1 Page 1 of 21 OWON Oscilloscope Installation and Use Guide Welcome to use Oscilloscope analysis software. The kind of communication software is use to acquire, store, analyze and display the data. The instruction book and the following helps are for your reference. I. PC Software USB Driver Install Guide USB with better rate and reliability is a kind of data transmission mode to widely use in connecting with PC.(ps: USB interface of Hand Hold type device series is mini USB. Please refer the device instruction) The serial port is another kind of transmission mode in some outdated PC as a supplementary in failure of USB transmission. The above two are both fit for Windows NT(2000, XP, Vista),while USB driver in Win 98 only for manual installation. (1). For Windows XP or Windows 2000 (1-1-1) Notice: for both x86 and x64. Plug into the running well device to open [Found New Hardware Wizard] dialog. Or you can right click [My Computer] and select [Manage], in the left area of opened [Computer Management] select [Device Manager], double click the item [USB Device] with? in the middle area to open the Wizard, In the Wizard, select [No, not this time],
2 Page 2 of 21 select [Install from a list or specific location(advanced)], select [Search for the best driver in these locations.], then select [Include this location in the search] and indicate a directory location for USB driver which is named as USBDRV and under the directory where you installed the program at,
3 Page 3 of 21 Then the installation is running, And complete,
4 Page 4 of 21 And prompt as installed, And show installed in [Device Management], Now you can use the program and use if for USB communication. If there is an early version of USB driver in your computer, you could try running reinstall.bat to fix, the file is under the directory of USBDRV. (2). For Windows Vista or Windows 7 The Microsoft Windows Systems since Windows Vista or Windows 7, change a lot, which require a new installation guide of USB driver. Here it is. During the whole installation, please assure that the device is running well and plugged into PC from USB.
5 Page 5 of 21 Right click [Computer], you can find it on the desktop, or in [Start] menu. in the pop up menu, click [Manage] and it will open a window named "Computer Management", as follow, in the left side click [Device Manager], it will show a devices tree in the middle, and then click the last one button "Scan for hardware changes" in tool bar as follow, and if the device is running well and plugged into PC, computer will detect an unknown device with a "!" icon. Right click the unknown device icon, in the pop up menu click [Update Driver Software...], In the open window, select [Browse my computer for driver software],
6 Page 6 of 21 The next window, select a directory path for the driver software location, and click "Next", Notice: the driver software location is a directory that is under the software setup folder named "USBDRV", and the contents inside are like these:
7 Page 7 of 21 OK, back to the driver installing, after last "Next" step, the system is installing driver software for you, as follow, In the course, It (for Windows XP x86&x64, Windows Vista x86&x64, Windows 7 x86) may open a window named "Windows Security" as below, and just select "Install this driver software anyway" to continue,
8 Page 8 of 21 Or sometimes it(for Windows 7 x64) may open a window named "Windows Security" as below, and just click "Install" to continue, And then continue installing, And finish. Now a successful installation window opens with information "Windows has successfully updated your driver software".
9 Page 9 of 21 Close the window, have a look at the "Computer Management" window, you will find a device under [LibUSB- Win32 Devices], it should be like this: Now the USB driver will work. (3). Serial Port connection Connect with serial port in PC directly. II. User Interface When the software enters into the main interface as Fig 2-2 which has eight parts:
10 Page 10 of 21 Fig Menu Bar: provide various functions introductions. 2. Tool Bar: provide usual shortcut bar of menu options. 3. Waveform displayed range: the main range waveforms displayed. 4. Cursor range: provide the following four cursor types to observe and measure waveforms: none, horizontal line, vertical line, double line(the horizontal line and the vertical line shows simultaneously) 5. Channel box range: in which provides information and set waveforms displayed and hidden, also set one channel as current waveform. 6. Time base division range: which is used for observe and set the time base of the current waveform. 7. Voltage division range: which is used for observe and set the voltage division of the current waveform. 8. Parameter displayed range: which shows the period, frequency and pk-pk of the current waveform. If the new waveform is non-periodicity, the period and the frequency are all 0. III. Menu: The following menus shows in the menu bar: 1- Files 2- View 3- Format 4- Communication 5- Language 6- Help The followings are the detailed description: (1). File (Fig 3-1)
11 Page 11 of 21 Fig Open: open the saved files with bin suffix. 2. Open recent: save the 10 open recent files 3. Save: save the current waveforms as pictures and support bmp, png, gif etc. 4. Print preview: preview the print effect 5. Print: by printer 6. Page setup: set the boundary value for printing 7. Exit: exit from the software (1). View menu(fig 3-2) Fig XY waveform: The voltage values of sampling point in CH1, CH2 are shown as X,Y of point coordinates. If only one of the channels, the function is not available. 2. Value list: The sequence of voltage value in sampling point of every channel is shown in the list which could be saved as.txt or.xls (Windows Office Excel) files exported to other documents. The list supply check all or check none, the right part as saved channel and click EXIT to close the list. Fig 3-2-2: Value list 1. Grids color: Bring out the color dialogue box and change the color.
12 Page 12 of Background color: double click waveform area of display to bring out the color dialogue box and change the background color. 3. Grids line: display or hide the grid scale of image background. (3). Format menu(fig 3-3) Fig Data line: draw the sampling point and connect by lines. 2. Data point: draw the sampling point discretely 3. Waveform inverse: inverse the channel voltage value. (4). Communications menu (Fig 3-4) Fig Set as USB and search USB port from connecting oscilloscopes then refresh connecting USB ports Fig Port setting: choose the port for communicate with oscilloscope Set as port with default parameters(115200,8,n,1). Fig 3-4-3: transport setting windows Fig The bottom of port setting window is the auto obtained area in which interval time and save catalogue of auto obtained could be set up.
13 Page 13 of data acquirement: acquire waveform data from instrument Note: You can get the data from instrument after connecting PC with USB cable and install driver and then choose the correct interface setting. Click "browse" to bring out the "save file" dialogue box and set the save path and file name. File type may be Vector or Bitmap corresponding suffix.bin or.bmp which will be acquired after correct setting in the instrument. On beginning the file format can be viewed from file types and choose "close window and load" when finishing, only bin file can load software and show, bmp open by picture tool. Fig auto acquirement: Gain data using interval time in interface set based on continuous "data acquirement" until clicking Stop.( the interval time should be 500ms or more to guarantee transmit and operation.) Vector data will be uploaded and shown immediately, recorded in hardware. Bitmap file recorded at the same way except for uploading.. The progress bar when auto acquiring data by interface will be shown at the right bottom as follows: Fig 3-4-5: Stop: Stop auto acquiring 4. Auto Player: automatically display the waveform data. Before using the function of Auto Player, you should make sure that you used the function of auto acquirement to record data files in hardware, which use a data record directory path. When you open the AutoPlayer Dialog, [Add] this directory path as a display [History], Choose [Play Mode] as [Turn] or [Reverse], set the [Time Delay(ms)], then you can click the Start Arrow Button to display the record waveform data, it also provides Stop, Next, Previous to operate, you can also drag the progress item to location. Fig oauto Play Window (5). Language(Fig 3-5) Fig 3-5: Language menu Chinese-English transfer (6). Help menu(fig 3-6)
14 Page 14 of 21 Fig Help: Open help file 2. Update: Find new version in server and download update as follows: IV. Toolbar: Fig 3-6-2: About: information Toolbar shortcuts common menu items about 12 numbers of knobs as follows: icon name function 1 open Open saved files with suffix bin 2 Print preview Preview the print effect 3 print Print by printer 4 Data display Display every voltage sequence of sampling point 5 Display/hide grids Display or hide grids scale of waveform background 6 Display linked waveform Draw the sampling point and connect in direct line 7 Display data point Draw the sampling point discretely 8 Inverse waveform inverse 9 Communication setting 10 Manual acquirement Set communication parameter 11 Auto acquirement Auto acquiring files Open waveform acquirement interface 12 stop Stop auto acquiring 13 AutoPlay Auto play the.bin file 14 help Open help files IIV. Relevant operations: Relevant operations (1).Operation for waveforms display range: (Fig 5-1) 1. operation for waveforms display range: (Fig ) (1). waveform moving up and down
15 Page 15 of 21 Firstly, the position as Fig (1->)is the zero voltage position of current waveform. 1) fine adjustment: drag "1->" and make the waveform move up and down on the screen 2) coarse adjustment: drag the ruler slider on the right and make the waveform move up and down, the moving area is the maximum movable area on time base division ( which could be check the current blocks on the right of cursor area, a block 1/8 of height of the screen.) (2). waveform move left and right: Move the mouse to the waveform, it becomes icon, then drag the mouse to make the waveform left and right as Fig Fig ). Change the color of waveform: double click "1->" to bring out color dialogue box to change the color. 2). The time base division of current waveform can be shown and adjusted in such area (as Fig 5-1-2) which is the time range about one scale on the vertical line (8 scales in the following), and adjust the scale in the come box, the waveform zoom to the corresponding division with the centre scale mark as centre shaft.(fig 5-1-3)
16 Page 16 of 21 Fig 5-1-2
17 Page 17 of 21 Fig In this aspect, voltage division can be use to show and adjust the current waveform which is one scale on the horizontal line (12 scales in the following, different series with different format which are only for reference.) and adjust the scale in the come box, the waveform zoom to the corresponding division with the zero voltage position as centre shaft.
18 Page 18 of 21 (2). Operation in cursor area(fig 5-2) Fig There are four cursor types for selection to measure and locate, such as: none, horizontal line, vertical line, double line (that is horizontal line and vertical line display at the same time) (Fig 5-2-1)
19 Page 19 of 21 Fig None: without measure cursor 2. Horizontal line: Drag the two horizontal lines in the waveform area up and down to measure every voltage value on the vertical of current waveform, as the above Fig the Y1,Y2 shows the actual voltage values of the two vertical lines corresponding to the zero point of current waveforms and dy=y1-y2 which is the voltage difference value of the two cursor lines. 3. Vertical line: Drag the two vertical lines in the waveform area left and right to measure every time value on the horizontal of current waveform, as the above Fig the x1,x2 shows the time value of the two vertical lines( the leftmost is zero value of time), dx=x1-x2 which is the time difference value of the two cursor lines. 4. Double lines (Fig 5-1-3): The vertical line and the horizontal line are all in the waveform area and be measured at the same time. (3). Operation in channel box (Fig 5-3) The user can click some channel in the channel box or some waveform in such area to set the current waveform ( select 1-> or sampling point in the waveform), then the selected channel would be high light marked and the corresponding data will be displayed in the time base division, voltage division and parameter area simultaneously.
20 Page 20 of 21 Fig It is index page... Select INDEX and CONTENTS of INSERT menu in the MS-Word, then select INDEX and click OK. (4). Operation for channel inverse Operation for inverse: inverse the channel voltage value. Before channel inversed:
21 Page 21 of 21 After channel inversed: Notice: the channel inversed status will keep on and apply for next waveform file's same channel as long as the software is not closed.
Power Log PC Application Software Users Manual May 2007 Rev. 1, 5/08 2007-2008 Fluke Corporation. All rights reserved. Specifications are subject to change without notice. All product names are trademarks
SA-9600 Surface Area Software Manual Version 4.0 Introduction The operation and data Presentation of the SA-9600 Surface Area analyzer is performed using a Microsoft Windows based software package. The
Prior to using the data modem features of your 9555 handset; you must first install the VCOMPORT 9555 USB Driver. This driver is compatible with: Windows XP Service Pack 2, Windows 2000 Service Pack 4,
Overview Downloading Images and Movies from a Digital Camera to OLYMPUS Master 2 This article describes how to download image and movie files from an Olympus digital camera 1 to the OLYMPUS Master 2 application.
Section 5: Installing the Print Driver In This Section Print Driver Installation for USB Connection Using the Ethernet Connection and a Crossover Cable Print Driver: The Epilog print driver is the software
Midland BT Updater After you downloaded the BT Updater setup application from the web site, double click on it and follow the installation procedure. The BTUpdater application is automatically installed
Windows XP.. ELITE Firmware Update procedures.. v1.41 Download and Extract the ELITE update. Note.. Please ensure you are logged in to Windows as an Administrator before carrying out the instructions in
Installing the Gerber P2C Plotter USB Driver 1 You can install a Gerber P2C plotter using a USB connection and communicate with it using compatible design software. The following procedures describe installing
DT-CONFIG SOFTWARE Config software for D2 systems USER S MANUAL CONTENTS 1. Introductions ------------------------------------------------------------------- 3 2. System Requirement and Connection ----------------------------------------
Centurion PLUS CPC4 Download Guide using C4 File Transfer Utility. 1010537 03 01 10 Section 50 1.0 Background: 1.1 The Centurion PLUS Control system consists of a Centurion PLUS Core (CPC4 1) and Color
User Manual Thermo Scientific Orion Orion Star Com Software Program 68X637901 Revision A April 2013 Contents Chapter 1... 4 Introduction... 4 Star Com Functions... 5 Chapter 2... 6 Software Installation
ScanShell.Net Install Guide Please install the software first - DO NOT PLUG IN THE SCANNER The scanner has been carefully packaged to avoid damage during transportation. Before operating the scanner, please
Install FileZilla Client Secure FTP is Middle Georgia State College s supported sftp client for accessing your Web folder on Webdav howeve you may use FileZilla or other FTP clients so long as they support
130 LogTag User Guide (1.8) Resolving USB Driver Problems Every USB device requires the appropriate driver files to be installed prior to them being successfully used on a computer. The installation process
File Management Utility T u t o r i a l Contents System Requirements... 2 Preparing Files for Transfer to GlobalMark... 2 Application Launch... 2 Printer Setup... 2 Communication Status... 4 Communication
User's Guide Using RFDBManager For 433 MHz / 2.4 GHz RF Version 1.23.01 Copyright Notice Copyright 2005 Syntech Information Company Limited. All rights reserved The software contains proprietary information
Legal Notes Unauthorized reproduction of all or part of this guide is prohibited. The information in this guide is subject to change without notice. We cannot be held liable for any problems arising from
HP LaserJet MFP Analog Fax Accessory 300 Send Fax Driver Guide Copyright and License 2008 Copyright Hewlett-Packard Development Company, L.P. Reproduction, adaptation, or translation without prior written
Trace.Net OTDR Test Management Software User s guide 2007-2009, AFL Telecommunications, all rights reserved. TRAC-01-1000 Revision B, 2009-06-10 Specifications are subject to change without notice. I Contents
SETTING UP YOUR REMOTE DESKTOP This section will assist you in setting you Remote Desktop Connection. This will allow you to create an icon for easy access to your virtual desktop. DISCLAIMER: All images
ScanWin Installation and Windows 7-64 bit operating system In order to run the ScanWin Pro install and program on Windows 7 64 bit operating system you need to install a Virtual PC and then install a valid,
How to Download Images Using Olympus Auto-Connect USB Cameras and Olympus Master Introduction Auto-Connect USB is a feature that allows Olympus digital cameras to emulate a Hard disk drive when connected
EasySuite User Guide V1.0 1 Launch EasySuite...3 Stop the EasySuite:...4 System Requirements...5 EasyCopy...6 Function Description...7 File Transfer...9 EasyMover...12 Function Descriptions of EasyMover...12
SW43W FlukeView Power Quality Analyzer Software Version 3.20 onwards Users Manual 4822 872 30778 December 2005, Rev.1, 04/06 2005 Fluke Corporation, All rights reserved. All product names are trademarks
USB DATA Link Cable USER MANUAL (Model: DA-70011-1) GO! Suite Quick Start Guide - GO! Suite Introduction - GO! Bridge Introduction - Remote Share Introduction - Folder Sync - Outlook Sync GO! Bridge for
OS X LION SET UP THE SYSTEM OS X Lion Set Up the System Last Edited: 2012-07-10 1 Personalize the Interface... 3 Organize the Desktop... 3 Configure Apple Trackpad... 4 Configure Apple Magic Mouse... 6
Trace600 OTDR Test Management Software User s guide T e s t & I n s p e c t i o n A Trace600 OTDR Test Management Software User s guide T e s t & I n s p e c t i o n 2002, AFL Telecommunications, all rights
Overview This application note describes the steps necessary to configure the SST DeviceNet OPC Server for use under Windows NT (Service Pack 3 or higher). This example shows how to set up a configuration
USB Driver Installation for Windows XP USB Serial Converter Driver Installation for Windows XP CAUTION: You must use the drivers on the CD-ROM supplied with your USB Device. DO NOT download drivers from
EigenLight S500 Power Monitor Software Manual Software Installation... 1 Installing S500 Power Monitor Software and LabVIEW Run-time Engine... 1 Install Drivers for Windows XP... 4 Install VISA run-time...
testo easyheat Configuration and Analysis software Instruction manual en 2 General Information General Information This documentation includes important information about the features and application of
Installing the OKI MCx61MFP USB Attached, in Windows 8 Preface Note: The following instruction uses screenshots of either the OKI MC561MFP for illustration purposes. However, these instructions apply to
Software User's Guide Brother QL-series The contents of this guide and the specifications of this product are subject to change without notice. Brother reserves the right to make changes without notice
SPLM DIGITAL SOUND PRESSURE LEVEL METER SoundLAB Software INSTALLATION & INSTRUCTION MANUAL Download Software from the Bogen website: www.bogen.com/software Specifications are subject to change without
QUICK START GUIDE SG2 Client - Programming Software SG2 Series Programmable Logic Relay SG2 Client Programming Software T he SG2 Client software is the program editor for the SG2 Series Programmable Logic
PT. KOROSI SPECINDO Corrosion Monitoring System Eng. Doc. No.: KS-MNL-KMS Korosi Monitoring System (KMS) Software Manual Revision Date Prepared Approved Description 0 23/05/2008 ES JH Issued for approval
CODESOFT Installation Scenarios NOTES: CODESOFT is a separate install from existing versions of CODESOFT. You will need to make note of your current settings (default directories, etc.) so you can duplicate
Legal Notes Unauthorized reproduction of all or part of this guide is prohibited. The information in this guide is subject to change for improvement without notice. We cannot be held liable for any problems
i -CEN S i -CEN'S USER S Manual 2007. 08. 13. i -CEN S Table of Contents Overview of i-cen S software... 4 1 Introduction of i-cen S... 4 2 Key Features... 5 3 Key Benefits... 5 4 System Specification...
NETWORK PRINT MONITOR User Guide Legal Notes Unauthorized reproduction of all or part of this guide is prohibited. The information in this guide is subject to change without notice. We cannot be held liable
USB DRIVER INSTALLATION GUIDE Use these instructions to install the USB drivers and Microsoft synchronization software for your Pathfinder 6057 printer. You need these utilities for your printer and computer
FreeCapture - A PC remote control program for GDS-800 SERIES (RS-232/GPIB) Version 2.0 Overview Thank you for using GDS-800 SERIES PC remote control program. This program is a demo tool for exploring our
Task Force on Technology EXCEL Basic terminology Spreadsheet A spreadsheet is an electronic document that stores various types of data. There are vertical columns and horizontal rows. A cell is where the
Please Install Your EASE Scan Tool DVD Before Installing Any Other Software That Came With Your Package. SECTION I - Installation ATTENTION: Do NOT connect an EASE Vehicle Interface Device to your Computer
DigiFox Professional Instruction Manual Ver1.0 Copyright Our company owns all rights of this unpublished work and intends to maintain it as a confidential work. We may also seek to maintain this work as
ProjectWise Explorer V8i User Manual for Subconsultants & Team Members submitted to Michael Baker International Subconsultants & Team Members submitted by Michael Baker International ProjectWise Support
Chapter 26 EasyPrinter... 2 26.1 Using EasyPrinter as a Printer Server... 3 26.1.1 Setup Procedure in EasyPrinter... 3 26.1.2 Setup Procedure in EasyBuilder... 4 26.2 Using EasyPrinter as a Backup Server...
SMS (Server Management Software) Digital Video Recorder User s Manual 1 Table of Contents Chapter 1 Overview.....03 1.1 In This Manual..03 1.2 Overview 03 1.3 SMS Features...03 1.4 Product Information..03
MULTIFUNCTIONAL DIGITAL SYSTEMS Network Fax Guide 2009 KYOCERA MITA Corporation All rights reserved Preface Thank you for purchasing Multifunctional Digital Color Systems. This manual explains the instructions
Fleet Maintenance Software Welcome Thank you for taking time to review FleetWise VB Maintenance Management Made Simple. This guide is intended to provide a quick overview of installing the software and
Concession FTP User Guide May 2011 Version 1.2 Concessions FTP User guide Page 1 of 41 Contents Objective... 3 FTP User Guide Microsoft Windows XP... 5 Anatomy of the Files... 9 Working on your Files....
26-1 26.EasyPrinter This chapter explains the setup steps of EasyPrinter. 26.1. Overview... 26-2 26.2. Using EasyPrinter as a Printer Server... 26-3 26.3. Using EasyPrinter as a Backup Sever... 26-5 26.4.
Transferring Scans from your Dolphin into Destiny Summary: This article is intended to provide Destiny users with an overview of the following tasks 1. Downloading the Dolphin Transfer Utility to their
KI6501 Data Manager Software User Manual CONTENTS 1. Installation of USB Virtual COM Port driver software... 2 2. Installation of KI6501 Manager Software... 2 3. Connecting KI6501 to PC (Personal Computer)
Select Correct USB Driver Windows often installs updated drivers automatically, and defaults to this latest version. Not all of these drivers are compatible with our software. If you are experiencing communications
BioWin Network Installation Introduction This document outlines the procedures for installing the network version of BioWin. There are three parts to the network version installation: 1. The installation
TPG MOBILE BROADBAND QUICK SETUP GUIDE TPG USB E1762 Modem Thank you for choosing TPG Mobile Broadband and USB E1762 modem. This User Guide provides step by step instructions that will take you through
PC Configuration Requirements: Configuration Requirements for ipc Operation The hardware and operating system of the PC must comply with a list of minimum requirements for proper operation with the ipc
Guide to Installing BBL Crystal MIND on Windows 7 Introduction The BBL Crystal MIND software can not be directly installed on the Microsoft Windows 7 platform, however it can be installed and run via XP
Picture Manager Picture Manager allows you to easily edit and organize the pictures on your computer. Picture Manager is an application that was included with Microsoft Office suite for Windows 2003 and
LOREX CLIENT Remote Software 4.0 Instruction Manual English Version 2.0 MODEL: L20WD800 Series www.lorexcctv.com Copyright 2008 LOREX Technology Inc. Table of Contents Table of Contents Software Installation...
SMS Database System Quick Start [Version 1.0.3] Warning ICP DAS Inc., LTD. assumes no liability for damages consequent to the use of this product. ICP DAS Inc., LTD. reserves the right to change this manual
Procedure to Install Printer to the LifeWindow 6000 Rev 3. The LifeWindow OS is Windows XP embedded. Printers with driver to XPe are supported. Some printers with driver to XP can also be installed. Please
Having the power to block or allow popup and popunder advertising www.contentwatch.com 2004 All rights reserved. 2369 West Orton Circle, Salt Lake City, UT 84119 Table of Contents Welcome and Installation
030.0051.01.0 Guide for Remote Control PDA For Use with Bluetooth and a PC Running Windows 7 Table of Contents A. Required Parts... 3 B. PC Software Installation... 3 C. Configure PC Software... 4 D. Testing