1 IOL Power Calculation After Myopic LASIK Hany Helaly, Lecturer of Ophthalmology, Faculty of Medicine, Alexandria University.
2 SUPERVISORS Prof. Dr. Mohammad El-Hifnawy Professor of Ophthalmology Faculty of Medicine Alexandria University Prof. Dr. Mohamed Shafik Professor of Ophthalmology Faculty of Medicine Alexandria University Dr. Amro Abou El-Kheir Assistant Professor of Ophthalmology Faculty of Medicine Alexandria University
3 IOL power calculation following myopic LASIK There is an increasing number of cataract surgeries in eyes after LASIK. IOL power calculation turned out to be problematic.
4 IOL power calculation following myopic LASIK Measured average k-reading + standard IOL power calculation formulas overestimation ( postop. hyperopia )
5 IOL power calculation following myopic LASIK After myopic LASIK : postop. hyperopia After hyperopic LASIK : postop. myopia Error 1 D in IOL power 0.71 D error at the spectacle plane Feiz V, Mannis MJ, Garcia-Ferrer F, et al. Intraocular lens power calculation after laser in situ keratomileusis for myopia and hyperopia: a standardized approach. Cornea 2001;20:792-7.
6 Causes of Error in estimating corneal power after excimer laser corneal surgery Keratometers & Topography devices measure anterior surface of central cornea. K in (D) is derived from r in mm measured using an effective refractive index to falsely consider a single refractive lens representing anterior and posterior surfaces.
7 Causes of Error in estimating corneal power after excimer laser corneal surgery This effective refractive index is considered valid as long as radii of anterior and posterior surfaces of the cornea are proportionate and resembles that of the model eye. RI is a standardized value ( in most keratometers and CVK devices).
8 Causes of Error in estimating corneal power after excimer laser corneal surgery r of anterior surface is considerably increased and distance between both refractive surfaces is decreased. Therefore, calculating keratometric diopters from anterior radius of curvature is not accurate.
9 Formula error The estimation of effective lens position (ELP) by the third- or fourth-generation formulas is not correct when the postoperative corneal power values are used.
10 Formula error Most 3 rd generation theoretical formulas (e.g. SRK/T, Holladay) use AL & corneal power values to predict ELP. Problem : two-variable prediction formulas rely on AL & central corneal power to predict ELP.
11 Formula error These formulas assume: The longer the axial length or steeper k values, the deeper the anterior chamber. The shorter the axial length or flatter k values, the shallower the anterior chamber.
12 Advances in IOL power calculation formulas after keratorefractive surgery The double-k formula. Holladay 2 formula. Haigis-L formula. Shammas-PL formula.
13 The double-k formula For 3 rd generation formulas: SRK/T. Holladay 1. Hoffer Q. This method uses 2 k-values: pre refractive for calc of ELP post refractive for the vergence formula that finally gives the IOL power
14 Holladay 2 formula Uses another innovative approach, which is to use measurements of corneal power. corneal diameter. anterior chamber depth (ACD). lens thickness. refractive error. axial length. to further refine the ELP calculation.
15 Haigis-L formula Using a correction curve to correct the current IOLMaster measurement of corneal radius to derive the effective equivalent corneal power. r corr = / ( X r meas ) Then enter it into the regular Haigis formula.
16 Shammas-PL formula Shammas described a corneal power correction method that uses only data available at the time of cataract surgery ( i.e. post-lasik K-readings). Kc = (1.14 X Kpost) - 6.8
17 Shammas-PL formula This corrected K-value is used in the Shammas post-lasik (Shammas-PL) formula In which the ELP does not vary with the corneal curvature that has been altered by the LASIK procedure.
18 There are many methods to calculate intraocular lens power after keratorefractive surgery, indicating that there is no single accurate measure that can be reliable to perform this task.
19 The aim of this work was to: Evaluate the accuracy of corneal power measurements for intraocular lens power calculation after myopic LASIK. Try to deduce a regression formula for IOL power calculation after myopic LASIK.
20 Subjects and Methods A clinical study that included 45 eyes of 37 patients with previous myopic LASIK who were scheduled for cataract surgery.
21 Exclusion criteria: Complicated cataract surgery. Corneal pathology e.g. corneal opacities. Refractive surgery complications e.g. flap related complications, abnormal healing.
22 Evaluation of the patients included 1-Complete History taking including refractive history. 2- Complete ophthalmologic examination. 3- Measuring k values by different methods. 4- AL & ACD were obtained from IOLMaster optical biometer.
23 Measuring k values by different methods 1- Manual keratometry. 2- Auto-keratometery. 3- Pentacam. 4- IOLMaster.
24 In this way, different k values were obtained Manual k values. Automated k values. IOLMaster k readings. Calculated k readings from clinical history method (if available data). Pentacam derived k values: Sim-k. Holladay EKR at 1 mm, 2 mm, 3 mm, 4 mm, and 4.5 mm. True net power.
25 Routine phacoemulsification + foldable IOL in the bag was performed through clear corneal tunnel incision
26 The implanted IOL type was either: Hydrophobic acrylic IOL (Acrysof SA60AT, Alcon Surgical Inc., Fort Worth, TX) Hydrophilic acrylic IOL with an A-constant value of 118: CIMflex42, Cima technology Inc., Pittsburg, PA SC25-FOLD, Eyekon Medical Inc., Clearwater, FL OcuFlex ANU6, Polymer Technologies International, Gujarat, India
27 Final refraction was obtained one to four months after the surgery. Most eyes (55 eyes) were examined during the second month, while the rest of the eyes (5 eyes) were examined during the third and the fourth months.
28 To calculate IOL power for post LASIK eyes All available k-readings were used in Haigis formula, SRK/T formula, and SRK/T formula with double k modification. Haigis-L formula. Shammas modified IOLMaster k-reading was used in both Shammas-PL formula and double k modification of Holladay formula.
29 To calculate IOL power for post LASIK eyes IOL power was determined by a regression formula deduced from the results of the study. Different combinations of means of multiple IOL powers calculated using various formulas were hypothetically calculated.
30 For the double-k modification of the thirdgeneration formulas, an average k value of D was used as the pre-refractive surgery k value.
31 The predictive accuracy of the calculation was analyzed by IOL prediction error = implanted IOL calculated IOL power with the actual postoperative MRSE as a target refraction Refraction prediction error = actual MRSE predicted MRSE For each of them, the arithmetic mean (mean arithmetic error) and absolute mean (mean absolute error) were calculated.
32 The predictive accuracy of the calculation was analyzed by In addition, percentages of correct refraction predictions within +/ D, +/-1.00 D, and +/ D were derived.
33 Statistical Analysis Comparisons of the means were performed with: t test for paired samples. ANOVA (Analysis Of Variance) test. Chi square and Fisher's exact tests were used to compare different frequencies.
34 Post-LASIK eyes 45 eyes of 37 patients. Age: years (33 to 65 years). Sex: Females: 26 (70%). Males: 11 (30%).
35 Post-LASIK eyes Mean AL: mm range to mm. Mean ACD: mm range 2.74 to 4.86 mm.
36 K-values of the included eyes Mean +SD (D) Range (D) Auto k Manual k IOLMaster k Pentacam Sim k EKR 1 mm EKR 2 mm EKR 3 mm EKR 4 mm EKR 4.5 mm True net k
37 Haigis formula using different k- values
38 Haigis formula using different k- values Eyes within + 2 D Eyes within + 1 D Eyes within D Number (%) Number (%) Number (%) EKR 3 mm 43 (95.6) 34 (75.6) 16 (35.6) EKR 4 mm 44 (97.8) 32 (71.1) 17 (37.8) Arithmetic refractive error (D) : EKR 3 mm EKR 4 mm (-2.44 to 1.90) (-1.86 to 2.50)
39 SRK/T formula using different k- values
40 SRK/T formula using different k- values Eyes within + 2 D Number (%) Eyes within + 1 D Number (%) Eyes within D Number (%) True net k 45 (100.0) 32 (71.1) 14 (31.1) Arithmetic refractive error (D) : (-1.90 to 1.67)
41 SRK/T formula using different k- values The result of the present study with tru net k was inferior to that of Kim et al using the same formula. + 1 D D Kim et al(2009) 93 % 70 % Present study 71.1% 31.1% Kim SW, Kim EK, Cho BJ, Song KY, Kim T. Use of the Pentacam true net corneal power for intraocular lens calculation in eyes after refractive corneal surgery. J Refract Surg 2009; 25:285-9.
42 dk SRK/T formula using different k- values
43 dk SRK/T formula using different k- values within + 2 D Number (%) within + 1 D Number (%) within D Number (%) IOLMaster k 41 (91.1) 27 (60.0) 13 (28.9) Pentacam Sim k 44 (97.8) 32 (71.1) 18 (40.0) EKR 4 mm 44 (97.8) 36 (80.0) 19 (42.2) EKR 4.5 mm 44 (97.8) 29 (64.4) 16 (35.6) EKR 4 mm Arithmetic refractive error (D) (-2.04 to 1.97)
44 Formulas using modified IOLMaster k- values Eyes within + 2 D Eyes within + 1 D Eyes within D Number (%) Number (%) Number (%) Haigis-L 41 (91.1) 33 (73.3) 20 (44.4) Shammas-PL 45 (100.0) 38 (84.4) 26 (57.8) Shammas-dk Holladay 43 (95.6) 37 (82.2) 25 (55.6)
45 Formulas using modified IOLMaster k values Arithmetic refractive error (D) Mean + SD (Range) Haigis-L (-2.79 to 1.62) Shammas-PL Shammas-dk Holladay (-1.98 to 1.29) (-1.64 to 2.26)
46 Formulas using modified IOLMaster k- values Shammas and Shammas showed the best results were with Shammas-PL formula. Mean absolute IOL prediction error Eyes within + 1 D Shammas et al (2003) D 93.3% Present study D 84.4% Shammas HJF, Shammas MC, Garabet A, Kim J, Shammas A, Labree L. Correcting the corneal power measurements for intraocular lens power calculations after myopic laser in situ keratomileusis. Am J Ophthalmol 2003;
47 Formulas using modified IOLMaster k- values Haigis published better results for Haigis-L formula in his series of 186 eyes. Mean arithmetic refraction prediction error Haigis W (2008) D (-2.30 to 2.40 D) Present study D (-2.79 to 1.62 D) - Within + 2 D - Within + 1 D - Within D 98.4% 84.0% 61.0% 91.1% 73.3% 44.4% Haigis W. Intraocular lens calculation after refractive surgery for myopia: Haigis-L formula. J Cataract Refract Surg 2008; 34:
48 Regression formula IOL power = (result of SRK/T formula using IOLMaster k) + (0.181 X axial length) 2.151
49 Regression formula Arithmetic refractive error (D) Mean + SD (Range) (-1.70 to 1.60) after excluding a case with
50 From above, the best results were with: Haigis formula using EKR 4 mm. SRK/T formula using true net power. dk SRK/T formula using EKR 4 mm. Shammas-PL. Haigis-L. The deduced regression formula.
51 The best combinations of formulas were: "Shammas-PL + Haigis using EKR 4 mm". "Shammas-PL + Haigis using EKR 4 mm + Haigis-L". "Shammas-PL + dk SRK/T using EKR 4 mm". "Shammas-PL + dk SRK/T using EKR 4 mm + Haigis- L". "Shammas-PL + SRK/T using true net k.
52 Various combinations of formulas The five combinations yielded excellent results. within +2 D : 100% within + 1 D : > 88% within D : > 55%
53 Various combinations of formulas Shammas-PL + dk SRK/T EKR 4 mm Arithmetic refractive error (D) Eyes within + 2 D Eyes within + 1 D Eyes within D (-1.04 to 1.30) 45 (100.0) 42 (93.3) 30 (66.7)
54 Clinical History method The pre-refractive surgery data (preoperative k and change in refraction) were available in five cases. No case was within + 1 D of intended refraction and only 2 out 5 cases were within + 2 D.
55 Subgroup Analysis Didn t show statistically significant difference between both subgroups (with different implanted IOL material) except when using Haigis formula with EKR 3 mm.
56 within + 2 D within + 1 D within D Arithmetic refractive error Haigis using EKR 3 mm (95.6) (75.6) (35.6) SRK/T true net k (100.0) (71.1) (31.1) dk SRK/T using sim k (97.8) (71.1) (40) dk SRK/T using EKR 4 mm (97.8) (80.0) (42.2) Shammas-PL (100.0) (84.4) (57.8) Haigis-L (91.1) (73.3) (44.4) Regression formula (97.8) (80.0) (48.9) Shammas-PL + dk SRK/T using EKR 4 mm (100.0) (93.3) (66.7) (-2.44 to 1.90) (-1.90 to 1.67) (-1.58 to 2.28) (-2.04 to 1.97) (-1.98 to 1.29) (-2.79 to 1.62) (-3.21 to 1.60) (-1.04 to 1.30)
58 Conclusion Manual k & automated k readings were steeper than that measured by the IOLMaster Their use in IOL power calculation formulas after myopic LASIK would have left the patient in a more hyperopic condition.
59 Conclusion The use of Pentacam Scheimpflug imaging and Holladay equivalent k readings improved the accuracy of IOL power calculation after myopic LASIK.
60 Conclusion The use of means of combinations of formulas yielded excellent results better than any single formula alone for post-lasik eyes.
62 For post-lasik eyes When using IOLMaster alone: Shammas-PL formula The deduced regression formula Haigis-L formula
63 For post-lasik eyes When using Pentacam without Holladay report: the combination of "Shammas-PL + SRK/T using true net k. When using Pentacam with Holladay report: the combination of "Shammas-PL + dk SRK/T using EKR 4 mm.
J CATARACT REFRACT SURG - VOL 33, JANUARY 2007 No-history method of intraocular lens power calculation for cataract surgery after myopic laser in situ keratomileusis H. John Shammas, MD, Maya C. Shammas,
the cornea (P) may be considered as the sum of the power of the anterior (Pa) and posterior (Pp) corneal surfaces, as shown in the formula: P = Pa + Pp = (n2 n1)/r1 + (n3 n2)/r2, where n1 is the refractive
COMPARISON OF INTRAOCULAR LENS POWER CALCULATION METHODS IN EYES THAT HAVE UNDERGONE LASER-ASSISTED IN-SITU KERATOMILEUSIS BY Li Wang MD PhD, Marc A. Booth MD, AND Douglas D. Koch MD* ABSTRACT Purpose:
Comparison of Corneal Power and Intraocular Lens Power Calculation Methods after LASIK for Myopia Seyed Mohammad Reza Taheri, MD 1 Azita Kheiltash, MD, MPH 2 Hassan Hashemi, MD 1,3 Abstract Purpose: To
eyetube.net The Creator s Forum: IOL Power Calculations for Postrefractive Surgery Eyes Olsen Formula By Thomas Olsen, MD, PhD The calculation of IOL power in an eye that has undergone corneal refractive
Intraocular Lens Power Calculation after Myopic Refractive Surgery Theoretical Comparison of Different Methods Giacomo Savini, MD, Piero Barboni, MD, Maurizio Zanini, MD Objective: To evaluate the reliability
Corneal Refractive Power Estimation and Intraocular Lens Calculation after Hyperopic LASIK Shady T. Awwad, MD, 1,2 Patrick S. Kelley, MD, 1 R. Wayne Bowman, MD, 1 H. Dwight Cavanagh, MD, PhD, 1 James P.
J CATARACT REFRACT SURG - VOL 32, MARCH 26 Simple regression formula for intraocular lens power adjustment in eyes requiring cataract surgery after excimer laser photoablation Samuel Masket, MD, Seth Everett
ASCRS ASOA Symposium & Congress Technicians & Nurses Program April 17-21, 2015 San Diego, California Update on Post-Refractive Surgery IOL Calculations ASCRS Technicians and Nurses Program April 18 th,
American Academy of Ophthalmology Annual Meeting Orlando, Florida, 21-25 October 2011 OCT-based IOL power calculation for eyes with previous myopic and hyperopic laser vision correction Weeks Professor
THE BEST OF BOTH WORLDS Dual-Scheimpflug and Placido Reaching a new level in refractive screening GALILEI G4 Clinical Applications Corneal Implant Planning The comes with a licensable corneal inlay software
articles Intraocular lens power calculation after corneal refractive surgery: Double-K method Jaime Aramberri, MD Purpose: To determine the accuracy of a method of calculating intraocular lens (IOL) power
eyetube.net The Creator s Forum: IOL Power Calculations for Postrefractive Surgery Eyes The minds behind today s commonly used formulas provide overviews and pearls for use. By Wolfgang Haigis, Ms, PhD;
Intraocular lens calculations after refractive surgery Robert A. Latkany, MD, Amit R. Chokshi, MD, Mark G. Speaker, MD, PhD, Jodi Abramson, MD, Barrie D. Soloway, MD, Guopei Yu, MD, MPH Purpose: To evaluate
ALL-IN-ONE Optical Biometry, Dual Scheimpflug Tomography and Placido Topography GALILEI G6 Clinical Applications High confidence premium IOL selection The GALILEI G6 offers a link to the ray-tracing software
J CATARACT REFRACT SURG - VOL 32, DECEMBER 2006 ARTICLES Estimation of true corneal power after keratorefractive surgery in eyes requiring cataract surgery: BESSt formula Edmondo Borasio, MedCBQ Ophth,
Corneal Power Measurements Using Scheimpflug Imaging in Eyes With Prior Corneal Refractive Surgery Jack T. Holladay, MD, MSEE, FACS; Warren E. Hill, MD, FACS; Andreas Steinmueller, MSc ABSTRACT PURPOSE:
ORBSCAN II ASSISTED INTRAOCULAR LENS POWER CALCULATION FOR CATARACT SURGERY FOLLOWING MYOPIC LASER IN SITU KERATOMILEUSIS (AN AMERICAN OPHTHALMOLOGICAL SOCIETY THESIS) BY Henry Gelender MD ABSTRACT Purpose:
Insert to January 2011 Comfort and confidence for all IOL calculations Advanced Optical Biometry and Keratometry with LENSTAR LS 900 CONTENTS Lenstar and Ray-Tracing Calculations...3 By Jaime Aramberri,
Refractive Surprises In Post Radial Keratotomy Cataract Surgery And Review Of Literature To Calculate Lens Power In Post Refractive Surgery Cases Dr. Devdatt. S Raut, MBBS. Dr. Suresh Ramchandani, MS.,DNB.,DOMS,.
How do we use the Galilei for cataract and refractive surgery? Douglas D. Koch, MD Mariko Shirayama, MD* Li Wang, MD, PhD* Mitchell P. Weikert, MD Cullen Eye Institute Baylor College of Medicine Houston,
Toric IOL An option or a must? ~ 15% cataract surgical patients >1.5 D Options: spectacles, CLs, Incision along steep axis, LRI, AK, toric IOL, Excimer Laser or a combination Walter J. Stark, MD Professor
Supplement to October 2014 Master Your Toric Planning and Improve Refractive Outcomes With the LENSTAR LS 900 Sponsored by Contents Lenstar and the T-Cone Toric Platform... xx Offering sophisticated planning
Chapter - 3 IOL : POWER CALCULATION & SELECTION Precise IOL power calculation is essential for optimal benefits of implant surgery. Prior to1975, IOL power was calculated on the basis of clinical history,
OPTICAL COHERENCE TOMOGRAPHY BASED CORNEAL POWER MEASUREMENT AND INTRAOCULAR LENS POWER CALCULATION FOLLOWING LASER VISION CORRECTION (AN AMERICAN OPHTHALMOLOGICAL SOCIETY THESIS) By David Huang MD PhD,
Cornea Total Corneal Power Estimation: Ray Tracing Method versus Gaussian Optics Formula Li Wang, 1 Ashraf M. Mahmoud, 2 Betty Lise Anderson, 3 Douglas D. Koch, 1 and Cynthia J. Roberts 2 PURPOSE. To evaluate
Premium Lenses in Problematic Patients How to be successful with presbyopic lenses in eyes that have already had LASIK surgery. Walter Bethke, Managing Editor If implanting a monofocal intraocular lens
BESSt 2.0 IOL Power Calculator - REFERENCE GUIDE Introduction BESSt 2.0 IOL Power Calculator was designed to allow more accurate corneal and IOL power estimation in eyes needing either cataract or clear
UPDATE/REVIEW Intraocular lens power calculation after myopic excimer laser surgery with no previous data Juan Carlos Mesa-Gutiérrez, MD, PhD, FEBO; Antonio Rouras-López, MD; Isabel Cabiró-Badimón, MD;
Laser Refractive Cataract Surgery with the LenSx Laser a Novartis company LenSx Laser 2 3 An Evolving Definition of Cataract Average Age of US Cataract Patient is Projected to Decline Today: earlier diagnosis
Cataract Testing What a Patient undergoes prior to surgery FINANCIAL DISCLOSURE I have no financial interest or relationships to disclose What do most Technicians find to be the most mundane yet very important
Jack T. Holladay, M.D., M.S.E.E., F.A.C.S. ADVANCED IOL POWER CALCULATIONS Jack T. Holladay, MD, MSEE, FACS I. Formulas and Measurements A. Variables Used to Predict ACD 1. Binkhorst 2-1981 - AL 2. Holladay
Comparison Combined LASIK Procedure for Ametropic Presbyopes and Planned Dual Interface for Post-LASIK Presbyopes Using Small Aperture Corneal Inlay Minoru Tomita, MD, PhD 1,2 1) Shinagawa LASIK, Tokyo,
Jack T. Holladay, M.D., M.S.E.E., F.A.C.S. ADVANCED IOL POWER CALCULATIONS Jack T. Holladay, MD, MSEE, FACS I. Formulas and Measurements A. Variables Used to Predict ACD 1. Binkhorst 2-1981 - AL 2. Holladay
Comparison of Residual Stromal Bed Thickness and Flap Thickness at LASIK and Post-LASIK Enhancement in Femtosecond Laser-Created Flaps Lingo Y. Lai, MD William G. Zeh, MD Clark L. Springs, MD The authors
Cataract Surgery in Small Eyes Richard S. Hoffman, MD Clinical Associate Professor of Ophthalmology Oregon Health & Science University No Financial Interests Anatomic Classification Short AC depth with
IOLMaster 500 Excellence in Biometry. Precision. Simplicity. Outcomes. 2 IOLMaster 500 The gold standard in biometry is evolving to meet today s challenges Mastery is achieved through continuous practice
3 Biometry for Refractive Lens Surgery Mark Packer, I. Howard Fine, Richard S. Hoffman CORE MESSAGES 2 Achieving emmetropia in refractive lens surgery depends on accurate biometry and IOL power calculation.
Intraocular lens power calculation after incisional and thermal keratorefractive surgery Mark Packer, MD, Laurie K. Brown, COMT, COE, Richard S. Hoffman, MD, I. Howard Fine, MD Purpose: To evaluate the
ejpmr, 2015,2(3), 436-440 EUROPEAN JOURNAL OF PHARMACEUTICAL AND MEDICAL RESEARCH www.ejpmr.com Tumram et al. SJIF Impact Factor 2.026 Research Article ISSN 3294-3211 EJPMR CLINICAL OUTCOME OF TORIC IOL
A New Method of Calculating Intraocular Lens Power After Photorefractive Keratectomy Nicola Rosa, MD; Luigi Capasso, MD; Antonio Romano, MD ABSTRACT PURPOSE: To find a method of calculating intraocular
sight see OLYMPIA EYE & LASER CENTRE Your one stop vision centre Our ophthalmic centre offers comprehensive eye management, which includes medical, At the Olympia Eye & Laser Centre, our vision is to improve
CORRECTION C O OF ASTIGMATISM DURING CATARACT SURGERY Abdallah dllh K. Hassouna, M.D. Sherein S. Wahba, M.D. Ain Shams University 2009 Main target What is the main target for all phaco surgeons? Main Target
Overview of Refractive Surgery Michael N. Wiggins, MD Assistant Professor, College of Health Related Professions and College of Medicine, Department of Ophthalmology Jones Eye Institute University of Arkansas
User Manual Panacea IOL & Toric Calculator Version 8(6.0) Content Introduction...3 Use instructions and generalities...7 IOL Power & Toric Calculator...7 IOL Power Calculator... 9 Add Multifocal Graphic
Refractive Surgery Over the last 25 years developments in medical technology and Refractive Surgery allow almost all need for glasses and contact lenses to be eliminated. Currently there are a number of
Anterior Elevation Maps as the Screening Test for the Ablation Power of Previous Myopic Refractive Surgery Soo Yong Jeong, MD, Hee-Seung Chin, MD, PhD, Jung Hyub Oh, MD, PhD Department of Ophthalmology,
ORIGINAL ARTICLE Calculation of Intraocular Lens Power Using Orbscan II Quantitative Area Topography After Corneal Refractive Surgery Carlos G. Arce, MD; Eduardo S. Soriano, MD; Robert W. Weisenthal, MD;
CLINICAL SCIENCES A Direct Method to Measure the Power of the Central Cornea After Myopic Laser In Situ Keratomileusis Sergio Sónego-Krone, MD; Gerson López-Moreno, MD; Oscar V. Beaujon-Balbi, MD; Carlos
Insert to Sponsored by an educational grant from Haag-Streit USA January 2010 Staying on Target With Better Biometry A roundtable discussion about the new LENSTAR LS900, the first biometer of the entire
Mark Packer, M.D. Informed Consent for Refractive Lens Exchange (Clear Lens Replacement) This surgery involves the removal of the natural lens of my eye, even though it is not a cataract. The natural lens
Curtin G. Kelley, M.D. Director of Vision Correction Surgery Arena Eye Surgeons Associate Clinical Professor of Ophthalmology The Ohio State University Columbus, Ohio Refractive Errors Myopia (nearsightedness)
Supplement to Sponsored by an educational grant from Oculus, Inc. February 2008 The Pentacam: Offering a Clearer View Densitometry software for cataracts Topography for premium-channel cataract surgery
Keratoconus Detection Using Corneal Topography Jack T. Holladay, MD, MSEE, FACS ABSTRACT PURPOSE: To review the topographic patterns associated with keratoconus suspects and provide criteria for keratoconus
4/2/2014 Making the Most of High-tech Biometry Today s advanced technology calls for a well-informed user in order to optimize results. Here s help. By Christopher Kent Senior Editor As cataract surgery
Informed Consent for Refractive Lens Exchange (Clear Lens Extraction) This form is designed to ensure that you have all the information you need to make a decision about whether or not you wish to undergo
ARTICLE Long-term stability of the posterior cornea after laser in situ keratomileusis Joseph B. Ciolino, MD, Stephen S. Khachikian, MD, Michael J. Cortese, OD, Michael W. Belin, MD PURPOSE: To study long-term
Hitting Refractive Targets The Lens Approach Mark Packer, MD, FACS Clinical Associate Professor Department of Ophthalmology Oregon Health & Science University Drs. Fine, Hoffman & Packer, LLC Eugene, Oregon
Insert to October 2013 THE COMPLETE PICTURE Experiences with the ALADDIN system Sponsored by Topcon THE COMPLETE PICTURE Advantages of The Aladdin How this all-in-one optical biometer edged out the IOLMaster
Monte Carlo Simulation of AcrySof Toric Results Warren Hill, MD, FACS 1*, Richard Potvin, MASc, OD 2* 1 East Valley Ophthalmology, 5620 East Broadway Road, Mesa, Arizona, USA 85206 2 Alcon Laboratories,
To toric or not to toric? A cautionary tale Amanda Maltry, MD; Anna Kitzmann, MD; Christine Sindt, OD 1/10/2013 Chief complaint: Decreased vision after cataract surgery History of present illness: 65 year
Original Article Comparing the Rate of Regression after Conductive Keratoplasty with or without Prior Laser-Assisted in situ Keratomileusis or Photorefractive Keratectomy Majid Moshirfar, Erik Anderson
ESOIRS 2013 Moderator: Alaa Ghaith, MD Faculty: Ahmed El Masri, MD Mohamed Shafik, MD Mohamed El Kateb, MD 1 A systematic approach to the management of Keratoconus through the presentation of different
When To Laser, When To Implant, When To Do Both Scott MacRae, MD Professor of Ophthalmology Professor of Visual Sciences StrongVision Refractive Surgery Center University of Rochester Eye Institute Refractive
We've developed this handbook to help our patients become better informed about the entire process of laser vision correction. We hope you find it helpful and informative. Dr. Booth received his medical
Shinagawa LASIK Center OCT-guided Femtosecond Laser for LASIK and Presbyopia Treatment Minoru Tomita, MD, Ph.D 1) Executive Medical Director at Shinagawa LASIK Center, Tokyo, Japan 2) Clinical Professor
Phacoemulsification: Considerations for Astigmatism Management Jason P. Brinton, MD and Thomas A. Oetting, MS, MD June 10, 2011 Introduction Patient expectations for cataract surgery have increased significantly
Management of Astigmatism in Cataract Surgery Jonathan B. Rubenstein, M Vice- Chairman and eutsch Family Professor of Ophthalmology irector of Refractive Surgery Rush University Medical Center Financial
Retreatment by Lifting the Original Laser in Situ Keratomileusis Flap after Eleven Years Hassan Hashemi, MD 1,2 Mehrdad Mohammadpour, MD 3 Abstract Purpose: To describe a case of successful laser in situ
CASE REPORT Bilateral loss in the quality of vision associated with anterior corneal protrusion after hyperopic LASIK followed by intrastromal femtolaser-assisted incisions Alain Saad, MD, Alice Grise-Dulac,
Aus dem Department für Augenheilkunde Universitäts-Augenklinik Tübingen Sektion Experimentelle Ophthalmochirurgie Leiter: Professor Dr. B. Jean A Comparative Analysis of Methods for Calculating IOL Power:
Diagnostic Technologies for Enhanced Cataract Surgery Outcomes Surgeons discuss their preferred tools for achieving superior results. BY WARREN E. HILL, MD; CARLO LACKERBAUER, MD; THOMAS OLSEN, MD, PHD;
Technical Data Onse veliquat praestie dipisci liquat eu feugait at non utatum do exerit lummodigna IOLMaster 500 Measurement range Axial length 14 38 mm Corneal radii 5 10 mm Anterior chamber depth 1.5
Cataract Surgery after Myopic Refractive Procedures Ray Guard Eye Center Huang Wei-Jen, MD Financial Disclosures : * No financial interest on products mentioned Cataract Surgery after Myopic Refractive
Drs. Fine, Hoffman and Packer, LLC PHYSICIANS AND SURGEONS, EyeMDs OPHTHALMOLOGY I. Howard Fine, M.D. Richard S. Hoffman, M.D. Mark Packer, M.D. 1550 Oak Street, Suite 5 www.finemd.com Eugene, OR 97401-7701
Archived at the Flinders Academic Commons http://dspace.flinders.edu.au/dspace/ This is the author s final corrected draft of this article. It has undergone peer review. Citation for the publisher s version:
Keratorefractive Surgery for Post-Cataract Refractive Surprise Moataz El Sawy Departmentof Ophthalmology, Faculty of Medicine,MenoufiyaUniversity, Egypt email@example.com Abstract: Purpose: To evaluate
Preliminary U.S. FDA Outcomes of SMILE D. REX HAMILTON, MD, MS, FACS HEALTH SCIENCES CLINICAL PROFESSOR OF OPHTHALMOLOGY MEDICAL DIRECTOR, UCLA LASER REFRACTIVE CENTER STEIN EYE INSTITUTE, GEFFEN SCHOOL
Ectasia Risk Factors: Lessons Learned Lewis R Groden, MD Execu-ve Medical Director, LasikPlus Vision Centers/LCA Vision Associate Professor University of South Florida, Department of Ophthalmology Tampa,