1 Digidata 1322A Operator s Manual Part Number Rev E March 2005 Printed in USA Copyright 2005 Axon Instruments / Molecular Devices Corp. No part of this manual may be reproduced, stored in a retrieval system or transmitted, in any form or by any means, electronic, mechanical, photocopying, microfilming, recording, or otherwise, without written permission from Molecular Devices Corp. QUESTIONS? Phone: 1 (800) Fax: +1 (510) Web:
3 If you encounter difficulties and need technical support, PLEASE read the TROUBLESHOOTING section on page 9 before you call Axon Instruments. i
5 iii COPYRIGHT The information in this manual is copyrighted and must not be reproduced in any form whatsoever without written permission from Axon Instruments / Molecular Devices Corp. VERIFICATION This instrument is extensively tested and thoroughly calibrated before leaving the factory. Nevertheless, researchers should independently verify the basic accuracy of the instrument using suitable test signals. DISCLAIMER This equipment is not intended to be used and should not be used in human experimentation or applied to humans in any way. CAUTION Do not open the unit. There are no user-serviceable parts inside. Copyright, Verification, Disclaimer, Caution
7 Declaration of Conformity v Declaration of Conformity Manufacturer: Type of Equipment: Model Number: Axon Instruments / Molecular Devices Corp Whipple Road Union City, CA USA Signal Conditioning Amplifier Digidata 1322A Year of Manufacture: 2000 Application of Council Directives: EC EMC Directive 89/336/EEC as amended EC Low Voltage Directive 73/23/EEC as amended Harmonized Standards to which Conformity is Declared: EMC: EN 55011: 1998, Class B EN : 1998 Safety: EN : 2001 I, the undersigned, hereby declare that the equipment specified above conforms to the above Directives and Standards. Authorized Signature and Date: (Signature on file) Declaration of Conformity
9 Table of Contents vii Table of Contents Chapter 1 Introduction...1 Introduction to the Digidata 1322A... 1 Components... 2 Minimum Computer Requirements... 2 Recommended Computer System... 2 Notes on the SCSI Adapter Card... 3 Rack Mount Kit... 3 Chapter 2 Installation...5 Instructions... 5 Chapter 3 Troubleshooting...9 Functional Checkout... 9 Step Step Step Troubleshooting Grounding and Minimizing Noise Before you call Chapter 4 Interface Description...15 Front Panel Analog Input Analog Output Digital Output Trigger In Rear Panel Telegraph Inputs Digital Outputs Trigger Output Line Frequency Output Digital Outputs (0-15) Digital Inputs (0-15) Peripherals Table of Contents
10 viii Table of Contents SCSI ID SCSI Ports SCSI Termination Power Supply Chapter 5 Specifications...21 Analog Input Analog Output Digital Inputs Digital Outputs Trigger In Warranty and Service...23 Index...25 Digidata 1322A Operator s Manual, Copyright 2005, Axon Instruments / Molecular Devices, Corp.
11 Introduction 1 Chapter 1 Introduction Introduction to the Digidata 1322A The Digidata 1322A is a high-resolution, low-noise digitizer intended for precision scientific applications. The Digidata 1322A digitizes at an aggregate speed of 500 khz. It provides sixteen multiplexed, 16-bit analog inputs and two nonmultiplexed, 16-bit analog outputs. There are sixteen digital input lines and sixteen digital output lines, as well as Tag and Start digital inputs. The Digidata 1322A communicates with the host computer through the SCSI bus. To minimize the likelihood of installation or performance problems, every Digidata 1322A is supplied from the factory with a SCSI adapter card that should be installed into the computer, although you may choose to use an existing SCSI port. The Digidata 1322A is compatible with both the Windows and Macintosh operating systems. It is a plug-and-play device; it is automatically recognized by Windows. The Digidata 1322A is supported by Clampex and AxoScope 8.2 and 9, and AxoGraph AxoScope is an easy-to-use, full-featured data acquisition program for Windows that is provided free with the Digidata. The Digidata 1322A has higher speed, lower noise and better resolution (greater dynamic range) than the previous Digidata 1200 models. New features also include better instrument telegraph support and line-frequency triggering. The Digidata 1322A is intended for benchtop use, but it is supplied with a simple rack-mount kit. Chapter 1
12 2 Introduction Components Digidata 1322A with power cord SCSI cable, 10 ft. length CD-ROM with AxoScope 9 (Windows only) Manual (includes Warranty Registration Card) SCSI adapter card and device driver CD-ROM Rack Mount Kit Minimum Computer Requirements IBM PC-compatible computer with a Pentium 400 MHz or faster, OR Power Macintosh One open PCI slot 128 MB RAM 200 MB hard disk Parallel port (if using with the parallel port key required by pclamp) 800 x 600 display system (small fonts) Windows 98, NT 4.0 operating system (PC), OR System 8 (Mac) Recommended Computer System IBM PC-compatible computer with a Pentium 2 GHz or faster, OR Power Macintosh G4 One open PCI slot 256 MB RAM (or higher) 1 GB hard disk (or more) Parallel port (if using with the parallel port key required by pclamp) 1024 x 768 display system (large or small fonts) Windows 2000 or XP Pro operating system (PC), OR System 9 (Mac) Digidata 1322A Operator s Manual, Copyright 2005, Axon Instruments / Molecular Devices, Corp.
13 Introduction 3 Notes on the SCSI Adapter Card One SCSI adapter card is provided with every Digidata 1322A. Be sure to read the SCSI Adapter Installation Guide. Some users may prefer to use an existing SCSI port on the computer. While it is possible to use other models of SCSI adapters, Axon Instruments can provide technical support only for the SCSI adapter bundled with the Digidata 1322A. The Digidata 1322A is compatible with the popular Adaptec SCSI adapter cards. The Digidata 1322A and cable use high-density 50-pin connectors. You may need a special adapter or cable to connect to a SCSI adapter with a wide-scsi connector (e.g., Adaptec 2940W or 2940UW). Rack Mount Kit The Digidata 1322A is designed to be a desktop unit. However, some users may want to mount the Digidata 1322A in a rack, so a Rack Mount Kit is provided. The Rack Mount Kit includes the following parts: One Rack Mount Panel Two Side Brackets with set screws Four Screws Refer to the figure on the next page and follow these instructions for installation: 1. Remove the bezel (the frame that surrounds the front panel) a. Use one finger to pull one side of the bezel outward (see figure) to unhook it. b. Remove the bezel. c. Do NOT remove the front panel! 2. Mount the Side Brackets a. Before inserting the Side Brackets, loosen the screws on the front panel (to allow clear access to the grooves on the side panels). b. Insert one Side Bracket on each side of the Digidata 1322A. c. Tighten the screws on the front panel. d. The Side Brackets should be free to move along the grooves of the side panels. Chapter 1
14 4 Introduction 3. Attach the Rack Mount Panel a. Slide the Rack Mount Panel around the front panel. b. Snap the bezel back in place around the front panel. c. Pull the Rack Mount Panel flush up against the bezel. d. Position the Side Brackets up against the Rack Mount Panel. e. Use the four Screws to secure the Rack Mount Panel to the Side Brackets. f. Tighten the Side Bracket set screws up against the side panels. 4. Secure the entire unit in the rack a. Place the unit (Digidata 1322A) plus the attached rack mount assembly in the rack. b. Secure the Rack Mount Panel to the rack with four rack screws (not included). ENCLOSURE FRONT PANEL MOUNTING BRACKET MOUNTING PANEL SMALL CREW MOUNTING BRACKET LARGE SCREW BEZEL Digidata 1322A Operator s Manual, Copyright 2005, Axon Instruments / Molecular Devices, Corp.
15 Installation 5 Chapter 2 Installation Instructions If you have a Digidata 1200A or 1200B system that you still wish to use, it is OK to leave the old Digidata board in the computer and to leave DriverLinx installed during the Digidata 1322A installation. Also, pclamp 9 users do not need to install the AxoScope CD that comes with the Digidata 1322A. AxoScope is automatically installed with pclamp 8 and above. 1. Shut down and turn off the computer. 2. Use the SCSI Host Adapter Installation Guide to install the included SCSI Host Adapter. If you lost the SCSI adapter installation guide, you can find a copy at the Axon web site on the Digidata 1320 Series technical support page: 3. Install AxoScope. If you have pclamp, install pclamp instead of AxoScope. (For Windows systems only. If you are installing on a Macintosh to use with AxoGraph 4.2 or above, see the AxoGraph instructions). a. Insert the AxoScope or pclamp CD-ROM into the CD-ROM drive. b. The setup dialog should automatically appear. If it does not, double-click on the My Computer icon on the Windows desktop. Double-click on the icon for your CD-ROM drive. The installation menu will appear. Chapter 2
16 6 Installation c. Install AxoScope or pclamp (press the third button). At the end of installation, a prompt asks if you want to reboot. Select No. d. Shut down and turn off the computer to connect the Digidata 1322A (it is not advisable to connect to the SCSI port while the computer is on). 4. Connect the Digidata 1322A to the SCSI adapter. a. Carefully connect the SCSI cable to the Digidata 1322A and to the SCSI port. b. Turn on the Digidata 1322A. c. Turn on the computer. d. New Hardware should be found and properly identified as the Digidata 1322A under Windows 98, 2000 and XP Pro. (NT 4.0 is not plug-andplay). 5. Configure the digitizer in AxoScope or pclamp. a. Disconnect all BNC cables. b. Open AxoScope (or Clampex). c. Go to Configure/Digitizer and select Digidata 132x Series in the Digitizer field. Digidata 1322A Operator s Manual, Copyright 2005, Axon Instruments / Molecular Devices, Corp.
17 Installation 7 d. Press the Configure button. In the Configure Digidata 132x dialog, press the Detect button. The word "Detected" will appear and the OK button will be enabled. e. Press the Calibrate button. A dialog will pop up warning you to disconnect any BNC cables from the Analog Outputs. Press OK, and in a couple of seconds it should say, Calibrated underneath the Calibrate button. Press the OK buttons until all the dialogs are closed. We recommend that the Digidata 1322A be calibrated whenever there is a change in the environment, such as moving it to another room or installing it on another computer. f. The Digidata 1322A is now ready to perform experiments. Chapter 2
19 Troubleshooting 9 Chapter 3 Troubleshooting Functional Checkout Step 1 If you are able to configure the Digidata 1322A in either AxoScope or Clampex then the Digidata 1322A is properly installed. Go to Configure/Digitizer. Choose Digidata 1320 in the Digitizer field and then press the Configure button. The Configure Digidata 132x dialog reports which digitizer is installed (1320A, 1321A, or 1322A). It also displays the SCSI ID number and the serial number of the device. The Power-on Holding Levels section is used to set the voltage on both Analog Out channels at the time that the Digidata 1322A is turned on. The initial Digital Out levels can also be set high or low by clicking on the check boxes above the numbers corresponding to each digital output (a check means high, blank means low). Step 2 Check to make sure that you have a proper ASPI driver installation on your system. This should have occurred during the installation of AxoScope or pclamp. Run aspichk.exe (found in the ASPIChk folder under the WinASPI folder on the AxoScope or pclamp CD). If you do not have the latest ASPI drivers installed on your computer, or if you have an improper configuration, ASPIChk will tell you. In this case, open the I386 folder (also under the WinASPI folder) and run aspiinst.exe to install the WinASPI files. Chapter 3
20 10 Troubleshooting Step 3 Check that the Digidata 1322A and SCSI card are properly installed. In Windows 98, go to Start Button/Settings/Control Panel/System/Device Manager tab. The figure shows an example of the Device Manager window. First, click on the plus (+) sign next to Axon Instruments. You should see Axon Digidata Then, click on the plus sign next to SCSI controllers. You should see the label for the SCSI adapter card shipped with the Digidata 1322A. Double-click on the SCSI adapter entry to see the device status. Windows NT 4.0 users can verify the correct installation of the SCSI adapter in a similar manner by going to Start Button/Settings/Control Panel/SCSI Adapters, and looking at the SCSI adapter list. In Windows 2000, go to Start Button/Settings/Control Panel/Administrative Tools/Computer Management/System Tools/Device Manager. The Device Manager is the same as in Windows 98. Digidata 1322A Operator s Manual, Copyright 2005, Axon Instruments / Molecular Devices, Corp.
21 Troubleshooting 11 Troubleshooting If you have any problem with the operation of the Digidata 1322A, follow this section to check out some possible problems and solutions. If your problem is still not resolved, please go through the "Before You Call" checklist below prior to contacting us. 1. If you suspect that the Digidata is not working properly, isolate the problem! Disconnect any external instruments and test the Digidata by itself. If you suspect problems with the analog or digital outputs, use a voltmeter or oscilloscope to monitor the signal. If you suspect a problem with the analog or digital inputs, hook up a known signal source such as a signal generator, or even the analog output if you know it is working properly. 2. The Power light on the front panel is off. Check the fuse. The fuse box is found in the middle of the Power Unit on the rear panel. Simply remove the power cord and then reach underneath the fuse box to open. 3. The screen shows a straight line instead of the input signal. Are all external connections properly made? Check cables for continuity problems. Make sure the SCSI cable is securely attached to both the card and the interface box. Try rebooting the computer and restarting the Digidata. 4. Noise is introduced when the data is digitized. If noise is added to the signal on the analog input, make sure that all cables are routed away from switching power supplies, power cords, monitors, or any other major sources of noise. Check for proper ground connections. See below for more information on proper grounding practices. 5. The Ready light on the front panel flashes continually during operation. The Digidata 1322A is in a problem state. Call Technical Support at Axon Instruments for a Service Request (SR) number before returning the unit for repairs. Chapter 3
22 12 Troubleshooting 6. The Digidata does not work at all or it locks up the computer when doing certain operations. Try resetting the Digidata by turning it off and then back on. Then reboot the computer. Refer to the SCSI Host Adapter Installation Guide to verify that the SCSI adapter card is installed properly. Call Technical Support at Axon Instruments if the problem persists. 7. In Clampex, the Digitizer Gain cannot be changed. This is possible with the 12-bit Digidata 1200A/B. The Digidata 1322A does not include a programmable digitizer gain. The higher 16-bit dynamic range and lower noise makes a programmable gain unnecessary. 8. When troubleshooting the Digidata or your software, it is best to simplify your software configuration by turning off all other programs. It may appear that there are no other programs running in the background. However, this may not be the case. To see if other programs are running in the background click on an empty place on the desktop and then press Ctrl-Alt-Del (holding these three keys down at the same time). This will open the Task Manager in Windows 98. In Windows NT 4.0 and 2000 you then need to click on the Task Manager button. Close all unnecessary programs running in the background. Hint: For optimal acquisition performance in Clampex and/or AxoScope do the following: 1. Configure/Lab Book Options select Never log any events 2. Configure/Program Options select Disable screen saver during data acquisition Digidata 1322A Operator s Manual, Copyright 2005, Axon Instruments / Molecular Devices, Corp.
23 Troubleshooting 13 Grounding and Minimizing Noise To avoid ground-loops, we recommend that you plug-in the Digidata 1322A to the same power-strip as the PC. Also, be aware that each Analog Input BNC on the Digidata 1322A is a single-ended input (all BNC shells are connected to signal ground). When noise in the system occurs, the first step is diagnosis. Take ALL instruments out of their racks, and connect them together with only ONE BNC connection. Observe if the hum (50 60 Hz noise) is eliminated. Also observe if the hum is produced by pickup from the headstage by shielding the headstage and watching the magnitude of the hum. If the hum is eliminated at this step, then connect the second BNC cable. If hum is now observed, this is probably a ground loop that is picking up an alternating magnetic field. Next try to eliminate the source of the alternating magnetic field: a cheap transformer, an electric motor, such as found in a nearby fan or refrigerator. Try to rearrange the two BNC cables to determine if their positioning tells you anything about the source of the alternating field. High frequency components (20 to 50 khz) may also appear if there is a ground loop. These originate from the switching power supply of the computer and can be picked up in the analog signal inputs of the Digidata 1322A. If removing the source of the alternating field is not possible, eliminate the ground loop by making one of the connections between the two instruments without a shield. Make this either with a naked unshielded wire, or with a BNC cable that has its shielding cut at one end. Make a break in the shielding away from the interface, near the connection on the instrument suspected of creating the ground loop. For users of Axon Instruments amplifiers, more information regarding noise reduction procedures can be found in the manuals for the Axopatch 200B, Axopatch-1D, and AxoClamp-2B. Chapter 3
24 14 Troubleshooting Before you call Please write down the following information before you call for technical support. This information can help us identify possible problems and will quickly expose known conflicts. 1. What is the serial number of the Digidata? The serial number is on a small barcode sticker on the back panel. 2. Gather the following computer information: The brand and model of computer you use; the type (e.g., Pentium II) and speed of the microprocessor (e.g., 2.1 GHz); how much RAM is installed; which operating system is loaded (e.g., Windows XP Pro SP2). 3. What other cards are installed in the computer (e.g., network, sound, SCSI, etc.)? If possible, find out what I/O addresses, DMA channels and IRQs are being used. 4. Clampex users: connect Analog Out 0 to Analog In 0. Then run an Episodic mode protocol with a waveform specified. Do you see the waveform when you hit the view button? 5. If you can reproduce a problem by following a series of steps using software, record these steps so that we can reproduce the problem. After following these steps, call Axon Instruments at or send to: Digidata 1322A Operator s Manual, Copyright 2005, Axon Instruments / Molecular Devices, Corp.
25 Interface Description 15 Chapter 4 Interface Description Front Panel The front panel connectors are all BNCs. There are two LEDs: POWER and READY. Upon power-up, the READY LED will blink for a few seconds, then it should remain on continuously. If it does not, please see TROUBLESHOOTING in Chapter 3. Chapter 4
26 16 Interface Description Analog Input There are sixteen 16-bit analog input channels. The BNC shields for the Analog Inputs are connected to the Analog ground. Each channel has a buffer amplifier to drive the analog multiplexer such that cross-talk between the inputs will be minimized. These input channels are used to digitize signals for Clampex or AxoScope. Analog Output The front panel has two 16-bit analog output channels. The average output noise is 150 µv peak-to-peak in a 10 khz bandwidth measured with a single pole filter (with no voltage command applied). Each channel has an operational amplifier to buffer the output signal of the D/A converter. The two analog output channels can be used simultaneously for waveform generation in Clampex. Digital Output Digital Outputs 0-3 are found on the front panel of the Digidata 1322A. The 5 V CMOS level can be used to trigger external devices and the control is provided in Clampex. Trigger In TAG and START are digital inputs compatible with TTL-level signals. The TAG input can be used to mark events (e.g., perfusion ON) in Clampex and AxoScope. The START input can be used to begin data acquisition. Digidata 1322A Operator s Manual, Copyright 2005, Axon Instruments / Molecular Devices, Corp.
27 Interface Description 17 Rear Panel There are several sets of connectors on the rear panel of the Digidata 1322A: ten BNCs, two 25-pin connectors, one 15-pin connector for peripherals, and two SCSI connectors. Telegraph Inputs The Digidata 1322A has a dedicated 12-bit A/D converter that provides four telegraph input channels on the rear panel. These telegraph input channels provide gain, frequency or capacitance values from Axon Instruments amplifiers (i.e., Axopatch 200B). These inputs are independent of the 16 analog input channels. Digital Outputs Digital Outputs 4-7 are found on the rear panel of the Digidata 1322A. Therefore, there are a total of eight Digital Output BNC connectors provided by the Digidata 1322A (four on the front panel and four on the rear panel). The CMOS level outputs are set to high (+ 5 V) or low (0 V). They can be used to trigger external devices and the control is provided in Clampex. Chapter 4
28 18 Interface Description Trigger Output This will support specific trigger output functions in pclamp 9. Line Frequency Output Line voltage frequency output is available from this BNC. This CMOS logic-level output provides a Hz digital signal (square pulse) that represents the frequency from the AC mains power. This signal can be used to trigger external devices at the line frequency. If you wish to trigger the Digidata synchronously with the line frequency, it is preferable to use the internal line frequency trigger which can be selected as one of the trigger sources in the Clampex or AxoScope software. Digital Outputs (0-15) This is a 25-pin female-type connector with the following pin assignments: pin Digital Output Line trigger 6 Analog ground 7 Analog ground Analog ground pin Digital Output 13 Analog ground Analog ground 19 Analog ground 20 Analog ground Analog ground Digidata 1322A Operator s Manual, Copyright 2005, Axon Instruments / Molecular Devices, Corp.
29 Interface Description 19 Digital Inputs (0-15) This is a 25-pin male-type connector with the following pin assignments: pin Digital Input No connect 6 Analog ground 7 Analog ground Analog ground pin Digital Input 13 Analog ground Analog ground 19 Analog ground 20 Analog ground Analog ground Peripherals This port provides a connection to future peripheral devices, such as a signal conditioner. SCSI ID This number identifies the SCSI device ID of the Digidata 1322A. The factory default setting is 4. All devices on the SCSI bus must have a unique ID. SCSI Ports On the rear panel there are two SCSI ports, which are identical. Either one is OK for connecting the Digidata 1322A to a SCSI host adapter card in the computer via the SCSI cable. The additional port is for connecting additional SCSI devices to the SCSI bus. Chapter 4
30 20 Interface Description SCSI Termination The Digidata 1322A s SCSI-2 interface comes with active termination on-board for reliable and high-speed data transfer. The switch on the rear panel provides for enabling and disabling termination. We recommend that this switch remain in the ON position. In a SCSI bus, only the devices at both ends of the bus require termination. If there is more than one SCSI device on the SCSI bus, we recommend that the Digidata 1322A be placed at the end of the SCSI chain. Power Supply The power supply is universal input. Therefore, both 110 and 220 V power sources are OK to use. Digidata 1322A Operator s Manual, Copyright 2005, Axon Instruments / Molecular Devices, Corp.
31 Specifications 21 Chapter 5 Specifications Analog Input Number of input channels: 16 single-ended Resolution: 16-bit, 1 in Acquisition rate (aggregate): 500 khz Input range: V to V Input resistance: 1 MΩ Gain value: 1 Integral linearity error: ±1% No Missing Codes: 16-bit Signal-to-(Noise+Distortion) (f IN =100 khz): 84 db (min) Chapter 5
32 22 Specifications Analog Output Number of channels: 2 Resolution: 16-bit Output voltage ranges: V to V Output impedance: < 0.1 Ω Output short circuit to signal ground: ±25 ma Integral nonlinearity: ±1% Digital Inputs Number of bits: 16 Input type: TTL compatible Digital Outputs Number of bits: 16 Output driver: HC compatible Output current: ±4 ma Trigger In Input type: TAG: START: TTL compatible always level sensitive 2 modes: edge and level Digidata 1322A Operator s Manual, Copyright 2005, Axon Instruments / Molecular Devices, Corp.
33 Warranty and Service 23 Warranty and Service Standard Warranty Axon Instruments / Molecular Devices warrants its non-consumable hardware products to be free from defects in materials and workmanship for 12 months from date of invoice. The warranty covers the cost of parts and labor to repair the product. Products returned to our factory for repair must be properly packaged with transportation charges prepaid and the shipment fully insured. We will pay for the return shipping of the product to the customer. If the shipment is to a location outside the United States, the customer will be responsible for paying all duties, taxes and freight clearance charges if applicable. The warranty is valid when the product is used for its intended purpose and does not cover products which have been modified without approval from Axon Instruments / Molecular Devices, or which have been damaged by abuse, accident or connection to incompatible equipment. To obtain warranty service, follow the procedure described in the Repair Service section. Failure to do so will cause long delays and additional expense to the customer. This warranty is in lieu of all other warranties, expressed or implied. Repair Service The company reserves the right to cease providing repair maintenance, parts and technical support for its non-consumable hardware products five years after a product is discontinued. If you purchased your instrument from a Distributor or OEM Supplier, contact them for repair service. Warranty and Service
34 24 Warranty and Service If you purchased your instrument from Axon Instruments / Molecular Devices, contact our Technical Support Department. If it is determined your instrument must return to the factory for repair, the Technical Support Representative will issue a Service Return (SR) number. Our Logistic Coordinator will contact you with specific instructions. Digidata 1322A Operator s Manual, Copyright 2005, Axon Instruments / Molecular Devices, Corp.
35 Index 25 Index 1200-BNC interface box, 15, 17 ADC Clock Output, 18 Analog Input, 13, 16, 21 Analog Output, 16, 22 AxoScope, 1, 2, 6, 9, 12, 16, 18 Clampex, 1, 6, 9, 12, 14, 16, 17, 18 Copyright, i Digital Input, 19, 22 Digital Output, 16, 17, 18, 22 Ground, 11 Grounding and minimizing noise, 13 Installation, 3, 5 Step-by-Step Instructions, 5 Interface Description Front Panel, 15 Rear Panel, 17 Introduction, 1 Components, 2 Minimum Computer Requirements, 2 Notes on the SCSI Adapter Card, 3 Recommended Computer System, 2 Line Frequency Output, 18 Peripherals, 19, 21 Power Supply, 20, 21 Rack Mount Kit, 3 SCSI ID, 9, 19, 21 SCSI Termination, 20, 21 Service Request Number, 24 Specifications, 21 Technical support, i, 14 Telegraph Inputs, 17 Trigger In, 16 Troubleshooting, i, 11, 15 Before you call, 14 Functional Checkout, 9 Grounding and Minimizing Noise, 13 Warranty, 23 Index
Serial ATA PCI Quick Installation Guide Introducing the Serial ATA PCI The Serial ATA PCI is an ultra high-speed dual channel Serial ATA controller board for use in Pentium-class computers. It achieves
Ikebana IEEE 1394 External DVD-R/RW Drive Quick Start Installation Guide for Windows PC Thank you for purchasing Ikebana IEEE 1394 External DVD-R/RW product. Please follow these installation instructions
Transmitter Interface Program Operational Manual Version 3.0.4 1 Overview The transmitter interface software allows you to adjust configuration settings of your Max solid state transmitters. The following
marian MARC 8 MIDI User's Guide November 2001, Marian. Hardware Design by Marian "SyncBus developed by Marian All rights reserved. No part of this User's Guide may be reproduced or transmitted in any form
SA-9600 Surface Area Software Manual Version 4.0 Introduction The operation and data Presentation of the SA-9600 Surface Area analyzer is performed using a Microsoft Windows based software package. The
Orbit PCI Mk 2 Network Card User Manual Part No. 502566 Issue 4 Information in this document is subject to change without notice. Companies, names and data used in examples herein are fictitious unless
UltraATA 133 PCI Quick Installation Guide Introducing the UltraATA 133 PCI The UltraATA 133 PCI is an ultra high-speed dual channel Ultra ATA/133 controller board for use in Pentium-class computer. Features
If you are installing the following models by means of the SCSI interface then please use this document. Fi-5530C2 Fi-6140 Fi-6670\A Fi-6240 Fi-6770\A Hardware requirements. You will need a PC system that
2940UWPro-ig.qxd 12/21/98 9:55 AM Page 1 INTRODUCTION With the SCSI Card 2940UW Pro, you can connect up to 15 SCSI devices to any IBM-compatible computer with PCI expansion slots. This installation guide
JTAGjet series SIGNUM SYSTEMS CORPORATION USB 2.0 Driver for JTAGjet and ADM51 Installation Instructions COPYRIGHT NOTICE Copyright (c) 2011 by Signum Systems Corporation. All rights are reserved worldwide.
EPSON Scan Server & EPSON TWAIN Pro Network EPSON Scan Server & EPSON TWAIN Pro Network SCANNER UTILITY PROGRAMS All rights reserved. No part of this publication may be reproduced, stored in a retrieval
Quick Operation Guide DAB, DAB + und DMB USB Digital Receiver Albrecht DR 403. 1 Note Thank you for purchasing the DAB/ DMB receiver. This Quick Guide is to guide you to the software installation of the
USB 3.0 to SATA 3.5" Enclosure Installation Guide Introduction The USB 3.0 to SATA 3.5" Enclosure is designed to support USB equipped computers. This enclosure provides large capacity mobile storage using
USB to RS-422/485 Serial Adapter (ID-SC0911-S1/SC0A11-S1) User s Manual 1 Copyright Statement No part of this publication may be reproduced in any form by any means without the prior written permission.
ADDENDUM BIODEX EMG/ANALOG SIGNAL ACCESS CONFIGURATION UTILITY SOFTWARE FOR SYSTEM 3 REVISION 2 AND SYSTEM 4 DYNAMOMETERS 830-000 835-000 840-000 850-000 BIODEX Biodex Medical Systems, Inc. 20 Ramsey Road,
Rev. D PRODUCTIVITY THROUGH INNOVATION 600 CONTROL DIRECT DRIVE TECHNICAL/OPERATION MANUAL 10 BORIGHT AVENUE, KENILWORTH NEW JERSEY 07033 TELEPHONE: 800-524-0273 FAX: 908-686-9317 TABLE OF CONTENTS Page
Zip ATAPI Drive Installation Guide Quick Start Information Installation Notes Technical Notes Troubleshooting Drive Specifications For Integrator/OEM Only http:www.iomega.com Table of Contents Quick Start
QUICK USER GUIDE v1.2.2 Berkeley Audio Design Alpha USB The Alpha USB is an asynchronous High Speed USB to digital audio interface designed to provide the highest possible audio quality from computer audio
Installation Guide N5101A Baseband Studio PCI Card Overview This guide provides instructions on how to install the N5101A Baseband Studio PCI card in a PC. The PCI card is installed in a PC and is used
Installing the Printer Software Checking the Package Contents Connecting the Printer Unpacking the Printer Installing the Ink Cartridges Plugging in the Printer Attaching the Printer Parts 4032216-00 EUL
USB-to-I2C Hardware User s Manual http://www.i2ctools.com/ Information provided in this document is solely for use with the USB-to-I2C product from SB Solutions, Inc. SB Solutions, Inc. reserves the right
NortechCommander Software Operating Manual MAN-00004 R6 If the equipment described herein bears the symbol, the said equipment complies with the applicable European Union Directive and Standards mentioned
Data Cable Modem Model# U10C018 User s Guide Rev. 1.02 July 26 st, 2004 Contents 1. BEFORE YOU BEGIN...4 Understand the Cable Modem s Features...4 Contact Your Local Cable Operator...4 Prepare Your Area
Intel RAID Controller Troubleshooting Guide A Guide for Technically Qualified Assemblers of Intel Identified Subassemblies/Products Intel order number C18781-001 September 2, 2002 Revision History Troubleshooting
SATA II-150 PCI Quick Installation Guide Introducing the SATA II-150 PCI The SATA II-150 PCI is an ultra high-speed dual channel Serial ATA Generation 2 controller for use in PCI enabled systems. Features
1 Unpack the box Check that you have all of the following items. If anything is missing, contact your supplier. SRS stick reader (the stick reader) USB flash drive/cd Contains Data Link software and reference
About the software The most recent versions of the applications contained on the accessory disc can be downloaded from the Korg website (http://www.korg.com). -MIDI Setup Guide Please note before use Copyright
USB2.0 EASY IDE ADAPTER INDEX Trademarks ---------------------------------------------------------------------------- Introduction ---------------------------------------------------------------------------
USER GUIDE USB 2.0 TO DVI-I ADAPTER Model: 30535 1 TABLE OF CONTENTS Introduction...2 Features...3 System Requirements...4 Package Contents...5 Device Overview...6 Device Driver Installation on Windows
PCMCIA 1 PORT RS422/485 1.2 EDITION OCTOBER 1999 Guarantee. FULL 36 MONTHS GUARANTEE. We guarantee your interface card for a full 36 months from purchase, parts and labour, provided it has been used in
Machine type: 10087/3110 Lenovo IdeaCentre Q180 Series User Guide Version 1.0 2011.09 31500665 Important Safety Information Before using this manual, it is important that you read and understand all of
User s Manual USB 1.1 Serial adapter Table of Contents Safety Instructions..... 1 Introduction...... 1 Features........2 System Requirements.......2 Package Contents...... 2 Specification..... 3 Driver
Introduction USB 3.0 to SATA 6Gb/s Adapter Quick Installation Guide The USB 3.0 to SATA 6Gb/s Adapter quickly adds SATA hard disk or SATA DVD/CD-ROM to your system through USB 3.0 interface. Features and
PL-2303 USB to Serial Adapter User Installation Manual (For Vendor Reference Only) For Windows 98/ME/2000/XP Release Version 1.0 (4/08/2002) Contents Introduction Features & Specifications System Requirements
USB 2.0 RG2304 4-Port USB 2.0 100m CAT 5e/6/7 Extender System User Guide Thank you for purchasing the USB 2.0 RG2304 Extender System. Please read this guide thoroughly. This document applies to Part Numbers:
Ultra ATA/133 Controller PCI Card user manual Model 158039 MAN-158039-UM-0508-02 Thank you for purchasing this MANHATTAN Ultra ATA/133 Controller PCI Card, Model 158039. Compatible with IDE and ATAPI drives
Serial ATA 2 Ports PCI Host This Manual is for the various Model of 2ports SATA HBA: Model A: Low Profile Serial ATA Internal 2 Ports Model B: Low Profile External 1 Port + Internal 1 Port Model C: Low
R Installation Guide APA-1460 SlimSCSI PCMCIA-to-SCSI Host Adapter Introduction This document explains how to install and use Adaptec s APA -1460 SlimSCSI PCMCIA-to-SCSI adapters. The SlimSCSI adapters
Messager USB Installation & User Manual Note: All units support MP3 playback of up to 128Kbps @ 48KHz. Units with serial numbers starting with 15 or above will support the addition of WAV playback done
Troubleshooting Tips Lifestyle SA-2 & SA-3 Amplifier Refer to the Lifestyle SA-2 & SA-3 Amplifier service manuals, part number 271720 for schematics, PCB layouts and parts lists. Preventative Repair Measures
USB to VGA Adapter USB2VGAE2 Instruction Manual USB 2.0 to VGA Multi Monitor External Video Adapter *Please ensure that the driver installation is complete prior to the physical installation of the USB
2.5" USB 2.0/eSATA to SATA Enclosure Quick Installation Guide Introducing the USB 2.0 2.5" Enclosure The 2.5" USB 2.0/eSATA to SATA Enclosure is designed to support USB or esata equipped computers. This
CONTROL SWITCHER User s Manual (Preliminary) www.voodoolab.com Copyright 2010 by Digital Music Corporation. This publication is protected by copyright and all rights are reserved. Voodoo Lab, Control Switcher,
Hi-Speed USB Flash Disk UFD Utility User s Manual For AP v188.8.131.52 Release For Windows 98/ME/2000/XP Version 1.1B (08/27/2004) Contents Contents...1 Introduction...1 Features & Specifications...2 System
User Guide External Hard Disk with USB 1.1/2.0 and IEEE 1394 connection Copyright 2003 All rights reserved. This manual is protected by Copyright. The Copyright is owned by Medion. Trademarks: MS-DOS and
3-Port FireWire 800/400 PCI Host Card 400 FireWire Low Included FireWire Upgrade to FireWire800! U S E R S M A N U A L Low FireWire 400 Included Package Contents : 1 3-Port FireWire 800/400 PCI Host Card
PCI-to-SATA RAID Adapter AEC-6890M User s Manual Version:1.0 Copyright 2003 ACARD Technology Corp. Release: August 2003 Copyright and Trademarks The information of the product in this manual is subject
OmniView 4-Port USB KVM Switch User Guide P73428 F1D104-USB Belkin Components F1D104-USB - OMNIVIEW SE 4-Port READ BEFORE INSTALLATION Tech Notes: 1) Windows 98/98 SE systems o HID (Human Interface Device)
IDE ATA/133 Controller 2 Port PCI ATA-133 IDE Adapter Card PCIIDE2 Actual product may vary from photo FCC Compliance Statement This equipment has been tested and found to comply with the limits for a Class
Laser Wireless Rechargeable Mouse Model #: LM8000WR User s Manual Rev. 060801 User s Record: To provide quality customer service and technical support, it is suggested that you keep the following information
SATA II-150 PCI RAID Quick Installation Guide Introducing the SATA II-150 PCI RAID The SATA II-150 PCI RAID is an ultra high-speed two channel Serial ATA Generation II RAID controller for use in PCI enabled
Distributed by: www.jameco.com 1-800-831-4242 The content and copyrights of the attached material are the property of its owner. User s Manual USB 2.0 DOCK Index INTRODUCTION...3 SYSTEM REQUIREMENTS...3
FireWire 800 + Hi-Speed USB Combo Quick Installation Guide Introducing the FireWire 800 + Hi-Speed USB Combo The FireWire 800+Hi-Speed USB Combo features the latest technology in FireWire 800 and USB 2.0.
Calibration Kit These instructions provide information on the installation, connection and operation of the Calibration Kit for use with SOR 800 Series pressure products, specifically the 805PT and 805QS.
Guide Port Interface Adapter User s Manual Model GPINT-PT Revision 2.0 Copyright 2004-2005, Shoestring Astronomy Page 1 Introduction The Shoestring Astronomy Guide Port Interface Adapter is designed specifically
Kramer Electronics, Ltd. USER MANUAL Models: VP-200N, 1:2 High Resolution XGA DA VP-300N, 1:3 High Resolution XGA DA VP-400N, 1:4 High Resolution XGA DA Contents Contents 1 Introduction 1 2 Getting Started
MICROGATE SERIAL COMMUNICATIONS RUN TIME KIT WINDOWS GUIDE MicroGate Systems, Ltd http://www.microgate.com MicroGate and SyncLink are registered trademarks of MicroGate Systems, Ltd. Copyright 2012-2016
2-Port RS232 Serial PCIe with 16950 UART Quick Installation Guide Introduction The 2-Port RS232 Serial PCIe with 16950 UART is a high-speed serial card that provides two additional serial ports to your
EPSON Stylus C84 EPSON Stylus Start Here C84 CPD-16675R1 4043482-00 XXX 1 Unpack 1 Remove everything from the box. Paper support Printer Manual Power cord Ink cartridges CD-ROM 2 Remove all the blue tape
Kramer Electronics, Ltd. USER MANUAL Models: VP-200N, 1:2 High Resolution XGA DA VP-300N, 1:3 High Resolution XGA DA VP-400N, 1:4 High Resolution XGA DA Contents Contents 1 Introduction 1 2 Getting Started
PROCESS FIELD BUS ProfiCore Installation manual Version 1.0-26 may 2004 Page 1-19 Copyright 2004 PROFIBUS Center Nederland Contact information: Tel. +31-(0)174-671800 Fax. +31-(0)174-671801 Email: email@example.com
NETWORK ADAPTER 10/100 Mbits/sec PCI Ethernet Network Card ST100S ST100SW ST100SLP Instruction Guide * Actual product may vary from photo The Professionals Source For Hard-to-Find Computer Parts FCC COMPLIANCE
PCI-to-SATA Adapter AEC-6290M User s Manual Version:1.0 Copyright 2003 ACARD Technology Corp. Release: November 2003 Copyright and Trademarks The information of the product in this manual is subject to
Using a Laptop Computer with a USB or Serial Port Adapter to Communicate With the Eagle System ECU DB9 USB 20-060_A.DOC Page 1 of 18 9/15/2009 2009 Precision Airmotive LLC This publication may not be copied
EXTERNAL HARD DRIVE Maxtor OneTouch User's Guide TM II P/N:20259600 i Contents 1 Introduction................................................. 1 Maxtor OneTouch II Drive Components............................
USB to Parallel Printer Cable (16898) and USB to db25 Parallel Printer Cable (16899) User s Manual Congratulations on your purchase of the Cables To Go USB-Parallel Cable adapter. This unique cable adapter
User's Guide 99 Washington Street Melrose, MA 02176 Phone 781-665-1400 Toll Free 1-800-517-8431 Visit us at www.testequipmentdepot.com Back to the Extech 407780 Product Page Integrating Sound Level Datalogger
HEX User s Manual (Preliminary) www.voodoolab.com Copyright 2014 by Digital Music Corporation. This publication is protected by copyright and all rights are reserved. Voodoo Lab, Pedal Power, Ground Control,
Introduction USB Over IP 1-Port Quick Installation Guide The USB Over IP 1-Port allows multiple users to share an external hard disk, Multi-Function printer, or other USB devices within a network. Features
FEATURES & APPLICATIONS Inexpensive 2 wire to 3 wire start controller for electric start high speed gas generators. Optimized for use with Outback Invertors. Supports three types of 3 wire generator control
PCI-to-PC Card Pro Quick Installation Guide Features and Benefits Compliant with PCI local bus specification 2.2 Compliant with PC card standard release 8.0 specification Supports CardBus (Card-32) and
T3 Mux M13 Multiplexer User Manual [Type the abstract of the document here. The abstract is typically a short summary of the contents of the document. Type the abstract of the document here. The abstract
User s Manual Home Series CR-H252 2-Bay Portable RAID Tower EXTERNAL 2-BAY SATA RAID USB3.0 ENCLOSURE 1 OF 13 Table of contents Chapter 1 Read me first... 3 1-1 Copyright and Trademark Notice...3 1-2 About
The RIDZ 8x2 Audio Switcher Engineering Manual Support Number 800-765-2930 International 712-852-2813 Table of Contents General Information for the RIDZ (8 x 2) Switcher..... 3 Input 9 on the RIDZ Switcher....6
Installing the Dionex PeakNet System Follow these installation instructions step-by-step, in the order specified. Failure to do so will prevent the PeakNet system from operating. 1999 Dionex Corporation
USER S MANUAL Model: PCI-16 HOST INTERFACE CARD FOR PCI BUS COMPUTERS InstruTECH Precision Instrumentation for Biological Research 20 Vanderventer Avenue, Suite 101E Port Washington, New York 11050-3752
CONVERTER DRIVE (Secure Digital Card Reader-Writer) ITEM NUMBER: CR-35TD Thank You! Thank you for purchasing the Digital Concepts Converter Drive. By inserting a Secure Digital card into your Converter
Installation Guide MAXTOR SERIAL ATA HARD DRIVE FCC Declaration of Conformance: This device complies with part 15 of the FCC Rules. Operation is subject to the following two conditions: (1) this device
Introduction USB to Serial Quick Installation Guide The USB to Serial adapter converts one USB port to multiple 9-pin (DB9) serial ports. Features and Benefits Full support for Windows-based software Low
MADRIX NEO Quick Start Guide & User Manual 4 th Edition, Last Update: March 2011 Thank You for Purchasing MADRIX NEO Please read this manual carefully and thoroughly before using the MADRIX NEO. Make sure