1 HOW TO CHOOSE THE AED FOR YOUR HOSPITAL Judy Boehm, RN, MSN In the last issue of Code Communications, I concluded that many hospitals are not yet able to deliver defibrillation to patients located outside of critical care units within the American Heart Association (AHA) recommended 3-minute goal. The number of hospitals who have instituted a twotiered system of defibrillation is increasing, but there are many institutions that need to change their practice and embrace the use of automated external defibrillators (AEDs). Survival of patients in ventricular fibrillation (VF) and pulseless ventricular tachycardia (VT) will not improve until Basic Life Support (BLS) trained first responders in areas outside of critical care are empowered to defibrillate prior to the arrival of the CPR team. In this issue of Code Communications I will discuss how to choose an AED for your hospital. I will present an overview of the selection process and then describe key quality characteristics for consideration when selecting an AED. The Process for Choosing an AED First, there must be a realization within the institution that change to two-tiered defibrillator technology is needed. With a two-tier model, manual defibrillators are utilized in critical care units and departments where staff are trained in Advanced Cardiac Life Support (ACLS) and arrhythmia recognition, and AEDs are operated by BLS-trained staff in other locations such as medical/surgical units, ambulatory care clinics, diagnostic/treatment areas, hallways, common areas such as information desks and the cafeteria, and outlying buildings. What may be driving the change to two-tiered defibrillator therapy? It may be that the life of the current defibrillators is exceeded based on their capital depreciation. Perhaps the institution has embraced the concept of early defibrillation and strives for best practice. If their mean time to defibrillation is measured, they may find that it exceeds 3 minutes when delivered outside of critical care units. Defibrillator technology has improved during the last decade, with biphasic waveforms resulting in improved conversion of ventricular and supraventricular arrhythmias - and less skin burn. If practitioners have tried to implement the 2005 AHA Guidelines for Cardiopulmonary Resuscitation and Emergency Cardiovascular Care, 1 they may realize that feedback on CPR technique, which is built into several AEDs, would be helpful. Or there may be a risk management concern with multiple models of defibrillators using a variety of accessories throughout the institution - and difficulty keeping staff competent in their use. Key leadership persons need to be convinced that a change to two-tiered defibrillation therapy fits within the organization s goals and that they can support the implementation program over the next year. It is helpful to frame the discussion as an extension of existing emergency response procedures rather than as a new program. The rationale and impact of this major practice change will need to be discussed with clinical and administrative leadership groups. Discussion topics might include the scientific background, efficacy, AHA 2005 Guidelines, staff acceptance, safety, training, liability, cost, etc. Now that AEDs are available in the community at airports, health clubs, grocery stores, sports arenas, etc., we should be able to move beyond the conviction that defibrillators are only operated by critical care nurses and physicians.
2 2 It is important to point out that BLS competency programs already incorporate general AED training, and now it can be made specific to one AED. References 2 through 13 at the end of this newsletter may help in this discussion with leadership. A decision-making group needs to oversee the process of changing to two-tiered defibrillator therapy. This group may be the Cardiopulmonary Resuscitation Committee, or a task force specifically selected for this purpose. Questions to consider when choosing group members include: Who are the leaders of resuscitation practice in your institution? Who will be affected by this change? Who sheperds product changes within the institution? Those persons who should be included in the decision-making group can be found in Table 1. Table 1. Persons to Include in the AED Decision-making Group Cardiologist; an electrophysiologist is especially helpful Intensivist since half of all cardiopulmonary arrests occur in critical care units Critical care nurses Pediatric representative, either a pediatric clinical nurse specialist or attending General care unit nurses since placing AEDs on med/surg units will be new for them both bedside provider and manager BLS-trained technical staff in diagnostic/treatment areas where AEDs will be placed both provider and manager Cardiac clinical nurse specialist with expertise in this area House staff representative Biomedical engineering representative, perhaps the manager or the technologist responsible for defibrillators Administrator Purchasing agent Information technology representative since all AED vendors have data management programs Educator since initial and ongoing competency in use of the AED will be needed Other members of the CPR team who perform defibrillation It is important that first responders who will newly use an AED be included in the decision-making group from the beginning so they can understand the change to two-tiered defibrillator therapy and help convince their peers that improved patient outcomes will result. Incorporation of defibrillation into their skill base will be frightening to general care nurses and technicians, who have only been responsible for a few minutes of CPR and then disappeared when the CPR team shows up. Consider where AEDs will be placed and who may be the greatest resistors to change in practice. For example, if it is predicted that security officers will carry an AED in their van, you may want to include a representative from this group since they do not think of themselves as providing medical care. Several authors writing about implementation of AEDs have mentioned that not including house staff in the decision-making group and in the AED training has lead to resistance, with remarks during codes such as Get that thing off and use the real defibrillator instead.
3 3 Discuss whether AED placement and use will extend to areas where staff has no medical training. For example, are there distant sites where education, risk management or medical record departments are located? Purchase of AEDs for this type of location may be postponed until later, since it involves a mind shift for the employees and commitment to BLS training. At Dartmouth-Hitchcock Medical Center, I saw that the cardiac arrest of a Financial Services employee in an outlying building soon brought a request from them to the CPR committee for an AED. Check with your local Emergency Medical Services to learn how long it takes for an ambulance to reach a victim in your outlying buildings. The institution may want to offer the same hope of survival to all persons throughout the medical center - whether inpatient, outpatient, visitor or employee. A champion should be chosen from the decision-making group who will guide the process from beginning to end. The champion should be passionate that this practice change is right for the institution and be given the time to make it happen. This will be the go-to person for planning, questions, and follow-up. It will be important to determine ahead of time how the decision of choosing an AED will be made. Will it be a consensus of the decision-making group, or does the final decision rest with a designated individual? The decision-making group should convene and discuss how the AED will be used within the institution. Will the defibrillator be used strictly as an automated device by BLS-trained first responders? If so, the AED chosen needs to be extremely easy to operate with simple and thorough prompts. Then a manual defibrillator would be transported to the scene for subsequent use by the ACLS team. Or will the defibrillator be used initially as an automated device by first responders and then also as a manual unit by members of the resuscitation team? If the AED will be used during the whole resuscitation, it may be desirable that the AED selected contain ECG monitoring, manual override, and even synchronized cardioversion. Does the institution desire their manual defibrillators to incorporate automatic advisory capacity and be available in some of the higher risk areas? I have found that BLS-trained staff are intimidated by these more complex defibrillators. Providing standalone AEDs is a simpler, less expensive option that will allow more defibrillators to be placed throughout the institution, even in non-traditional areas. If the AED will be used for bedside monitoring of patients not in full arrest, then consider selecting an AED with 3-lead ECG monitoring capability. There are even a few AEDs that have a chart recorder. If the institution currently delivers manual defibrillation with paddles, consider switching these devices to hands-free defibrillation with disposable electrodes when the AED is introduced. It is helpful when the manual defibrillator and the automated defibrillator at an institution are from the same vendor, since electrodes may be shared, the data management program may be similar, and the sales rep is consistent. You may also want to determine what defibrillators are being used by the Emergency Medical Services in your region so that compatibility can be considered. Before proceeding further, the group should make sure that budgetary support is available for this change, and build this into the timeline. Now you need to select the vendors who produce AEDs that meet your institutional needs. Using ECRI services is a good place to start. ECRI (formerly the Emergency Care Research Institute) is a nonprofit health services research agency and a Collaborating Center of the World Health Organization. It is designated as an Evidence-based Practice Center by the U. S. Agency for Healthcare Research and Quality. ECRI s mission is to improve the safety, quality, and cost-effectiveness of healthcare. Within their Healthcare Product Comparison System, they have technology overviews, the ability to create a customized comparison chart with side-by-side specifications, and links to supplier web sites and addresses. Laboratory, clinical and user evaluations of medical products are routinely published in their monthly Health Devices. AEDs were discussed in their June issues for 2003 and I found the section AEDs in Your Facility, A Risk Management Perspective in the June 2003 issue especially
4 4 helpful on this topic. The information from ECRI may help you narrow the field of AEDs that meet the needs of your institution. Their web site is Companies who manufacture AEDs are found in the table below along with their respective products that are suggested for use in the hospital setting. AEDs targeted for public access use out of the hospital are not included. Table 2. Suppliers of AEDs Burdick Inc, A Quinton Co defibrillator.htm CardioVive AT CardioVive DM Cardiac Science Inc Powerheart AED G3 Powerheart AED G3 Pro Progetti srl PG DEF Rescue 2 Schiller America Inc nodeid,910,_country,hq,_language,en.html FRED FRED easy Defibtech LLC LifeLine AED Medtronic Emergency Response Systems Lifepak CR Plus Lifepak 500 Philips Medical Systems HeartStart FR2+ HeartStart FRx Welch Allyn Inc catalog/type.asp?id=33392 AED 20 AED 10 ZOLL Medical Corporation AED Plus AED Pro It is helpful to have a vendor fair in order to compare AEDs side by side. Staff who currently operates defibrillators and potential new users of an AED should be invited to the fair. Each vendor could be given a designated time(s) to present their AED(s) to a seated audience. In addition the representative would present their AED at a booth as reviewers stop by. Product literature and educational materials should be available. It may be helpful to have targeted small group meetings to discuss specific aspects of the AEDs. For example, BioMed may want their own group discussion regarding set-up, maintenance and reliability. Educators will be interested in teaching methods and support. The electrophysiologist may desire a group to discuss the different biphasic waveforms, efficacy, and energy settings. Do not be surprised if this issue becomes contentious with heated debate about the importance of the waveform and maximum energy.
5 5 Evaluating the Quality of AEDs An evaluation form should be developed to obtain feedback from the reviewers during the vendor fair and additional small group discussions. Key AED quality characteristics should be evaluated from several aspects: Clinical Analysis algorithm, waveform, and energy delivery Technical Education Data management Cost Each institution needs to decide what quality aspects are important for them to evaluate and incorporate these into their evaluation tool. Only select characteristics would be evaluated by each group. For example, potential new users may evaluate only the clinical characteristics. Those who will orient staff to the AED and track competence would evaluate education as well as clinical characteristics. The decisionmaking group will probably evaluate all characteristics. It may be helpful to rank the importance of the characteristics prior to summarizing scores from all the evaluations. The format for an evaluation tool may look something like this. Any Hospital AED Evaluation November, 2006 Name: Date: Department: Position: RN MD Other: Please use numbers 1 through 5 to indicate whether you believe the quality characteristics are present for each AED. Strongly agree = 1 Agree = 2 Disagree = 3 Strongly disagree = 4 Cannot assess/ = 5 Does not apply Clinical Quality Characteristics AED #1 AED #2 AED #3 The AED is lightweight, compact and easy to carry. The AED provides clear indication of its readiness for use. All needed components are conveniently accessible within the carrying case. etc My Choice AED #1 AED #2 AED #3 Place a checkmark in the box that corresponds with the AED that you believe is best for our institution.
6 6 In Table 3 is my list of quality characteristics that you may want to incorporate into an evaluation tool for the ideal AED. Table 3. Key Quality Characteristics of the Ideal AED Clinical Quality Characteristics The AED is lightweight, compact and easy to carry. The AED provides clear indication of its readiness for use. All the needed components are conveniently accessible within the carrying case. The AED is quick and easy to set up for use on a patient. The AED electrodes are pre connected to ease set up. The AED adapts easily for use with children. The diagrams for placement of the electrodes are helpful (both adult and children). The AED is compliant with 2005 American Heart Association Guidelines. Coaching is provided on CPR technique, i.e. rate, depth, and interruptions of compressions, timing of breaths. The voice prompts are helpful (appropriate, loud, clear). The text messages and/or graphic icons are helpful (appropriate, clear). The screen is easy to read in daylight and darkness. The screen displays a clear ECG. The AED is simple and intuitive to operate. The AED operates in a manner that promotes safety of care providers. The AED provides useful prompts of problems, e.g. poor electrode contact, disconnect, low battery. The AED can be set up to monitor the ECG in 3 leads. The AED manual override is easy to identify and operate. The AED is able to perform synchronized shock. A spare AED battery is easy to install. The wall mounting bracket provides easy access to the AED. Analysis Algorithm, Waveform and Energy Delivery Quality Characteristics The first shock conversion rate for ventricular fibrillation/pulseless ventricular tachycardia and the average number of shocks for conversion is acceptable with this waveform. The scientific rationale for the energy output settings used by the vendor is acceptable. The AED automatically compensates for patient impedance when delivering the voltage. The voltage is easily attenuated when the device is used with children The charge time for the AED is within 10 seconds. The AED's algorithm ensures good sensitivity and specificity when analyzing rhythms in both adults and children.
7 7 Analysis Algorithm, Waveform and Energy Delivery Quality Characteristics (continued) The AED s algorithm is accurate given pacemaker artifact and electrical artifact. The AED is able to analyze VF waveform frequency and advise whether shock or CPR is initially desirable. Technical Quality Characteristics The AED performs automatic self tests at an appropriate frequency for the internal circuitry, battery, pads, and discharge. The AED provides clear indication of its readiness/unreadiness to perform. AED maintenance to be performed by biomedical engineering is minimal. The warranty is competitive. The AED has a low problem/repair history. Programming the AED is easy and efficient. Drift of the AED clock is minimal. The time on the AED can be synchronized to a designated clock. The defibrillator, accessories, and battery are reliable. Current battery capacity is easy to identify. It is easy to see when the electrodes and battery are expired. The shelf life of the battery is acceptable. The environmental specifications for temperature, humidity, altitude, vibration, shock/drop abuse, water resistance, and electromagnetic interference are acceptable. The operating time is competitive (minutes of ECG monitoring, number of shocks over time). Heart rate and elapsed time are clearly visible on the screen. A chart recorder is available with the AED. Audio recording is available on the AED. The wall mounting cabinet for the AED provides good security for the device. The wall mounting option for the AED can be wired to a central operator. The vendor rep is approachable, knowledgeable, and available. The vendor provides onsite assessment for AED needs of the institution. The manufacturer has a history of producing good quality defibrillators. The company has a history of reliable and speedy customer service. There are no recent recalls or safety alerts that influence operation of the AED and its accessories.
8 8 Education Quality Characteristics The AED training device and supplies are similar to the real AED. The real AED can be adapted for training. The training equipment and supplies are easy to use. The training materials are in a variety of formats, i.e. videos, DVDs, handouts, computerized learning programs. A training program is available which is interactive and tests competency. The training scenarios reflect current AHA standards. Competency skill checklists and written tests are available from the vendor. Templates for AED protocols in the hospital are available from the vendor. The company provides support for training. Data Management Quality Characteristics The software requirements for installing the vendor s data management program are available at my institution. The hard disk storage space needed for the data management program and its events is reasonable and manageable. Transferring data from the AED to the data management software program is easy and minimizes/eliminates down time of the defibrillator. Data entered into the software program for resuscitation is that required by AHA, National Registry of CPR, and the institution. The internal memory capacity for storage of resuscitation data is acceptable. The AED data management program is intuitive and easy to use. The data management program produces meaningful case reports that include ECG waveform, event log, case data, and CPR performance. The data management program produces meaningful aggregate reports. Expert technical support is available for use of the data management program. Cost The cost of the AED device is competitive. The cost of the electrodes (adult and pediatric), supply kit, and batteries are competitive. The AED electrodes interface well with the manual defibrillator from the same vendor. The AED electrodes are multifunctional, and support ECG monitoring, defibrillation and pacing. The shelf life of the electrodes is competitive. Rechargeable batteries are an option.
9 9 Cost (continued) The cost of the training device and training electrodes are competitive. The data management program is cost effective. The vendor offers a trade-in discount for defibrillators taken out of service. Upgrades are available for the AED and the data management program at a competitive cost. Conclusion As you are deliberating about which AED to purchase, talk with other users of the various AEDs. The vendor representatives can provide you with contact names. I have found it helpful to keep for a short time within the institution those AEDs seriously being considered so they can be investigated further. Once the AED is selected for the institution, work with the representative to make an assessment of target areas for AED placement, incorporating the 3-minute time goal. Use a stopwatch to calculate fast walking time so that an AED (or manual defibrillator) will be no further than one minute away. It should take no more than one minute to reach the AED, one minute to return to the victim, and one minute to operate the AED until a shock is delivered. The rep will have valuable suggestions about how to implement the AED program and will provide educational support. Talk with others who have recently introduced this AED into their institution and learn what implementation methods worked best for them, and what stumbling blocks they encountered. Moving into two-tiered defibrillator therapy is a culture shift for an institution, but if leadership is behind the change and you select a good quality AED that will meet the needs of your organization, then implementation flows smoothly. Don t forget to track patient survival from VF/pulseless VT resuscitations and compare the before AED to the after AED rate for return of spontaneous circulation and hospital discharge. You will see a significant improvement that can be proudly shared within your organization.
10 References 1 American Heart Association American Heart Association Guidelines for Cardiopulmonary Resuscitation and Emergency Cardiovascular Care. Circulation 2000; 112:IV-1-IV-211 (Suppl). 2 Destro, A., Marzaloni, M., Sermasi, S. & Rossi, F. Automatic external defibrillators in the hospital as well? Resuscitation 1996; 31: Gombotz, H., Weh, B., Mitterndorfer, W. & Rehak, P. In-hospital cardiac resuscitation outside the ICU by nursing staff equipped with automated external defibrillators The first 500 cases. Resuscitation 2006; 70: Hanefeld, C., Lichte, C., Mentages-Schroter, I., Sirtl, C. & Mugge, A. Hospital-wide first-responder automated external defibrillator programme: 1 year experience. Resuscitation 2005; 66: Kaye, W., Mancini, M. E., & Richards, N. Organizing and implementing a hospital-wide firstresponder automated external defibrillation program: strengthening the in-hospital chain of survival. Resuscitation 1995; 30: Kaye, W., Richards, N., Yanus, A. & Sawyer-Siva, S. Can prompt recognition and rapid firstresponder automated external defibrillation in the hospital improve outcome from cardiac arrest? Preliminary results. Critical Care Medicine 1997; 25:A57 (Suppl). 7 Kay, W., et al. Strengthening the in-hospital chain of survival with rapid defibrillation by first responders using automated external defibrillators: Training and retention issues. Annals of Emergency Medicine 1995; 25: Kenward, G., Castle, N. & Hodgetts, T.J. Should ward nurses be using automatic external defibrillators as first responders to improve the outcome from cardiac arrest? A systematic review of the primary research. Resuscitation 2002; 52: Kyller, M. & Johnstone, D. A 2-tiered approach to in-hospital defibrillation. Nurses respond to a trial of using automated external defibrillators as part of a code-team protocol. Critical Care Nurse 2005; 25(4): Mancini, M. E. AEDs: Changing the way you respond to cardiac arrest. American Journal of Nursing 1999; 99(5): Mancini, M. E. & Kay, W. In-hospital first-responder automated external defibrillation: What critical care practitioners need to know. American Journal of Critical Care 1998; 7: Martin, N.K., & Ahrens, T.S. Influence of automatic external defibrillators (AEDs) on in-hospital cardiopulmonary arrests. American Journal of Critical Care 2004; 13:269 (Abstract). 13 Powers, C. C., & Martin, N.K. When seconds count, use an AED. American Journal of Nursing May, 2002:8-10 (Suppl). 10
19.109 AUTOMATED EXTERNAL DEFIBRILLATOR References: American Heart Association Healthcare Provider (CPR/AED) Ohio Revised Code Chapter 2305.235 Immunity of Person Involved with Providing Automated External
LIFEPAK 500 automated external defibrillator AED QUIZ 1. Defibrillation must be delivered in order to give the victim the best chances of survival. a. after CPR b. early c. late d. many times 2. When a
ACLS Study Guide The American Heart Association released new resuscitation science and treatment guidelines on October 19, 2010. Please read the below information carefully This letter is to confirm your
1 MAKING CODE DOCUMENTATION WORK FOR YOU THE ELECTRONIC WAY Judy Boehm, RN, MSN Introduction As the cardiac clinical nurse specialist at a major tertiary medical care center, I was responsible for managing
( www.nanoomtech.co.kr ( 4 minutes start brain damage 6 minutes brain death begins 10 minutes biological death Resuscitate within 5 Minutes do we need AEDs? Sudden Cardiac Arrest (SCA) is the leading cause
Emergency Cardiac Care: Decision Support Tool #1 RN-Initiated Emergency Cardiac Care Without Cardiac Monitoring/Manual Defibrillator or Emergency Cardiac Drugs Decision support tools are evidence-based
ACLS Provider Manual Comparison Sheet Based on 2010 AHA Guidelines for CPR and ECC CPR Chest compressions, Airway, Breathing (C-A-B) BLS Changes New Old Rationale New science indicates the following order:
2015 Interim Resources for BLS Original Release: November 25, 2015 Starting in 2016, new versions of American Heart Association online courses will be released to reflect the changes published in the 2015
AED Automated External Defibrillator Implementing an AED Program Mission The mission of the American Heart Association is to build healthier lives, free of cardiovascular diseases and stroke. That single
2015 Interim Resources for HeartCode ACLS Original Release: November 25, 2015 Starting in 2016, new versions of American Heart Association online courses will be released to reflect the changes published
2011 Pediatric Advanced Life Support (PALS) Classroom Course & Materials Frequently Asked Questions (FAQs) As of November 3, 2011 Course Information Q: What is the PALS Course? A: The American Heart Association
AUTOMATIC EXTERNAL DEFIBRILLATOR PROGRAM PURPOSE This program provides a guide for the purchase, training, placement and usage of Automated External Defibrillators (AED s) in any University facility or
ACLS Study Guide The ACLS Provider exam is 50-mutiple choice questions. Passing score is 84%. Student may miss 8 questions. For students taking ACLS for the first time or renewing students with a current
How you can help save lives Through Life Support Training Courses with THE INTERNATIONAL LIFE SUPPORT TRAINING CENTER (ILSTC) TABLE OF CONTENTS Introduction Page 3 Basic Life Support for Healthcare Provider
BELMONT PUBLIC SCHOOLS Automated External Defibrillator Program (AED) The Belmont Public Schools will have a limited Public Access Defibrillation Program with Automatic External Defibrillators (AED) located
Multifunction Defibrillator Pads Product information Innovative solutions in resuscitation therapy With pads for emergency use, monitoring, as well as special procedures, there are Pads sized and designed
Advanced Cardiac Life Support Provider & Provider Renewal Courses (ACLS & ACLS-R) Baptist Health is an authorized American Heart Association (AHA) provider and has approved these courses for Continuing
Adding IV Amiodarone to the EMS Algorithm for Cardiac Arrest Due to VF/Pulseless VT Introduction Before the year 2000, the traditional antiarrhythmic agents (lidocaine, bretylium, magnesium sulfate, procainamide,
Date: May 31, 2012 Page 1 of 5 Cardiac Arrest General This protocol should be followed for all adult cardiac arrests. Medical cardiac arrest patients undergoing attempted resuscitation should not be transported
Instructions For Use AED Trainer PART NUMBER Copyright 2007 Cardiac Science Corporation. All rights reserved. The Automated External Defibrillator (AED) Trainer is a device for teaching basic AED rescue
Resuscitation Quality Improvement Frequently Asked Questions Below are answers to frequently asked questions about the Resuscitation Quality Improvement (RQI) Program. If you have a question that is not
ACCESSORIES LIFEPAK 1000 DEFIBRILLATOR Genuine accessories from Physio-Control POWER OPTIONS / ECG Monitoring accessories Thank you for choosing Physio-Control as your partner in helping you save lives
Question-and-Answer Document 2010 AHA Guidelines for CPR & ECC As of October 18, 2010 Q: What are the most significant changes in the 2010 AHA Guidelines for CPR & ECC? A: Major changes for all rescuers,
X Series Extensive Capabilities for Patient Transport Transporting Patients throughout the Hospital Requires the Right Equipment Full Featured yet Compact When transporting critically ill patients, you
FINANCE & OPERATIONS POLICY LIBRARY Revised: 01.17.07 TABLE OF CONTENTS Automated External Defibrillator (AED) Program I. INTRODUCTION...2 II. PURPOSE...2 III. POLICY...2 IV. REFERENCES...2 V. DEFINITIONS...3
Connected Philips HeartStart connectivity and data management for ALS professionals Flexible and reliable, Philips offers a wide choice of dependable networking and software solutions for capturing, transmitting,
What Are Arrhythmias? Many people have questions about what the word arrhythmia means, and arrhythmias can be a difficult subject to understand. The text below should give you a better understanding of
3 How to Use the LIFEPAK 1000 Defibrillator HOW TO USE THE LIFEPAK 1000 DEFIBRILLATOR 3 This section provides an overview of information and instructions for using the LIFEPAK 1000 defibrillator. Modes
Cardiac Arrest Pediatric Ventricular Fibrillation / Pulseless Ventricular Tachycardia Protocol revised October 2008 Preamble In contrast to cardiac arrest in adults, cardiopulmonary arrest in pediatric
Getting Started A Guide to Public Access Defibrillation 2010 Edition Dear Client, Thank you for your efforts in helping to make your community Cardiac Safe! Since it s launch in 1997, the Toronto EMS Cardiac
ACCESSORIES for use with the LIFEPAK DEFIBRILLATOR 1000 Genuine accessories from Physio-Control, a division of Medtronic. POWER OPTIONS / ECG Monitoring accessories Thank you for choosing Physio-Control
ACCESSORIES for use with the LIFEPAK DEFIBRILLATOR 1000 Genuine accessories from Physio-Control POWER OPTIONS / ECG Monitoring accessories Thank you for choosing Physio-Control as your partner in helping
HEART SAFE SOLUTIONSM Protecting your people means being ready. AED program implementation and management Kelly Kettler, RN-BSN, Cabela s Claims Manager Physio-Control Heart Safe Solution customer since
Automated External Defibrillator 1. INTRODUCTION An automated external defibrillator (AED) is used to treat victims who experience sudden cardiac arrest. It must only be applied to victims who are unconscious,
PRODUCT BROCHURE 1 Introducing the first 12-lead for the BLS team. You know time is of the essence when responding to a chest pain emergency. At that critical moment, what if you could do more for patients?
FOUNDATIONHTSI Healthcare Technology Safety Institute Safety Innovations Simple Solutions for Improving Patient Safety In Cardiac Monitoring Eight Critical Elements to Monitor Alarm Competency University
Administrative Procedure Automated External Defibrillators (AEDs) AP 400.37 Procedure for: Principals, Maintenance and Custodial Staff Adopted: April 12, 2013 Submitted by: Superintendent of Business &
Cardiac Arrest - Ventricular Fibrillation / Pulseless Ventricular Tachycardia Protocol revised October 2008 Preamble Survival from cardiorespiratory arrest for patients who present with ventricular fibrillation
Pacemakers and Defibrillators Andrew Gentilin, RN, BScN, MN-ACNP Nurse Practitioner University Health Network Clinical Cardiac Electrophysiology Outline 1) What is a pacemaker/defibrillator? Parts of the
Automated External Defibrillator (AED) Program BEREA COLLEGE AUTOMATED EXERNAL DEFIBRILLATOR (AED) POLICY Sudden Cardiac Arrest (SCA) is a condition that occurs when the electrical impulses of the human
EMRI EMRI Emergency Medicine Risk Initiative EMRI is a continuous cycle of Risk, Safety, and Quality (TSGRSQ). It is designed to reduce medical errors, the risk of misdiagnosis, and improper management
Success Manual and Cheat Sheet Notes to Pass Your Basic Life Support (BLS) Course Written by: Jay Snaric, MS And Kimberly Hickman, RN CPR St. Louis 44 Meramec Valley Plaza St. Louis MO 63088 www.stlcpr.com
AHA C LASSES Basic Life Support ACLS PALS FALL 2007 AHA Classes Basic Life Support ACLS PALS ORIENTATION Mercy Education Department Class/Course Catalog New Employee Nursing New Manager CONTINUING ED Pain
2011 Advanced Cardiovascular Life Support (ACLS) Classroom Course & Materials Frequently Asked Questions (FAQs) As of July 21, 2011 Course Information Q: What is the Advanced Cardiovascular Life Support
Automated External Defibrillators (AED) Program Oversight DISTRICT AED/HEALTH COORDINATOR 1. Coordinate school/district CPR/AED training 2. Work with schools to select and maintain CPR/AED trained personnel
February 2015 NATIONAL AMBULANCE SERVICE ONE LIFE PROJECT Improving patient outcomes from Out Of Hospital Cardiac Arrest David Hennelly AP MSc Jan 2015 THE ONE LIFE PROJECT IS BEING LED BY THE NATIONAL
MANAGED TRAINING SERVICES INNOVATIVE TRAINING AND MANAGEMENT SYSTEM WITH FIXED PRICE FEE STRUCTURE Quality training by industry experts Training qualifications and recertifications tracked through centralised
Effect of Frequent Application of Code Blue Training Program on the Performance of Pediatric Nurses Ghada Saeed AL-Ghamdi 1 ; Magda Aly Essawy 2 and Dr. Mohammad Al-Qahtani 3 1 College of Nursing, University
CPR (Cardio Pulmonary Resuscitation) and AED (Automated External Defibrillator) Information August 29, 2013 Today s Session will begin at 3:30 PM While you are waiting, please do the following: Configure
Automatic External Defibrillator Program 1.0 Introduction This policy serves as guidance to outline the availability, maintenance, and proper use of Automated External Defibrillators (AEDs) at the University
Connected Care Philips HeartStart connectivity and data management for EMS professionals Flexible and reliable, like Connected Care is the Philips approach to open data management solutions that simplify
Basic Life Support Study Guide Critical Care Training Center www.acls123.com 818-909-7234 Course Overview This study guide is an outline of content that will be taught in the American Heart Association
Software Upgrade Instructions for Powerheart, FirstSave, and CardioVive Automated External Defibrillators (AEDs) Copyright 2010 Cardiac Science Corporation. All rights reserved. Contents About the upgrade
ACCESSORIES BROCHURE LIFEPAK 1000 Defibrillators Genuine Accessories from Physio-Control LIFEPAK 1000 Defibrillators 1 LIFEPAK 1000 Genuine Accessories from Physio-Control Ensure the safety of your staff
HEARTSHARE TRAINING SERVICES, INC. www.heartsharetraining.com 1101 S. Winchester Blvd. Suite #K-230 San Jose, CA 95128 (408) 246-0311 A.C.L.S. Recertification Course Dear ACLS Recertification Participant:
DEFIBRILLATOR PRODUCT BROCHURE 1 Not every cardiac emergency is the same. Neither is every responder. Your world demands flexibility and that s exactly what the defibrillator from Physio-Control delivers.
Update on Small Animal Cardiopulmonary Resuscitation (CPR)- is anything new? DVM, DACVA Objective: Update on the new Small animal guidelines for CPR and a discussion of the 2012 Reassessment Campaign on
Law Enforcement AED IMPLEMENTATION GUIDE Works like you work. After the third shock, her heart started right up. By then, the paramedics arrived; I knew she was doing all right when she started arguing
RESUSCITATION RFACT SHEET Get With The Guidelines -Resuscitation is the American Heart Association s collaborative quality improvement program demonstrated to improve adherence to evidence-based care of
Continuing Education Catalog An Affiliate of the National University System www.westmedcollege.edu Continuing Education Catalog Notice for Students This catalog is the official publication of the programs,
CodeNet Administrator s Guide Software version 4.0 Manual 1.0 revision 0 ZOLL Data Systems 2000-2007 by ZOLL Data Systems. All rights reserved. ZOLL Data Systems is a subsidiary of ZOLL Medical Corporation.
6 In-hospital resuscitation Introduction These guidelines are aimed primarily at healthcare professionals who are first to respond to an in-hospital cardiac arrest and may also be applicable to healthcare
Carts, Wall Mounts or Both? Exploring the Best Criteria when Implementing EMR Equipment INTRODUCTION Healthcare IT News reports that hospitals are in an EMR (electronic medical record) buying frenzy due
PALS/Pediatric Advanced Life Support Provider Course Dates: January 26, 2015 July 27, 2015 February 23, 2015 August 24, 2015 March 23, 2015 September 28, 2015 April 29, 2015 October 26, 2015 May 27, 2015
Trusted by those who save lives Philips HeartStart FR2+ Defibrillator Product information Tools for the pros; simplicity for the lay responder The HeartStart FR2+ has come to be a trusted tool of professional
BASIC LIFE SUPPORT - ADULT First Issued by/date BKW PCT May 2003 Issue Version Purpose of Issue/Description of Change 3 Policy reviewed. No changes in procedure or legislation since policy revised 12 months
University of Kentucky / UK HealthCare Policy and Procedure Policy # NR08-03 Title/Description: Care of Patients with Pacemakers Purpose: To outline the care of patients with permanent or temporary pacemakers.
Risk Management for Medical Devices CĂLIN CORCIOVĂ, MARIUS TURNEA, RADU CIORAP Department Biomedical Science Faculty of Medical Bioengineering, Grigore T. Popa University of Medicine and Pharmacy - Iasi
H i g h l i g h t s o f t h e 2010 American Heart Association Guidelines for CPR and ECC Contents Major Issues Affecting All Rescuers 1 Lay Rescuer Adult CPR 3 Healthcare Provider BLS 5 Electrical Therapies
University of California Irvine Medical Education Simulation Center HeartCode BLS and ACLS Frequency Asked Questions Q: What is HeartCode Part 1? Q: What content is taught in HeartCode BLS Part 1? Q: Approximately
Frequently Asked Questions FirstVoiceTraining.com Is online training an approved way to receive CPR or First Aid certification? Yes. Online training is now an approved method of CPR and First Aid certification
PATIENT CARE SERVICES POLICY AND PROCEDURE Title: within CHN Facilities Section: Provision of Care, Treatment, and Services Dept. of Origin: Patient Care Services Effective Date: February 02, 1999 Last
Philips HeartStart MRx Monitor/Defibrillator and Philips Supplies They are Made for Each Other Supplies Key Selling Attributes What to say to your Customers about Philips Monitor/Defibrillator Supplies:
Connecting Remote Cardiac Monitoring Issues with Care Areas ABSTRACT Patients in noncritical settings may have underlying cardiac conditions or demonstrate unexpected symptoms and condition changes that
Ready to save lives Philips HeartStart XL+ Defibrillator/Monitor Ready to respond, revive, Hospital cardiac emergencies are typically stressful and chaotic events. The last thing you want to worry about
ACLS CHALLENGE FAST TRACK CNE: 0.0 CME: 2.0 This Challenge Fast Track, taught according to the 2010 guidelines, is designed for healthcare providers who either direct or participate in the resuscitation
ACCESSORIES LIFEPAK CR Plus and LIFEPAK EXPRESS Defibrillators Genuine accessories from Physio-Control BATTERY AND ELECTRODE OPTIONS LIFEPAK EXPRESS AED Thank you for choosing Physio-Control as your partner
Basic Life Support & Automated External Defibrillation Check response Shake gently Ask loudly: Are you all right? If not responsive Open airway & check for breathing If not breathing normally or not breathing
Maximizing defibrillator performance Philips HeartStart FR2-Series Defibrillator supplies and accessories Carry cases There are four carry cases available for the HeartStart FR2-Series Defibrillator: the
Guidelines for the perioperative management of patients with implantable pacemakers or implantable cardioverter defibrillators, where the use of surgical diathermy/electrocautery is anticipated. Introduction
ACLS Megacode Case 1: Sinus Bradycardia (Bradycardia VF/Pulseless VT Asystole Out-of-Hospital Scenario You are a paramedic and arrive on the scene to find a 57-year-old woman complaining of indigestion.
Tips for Success Defining EHR System Requirements The number, variety and complexity of EHR systems in today s market has made the search for a system complex and somewhat daunting unless an organized,
POLICIES AND PROCEDURES Title: CODE BLUE Adult / Pediatric I.D. Number: 1012 Authorization: [x] Advanced Life Support Committee [x] SHR Nursing Practice Source: Advanced Life Support Committee Cross Index: