1 PS3.8 DICOM PS c - Network Communication Support for Message Exchange
2 Page 2 PS3.8: DICOM PS c - Network Communication Support for Message Exchange Copyright 2015 NEMA
3 DICOM PS c - Network Communication Support for Message Exchange Page 3 Table of Contents Notice and Disclaimer... 9 Foreword Scope and Field of Application Normative References International Standards Other Documents Definitions Reference Model Definitions Naming and Addressing Definitions Service Conventions Definitions Presentation Service Definitions ACSE Service Definitions DICOM Introduction and Overview Definition DICOM Communication Support Definitions Symbols and Abbreviations Conventions Network Communication Support Environment OSI Upper Layer Service for DICOM Application Entities A-ASSOCIATE Service A-ASSOCIATE Parameters Mode (Fixed) Application Context Name Calling AE Title Called AE Title Responding AE Title (Fixed) User Information Result Result Source Diagnostic Calling Presentation Address Called Presentation Address Responding Presentation Address Presentation Context Definition List Presentation Context Definition Result List Presentation Requirements (Fixed Value) Session Requirements (Fixed Value) Other Parameters A-ASSOCIATE Service Procedure A-RELEASE Service A-RELEASE Parameters Reason (Fixed) Result (Fixed) A-RELEASE Service Procedure A-ABORT Service A-ABORT Parameters Abort Source A-ABORT Service Procedure A-P-ABORT Service A-P-ABORT Parameter A-P-ABORT Service Procedure Sequencing Information P-DATA Service P-DATA Parameters DICOM OSI Upper Layer Profile DICOM Upper Layer Protocol for TCP/IP Use of the Transport Service Provided By TCP General... 37
4 Page 4 DICOM PS c - Network Communication Support for Message Exchange Opening a TCP Transport Connection Transferring Data On a TCP Connection Closing a TCP Transport Connection ARTIM Timer DICOM Upper Layer Protocol for TCP/IP State Machine Machine States Definition State Machine Actions Definition DICOM Upper Layer Protocol for TCP/IP State Transition Table DICOM Upper Layer Protocol for TCP/IP Data Units Structure General A-ASSOCIATE-RQ PDU Structure Application Context Item Structure Presentation Context Item Structure Abstract Syntax Sub-Item Structure Transfer Syntax Sub-Item Structure User Information Item Structure A-ASSOCIATE-AC PDU Structure Application Context Item Structure Presentation Context Item Structure Transfer Syntax Sub-Item Structure User Information Item Structure A-ASSOCIATE-RJ PDU Structure P-DATA-TF PDU Structure Presentation Data Value Item Structure A-RELEASE-RQ PDU Structure A-RELEASE-RP PDU Structure A-ABORT PDU Structure Conformance Conformance Requirements Retired TCP/IP Network Communication Support Conformance Statement A. Application Context Names (Informative) A.1. Application Context Definition A.2. DICOM Application Context Name Encoding and Registration A.2.1. DICOM Registered Application Context Names B. Abstract and Transfer Syntaxes (Informative) B.1. Abstract Syntax Definition B.2. Transfer Syntax Definition B.3. DICOM Abstract and Transfer Syntax Names Encoding and Registration B.3.1. DICOM Registered Abstract and Transfer Syntax Names B.3.2. Privately Defined Abstract and Transfer Syntax Names C. DICOM Addressing (Normative) C.1. DICOM Application Entity Titles C.2. Naming and Addressing Usage Rules D. Use and Format of the A-ASSOCIATE User Information Parameter (Normative) D.1. Maximum Length Negotiation D.1.1. Maximum Length Sub-Item Structure (A-ASSOCIATE-RQ) D.1.2. Maximum Length Sub-Item Structure (A-ASSOCIATE-AC) D.2. Extended User Information Negotiation E. Usage of the P-DATA Service By the DICOM Application Entity (Normative) E.1. Encapsulation Rules E.2. Message Control Header Encoding F. DICOM UL Encoding Rules for Application Contexts, Abstract Syntaxes, Transfer Syntaxes (Normative) F.1. Encoding Rules G. Overview of the OSI Layer and Services Concepts (Informative) H. Index of Item and PDU Types (Informative)... 71
5 DICOM PS c - Network Communication Support for Message Exchange Page 5 List of Figures 1-1. ISO OSI Basic Reference Model DICOM Network Protocol Architecture Associate Request Association Release Association User Initiated Abort Provider Initiated Abort Data Transfer Protocol Data Units Structure and Encoding Protocol Data Units Structure and Encoding E.2-1. Presentation Data Value and the Message Control Header G-1. Relationship of Services to Protocol G-2. Service Description Conventions... 70
6 Page 6 DICOM PS c - Network Communication Support for Message Exchange
7 DICOM PS c - Network Communication Support for Message Exchange Page 7 List of Tables 7-1. Upper Layer Services Key A-ASSOCIATE Service Parameters A-ASSOCIATE Service Parameter (Fixed or Not Used) A-RELEASE Service Parameters A-ABORT Service Parameters A-P-ABORT Service Parameters P-DATA Service Parameter No Association Association Establishment Data Transfer Association Release Waiting for Transport Connection Close Association Establishment Related Actions Data Transfer Related Actions Association Release Related Actions Association Abort Related Actions DICOM Upper Layer Protocol State Transition Table ASSOCIATE-RQ PDU Fields Application Context Item Fields Presentation Context Item Fields Abstract Syntax Sub-Item Fields Transfer Syntax Sub-Item Fields User Information Item Fields ASSOCIATE-AC PDU Fields Presentation Context Item Fields Transfer Syntax Sub-Item Fields User Information Item Fields ASSOCIATE-RJ PDU Fields P-DATA-TF PDU Fields Presentation-Data-Value Item Fields A-RELEASE-RQ PDU Fields A-RELEASE-RP PDU Fields A-ABORT PDU Fields D.1-1. Maximum Length Sub-Item Fields (A-ASSOCIATE-RQ) D.1-2. Maximum Length Sub-Item Fields (A-ASSOCIATE-AC)... 63
8 Page 8 DICOM PS c - Network Communication Support for Message Exchange
9 DICOM PS c - Network Communication Support for Message Exchange Page 9 Notice and Disclaimer The information in this publication was considered technically sound by the consensus of persons engaged in the development and approval of the document at the time it was developed. Consensus does not necessarily mean that there is unanimous agreement among every person participating in the development of this document. NEMA standards and guideline publications, of which the document contained herein is one, are developed through a voluntary consensus standards development process. This process brings together volunteers and/or seeks out the views of persons who have an interest in the topic covered by this publication. While NEMA administers the process and establishes rules to promote fairness in the development of consensus, it does not write the document and it does not independently test, evaluate, or verify the accuracy or completeness of any information or the soundness of any judgments contained in its standards and guideline publications. NEMA disclaims liability for any personal injury, property, or other damages of any nature whatsoever, whether special, indirect, consequential, or compensatory, directly or indirectly resulting from the publication, use of, application, or reliance on this document. NEMA disclaims and makes no guaranty or warranty, expressed or implied, as to the accuracy or completeness of any information published herein, and disclaims and makes no warranty that the information in this document will fulfill any of your particular purposes or needs. NEMA does not undertake to guarantee the performance of any individual manufacturer or seller's products or services by virtue of this standard or guide. In publishing and making this document available, NEMA is not undertaking to render professional or other services for or on behalf of any person or entity, nor is NEMA undertaking to perform any duty owed by any person or entity to someone else. Anyone using this document should rely on his or her own independent judgment or, as appropriate, seek the advice of a competent professional in determining the exercise of reasonable care in any given circumstances. Information and other standards on the topic covered by this publication may be available from other sources, which the user may wish to consult for additional views or information not covered by this publication. NEMA has no power, nor does it undertake to police or enforce compliance with the contents of this document. NEMA does not certify, test, or inspect products, designs, or installations for safety or health purposes. Any certification or other statement of compliance with any health or safety-related information in this document shall not be attributable to NEMA and is solely the responsibility of the certifier or maker of the statement.
10 Page 10 DICOM PS c - Network Communication Support for Message Exchange
11 DICOM PS c - Network Communication Support for Message Exchange Page 11 Foreword This DICOM Standard was developed according to the procedures of the DICOM Standards Committee. The DICOM Standard is structured as a multi-part document using the guidelines established in [ISO/IEC Directives, Part 3].
12 Page 12 DICOM PS c - Network Communication Support for Message Exchange
13 DICOM PS c - Network Communication Support for Message Exchange Page 13 1 Scope and Field of Application The Communication Protocols specified in this part of PS 3 closely fit the ISO Open Systems Interconnection Basic Reference Model (ISO , see Figure 1-1). They relate to the following layers: Physical, Data Link, Network, Transport, Session, Presentation and the Association Control Services (ACSE) of the Application layer. The communication protocols specified by this part are general purpose communication protocols (TCP/IP) and not specific to this standard. The other aspects of the Application Layer protocols are addressed in other parts of this standard as discussed in PS3.1 PS3.1. Layer OSI End-System Peer Protocol OSI End-System Application Presentation Session Transport Network Data Link Physical Physical Media Figure 1-1. ISO OSI Basic Reference Model
14 Page 14 DICOM PS c - Network Communication Support for Message Exchange
15 DICOM PS c - Network Communication Support for Message Exchange Page 15 2 Normative References [ISO/IEC Directives, Part 3] ISO/IEC Drafting and presentation of International Standards. 2.1 International Standards ISO , Information Processing Systems - Open Systems Interconnection - Basic Reference Model ISO , OSI Basic Reference Model - Part 3: Naming and Addressing ISO 8327:1987, Information Processing Systems - Open Systems Interconnection - Connection Oriented Session Protocol Specification ISO 8327/AM 2, Information Processing Systems - Open Systems Interconnection - Connection Oriented Session Protocol Specification - Amendment 2: Incorporation of Unlimited User Data ISO 8649:1987, Information Processing Systems - Open Systems Interconnection - Service Definition for the Association Control Service Element ISO 8650:1987, Information Processing Systems - Open Systems Interconnection - Protocol Specification for the Association Control Service Element ISO TR 8509, Information Processing Systems - Open Systems Interconnection - Service Conventions ISO 8822:1988, Information Processing Systems - Open Systems Interconnection - Connection-Oriented Presentation Service Definition ISO 8823:1988, Information Processing Systems - Open Systems Interconnection - Connection Oriented Presentation Protocol Specification ISO 8824:1990, Information Processing Systems - Open Systems Interconnection - Specification of Abstract Syntax Notation One (ASN.1) ISO 8825:1990, Information Processing Systems - Open Systems Interconnection - Specification of Basic Encoding Rules for Abstract Syntax Notation One (ASN.1) ISO/IEC 9545, Information Processing Systems - Open Systems Interconnection - Application Layer Structure ISO/IEC , Information technology - Open Systems Interconnection - Procedures for the operation of OSI Registration Authorities: General procedures and top arcs of the ASN.1 Object Identifier tree ISO/IEC TR , Information Processing Systems - Open Systems Interconnection - International Standardized Profiles, Part 1: Taxonomy Framework 2.2 Other Documents NIST Special Publication Stable Implementation Agreements for Open Systems Interconnection Protocols RFC 791, Internet Protocol - DARPA Internet Protocol Specification RFC 792, Internet Control Message Protocol - DARPA Internet Program Protocol Specification RFC 793, Transmission Control Program - DARPA Internet Protocol Specification RFC 950, Internet Subnetting RFC 1881, IPv6 Address Allocation Management RFC 2460, Internet Protocol, Version 6 (IPv6) Specification
16 Page 16 DICOM PS c - Network Communication Support for Message Exchange
17 DICOM PS c - Network Communication Support for Message Exchange Page 17 3 Definitions 3.1 Reference Model Definitions This part of the Standard is based on the concepts developed in ISO and makes use of the following terms defined in it: a. application entity b. application layer c. application process d. data link layer e. layer entity f. network layer g. physical layer h. presentation layer i. presentation service j. protocol or layer protocol k. protocol data unit or layer protocol data unit l. service or layer service m. service access point n. session layer o. transfer syntax p. transport layer q. transport protocol r. transport connection 3.2 Naming and Addressing Definitions This part of the Standard makes use of the following terms defined in ISO : a. calling presentation address b. called presentation address c. responding presentation address d. called application entity title e. calling application entity title 3.3 Service Conventions Definitions This part of the Standard makes use of the following terms defined in ISO/TR 8509: a. service provider
18 Page 18 DICOM PS c - Network Communication Support for Message Exchange b. service user c. confirmed service d. non-confirmed service e. primitive f. request (primitive) g. indication (primitive) h. response (primitive) i. confirmation (primitive) 3.4 Presentation Service Definitions This part of the Standard makes use of the following terms defined in ISO 8822: a. abstract syntax b. abstract syntax name c. presentation context d. presentation data values e. transfer syntax f. transfer syntax name 3.5 ACSE Service Definitions This part of the Standard makes use of the following terms defined in ISO 8649: a. association or application association b. application context c. association control service element d. association initiator 3.6 DICOM Introduction and Overview Definition a. message 3.7 DICOM Communication Support Definitions The following definitions are commonly used in this part of the Standard: Unique identifier (UID) :The scheme used to provide global unique identification for objects. It uses the structure defined by ISO 8824 for OSI Object Identifiers. DICOM upper layer:the Upper Layer protocols are related to the Session, Presentation and part of the Application Layer of the ISO reference model. These protocols provide the Upper Layer Service. This Service is a proper subset of the ACSE Service and OSI Presentation Layer Service.
19 DICOM PS c - Network Communication Support for Message Exchange Page 19 4 Symbols and Abbreviations The following symbols and abbreviations are used in this part of the Standard. ACR ACSE ASCII AE ANSI AP ASE ARTIM CEN TC251 DICOM FDDI American College of Radiology Association Control Service Element American Standard Code for Information Interchange Application Entity American National Standards Institute Application Process Application Service Element Association Request/Reject/Release Timer Comite Europeen de Normalisation-Technical Committee Medical Informatics Digital Imaging and Communications in Medicine Fiber Distributed Data Interface HL7 Health Level 7 IEC IEEE ISDN ISO ISP JIRA LAN NEMA NIST OSI PDU PDV SAP TCP/IP UID UL WAN International Electrotechnical Commission Institute of Electrical and Electronics Engineers Integrated Services Digital Network International Organization for Standardization International Standardized Profile Japan Medical Imaging and Radiological Systems Industries Association Local Area Network National Electrical Manufacturers Association National Institute of Standards and Technology Open Systems Interconnection Protocol Data Unit Presentation Data Values Service Access Point Transmission Control Program/Internet Protocol Unique Identifier Upper Layers Wide Area Network
20 Page 20 DICOM PS c - Network Communication Support for Message Exchange
21 DICOM PS c - Network Communication Support for Message Exchange Page 21 5 Conventions The following conventions are used for the service description tables shown in this part of the Standard. (=) The same parameter value used in the indication or confirmation shall be the same as used in the request or response respectively. C M MF NU P U UF Conditional (on User Option) Mandatory usage Mandatory with a fixed value Not Used Provider Initiated User Option User Option with a fixed value Blank entries are not applicable.
22 Page 22 DICOM PS c - Network Communication Support for Message Exchange
23 DICOM PS c - Network Communication Support for Message Exchange Page 23 6 Network Communication Support Environment The Network Communication Services specified in PS3.8 are a set of generic services provided to support the communication of DICOM Application Entities. They are a proper subset of the services offered by the OSI Presentation Service (ISO 8822) and of the OSI Association Control Service Element (ACSE) (ISO 8649). They shall be referred to as the Upper Layer Service or UL Service. The DICOM UL Service is specified in Section 7. This UL Service is provided by the Upper Layer Protocol for TCP/IP (see Section 9). Figure 6-1 shows the TCP/IP protocol stack that is available to support the communication of DICOM Application Entities. Medical Imaging Application DICOM Application Message Exhange BOUNDARY: DICOM Upper Layer Service DICOM Upper Layer Protocol for TCP/IP TCP/IP Transport Layer Network Figure 6-1. DICOM Network Protocol Architecture
24 Page 24 DICOM PS c - Network Communication Support for Message Exchange
25 DICOM PS c - Network Communication Support for Message Exchange Page 25 7 OSI Upper Layer Service for DICOM Application Entities This section provides a description of how to use the OSI Association Control Service Element (ACSE) and OSI Presentation Layer to provide the Upper Layer Service necessary to support the communication of DICOM Application Entities. This Upper Layer Service is a fully conformant subset of the services offered by the ACSE and the OSI Presentation Layer. The UL Services are listed in Table 7-1. Table 7-1. Upper Layer Services A-RELEASE SERVICE A-ASSOCIATE A-ABORT A-P-ABORT P-DATA TYPE Confirmed Confirmed Non-Confirmed Provider-initiated Non-Confirmed In addition to the Upper Layer Service specification, this section defines at the parameter level the use of each element of this Upper Layer Service by DICOM Application Entities. The rules guiding the use of this Upper Layer Service by the DICOM Application Entities are addressed in PS A-ASSOCIATE Service The establishment of an association between two AEs shall be performed through ACSE A-ASSOCIATE request, indication, response and confirmation primitives. The initiator of the service is hereafter called a requestor and the service-user that receives the A-ASSO- CIATE indication is hereafter called the acceptor. It shall be a confirmed service. The A-ASSOCIATE service supports the equivalent of a channel establishment in a point-to-point interface (see the retired PS3.9). Figure 7-1 illustrates the association establishment between two AEs. A-ASSOCIATE request Requestor DICOM UL Service Provider Acceptor A-ASSOCIATE indication A-ASSOCIATE response A-ASSOCIATE confirmation ( SAP ) ( SAP ) Figure 7-1. Associate Request
26 Page 26 DICOM PS c - Network Communication Support for Message Exchange A-ASSOCIATE Parameters Table 7-2 lists the parameters that shall be required for the A-ASSOCIATE service used by DICOM Application Entities in this standard. Table 7-2. Key A-ASSOCIATE Service Parameters A-ASSOCIATE parameter name Request Indication Response Confirmation application context name M M(=) M M(=) calling AE title M M(=) M M(=) called AE title M M(=) M M(=) user information M M(=) M M(=) result M M(=) result source M diagnostic U C(=) calling presentation address M M(=) called presentation address M M(=) presentation context definition list M M(=) presentation context definition list result M M(=) See Section 5 of this part for table conventions. Table 7-3 lists the parameters for the A-ASSOCIATE service that shall contain fixed values or shall not be used by DICOM Application Entities in this standard. Table 7-3. A-ASSOCIATE Service Parameter (Fixed or Not Used) A-ASSOCIATE parameter name Request Indication Response Confirmation mode UF MF(=) responding AE title MF MF(=) calling/called/responding AE qualifier NU NU NU NU calling/called/responding AP invoc-id NU NU NU NU calling/called/responding AE invoc-id NU NU NU NU responding presentation address MF MF(=) default context name/result NU NU NU NU presentation & session requirements UF UF(=) UF UF(=) other parameters (see ISO 8822 & 8649) NU NU NU NU Mode (Fixed) This parameter allows the negotiation of the optional Mode OSI-ACSE Service parameter. Only the default value of "normal" is used by DICOM Application Entities. Therefore, this parameter shall always specify the value "normal." Application Context Name This parameter identifies the application context proposed by the requestor. The acceptor shall return either the same or a different name. The returned name shall specify the application context to be used for this association. Further discussion on Application Context Names can be found in Annex A.
27 DICOM PS c - Network Communication Support for Message Exchange Page 27 An application context is an explicitly defined set of application service elements, related options, and any other information necessary for the interworking of application entities on an association. The offer of an alternate application context by the acceptor provides a mechanism for limited negotiation. If the requestor cannot operate in the acceptor's application context, it shall issue an A-Abort request primitive. Application Context Names for the DICOM Application Entity as well as Application Context Names usage rules are defined in PS Calling AE Title This parameter identifies the Application Entity (AE) that shall contain the requestor of the A-ASSOCIATE service. It is based on the Source DICOM Application Name. The relationship between DICOM Application Names and AE titles is specified in Annex C. The Calling AE title may or may not be the same as the Initiator Address present in DICOM Messages exchanged over the association. It is the responsibility of the UL User that received the A-ASSOCIATE-RQ to verify whether the Calling AE Title is one of its known remote DICOM Application Names Called AE Title This parameter identifies the Application Entity that shall contain the intended acceptor of the A-ASSOCIATE service. It is based on the Destination DICOM Application Name. The relationship between DICOM Application Name and AE titles is specified in Annex C. The Called AE title may or may not be the same as the Receiver Address present in DICOM Messages exchanged over the association. It is the responsibility of the UL User that received the A-ASSOCIATE-RQ to verify whether the Called AE Title is its (or one of its) DICOM Application Name(s) Responding AE Title (Fixed) This parameter identifies the AE that shall contain the actual acceptor of the A-ASSOCIATE service. In this standard it shall always contain the same value as the Called AE Title of the A-ASSOCIATE indication User Information This parameter shall be used by the requestor and the acceptor of the association to include DICOM Application Entity user information. Its meaning shall depend on the application context that accompanies the primitive. The usage of this parameter is specified in Annex D. 1. This parameter is used to carry initialization information for the DICOM Application Entities as defined in the application context specified by the value of the accompanying Application Context Name parameter. 2. Annex D specifies some user information sub-items, and references PS3.7 for the specification of additional sub-items. PS3.7, in turn, references PS3.4 for the specification of Service-class-application-information used in some sub-items Result This parameter shall be provided either by the acceptor of the A-ASSOCIATE request, by the UL service-provider (ACSE related function), or by the UL service-provider (Presentation related function). It shall indicate the result of using the A-ASSOCIATE service. It shall take one of the following symbolic values: a. accepted; b. rejected (permanent); c. rejected (transient).
28 Page 28 DICOM PS c - Network Communication Support for Message Exchange The rejected (permanent) implies that the association calling UL user (when returning such a result to an association request) does not need to "call later." A permanent situation exists that prevents the association establishment (e.g., remote DICOM Application Name unknown) Result Source The value of the parameter is supplied by the UL service-provider. It identifies the creating source of the Result parameter and the Diagnostic parameter, if present. It shall take one of the following symbolic values: a. UL service-user; b. UL service-provider (ACSE related function);c) UL service-provider (Presentation related function). If the Result parameter has the value "accepted," the value of this parameter is "UL service-user." Diagnostic This parameter shall only be used if the Result parameter has the value of "rejected (permanent) " or "rejected (transient)." It shall be used to provide diagnostic information about the result of the A-ASSOCIATE service. If the Result Source parameter has the value "UL service-user," it shall take one of the following symbolic values: a. no-reason-given b. application-context-name not supported c. calling-ae-title not recognized d. called-ae-title not recognized e. calling-ae-qualifier not recognized (see note) f. calling-ap-invocation-identifier not recognized (see note) g. calling-ae-invocation-identifier not recognized (see note) h. called-ae-qualifier not recognized (see note) i. called-ap-invocation-identifier not recognized (see note) j. called-ae-invocation-identifier not recognized (see note) If the Result Source parameter has the value "UL service-provider" (ACSE related function), it shall take one of the following symbolic values: a. no-reason-given b. no-common-ul version If the result source has the value "UL service-provider" (Presentation related function), it shall take the following symbolic values: a. no-reason-given b. temporary-congestion c. local-limit-exceeded d. called-(presentation) -address-unknown
29 DICOM PS c - Network Communication Support for Message Exchange Page 29 e. Presentation-protocol version not supported f. no-(presentation) Service Access Point (SAP) available Even though some of the above symbolic values correspond to parameter errors not used in this standard, they are included to allow the notification of errors resulting from the unauthorized use of these parameters Calling Presentation Address This parameter shall contain a structured destination address unambiguous within the global network address structure. This shall be a TCP/IP Address. See Annex C Called Presentation Address This parameter shall contain a structured destination address unambiguous within the global network address structure. This shall be a TCP/IP Address. See Annex C Responding Presentation Address In this standard, a responding presentation address shall always contain the same value as the called Presentation Address of the A-ASSOCIATE indication. This parameter shall contain a structured destination address unambiguous within the global network address structure Presentation Context Definition List This parameter used in an A-ASSOCIATE request or indication shall consist of a list containing one or more presentation contexts. Each item shall contain three components, a presentation context identification, an Abstract Syntax Name, and a list of one or more Transfer Syntax Names. The presentation context identification components of this parameter exist to distinguish presentation contexts in communication. Such an identification of presentation context(s) applies only within the context of a given association (i.e., different presentation contexts may be identified by the same presentation context identification on different associations). It is the association-requestor's responsibility to assign an arbitrary, but unused identifier for each proposed presentation context on a given association. There is no restriction on the ordering of the presentation contexts in relation to their identifiers. A separate presentation context will be associated with each Abstract Syntax Name in each of the elements of the Presentation Context Definition List parameter. If the same Abstract Syntax Name occurs more than once, a separate and distinctly identified presentation context will be generated for each occurrence (as only one Transfer Syntax per presentation context can be accepted). Abstract Syntaxes defined by this standard and used by DICOM Application Entites are defined in PS3.4. Transfer Syntaxes defined by this standard and used by DICOM Application Entities are defined in PS3.5. Further discussion on Abstract Syntaxes and Transfer Syntaxes can be found in Annex B Presentation Context Definition Result List This parameter used in the A-ASSOCIATE Response and Confirmation indicates the acceptance or rejection of each of the presentation context definitions proposed in the presentation context definition list parameter (Section ). The Presentation Context Definition Result List parameter shall take the form of a list of result values. There is a one to one correspondence between each one of these result values and each of the presentation contexts proposed in the Presentation Context Definition List parameter. Each result value represents either "acceptance," "user-rejection," or "provider-rejection." The values of the results are assigned by the UL user on the response service primitive. The result values may be sent in any order.
Digital Imaging and Communications in Medicine (DICOM) Part 8: Network Communication Support for Message Exchange Published by National Electrical Manufacturers Association 1300 N. 17th Street Rosslyn,
1 2 3 4 5 Digital Imaging and Communications in Medicine (DICOM) 6 7 8 Supplement 44: Clarification of network addressing and the retirement of non-tcp/ip communication. 9 10 11 12 13 14 15 16 17 18 VERSION:
Digital Imaging and Communications in Medicine (DICOM) Part 1: Introduction and Overview Published by National Electrical Manufacturers Association 1300 N. 17th Street Rosslyn, Virginia 22209 USA Copyright
Digital Imaging and Communications in Medicine (DICOM) Part 10: Media Storage and File Format for Media Interchange Published by National Electrical Manufacturers Association 1300 N. 17th Street Rosslyn,
Digital Imaging and Communications in Medicine (DICOM) Part 4: Service Class Specifications Published by National Electrical Manufacturers Association 1300 N. 17th Street Rosslyn, Virginia 22209 USA Copyright
Digital Imaging and Communications in Medicine (DICOM) Part 10: Media Storage and File Format for Media Interchange Published by National Electrical Manufacturers Association 1300 N. 17th Street Rosslyn,
Digital Imaging and Communications in Medicine (DICOM) Part 1: Introduction and Overview Published by National Electrical Manufacturers Association 1300 N. 17th Street Rosslyn, Virginia 22209 USA Copyright
American National Standard for Lighting Equipment Voltage Surge Requirements Secretariat: National Electrical Manufacturers Association Approved: June 19, 2015 American National Standards Institute, Inc.
NEMA XR 29-2013 Standard Attributes on CT Equipment Related to Dose Optimization and Management Published by: National Electrical Manufacturers Association 1300 North 17th Street Rosslyn, Virginia 22209
NEMA Standards Publication A Brief Comparison of NEMA 250 Enclosures for Electrical Equipment (1000 Volts Maximum) and IEC 60529 Degrees of Protection Provided by Enclosures (IP Code) Published by National
NEMA ICS 61131-1-2005 (R2013) IEC Publication 61131-1 Programmable Controllers Part 1: General Information Published by: National Electrical Manufacturers Association 1300 North 17 th Street, Suite 900
NETWORK MANAGEMENT FOR PICTURE ARCHIVING AND COMMUNICATION SYSTEMS by Edlic Nga-Lik Yiu B.A.Sc., Simon Fraser University, 2000 Edwood Nga-Wood Yiu B.A.Sc., Simon Fraser University, 2000 PROJECT SUBMITTED
NEMA SSL 3 HIGH-POWER WHITE LED BINNING FOR GENERAL ILLUMINATION NEMA Standards Publication SSL 3-2011 High-Power White LED Binning for General Illumination Published by: National Electrical Manufacturers
Revision of ANSI C84.1-2006 American National Standard For Electric Power Systems and Equipment Voltage Ratings (60 Hertz) Secretariat: National Electrical Manufacturers Association Approved December 6,
NTCIP 2301 v02 National Transportation Communications for ITS Protocol Simple Transportation Management Framework (STMF) Application Profile (AP) (AP-STMF) A Joint Standard of AASHTO, ITE, and NEMA version
ISO/IEC 14543-4-2 INTERNATIONAL STANDARD Edition 1.0 2008-05 Information technology Home electronic system (HES) architecture Part 4-2: Communication layers Transport, network and general parts of data
18821 Bardeen Ave. Irvine, CA 92612 Phone: 800.800.8600 Fax: 949.752.7317 Candelis, Inc. DICOM Conformance Statement ImageGrid Storage Server 0 INTRODUCTION ImageGrid Storage Server is a complete hardware/software,
Revision of ANSI C12.10-2004 American National Standard Physical Aspects of Watthour Meters Safety Standard Secretariat: National Electrical Manufacturers Association Approved June 28, 2011 American National
Digital Imaging and Communications in Medicine (DICOM) Supplement 113 Email Transport Prepared by: DICOM Standards Committee, Working Group 1300 N. 17th Street Suite 175 Rosslyn, Virginia 09 USA VERSION:
IEC 62304 CONSOLIDATED VERSION Edition 1.1 2015-06 colour inside Medical device software life cycle processes INTERNATIONAL ELECTROTECHNICAL COMMISSION ICS 11.040 ISBN 978-2-8322-2765-7 Warning! Make sure
INTERNATIONAL TELECOMMUNICATION UNION ITU-T X.690 TELECOMMUNICATION STANDARDIZATION SECTOR OF ITU (07/2002) SERIES X: DATA NETWORKS AND OPEN SYSTEM COMMUNICATIONS OSI networking and system aspects Abstract
INTERNATIONAL STANDARD ISO/IEC 9040 Second edition 1997-06-I 5 Information technology - Open Systems Interconnection - Virtual Terminal Basic Class Service Technologies de / information - In terconnexion
ANSI/NEMA C29.2-2012 American National Standard For Insulators Wet Process Porcelain and Toughened Glass Suspension Type Secretariat: National Electrical Manufacturers Association Approved: October 2012
Objectives of Lecture Network Architecture Show how network architecture can be understood using a layered approach. Introduce the OSI seven layer reference model. Introduce the concepts of internetworking
INTERNATIONAL STANDARD IEC 62056-21 First edition 2002-05 Electricity metering Data exchange for meter reading, tariff and load control Part 21: Direct local data exchange This English-language version
TECHNICAL SPECIFICATION IEC/TS 62351-7 Edition 1.0 2010-07 colour inside Power systems management and associated information exchange Data and communications security Part 7: Network and system management
HP Medical Archive Solutions DICOM Conformance Statement January 2007 (Third Edition) Part Number Copyright 2007, 2007 Hewlett-Packard Development Company, L.P. Confidential computer software. Valid license
IEC/PAS 62443-3 PUBLICLY AVAILABLE SPECIFICATION PRE-STANDARD Edition 1.0 2008-01 Security for industrial process measurement and control Network and system security INTERNATIONAL ELECTROTECHNICAL COMMISSION
IEC/TR 80001-2-3 TECHNICAL REPORT Edition 1.0 2012-07 colour inside Application of risk management for IT-networks incorporating medical devices Part 2-3: Guidance for wireless networks INTERNATIONAL ELECTROTECHNICAL
INTERNATIONAL TELECOMMUNICATION UNION ITU-T X.680 TELECOMMUNICATION STANDARDIZATION SECTOR OF ITU Amendment 1 (06/99) SERIES X: DATA NETWORKS AND OPEN SYSTEM COMMUNICATIONS OSI networking and system aspects
NEMA Standards Publication No. WC 26-2008 EEMAC Standards Publication 201-2008 Binational Wire and Cable Packaging Standard Published by National Electrical Manufacturers Association 1300 North 17th Street,
NEMA ICS 61131-4-2005 (R2013) IEC Publication 61131-4 Programmable Controllers Part 4: User Guidelines Published by: National Electrical Manufacturers Association 1300 North 17 th Street, Suite 900 Rosslyn,
Application Note IP Addressing A Simplified Tutorial July 2002 COMPAS ID 92962 Avaya Labs 1 All information in this document is subject to change without notice. Although the information is believed to
Digital Imaging and Communications in Medicine (DICOM) Part 5: Data Structures and Encoding Published by National Electrical Manufacturers Association 1300 N. 17th Street Rosslyn, Virginia 22209 USA Copyright
Xerox Multifunction Devices July 22, 2003 for the user Simple Network Management Protocol (SNMP) Primer Purpose This document introduces the history, purpose, basic functionality and common uses of SNMP
Overview Dipl.-Ing. Peter Schrotter Institute of Communication Networks and Satellite Communications Graz University of Technology, Austria Fundamentals of Communicating over the Network Application Layer
COMPUTER NETWORKS NETWORK ARCHITECTURE AND PROTOCOLS The Need for Standards Computers have different architectures, store data in different formats and communicate at different rates Agreeing on a particular
1. Introduction to IP... 1 2. IP Networks, IP Hosts and IP Ports... 1 3. IP Packet Structure... 2 4. IP Address Structure... 2 Network Portion... 2 Host Portion... 3 Global vs. Private IP Addresses...3
Introduction Computer Network. Interconnected collection of autonomous computers that are able to exchange information No master/slave relationship between the computers in the network Data Communications.
Chapter 2: Communicating over the 51 Protocol Units and Encapsulation For application data to travel uncorrupted from one host to another, header (or control data), which contains control and addressing
Protocols and Architecture Protocol Architecture. Layered structure of hardware and software to support exchange of data between systems/distributed applications Set of rules for transmission of data between
CLASS C PROFILE Prepared by: NTCIP Steering Group May 1996 Draft March 1998 Table of Contents FOREWORD...i Section 1: GENERAL...1-1 1.1 SCOPE...1-1 1.1.1 Background...1-1 1.1.2 Purpose of Document...1-1
Ethernet Babak Kia Adjunct Professor Boston University College of Engineering ENG SC757 - Advanced Microprocessor Design Ethernet Ethernet is a term used to refer to a diverse set of frame based networking
Technical Publication DICOM Conformance Statement Document Revision 3 October 20, 2010 2010 Copyright Brainlab AG 1 Conformance Statement Overview This is a conformance statement for the Brainlab software
DICOM 3.0 CONFORMANCE STATEMENT JETStream Workspace Version 1.0 9705-0067, Rev A October, 004 Copyright Statement Philips Medical Systems has taken care to ensure the accuracy of this document. However,
DOLPHIN DICOM IMAGING DICOM CONFORMANCE STATEMENT NOTICE The software described in this document has been validated in accordance with the governing DICOM standard at the time of this document's release.
Module 1 Introduction Lesson 2 Layered Network Architecture Specific Functional Objectives On Completion of this lesson, the students will be able to: State the requirement for layered approach Explain
Computer Networks - Xarxes de Computadors Teacher: Llorenç Cerdà Slides: http://studies.ac.upc.edu/fib/grau/xc Outline Course Syllabus Unit 2. IP Networks Unit 3. TCP Unit 4. LANs Unit 5. Network applications
DICOM Conformance Statement Merge Eye Care PACS v. 4.0 Merge Healthcare 900 Walnut Ridge Drive Hartland, WI 53029 USA 877.44.MERGE 2012 Merge Healthcare. The information contained herein is confidential
INTERNATIONAL STANDARD ISO/IEC 14662 First edition Information Technologies - Open-edi reference model Technologie de l'information - Modèle de référence EDI-ouvert Reference number Page 2 Contents Foreword...
Communication Networks MAP-TELE 2011/12 José Ruela Network basic mechanisms Network Architectures Protocol Layering Network architecture concept A network architecture is an abstract model used to describe
A Layered Approach to Computer Networks Physical Layer Data Link Layer Network Layer Transport Layer Session Layer Presentation Layer Application Layer Different layer of abstraction Different error control
Networking Test 4 Study Guide True/False Indicate whether the statement is true or false. 1. IPX/SPX is considered the protocol suite of the Internet, and it is the most widely used protocol suite in LANs.
ΕΠΛ 674: Εργαστήριο 5 Firewalls Παύλος Αντωνίου Εαρινό Εξάμηνο 2011 Department of Computer Science Firewalls A firewall is hardware, software, or a combination of both that is used to prevent unauthorized
Computer Networks Chapter 5 Transport Protocols Transport Protocol Provides end-to-end transport Hides the network details Transport protocol or service (TS) offers: Different types of services QoS Data
Version 2.7 DICOM 3.0 Conformance Statement 1 1. Conformance Statement Overview SOP Classes Query and Retrieve Study Root Query/Retrieve Information Model Find Study Root Query/Retrieve Information Model
NICC Document Requirements on Communications Providers in relation to Customer Line Identification display services and other related services Michael Faraday House, Six Dials Way, Stevenage SG1 2AY Tel.:
Technical Report TR-010 Requirements & Reference Models for ADSL Access Networks: The SNAG Document June 1998 Abstract: This document outlines architectural requirements and reference models for ADSL services
INTERNATIONAL STANDARD IEC 60870-5-103 First edition 1997-12 Telecontrol equipment and systems Part 5-103: Transmission protocols Companion standard for the informative interface of protection equipment
14 June 2002 Clause 5 Clause 5 reprinted with permission from: IEEE Std 802.15.1-2002 IEEE Standard for Information technology Telecommunications and information exchange between systems Local and metropolitan
INTERNATIONAL STANDARD ISO/IEC 12207 IEEE Std 12207-2008 Second edition 2008-02-01 Systems and software engineering Software life cycle processes Ingénierie des systèmes et du logiciel Processus du cycle
TECHNICAL REPORT IEC/TR 62443-3-1 Edition 1.0 2009-07 colour inside Industrial communication networks Network and system security Part 3 1: Security technologies for industrial automation and control systems
g GE Healthcare Technical Publications Direction 1007228 Revision 0 Navigation Plus System Version 5.2 For DICOM Copyright 2004 by General Electric Co. Do not duplicate THIS PAGE LEFT INTENTIONALLY BLANK
DICOM Conformance Statement CBS Images and Worklist Version 2.01 Compulink Inc. 1100 Business Center Circle Thousand Oaks, CA 91320 Internal document number: 4226-000-000-001 rev 1 Creation Date: 20131101
DICOM Correction Proposal STATUS Assigned Date of Last Update 20165/01/08 Person Assigned Submitter Name Harry Solomon Harry Solomon Submission Date 2015/09/21 Correction Number CP-1550 Log Summary: Refactor
Application Protocol Reference Architecture Chapter 7 Application Protocol Reference Architecture This chapter proposes an alternative reference architecture for application protocols. The proposed reference
A Seminar Report on Simple Network Management Protocol Submitted in partial fulfillment of the requirement for the award of degree Of Computer Science SUBMITTED TO: SUBMITTED BY: www.studymafia.org www.studymafia.org
EPJ Web of Conferences 68, 000028 (2014) DOI: 10.1051/ epjconf/ 20146800028 C Owned by the authors, published by EDP Sciences, 2014 Implementation of DICOM Modality Worklist at Patient Registration Systems
100-101: Interconnecting Cisco Networking Devices Part 1 v2.0 (ICND1) Course Overview This course provides students with the knowledge and skills to implement and support a small switched and routed network.
- System and Network Management Network Management : ability to monitor, control and plan the resources and components of computer system and networks network management is a problem created by computer!
TECHNICAL SPECIFICATION IEC/TS 62443-1-1 Edition 1.0 2009-07 colour inside Industrial communication networks Network and system security Part 1-1: Terminology, concepts and models INTERNATIONAL ELECTROTECHNICAL
ΕΠΛ 475: Εργαστήριο 9 Firewalls Τοίχοι πυρασφάλειας Department of Computer Science Firewalls A firewall is hardware, software, or a combination of both that is used to prevent unauthorized Internet users
INTERNATIONAL STANDARD ISO 18185-4 First edition 2007-05-01 Freight containers Electronic seals Part 4: Data protection Conteneurs pour le transport de marchandises Scellés électroniques Partie 4: Protection