1 page 1 of 68 Technology for Vacuum Systems Instructions for use BVC control BVC control G BVC professional BVC professional G BioChem-VacuuCenter
2 Dear customer, page 2 of 68 Your VACUUBRAND diaphragm pumps are designed to provide you with many years of trouble-free service with optimal performance. Our many years of practical experience allow us to provide a wealth of application and safety information. Please read these instructions for use before the initial operation of your pump. VACUUBRAND diaphragm pumps combine our many years of experience in design, construction and practical operation, with the latest developments in material and manufacturing technology. Our quality maxim is the zero defect principle: Every diaphragm pump, before leaving our factory, is tested intensively, including an endurance run of 14 hours. Any faults, even those which occur rarely, are identified and can be eliminated immediately. After completion of the endurance run, every pump is tested, and must achieve specifications before shipment. We are committed to providing our customers only pumps that meet this high quality standard. While our pumps cannot eliminate all of your work, we design, manufacture and test them to ensure that they will be an effective and trouble-free tool to assist you in that work. Yours, VACUUBRAND GMBH + CO KG After sales service: Contact your local dealer or call Trademark index: VACUU LAN (US-Reg.No 3,704,401), VACUU BUS, VACUU CONTROL TM, chemistry-hybrid TM, Peltronic, TURBO MODE TM, VARIO (US-Reg.No 3,833,788), VARIO-SP TM, VACUUBRAND (US-Reg.No 3,733,388) and also the shown company logos are trademarks of VACUUBRAND GMBH + CO KG in Germany and/or other countries.
3 page 3 of 68 Contents Safety information!...4 Important information!... 4 General information... 6 Intended use... 6 Setting up and installing the system... 7 Ambient conditions... 8 Operating conditions... 9 Safety during operation Maintenance and repair Technical data...15 Wetted parts System parts Use and operation...21 First steps: Installation Operation BVC control / BVC control G Operation BVC professional / BVC professional G Level sensor of BVC professional / BVC professional G During operation Filtration Filter and collection bottle Disinfection routine Cleaning and decontaminating...35 Adjustment of the BVC level sensor to work properly with a new bottle, or to correct false alarms...38 Assembling of components...42 Replacing the filter Assembling a second VHC pro connection set Assembling quick coupling bottle - pump unit Assembling a hose nozzle at the bottle head of the BVC Quick coupling sets Accessories - spare parts...48 Troubleshooting...50 Replacing diaphragms and valves...52 Cleaning and inspecting the pump heads Replacing the diaphragm Assembling the pump head Replacing the fuse Notes on return to the factory...62 Health and safety clearance form...65 EC Declaration of Conformity of the Machinery...67
4 page 4 of 68 Safety information! Important information! + Keep this manual complete and accessible to personnel at all times! + Read this manual carefully before installing or operating the equipment. Observe the instructions contained in this manual. + Do not modify the equipment without authorization. NOTICE This manual is an integral part of the equipment described therein. It describes the safe and proper use of the vacuum pump. Make operating personnel aware of dangers arising from the pump and the pumped substances. VACUUBRAND disclaims any liability for inappropriate use of these pumps and for damage from failure to follow instructions contained in this manual. This manual is only to be used and distributed in its complete and original form. It is strictly the users responsibility to check carefully the applicability of the guidance in this manual with respect to his product. Manual-no.: / 09/04/2014 The following signal word panels and safety symbols are used throughout this manual: This is the safety alert symbol. It is used to alert you to potential personal injury hazards. Obey all safety messages that follow this symbol to avoid possible injury and death.
5 page 5 of 68 DANGER indicates a hazardous situation which, if not avoided, will result in death or serious injury. + WARNING indicates a hazardous situation which, if not avoided, could result in death or serious injury. CAUTION indicates a hazardous situation which, if not avoided, could result in minor or moderate injury. NOTICE NOTICE is used to address practices not related to personal injury. Caution! Hot surface! Disconnect equipment from AC power. Formatting used in this manual: Note: The signal word panels in all sections of this manual always refer to all paragraphs of the same format ( / + / / plain text) following each signal word panel. The document Safety information for vacuum equipment is part of this manual! Read the Safety information for vacuum equipment and observe the instructions contained therein!
6 page 6 of 68 General information Read and comply with this manual before installing or operating the equipment. Before operating the equipment read and comply with the section Cleaning and decontamination! NOTICE Remove all packing material, remove the product from its packing-box, remove the protective covers from the inlet and outlet ports and keep, inspect the equipment. If the equipment is damaged, notify the supplier and the carrier in writing within three days; state the item number of the product together with the order number and the supplier s invoice number. Retain all packing material for inspection. Do not use the equipment if it is damaged. Lift and transport the equipment at the provided handle. Intended use + Prevent any part of the human body from coming in contact with the vacuum. Make sure that the individual components are only connected, combined and operated according to their design and as indicated in the instructions for use. Use only original VACUUBRAND accessories. Comply with notes on correct vacuum and electrical connections, see section Use and operation. The systems are designed for ambient temperatures during operation between +10 C and +40 C. Check the maximum temperatures if installing the system in a cabinet or a housing and make sure ventilation is adequate. Install an external automatic ventilation system if necessary.
7 NOTICE page 7 of 68 Ensure that the equipment and all components are suitable for the intended application. Use the system only for aspiration of liquids and filtration. Setting up and installing the system Equipment must be connected only to a suitable electrical supply and a suitable ground point. Failure to connect the motor to ground may result in deadly electrical shock. The supply cable may be fitted with a molded European IEC plug or a plug suitable for your local electrical supply. If the plug has been removed or has to be removed, the cable will contain wires color coded as follows: green or green and yellow: earth; blue or white: neutral; brown or black: live. The device has an internal fuse. Do not permit any uncontrolled pressurizing (e.g., make sure that an exhaust tubing cannot become blocked). Risk of bursting! Due to the high compression ratio of the pumps, pressure at the outlet port may be generated that is higher than the maximum permissible pressure compatible with the mechanical stability of the system. Always provide a free and pressureless exhaust line. Comply with maximum permissible pressures and pressure differences; see section Technical data. Do not operate the pumping unit with overpressure at the inlet or outlet. Check that line voltage and current conform with the equipment. (See rating plate.) Avoid overpressure of more than 0.2 bar in instances in which inert gas is connected.
8 NOTICE page 8 of 68 Provide a firm level platform for the equipment. Ensure a stable position of the pump without any mechanical contact except of the pump feet. Comply with all applicable safety regulations. Check fan regularly for dust/dirt, clean if necessary to avoid reduced ventilation. Avoid high heat supply. If the equipment is brought from cold environment into a room for operation, allow the equipment to warm up. (Watch for water condensation on cold surfaces.) The diameter of the an outlet tubing should be at the least as large as the diameter of the pump connections. Comply with all applicable and relevant safety requirements (regulations and guidelines), implement the required actions and adopt suitable safety measures. Ambient conditions NOTICE To the best of our knowledge the equipment is in compliance with the requirements of the applicable EC-directives and harmonized standards (see Declaration of conformity ) with regard to design, type and model, especially directive DIN EN This directive gives, in detail, conditions, under which the equipment can be operated safely (see also IP degree of protection). Adopt suitable measures in case of differences, e. g. using the equipment outdoors, installation in higher altitudes, conductive pollution or bedewing. Pay attention to the permissible maximum ambient temperatures (see Technical data ).
9 page 9 of 68 Operating conditions The devices have no approval for operation in or for pumping of potentially explosive atmospheres. The devices are not suitable to pump: - unstable substances and substances which react explosively under impact (mechanical stress) and/ or when being exposed to elevated temperatures without air, - self inflammable substances, - substances which are inflammable without air and - explosive substance The pumps have no approval for operation below ground. The pumps are not suitable for pumping dust. The devices are not suitable for pumping substances which may form deposits inside the pump. Deposits and condensate in the pump may lead to increased temperatures even to the point of exceeding the maximum permitted temperatures! Increased temperatures may cause ignition of inflammable substances that may have been deposited inside the pump during use. If there is a danger of the formation of deposits in the pump chamber (check inlet and outlet of the pump), inspect the pump chamber regularly and clean if necessary. Take into consideration interactions and chemical reactions of the pumped media. Ensure that the materials of the wetted parts are compatible with the pumped substances, see section Technical data. NOTICE If pumping different substances, it is recommended that the pump be purged with air or inert gas prior to changing the pumped media in order to pump out residues and to avoid reactions of the pumped substances with each other and with the pump materials.
10 page 10 of 68 Safety during operation Avoid interactions of the media in the collection bottle absolutely. Comply with material safety data sheets and notes on safe use of the manufacturer. Do not mix incompatible disinfectants and/or incompatible reagents / solvents as well as unknown substances. For example sodium hypochlorite (chlorine bleach) Incompatible chemicals and agents Acids or acidic compounds (e. g. hydrochloric acid, aluminium chloride) Ammonia containing compounds (e. g. ammonium hydroxide, quarternary ammonium salts) Organic chemicals (e. g. solvents, polymers, amines, oils) Metals (e. g. copper, iron) Hydrogen peroxide Reducing agents (e. g. sodium thiosulfate) Guanidine salts (e. g. guanidine hydrochloride, guanidine thiocyanate) Possible results of mixing with chlorine bleach Release of chlorine gas Formation of explosive compounds, release of chlorine gas and other hazardous gases Formation of chlorinated organics, release of chlorine gas and other hazardous gases Release of oxygen, overpressure, rupture of a closed system Production of heat, boiling Release of toxic gases, e. g. chlorine, chloramine, hydrogen cyanide Avoid the formation of dangerous reactions / gases in the BVC, especially in the collection bottle. If this is impossible dispose of dangerous gases at the outlet of the pump appropriately.
11 page 11 of 68 Adopt suitable measures to prevent the release of dangerous, toxic, explosive, corrosive, noxious or polluting fluids, vapors and gases. In such cases, install an appropriate collecting and disposal system and take protective action for pump and environment. If the equipment is used in combination with hazardous materials (e. g., in medical-microbiological laboratories) check all relevant safety and health requirements and determine the applicability of regulatory limitations prior to use if necessary. Use appropriate safety work materials and methods if necessary, e. g., steam sterilization, sterilization indicators or germicides. For notes concerning the sterilization of wetted parts, see the Technical Data section and Cleaning and decontaminating. It is the user s responsibility to ensure effective sterilization. Never operate a defective or damaged device. The user must take suitable precautions to prevent any formation of explosive mixtures in the expansion chamber or in the housing. In case of a diaphragm crack, for example, mechanically generated sparks, hot surfaces or static electricity may ignite these mixtures. Use inert gas for venting if necessary. Potentially explosive mixtures at the outlet of the pump have to be drained appropriately evacuated, or diluted with inert gas to non-explosive mixtures. Check the collection bottle regularly for cracks. Do never use a collection bottle with cracks nor expose it to vacuum. Use the system only in combination with the integrated hydrophobic filter (sterilizable) to protect pump and environment (people).
12 page 12 of 68 Comply with applicable regulations when disposing of chemicals. Take into consideration that chemicals may be polluted. Take adequate precautions to protect people from the effects of dangerous substances (chemicals, thermal decomposition products of fluoroelastomers), wear appropriate safety-clothing and safety glasses. Use only OEM spare parts and accessories; otherwise safety and performance of the equipment as well as the electromagnetic compatibility of the equipment might be reduced. The CE mark or the ctuvus mark (see rating plate) may become void if original equipment manufacturer parts are not used. In case of leaks in the manifold, aspirated substances might be released into the environment. Comply especially with notes on operation and use and maintenance. Due to the residual leak rate of the equipment, there may be an exchange of gas, albeit extremely slight, between the environment and the vacuum system.adopt suitable measures to prevent contamination of the pumped substances or the environment. NOTICE Maximum liquid level in collection bottle: approx. 80 %, depending on the application. (When working with lowboiling liquids, or liquids that tend to foam, the maximum liquid level might be reduced). Ensure that, in case of failure, the system will always return to a safe status. Provide appropriate protective measures (i.e., precautions which allow for the requirements of the respective application) even for the case of failure and malfunction. When operating the pump, the loss of pumping or venting must not lead to a critically dangerous situation under any circumstances.
13 NOTICE page 13 of 68 Electronic equipment is never 100% fail-safe. This may lead to an indefinite status of the equipment. Failure of the pumping unit (e. g., by power failure) or connected components, or change of parameters must not lead to a critically dangerous situation under any circumstances. In case of diaphragm cracks or leaks in the manifold, pumped substances might be released into the environment or into the pump housing. Comply especially with notes on operation and use and maintenance. Pay attention to the safety symbol hot surfaces on the equipment. Hot parts may cause burns if touched. Adopt suitable measures to prevent any danger arising from hot surfaces or electric sparks. Ensure that hot surfaces of the pump do not cause burns. Provide a suitable contact guard if necessary. In case of overload, the motor is shut down by a self-hold thermal cutout in the winding. Attention: Reset possible only manually. Switch off the pump or isolate the equipment from power supply. Identify and eliminate the cause of failure. Wait approx. five minutes before restarting the pump. Attention: In case of supply voltage below 100V ( V version), the lock of the cutout might be restricted and the pump might restart on its own after sufficient cooling down. Take suitable precautions if an automatic restart of the pump may lead to a critical dangerous situation. Maintenance and repair NOTICE Wear parts have to be replaced regularly. Under typical operating conditions, the projected lifetime of the diaphragms and valves is > operating hours. Bearings have a typical durability of h. Motor capacitors have a typical durability in the range of to h depending strongly on the operating conditions like ambient temperature, humidity or load.
14 page 14 of 68 Never operate the pump if covers or other parts of the pump are disassembled. Ensure that the pump cannot be operated accidentally. Isolate equipment from power supply before removing the cover! Before starting maintenance, unplug the equipment and wait 5 seconds to allow the capacitors to discharge. Attention: The pump might be contaminated with noxious or otherwise dangerous process chemicals that have been pumped during operation. Ensure that the pump is decontaminated before maintenance. Take adequate precautions to protect people from the effects of dangerous substances if contamination has occurred. Wear appropriate safety-clothing when you come in contact with contaminated components. Wear parts have to be replaced regularly. Never operate a defective or damaged pump. Check every motor capacitor regularly by measuring its capacity and estimating its operation time. Exchange old capacitors early enough to prevent a failure. When motor capacitors fail, they can get hot enough to melt or cause a flame which could be dangerous for persons and equipment in the vicinity. Motor capacitors have to be replaced by an electrician. Before starting maintenance, vent the system and allow sufficient cooling of the pump. Drain condensate, if applicable. Avoid the release of pollutants. NOTICE Ensure that maintenance is done only by suitably trained and supervised technicians. Ensure that the maintenance technician is familiar with the safety procedures which relate to the products processed by the pumping system. In order to comply with law (occupational, health and safety regulations, safety at work law and regulations for environmental protection) vacuum pumps, components and measuring instruments returned to the manufacturer can be repaired only when certain procedures (see section Notes on return to the factory ) are followed.
15 Technical data page 15 of 68 Type Maximum pumping speed (ISO 21360) at 50/60 Hz Ultimate vacuum (absolute) Maximum permissible inlet pressure (absolute) Maximum permissible outlet pressure (absolute) Rated power cfm (m 3 /h) Torr (mbar) psi (bar) psi (bar) hp (kw) BVC control / G BVC professional / G Pump 0.4/ 0.5 (0.7 / 0.85) 112 (150) 16 (1.1) 16 (1.1) (0.04) No-load speed 50/60 Hz rpm 1500 / 1800 Control range Permissible ambient temperature storage / operation Torr (mbar) F ( C) ( ) 14 to 140 / 50 to 104 (-10 to +60 / +10 to +40) Maximum installation altitude m 2000 above mean sea level Permissible relative atmospheric moisture during operation (no condensation) Maximum permissible range of supply voltage: Attention: Observe specifications of rating plate! Rated current at: V~ 50/60 Hz 230 V~ 50/60 Hz Fuse Motor protection % 30 to 85 A A V~ ±10% 50/60 Hz 230 V~ ±10% 50/60 Hz 1.9 / / x 2.5 AT, 0.19 x 0.78 in (5 x 20 mm) thermal cutout Degree of protection IEC 529 IP 40 A-weighted emission sound pressure level* (uncertainty K pa : 3 db(a)) db(a) 45 * Measurement according to EN ISO 2151:2004 and EN ISO 3744:1995 at 230V/50Hz and ultimate vacuum with exhaust tube at outlet.
16 page 16 of 68 Type Volume of collection bottle Connection at the inlet Connection at the outlet Hydrophobic filter Type Manufacturer Nominal pore diameter Dimensions L x W x H approx. BVC control BVC control G BVC professional BVC professional G BVC control BVC control G BVC professional BVC professional G Weight approx. BVC control / professional BVC control G / professional G BVC control / professional BVC control G / professional G gal (l) in in in in (mm) (mm) (mm) (mm) lbs. lbs. (kg) (kg) BVC control / G BVC professional / G System 1 (polypropylene) / 0.5 (borosilicate glass) (4 (polypropylene) / 2 (borosilicate glass)) hose nozzle DN 3/5mm hose nozzle DN 8 mm / silencer Midisart 2000 Sartorius stedim 0.2 µm 16.1 x 7.6 x x 7.6 x x 7.6 x x 7.6 x 16.9 (408 x 194 x 500) (408 x 194 x 430) (408 x 194 x 500) (408 x 194 x 430) 16.1 (230V) / 17.0 (120V) 17.0 (230V) / 17.9 (120V) (7.3 (230V) / 7.7 (120V)) (7.7 (230V) / 8.1 (120V)) We reserve the right for technical modifications without prior notice!
17 page 17 of 68 Wetted parts Components Pump Housing cover insert Head cover Diaphragm clamping disc Diaphragm Valve Inlet Outlet Silencer Filter Diaphragm Housing Tubing Collection bottle 4l Bottle / screw cap Hose nozzle at bottle head (connection VHC pro ) Hose in bottle Hose nozzle (connection filter) Counter nut Closing screw Collection bottle 2l Bottle Cap insert Flat seal Hose nozzle at bottle head (connection VHC pro ) Hose in bottle Hose nozzle (connection filter) Closing screw Wetted materials PTFE, carbon reinforced ETFE, carbon fibre reinforced ETFE, carbon fibre reinforced PTFE PTFE / FFKM ETFE ETFE silicone rubber PTFE PP silicone rubber PP PP PTFE PPS, glass fibre reinforced PP PPS, glass fibre reinforced Borosilicate glass PP EPDM PP PTFE PPS, glass fibre reinforced PPS, glass fibre reinforced We reserve the right for technical modifications without prior notice!
18 System parts page 18 of 68 Position Designation 1 Pump ME 1C 2 Mains connection 3 Fuse holder 4 On / Off switch 5 Touch panel 6 Rating plate 7 Outlet 8 Connection tubing 9 Hydrophobic protection filter 10 Connection filter 11 Connection VacuuHandControl VHC pro 12 Closing screw (optional: connection second VHC pro ) 13 Screw cap / bottle cap with insert 14 Collection bottle 15 Handle 16 Level sensor BVC control
19 page 19 of 68 BVC control G BVC professional
20 page 20 of 68 BVC professional G On / Off switch System switched off System switched on Attention: Operate the on/off switch only by hand.
21 Sicherheitshinweise für Vakuumgeräte Safety information for vacuum equipment Avis de sécurité pour des dispositifs à vide Указания за безопасност за вакуумни уреди 真 空 设 备 的 安 全 信 息 Bezpečnostní upozornění pro vakuové přístroje Sikkerhedsregler for vakuumudstyr Ohutusnõuded vaakumseadmetele Notas sobre la seguridad para equipos de vacío Vakuumilaitteen turvallisuustiedot Υποδείξεις ασφάλειας για αντλίες κενού Sigurnosne napomene za vakuumske uređaje A vákuum-készülékekkel kapcsolatos biztonsági tudnivalók Istruzioni di sicurezza per apparecchi a vuoto 真 空 装 置 を 安 全 に 取 り 扱 うために 진공 장비에 대한 안전 정보 Vakuuminės įrangos saugos informacija Vakuuma iekārtu drošības noteikumi Veiligheidsvoorschriften voor vacuümapparaten Wskazówki bezpieczeństwa do urządzeń próżniowych Informação de Segurança para Equipamento que funciona a Vácuo Instrucţiuni de siguranţă pentru aparatele de vidare Указания по технике безопасности при работе с вакуумными устройствами Säkerhetsinformation för vakuumutrustning Varnostni nasveti za vakuumske naprave Bezpečnostné pokyny pre vákuové zariadenia Vakumlu cihazlar için güvenlik uyarıları DE EN FR BG CN CZ DA EE ES FI GR HR HU IT JP KR LT LV NL PL PT RO RU SE SI SK TR 1 Use and operation First steps: Installation page 21 of 68 Unpack the equipment. Safety information for vacuum equipment Read and comply with the document Safety information for vacuum equipment - Sicherheitshinweise für Vakuumgeräte! Assemble tubing with filter. Setting up the BVC. Connect to power supply. Check line voltage and current prior to switching on! Attach tubing of the VacuuHandControl VHC pro to the hose nozzle of the bottle head.
22 page 22 of 68 Operation BVC control / BVC control G Do not press the keys. Only touch the keys of the touch panel. The keys + and - have to be touched > 0.25 seconds to be actuated. A successful action is confirmed by a blip and the flashing of LEDs Position Designation 1 Key to reduce suction power 2 Display suction power 3 Key to increase suction power - + Use the keys to set suction power of the system. The suction power can be set linearly in a range from 150 mbar (1 LED flashes) to 850 mbar (8 LEDs flash) underpressure (relative to atmospheric pressure). From higher software versions A flashing LED indicates that the actual available suction power differs from the preset suction power. Attention: If the collection bottle is under vacuum and the vacuum demand is reduced, the existing underpressure inside the bottle remains until the vacuum is reduced (pressure rises) by further aspiration with the VHC pro.
23 NOTICE page 23 of 68 Depending on ambient atmospheric pressure (depending on altitude or weather conditions) and on the vapor pressure of the media inside the collection bottle, the setting of a high underpressure (e. g., 850 mbar, 8 LEDs flash) may lead to continuous operation of the pump. In this case a reduction of the suction power is recommended.
24 page 24 of 68 Operation BVC professional / BVC professional G Do not press the keys. Only touch the keys of the touch panel. The keys + and - have to be touched > 0.25 seconds to be actuated. The other keys have to be touched > 0.5 seconds. Touch the keys with LED below the LED. A successful action is confirmed by a blip and the flashing of LEDs Position Designation 1 Key to select bottle size and level sensor 2 Key to reduce suction power 3 Key bottle change 4 Display suction power 5 Key to increase suction power - + Use the keys to set suction power of the system. The suction power can be set as underpressure (relative to atmospheric pressure) linearly in a range of 150 mbar (1 LED flashes) to 850 mbar (8 LEDs flash). From higher software versions A flashing LED indicates that the actual available suction power differs from the preset suction power. Attention: If the collection bottle is under vacuum and the vacuum demand is reduced, the existing underpressure inside the bottle remains until the vacuum is reduced (pressure rises) by further aspiration with the VHC pro.
25 NOTICE page 25 of 68 Depending on ambient atmospheric pressure (depending on altitude or weather conditions) and on the vapor pressure of the media inside the collection bottle, the setting of a high underpressure (e. g., 850 mbar, 8 LEDs flash) may lead to continuous operation of the pump. In this case, a reduction of the suction power is recommended. Use the keys to select the bottle size (2 liter glass bottle or 4 liter polypropylene bottle), and with it the activation of the corresponding level sensor. To operate the key 2l Glass or 4l PP touch the key > 1 second. The LED of the selected bottle flashes blue. Use the key to turn off the level sensor alarm and to start or stop the pump during bottle change. To operate the key bottle change, touch the key below the LED > 0.5 seconds.
26 page 26 of 68 Level sensor of BVC professional / BVC professional G The sensor foil is located on the bottle support. The level sensor gives an alarm and switches off the pump to avoid an overfilling of the collection bottle, if the liquid level in the collection bottle reaches the height of the level sensor, approximately 80% of the maximum bottle capacity (grey marked range with bottle symbols on the sensor foil, both for the 2l or 4l bottle). We recommend checking the function of the level sensor regularly (depending on operating conditions): Fill bottle with liquid, and confirm that the level sensor alarms. The initiation time of the level sensor is up to 10 seconds. NOTICE In case of false alarm (empty bottle or bottle less than 80% filled) it might be necessary to adjust the sensor using the individual bottle. (See section Adjustment of the level sensor.) Strong electromagnetic fields may influence the alarm level of the capacitive measurement element. Level sensor range when selecting the 4l bottle Level sensor range when selecting the 2l bottle
27 NOTICE page 27 of 68 Use only the original VACUUBRAND BVC bottle or the original spare bottle (see Accessories - Spare parts ). Do not fix adhesive foil or something near it at the bottle side next to the sensor foil or directly at the sensor foil.
28 page 28 of 68 During operation Attention: When removing the aspiration controller VHC pro liquid in the tube may leak! Adopt suitable measures to prevent the release of dangerous, toxic, explosive, corrosive, noxious or polluting fluids, vapors and gases. In such cases, use an appropriate collecting and disposal system and take protective action for personnel, equipment and environment. max. 40 C Max. ambient temperature: 40 C Use the system only with the integrated hydrophobic filter to protect the vacuum supply from aspirated liquids and aerosols, and to protect the environment/user from contamination risk. 1 2 NOTICE The BVC professional (with 4l polypropylene bottle) is equipped with quick couplings at the bottle head (1) and at the inlet of the pump (2). Quick couplings for other versions are available as accessories. (See section Accessories - Spare parts.) Quick coupling set (1): VHC pro to bottle: When disconnected, the collection bottle closes vacuum tight. Quick coupling set (2): Bottle to pump unit: When disconnected, the collection bottle closes vacuum tight.
page 1 of 75 Technology for Vacuum Systems Instructions for use MZ 2C EX MD 4C EX MV 10C EX MZ 2C EX + AK + EK MZ 2C EX + IK + EK MD 4C EX + AK + EK MV 10C EX + AK + EK Chemistry diaphragm pumps with ATEX
Register your instrument! www.eppendorf.com/myeppendorf Operating manual Copyright 2014 Eppendorf AG, Germany. All rights reserved, including graphics and images. No part of this publication may be reproduced
page 1 of 63 Technology for Vacuum Systems Instructions for use PC 510 NT / PC 511 NT / PC 520 NT PC 610 NT / PC 611 NT / PC 620 NT Chemistry pumping unit page 2 of 63 Dear customer, Your VACUUBRAND diaphragm
IMPORTANT SAFETY NOTICE To: Our Valued Customers User safety is a major focus in the design of our products. Following the precautions outlined in this manual will minimize your risk of injury. ITT Goulds
Industrial Process Pump Safety Manual IMPORTANT SAFETY NOTICE To: Our Valued Customers User safety is a major focus in the design of our products. Following the precautions outlined in this manual will
SunMaxx Solar Filling Operating Instructions Content 1. Declaration of conformity... 2 2. Introduction... 2 3. Transportation and unpacking... 4 4. Mounting and commissioning... 5 5. End of operation...
1in. Clear Water Pump OWNER'S MANUAL WARNING: Read carefully and understand all INSTRUCTIONS before operating. Failure to follow the safety rules and other basic safety precautions may result in serious
Infrarot-Bewegungsmelder IP44 infrared motion sensors IP44 ODA (weiß) slim ODA (schwarz) slim 95174 96000 ODA (weiß) ODA (schwarz) 95175 96001 Betriebsanleitung User s Manual User s Manual infrared motion
The device type and date of manufacture (week/year) can be found on the device rating plate. In the event of any queries about the device, please quote all the details given on the rating plate. For further
Oil and Coolant Circulating Heating System Model - OCSM Installation & Operation Manual 216280-000 REV 2 Identifying Your System The HOTSTART heating system is designed to heat fluids for use in marine
Meaco 30L and Meaco 40L dehumidifier instruction manual Please read this instruction manual before using the dehumidifier and keep safe for future reference SAFETY INSTRUCTIONS PLEASE READ ALL INSTRUCTIONS
1. Description This unit is a pump that will clear the waste from a domestic shower and washbasin, bath, washing machine, glass washer and a sink. The unit will start automatically. Please pay particular
www.burcam.com 2190 Dagenais Blvd. West TEL: 514.337.4415 LAVAL (Quebec) FAX: 514.337.4029 CANADA H7L 5X9 infoburcam.com This product has been carefully packaged at the factory to prevent damage during
OASIS-PLUS 120V INFORMATION & OPERATING INSTRUCTIONS READ ALL INSTRUCTIONS BEFORE OPERATING READ ALL INSTRUCTIONS BEFORE OPERATING OZONE IS A POWERFUL OXIDIZER AND MUST BE USED WITH CARE 56041852 WARNING:
Portable Air Conditioner Owner's Manual Model:3 in 1 12,000 Btu/h Series 3 Please read this owner s manual carefully before operation and retain it for future reference. CONTENTS 1. SUMMARY...1 2. PORTABLE
6262100/6 IM-P626-01 ST Issue 6 PBX Series Quick Fit Connector Bimetallic Steam Traps Installation and Maintenance Instructions 1. Safety information 2. General product information 3. Installation 4. Commissioning
Rp 3/4 Ergo-Pro Single Line Solar Station Installation and Operating Instructions Item No Pump type 677.21.70 WILO ST 15/6 13 2 4 11 Technical Specifications Max. operating pressure: 6 bar Max. operating
24L OIL FREE AIR COMPRESSOR MODEL NO: RANGER 6/240 PART NO: 2242010 OPERATION & MAINTENANCE INSTRUCTIONS LS1013 INTRODUCTION Thank you for purchasing this CLARKE 24L Oil Free Air Compressor. Please read
280C AIR COMPRESSOR KIT PART NO. 28021 IMPORTANT: It is essential that you and any other operator of this product read and understand the contents of this manual before installing and using this product.
Art.S001-S002 SOLAR MODULE WITH DELIVERY AND RETURN CONNECTIONS Technical Information sheet 0001/07/Rev01 ENG FUNCTION Series S001 and S002 circulation units are applied to the primary circuit of solar
Air conditioning service unit CAAS500 OPERATING INSTRUCTIONS TES1453 09/2004 C A A S 5 0 0 O P E R A T I N G I N S T R U C T I O N S P a g e 2 IMPORTANT THIS MANUAL CONTAINS INFORMATION PERTINENT TO OPERATOR
Type T-CUT Tubular/pipe modules for micro and ultrafiltration Tubular-/Rohrmodule für die Mikro- und Ultrafiltration Modules tubulaires pour micro- et ultrafiltration Quickstart English Deutsch Français
XSENTIAL X-301 INSTANT HOT WATER DISPENSER INSTALLATION & OPERATING GUIDE 1.0 Components: When you purchase the Diana Instant Hot Water tank, contents include: 1. X-301 boiler 2. DLX-580 faucet 3. Feed
Freedom8 ShoeBox Compressor Manual Warning!! This product is not a toy! Use or misuse can cause severe injury or death! Use only with adult supervision. This unit is only to be used with tanks, hoses and
Operation Manual Diaphragm pumps 1-headed Types MP 301 E MPC 301 E 4000732 2008-08-19 We are constantly working on the further development of all our product types. Reprinting or reproduction of this manual,
Tankless Water Heater USER S INFORMATION MANUAL Models WGRTNG199 / WGRTLP199 WGRTCNG199 / WGRTCLP199 NOTICE: Westinghouse reserves the right to make product changes or updates without notice and will not
Service Guide Evaporator Coil Cleaning and Condensate Drain Maintenance Guidelines ORDER No: UN-SVG001A-EN DATE: October 9, 2007 Introduction This document defines American Standard and Trane s recommendations
325C AIR COMPRESSOR KIT PART NO. 32530-32533 - 32534-32538 350C AIR COMPRESSOR KIT PART NO. 35030-35033 400C AIR COMPRESSOR KIT PART NO. 40040 450C AIR COMPRESSOR KIT PART NO. 45040 325C 350C 400C 450C
Autoclave Safety Purpose: Sterilization refers to the complete killing of all living organisms, including spores. Common sterilization techniques include the application of wet heat, dry heat, chemicals,
NATURAL GAS TO PROPANE CONVERSION KIT 090 INSTALLATION INSTRUCTIONS FOR ALTITUDES 0 -,00 FT. ONLY PROPANE CONVERSION KIT SAFETY... INSTALLATION REQUIREMENTS... Tools and Parts... LP Gas Requirements...
TM Water Dispenser Owner s Manual 500 Series IMPORTANT: This manual includes important safety material; please read all instructions before plugging power supply cord into receptacle. STOP & READ DIRECTIONS
MODE ALARM Portable Air Conditioner OWNER S MANUAL Read these instructions before use 8 Model: MF08CESWW Voltage rating: 115V~60Hz Power rating : 800W Customer Support : 1-800-474-2147 For product inquiries
WWW.BURCAM.COM 2190 Boul. Dagenais West TEL: 514.337.4415 LAVAL (QUEBEC) FAX: 514.337.4029 CANADA H7L 5X9 firstname.lastname@example.org Your pump has been carefully packaged at the factory to prevent damage during shipping.
LUFTIG BF570 IMPORTANT SAFETY INSTRUCTIONS YOUR SAFETY AND THAT OF OTHERS IS PARAMOUNT This manual and the appliance itself provide important safety warnings, to be read and observed at all times. This
Instructions - Parts List Leak Detection System 311200A For detecting leaks in your diaphragm pump system. Not for use in explosive atmospheres. Part No. 234576 120 psi (0.8 MPa, 8 bar) Maximum Air Input
Trouble Shooting Pump Trouble Possible Cause Remedy Oil leaking in the area of water pump crankshaft Worn crankshaft seal, bad bearing, grooved shaft, or failure of retainer o-ring. Excessive play on crankshaft
Table of Contents Important Safeguards...2 Product Layout...3 Preparing for Use...4 Air-conditioning without installation...4 Air-conditioning with installation...5 Control Panel...6 Operating from the
Fan Coil Replacement Coil Kit EBX & EBXX These instructions must be read and understood completely before attempting installation. These instructions covers the installation of replacement coil kit into
SALTER AIRE Plus COMPRESSOR Model 8350 / 8352 / 8353 Instruction Manual Page 1 of 9 TABLE OF CONTENTS Introduction...3 Important Safeguards..3 System Components...5 Setup and Operation...5 Setup...5 Operation....6
Series 150S and 157S (Snap Switch, All Models except 157S-RB-P) Low Water Cut-Off/Pump Controllers For Steam Boilers and Other Level Control Applications McDonnell & Miller Installation & Maintenance Instructions
USER INSTRUCTIONS FOR GET PORTABLE 12k BTU AIR CONDITIONER MODEL No. GPACU12HR CONTENTS Introduction Safety Notes Identification of parts Installation instructions Operation instructions Maintenance Troubleshooting
Ultrasonic support removal tank User guide and safety manual See inside for use and safety information. The Ultrasonic support removal tank automatically removes PLA support material from a printed ABS
INSTALLATION MANUAL AIR CONDITIONER Please read this installation manual completely before installing the product. Installation work must be performed in accordance with the national wiring standards by
1263050/6 IM-P126-07 ST Issue 6 BPW32 DN15, DN20 and DN25 Balanced Pressure Wafer Steam Trap Installation and Maintenance Instructions 1. Safety information 2. General product information 3. Installation
SMD Rework Station Thank you for purchasing the Hakko 50B SMD Rework Station. The Hakko 50B is designed to solder and desolder surface mounted devices with hot air. Please read this manual before operating
876124 Edition B April 2013 Pneumatic Filters, Regulators, Lubricators, Filter/Regulators and Combination Units 849XXX Maintenance Information Save These Instructions Product Description The air line Filters,
NewAir AC-10100E / AC-10100H Portable Air Conditioner Owner s Manual PLEASE READ AND SAVE THESE INSTRUCTIONS ELECTRICAL SAFETY This appliance is for indoor use only. Always turn off the unit and unplug
Website: www.andico.com.au Service manual CAUTION - BEFORE SERVICING THE UNIT, READ THE SAFETY - PRECAUTIONS IN THIS MANUAL. - ONLY FOR AUTHORISED SERVICE PERSONNEL. MODELS: MPK1-09CR-QB8 MPK1-12ER-QB6
Instruction Sheet 07-2013 Installation Instructions for Alarm Module Kit A043F059 1 Introduction The information contained within is based on information available at the time of going to print. In line
2 Contents Components Attachments Guidance Installation Operation Maintenance Service Technical Troubleshooting Symbols used to mark instructions...3 Included Attachments...4 Congratulations...5 Getting
Int968 DE LUXE REPAIRING SYSTEM INSTRUCTION MANUAL Thank you for purchasing Aoyue Int968 Repairing System. It is important to read the manual before using the unit. Please keep manual in accessible place
REITEL Feinwerktechnik GmbH Senfdamm 20 49152 Bad Essen Tel. (++49) 54 72/94 32-0 Fax: (++49) 5472/94 32-40 Operating Instructions Supersteam Dear Customer! With the purchase of this SUPERSTEAM you have
OPERATOR S MANUAL 18 VOLT, 1 HOUR CHARGER P110 Your battery charger has been engineered and manufactured to Ryobi s high standard for dependability, ease of operation, and operator safety. When properly
Page 1 of 5 COMPRESSED GASES A compressed gas is defined as any mixture of gases in a container with a pressure exceeding 40 psi. at 70 o F, or 104 psi. at 130 o F; or any flammable liquid with an absolute
3HP DIRECT DRIVE COMPRESSOR MODEL NO: PED14A50,PED14A100, SED14A100 PART NO: 1499200, 1499202, 1499230 OPERATION & MAINTENANCE INSTRUCTIONS 0210 INTRODUCTION Thank you for purchasing this CLARKE 3HP Direct
January 6, 2012 Preaction 332a 1. Description Viking Supervised Surefire Preaction Systems Utilize the Viking G-2000P Valve. The small profile, lightweight, pilot operated Viking G-2000P Valve comes complete
Installation and Service Instructions for use by heating contractors Neutralization System Part no. 7441823 Grünbeck part no. 410480 Safety and Installation Requirements Please ensure that these instructions
6131 Series User Guide v1.1-06.09.10:user guide 09/09/2010 11:42 Page 1 User Guide Vax Careline: (UK) 0844 412 8455 Multifunction Get Cleaning... What s your Vax model number? 6 1 3 1 What s your serial
AIRREX AIR CONDITIONER USER S MANUAL HSC-24A Thank you for purchasing an AIRREX AIR CONDITIONER. BEFORE operation please read this user s manual carefully. Keep this manual readily available. It is ESSENTIAL
Please read this owner s Manual carefully before operating the unit. - Cooling - Heating - Dehumidifying - Fan TABLE OF CONTENTS INTRODUCTION 2 IMPORTANT SAFEGUARDS...2 PACKAGE CONTAINS..2 NAMES OF PARTS.3
Portable Air Conditioner OWNER S MANUAL Read these instructions before use Model: MM14CCS Customer Support : 1-800-474-2147 Voltage rating: 115V~60Hz Power rating : 1400W For product inquiries or support
USER S MANUAL MaxPower 400-600 UPS Uninterruptible Power System 28-2MAXPO0018 IMPORTANT SAFETY INSTRUCTIONS SAVE THESE INSTRUCTIONS This manual contains important instructions for models MaxPower 400 and
A/C MANIFOLD GAUGE SET R134A 92649 ASSEMBLY AND OPERATING INSTRUCTIONS All technicians opening the refrigeration circuit in automotive air conditioning systems MUST be certified in refrigerant recovery
DIESEL PROTECTION SYSTEMS Automatic Diesel Engine Shut Down System for Safe Area Applications SELECTION, APPLICATION AND MAINTENANCE Series 300 Series 310 SYSTEM DESCRIPTION Suitable for attended engine
Instruction Manual Alfa Laval SB Membrane Sample Valve ESE02963-EN1 2015-10 Original manual Table of contents The information herein is correct at the time of issue but may be subject to change without
Portable Air Conditioner OWNER S MANUAL Read these instructions before use 8 Model: MN12CES / MN10CESWW Voltage rating: 120V~60Hz Power rating : 1100W (MN12CES) Power rating : 900W (MN10CESWW) Customer
Operating Instructions Sake Warmer Ti-2 CONTENTS Safety Precautions... 3 Purpose of Use... 4 Operation... 5 Names of Parts... 5 Inspection and Installation Before Use... 6 Instructions for Use... 7 After
1.0 IMPORTANT RECEIVING INSTRUCTIONS Visually inspect all components for shipping damage. Shipping Damage is not covered by warranty. If shipping damage is found, notify carrier at once. The carrier is
USER INSTRUCTIONS FOR 10 LITRE PORTABLE DEHUMIDIFIER MODEL NO. DHMD102 THANK YOU FOR CHOOSING YOUR NEW DEHUMIDIFIER. BEFORE USING THE UNIT READ THESE INSTRUCTIONS FULLY AND RETAIN THEM FOR FUTURE REFERENCE
INSTRUCTION MANUAL PRESSURE TANK PET-10 This manual contains IMPORTANT WARNINGS and INSTRUCTIONS. Equipment in this manual is exclusively for painting purposes. Do not use for other purposes. The operator
9-27-2010 National- Spencer Inc. 19.2V HEAVY DUTY GREASE GUN PRODUCT SPECIFICATION Charger Input Power 110 VAC Battery Output Power 19.2V Battery Capacity 1500 MAH Battery Pack Charge Time 1 Hour Maximum
ARCO Electric Products Installation and Maintenance Manual Low Voltage Automatic Power Factor Correction Capacitor Systems 2013 READ CAREFULLY These instructions are intended to cover good practices in
BRAVER UPS (Uninterruptible Power System) User s Manual Safety CAUTION! This UPS utilizes voltages that may be hazardous. Do not attempt to disassemble the unit. The unit contains no user replaceable parts.
WWW.BURCAM.COM 2190 Boul. Dagenais West TEL: 514.337.4415 LAVAL (QUEBEC) FAX: 514.337.4029 CANADA H7L 5X9 email@example.com Your pump has been carefully packaged at the factory to prevent damage during shipping.
FLUORESCENT UV- RING LIGHT OPERATING INSTRUCTION Caution! UV-radiation of this device is in the range of UV-A (320-400 nm). Direct exposure to eyes shall therefore be avoided. UV protection glasses shall