrator
Instructions for use Anti-Zona Pellucida Antibody Latex Agglutination Test Cat. No.: BS-20-10 Size: 50 Determinations Storage: 2 C 8 C (36 F 46 F) Screening test for the determination of anti-zona pellucida
ABORhCard Package Insert ABO and Rh Blood Grouping Device ABORhCard Intended Use The ABORhCard is a qualitative in vitro test that provides a simultaneous ABO and Rh determination of an individual s ABO
CANINE BLOOD TRANSFUSION Mary Anne Zanetos October 16, 2008 Great Dane Club of America TRANSFUSION TOPICS What are blood types and why are they important? What systems are used to classify blood types?
Rat Creatine Kinase MB isoenzyme,ck-mb ELISA Kit Catalog No: E0479r 96 Tests Operating instructions www.eiaab.com FOR RESEARCH USE ONLY; NOT FOR THERAPEUTIC OR DIAGNOSTIC APPLICATIONS! PLEASE READ THROUGH
KAMIYA BIOMEDICAL COMPANY Mouse Creatine Kinase MB isoenzyme (CKMB) ELISA For the quantitative determination of mouse CKMB in serum, plasma, cell culture fluid and other biological fluids Cat. No. KT-57681
Canine creatine kinase MB isoenzyme (CK-MB)ELISA Kit Catalog No. CSB-E15852c (96T) This immunoassay kit allows for the in vitro quantitative determination of canine CK-MB concentrations in serum and plasma.
Helicobacter pylori IgG ELISA Kit Catalog Number KA0219 96 assays Version: 02 Intended for research use only www.abnova.com Table of Contents Introduction... 3 Intended Use... 3 Background... 3 Principle
Blood Physiology Practical 4 Contents Erythrocytes The blood types Practical tasks Determination of blood groups of the ABO system Determination of the Rhesus system (Rh factor) The cross matching test
PR110 G-Biosciences 1-800-628-7730 1-314-991-6034 firstname.lastname@example.org A Geno Technology, Inc. (USA) brand name Dot Blot Analysis Teacher s Guidebook (Cat. # BE 502) think proteins! think G-Biosciences
Helicobacter pylori IgG ELISA Kit Catalog Number KA0220 96 assays Version: 02 Intended for research use only www.abnova.com Table of Contents Introduction... 3 Intended Use... 3 Background... 3 Principle
Blood Typing Laboratory Exercise 40 Background Blood typing involves identifying protein substances called antigens that are present in red blood cell membranes. Although there are many different antigens
K-ASSAY Mouse IgM ELISA For the quantitative determination of IgM in mouse biological samples Cat. No. KT-407 For Research Use Only. 1 Rev. 072309 K-ASSAY PRODUCT INFORMATION Mouse IgM ELISA Cat. No. KT-407
The Biotechnology Education Company Blood-Based Cancer Diagnostics EDVO-Kit 141 Store entire experiment at room temperature. EXPERIMENT OBJECTIVE: The objective of this experiment is to learn and understand
STANDARD OPERATING PROCEDURE FOR RESOLVING ABO & Rh BLOOD GROUP DISCREPANCIES TITLE: STANDARD OPERATING PROCEDURE FOR RESOLVING ABO & RH BLOOD GROUP DISCREPANCIES 1.0 Principle 1.1 To resolve ABO and Rh
Rat Creatine Kinase MB Isoenzyme (CKMB) ELISA For the quantitative determination of rat CKMB in serum, plasma, tissue homogenates and other biological fluids. Cat. No. KT-12247 For Research Use Only. Not
Artificial Blood Typing Background: Human blood may be classified according to the presence or absence of certain antigens or factors, that are attached to the surface of the red blood cells, or erythrocytes.
Rat creatine kinase MM isoenzyme (CK-MM) ELISA Kit Catalog Number. CSB-E14405r For the quantitative determination of rat creatine kinase MM isoenzyme (CK-MM) concentrations in serum, plasma, tissue homogenates.
Exercise 8 Direct Antiglobulin Test (DAT) Objectives: 1. State the purpose for performing the DAT. 2. State what a positive DAT indicates. 3. List the reagents which are used for performing the DAT. 4.
Mouse Insulin ELISA For the quantitative determination of insulin in mouse serum and plasma Please read carefully due to Critical Changes, e.g., Calculation of Results. For Research Use Only. Not For Use
Human glycated hemoglobin A1c (GHbA1c)ELISA Kit Catalog No. CSB-E08139h (96T) This immunoassay kit allows for the in vitro quantitative determination of human GHbA1c concentrations in serum, plasma and
ABORhCard Applications and Performance Studies White Paper Contents Background....................................... 2 ABORhCard Simple, Accurate, Fast Blood Typing....... 2 Product Design....................................
KAMIYA BIOMEDICAL COMPANY Mouse Testosterone ELISA For the quantitative determination of mouse Testosterone in biological samples. Cat. No. KT-52637 For Research Use Only. Not for use in diagnostic procedures.
KAMIYA BIOMEDICAL COMPANY Mouse C-Peptide I EIA For the quantitative determination of C-Peptide I in mouse plasma, serum and urine. Cat. No. KT-367 For Research Use Only. 1 Rev. 091207 PRODUCT INFORMATION
IV. Principles of Serological Testing in Immunohematology A. Blood Group Antigens and Antibodies 1. Blood group antigens are structures embedded in or protruding from RBCs, WBCs and Platelets. 2. Exposure
ABO-Rh Blood Typing Using Neo/BLOOD Objectives Determine the ABO and Rh blood type of unknown simulated blood samples. Prepare a simulated blood smear. Examine a prepared blood smear under the microscope
KAMIYA BIOMEDICAL COMPANY Mouse krebs von den lungen 6 (KL-6) ELISA For the quantitative determination of mouse KL-6 in serum, plasma, cell culture supernatants, body fluid and tissue homogenate Cat. No.
EXERCISE 5: ERYTHROCYTES SEDIMENTATION RATE - ESR, SED RATE Textbook: Skills: None 10 points Objectives: 1. State the principle of the Erythrocytes Sedimentation Rate - ESR. 2. List two factors which may
Before performing testing, all operators MUST read and become familiar with Universal Precautions for Prevention of Transmission of Human Immunodeficiency Virus, Hepatitis B Virus, and other Blood-borne
Comparing the Tube and Gel Techniques for ABO Antibody Titration, as Performed in Three European Centers 林 良 豐 100.07.18 Introduction Data from 60 consecutive ABO-incompatible kidney transplantations performed
SMF Awareness Seminar 2014 Clinical Evaluation for In Vitro Diagnostic Medical Devices Dr Jiang Naxin Health Sciences Authority Medical Device Branch 1 In vitro diagnostic product means Definition of IVD
Revision No.: ZJ0002 Issue Date: Aug 7 th, 2008 HBV Quantitative Real Time PCR Kit Cat. No.: HD-0002-01 For Use with LightCycler 1.0/LightCycler2.0/LightCycler480 (Roche) Real Time PCR Systems (Pls ignore
ab185915 Protein Sumoylation Assay Ultra Kit Instructions for Use For the measuring in vivo protein sumoylation in various samples This product is for research use only and is not intended for diagnostic
Understanding DARZALEX Interference with Blood Compatibility Testing DARZALEX (Daratumumab) Results in a False Positive Indirect Coombs Test DARZALEX is a human monoclonal antibody for the treatment of
Exercise 9 Elution Study Objectives: 1. State when an elution procedure may be performed. 2. List four situations in which the performance of an elution may provide helpful information. 3. List three types
immunisation Bed side test Antibody tests RhD typing Practice for IV year medical students Zita Csernus MD National Blood Transfusion Service Blood Transfusion Centre Pécs Rh Blood Group System Discovery:
HiPer RA Test Teaching Kit Product Code: HTI019 Number of experiments that can be performed: 20 Duration of Experiment: 1hour Storage Instructions: The kit is stable for 6 months from the date of receipt
Standardization and Evaluation of IgA and IgM Gel Column Agglutination for Direct Antiglobulin Testing ABSTRACT BACKGROUND: Diagnosis of autoimmune hemolytic anemia is commonly confirmed with a positive
IgE (Human) ELISA Kit Catalog Number KA0216 96 assays Version: 03 Intended for research use only www.abnova.com Table of Contents Introduction... 3 Intended Use... 3 Background... 3 Principle of the Assay...
FOR IN VITRO AND RESEARCH USE ONLY NOT FOR USE IN CLINICAL DIAGNOSTIC PROCEDURES CEA458Ge 96 Tests Enzyme-linked Immunosorbent Assay Kit For Testosterone (Testo) Organism Species: General Instruction manual
BLOOD PHYSIOLOGY Lecture 3 BLOOD TYPES AND BLOOD TRANSFUSION 2nd year Dentistry 2008 1 Objectives 1. Describe the basis of blood typing 2. Know the ABO and Rh systems for blood typing and their clinical
Calibration Verification Technical Bulletin Number: 054-869A Date Prepared: 6/97 Purpose This technical bulletin is intended to provide a basic protocol for performing calibration verification of the SureStepPro
Unit 8 Pretransfusion Compatibility Testing A. The purpose of pretransfusion testing is to select, for each recipient, blood components that, when transfused, will have acceptable survival and will not
Blood Grouping Reagent DG Gel 8 ABO/Rh + Kell REF 210383 3034946 Instructions for Use INTENDED USE The DG Gel 8 ABO/Rh + Kell card is for the determination of ABO forward and reverse group, and D and K
BLOOD BANK SPECIMEN COLLECTION PROCEDURE INTRODUCTION Scientific and technical advances in blood group serology have made the transfusion of blood a relatively safe procedure, but serious adverse effects
Instructions for use PCR KIT FOR EXTRACTION OF RNA AND REAL TIME PCR DETECTION KIT FOR HEPATITIS C VIRUS RNA Research Use Only Qualitative Uni Format VBD0798 48 tests valid from: December 2013 Rev11122013
Canine inositol 1,4,5,-trisphosphate,IP3 ELISA Kit Catalog No: E2037c 96 Tests Operating instruction www.eiaab.com FOR RESEARCH USE ONLY; NOT FOR THERAPEUTIC OR DIAGNOSTIC APPLICATIONS! PLEASE READ THROUGH
KAMIYA BIOMEDICAL COMPANY Dog Insulin Like Growth Factor 1 (IGF1) ELISA For the quantitative determination of dog IGF1 in serum, plasma and other biological fluids Cat. No. KT-18280 For Research Use Only.
Director s Signature: Date: Clinical Laboratories Point of Care Testing Quidel QuickVue + One-Step hcg Pregnancy Test PURPOSE The QuickVue+ One-Step hcg Combo test is a one-step immunoassay intended for
Summary of Significant Changes at this Revision 1. New version of test strips - test wait time revised to 5 minutes from 10 minutes. 2. New company name:- Nantong Egens Biotechnology Co. Ltd.,China. COPY
MODULE Rhesus Blood Group System 12 RHESUS BLOOD GROUP SYSTEM 12.1 INTRODUCTION The Rhesus blood group system is the second most important system in transfusion practice. Rh typing is routinely performed
1 History of Innovation Only Immucor consistently delivers the next-generation of blood bank automation to labs worldwide ABS HV Galileo NEO 1998 1999 2004 2007 2010 ABS2000 ABS Precis Echo 2 Innovation
Immunologic Methods Part One Definitions Part Two Antigen-Antibody Reactions CLS 420 Clinical Immunology and Molecular Diagnostics Kathy Trudell MLS (ASCP) CM SBB CM email@example.com Discuss the
ab108653 Cancer Antigen CA125 Human ELISA Kit Instructions for Use For the quantitative measurement of Human Cancer Antigen CA125 concentrations in serum. This product is for research use only and is not
Seahorse XF Cell Mito Stress Test Kit Part # 103015-100 User Guide For use with Seahorse XF e and XF Extracellular Flux Analyzers For Research Use Only 1 Table of Contents Product Description... 2 Introduction...
FOR IN VITRO AND RESEARCH USE ONLY NOT FOR USE IN CLINICAL DIAGNOSTIC PROCEDURES SEA479Ra 96 Tests Enzyme-linked Immunosorbent Assay Kit For Creatine Kinase MB Isoenzyme (CKMB) Organism Species: Rattus
FOR IN VITRO AND RESEARCH USE ONLY NOT FOR USE IN CLINICAL DIAGNOSTIC PROCEDURES SEB268Mu 96 Tests Enzyme-linked Immunosorbent Assay Kit For Phenol Sulfotransferase (PST) Organism Species: Mus musculus
JOINT COMMISSION INTERNATIONAL ACCREDITATION STANDARDS FOR CliniCAl laboratories 2nd Edition Effective 1 April 2010 International Patient Safety Goals (IPSG) Goals The following is a list of all goals.
Technical Manual Maxwell 16 Blood DNA Purification System Caution: Handle cartridges with care; seal edges may be sharp. 2800 Woods Hollow Rd. Madison, WI USA In Vitro Diagnostic Medical Device MDSS GmbH
INMA LABORATORY MANUAL NUTRITION BIOMARKERS Cohorte-VALENCIA This manual describes the collection, separation, storage and transport of blood samples for the INMA study. Careful collection of samples is
QuantiFERON TB Gold In Tube Method (QGIT Test) Manufacturer: Cellestis Inc. (USA) a QIAGEN Company P1078 TB APPRISE TB Ante vs. Postpartum Prevention with INH in HIV Seropositive mothers and their Exposed
Haemostasis Laboratory, University of Glasgow, Glasgow Royal Infirmary. Prepared by Ann Rumley, Paul Welsh. 03/05/05 Standard Operating Procedure for High sensitivity CRP assay Principle of the method: