1 Lesson 02: Sound Wave Production This lesson contains 26 slides plus 11 multiple-choice questions. This lesson was derived from pages 2 through 7 in the textbook:
2 ULTRASOUND imaging: a diagnostic ultrasound classification in which graphic displays of the patient s internal structures are produced Ultrasound scanners provide images of soft tissue for abdominal, obstetric, gynecologic, ophthalmic, musculoskeletal, cardiac, and vascular evaluations. They are frequently used alone or to complement other diagnostic imaging modalities. Ultrasound has many advantages over other diagnostic imaging modalities. Ultrasound is generally noninvasive, and unlike radiographic and radionuclide diagnostic imaging techniques, it utilizes no radiation.
3 Sound Wave Production
5 CATEGORIES OF SOUND INFRASOUND = below 20 Hz AUDIBLE SOUND = 20 Hz to 20 khz ULTRASOUND = above 20 khz frequency: the number of vibrations per second of a waveform of energy cycle: a complete variation of an acoustic variable, containing one condensation and one rarefaction hertz (Hz): one cycle per second Sound waves are mechanical vibratory disturbances resulting from molecular condensations (compressions) and rarefactions (decompressions). The frequency of the sound, or number of cycles per second, determines its category. The basic unit of frequency is the hertz. Sound categories include infrasound (frequencies below 20 Hz), audible sound (frequencies between 20 Hz and 20,000 Hz), and ultrasound (frequencies above 20,000 Hz).
6 MEDICAL DIAGNOSTIC ULTRASOUND ABOVE 1 MHz megahertz (MHz): one million hertz matter: anything that takes up space Ultrasound systems that are used for medical diagnostic applications operate at frequencies above one megahertz (1 MHz or 1,000,000 Hz). Sound waves propagate through matter by causing molecules to vibrate successively along the sound path. Sound waves carry energy, not matter, from one point to another.
7 SOUND VELOCITY STIFFNESS (velocity increases with stiffness) DENSITY density: density: the amount of mass in a substance for a given volume (velocity decreases with density) Stiffness and density properties of a material determine the velocity (speed of sound propagation), which in a patient typically averages 1540 meters per second. Although an increase in density results in a slight decrease in the velocity, the velocity of sound is significantly higher in materials that are very stiff. Examples of stiff materials that could significantly affect sound velocity in a patient are bones, gallstones, and hard plaque. The velocity of sound is not affected by frequency.
8 SOUND VELOCITIES Material Meters per second Air 330 Pure Water 1430 Fat 1450 Soft Tissue 1540 Muscle 1585 Bone 4080
9 SOUND VELOCITIES STIFFNESS OF MEDIUM DENSITY OF MEDIUM SOUND VELOCITY Increase Increase Decrease Decrease Increase Decrease Decrease Increase
10 PIEZOELECTRIC EFFECT TRANSMIT electrical energy to mechanical energy RECEIVE mechanical energy to electrical energy piezoelectric: refers to the conversion of electrical energy (voltage) into mechanical (sound) energy or the conversion of mechanical energy into electrical energy Ultrasound is produced by a transducer, which is a device that converts one form of energy into another. The transducer s main component is a vibrating piezoelectric element, or crystal, which can be energized by electrical energy to produce mechanical ultrasound energy, or acoustic pressure. The electrical energy typically applied to the element is in the range of 100 volts. Pressure is the amount of force per unit area and is often measured in Pascals.
11 RESONANT FREQUENCY The fundamental frequency of a transducer resonant: the ability of an object to either operate at or generate energy of a specific frequency only continuous mode: transducer excitation by repeating cycles of alternating driving voltage Transducers may be energized to produce ultrasound pulses within the range of the element s nominal, or resonant frequency. Some transducers may be operated to produce ultrasound pulses equal to one of two or three selected frequencies (multi-hertz) within the range of the resonant frequency of a broadband transducer. The continuous mode results in a non-pulsed wave with cycles that repeat indefinitely at the transducer s resonant frequency. The resonant frequency of a transducer is mainly related to the piezoelectric element s thickness. The thinner the element, the higher the resonant frequency of the transducer.
12 RESONANT FREQUENCY PIEZOELECTRIC ELEMENT THICKNESS RESONANT FREQUENCY Increase Decrease Decrease Increase
13 PIEZOELECTRIC CERAMICS lead zirconate titanate barium titanate lead metaniobate lead titanate Many piezoelectric elements are ceramic. Typical ceramic materials are lead zirconate titanate, barium titanate, lead metaniobate, and lead titanate. Lead zirconate titanate is the most commonly used material. The ceramic material in a transducer has piezoelectric characteristics because it was polarized while at a temperature above its Curie point. If a transducer is sterilized by heating it above the Curie point, depolarization may result.
14 PIEZOELECTRIC EFFECT Electrical energy Mechanical energy array: a transducer assembly containing several piezoelectric elements A transducer may transmit sound from a single piezoelectric element or an array.
15 PIEZOELECTRIC EFFECT Electrical energy Mechanical energy Another function of the piezoelectric element is the reception of echoes that return from the object being studied. The piezoelectric element, upon receiving a returning echo, converts the acoustic pressure into an electrical signal.
16 PIEZOELECTRIC EFFECT Electrical energy Mechanical energy Although sound may be transmitted and received using a single piezoelectric element, both functions cannot be performed simultaneously. Pulsed transducers utilize the same piezoelectric element(s) during separate transmit and receive intervals
17 Huygens Principle Sound waves produced by imaging transducers originate as numerous points on the surface of a piezoelectric element and serve as sources of small individual sound wavelets. Acoustic interference occurs when two or more wavelets are produced at the same time. Their amplitudes at any point in space may be added together to determine their combined effect. If the amplitude of the resultant wave is larger than either of the original waves, the waves are said to interfere constructively. If the resultant amplitude is smaller than one of the individual waves, the waves are interfering destructively. With constructive interference, the individual waves become tangent to each other and have the same phase relationship. This normally results in a sound beam with most of the useful energy present along the primary path of propagation. Constructive interference may be present off-axis of the primary beam path, resulting in side lobes.
18 Huygens Principle Side lobes often contribute to image artifacts. The multi-element structure of transducer arrays also contribute to side lobes (often called grating lobes). Apodization is a technique that applies variable-amplitude voltages across individual array elements. This changes the relative power from the individual elements to reduce the intensity of grating lobes. Subdicing is an additional technique used to reduce or eliminate grating lobes. This is accomplished by dividing each element of an array into many smaller subelements with each subelement group wired together to operate electrically as a single element.
19 LONGITUDINAL WAVE PROPAGATION Sound waves in a material produce particle motion that is back and forth along the direction of travel. This back and forth motion, termed longitudinal wave propagation, may be represented graphically. The positive peaks on the graph represent the condensations (where there is a high molecular concentration). The negative peaks on the graph represent the rarefactions (where there is a low molecular concentration).
20 WAVE PARAMETERS The period (P) of a sound wave is the time of one cycle. The wavelength (l) is the distance between two successive points in a pulse that are in the same state of compression. It is the actual distance traveled during a complete cycle. The amplitude (A) is an indication of the relative intensity (loudness) of the sound. The higher the frequency, the shorter the period and wavelength. The pulse duration is the period multiplied by the number of cycles in the pulse. The spatial pulse length is the wavelength multiplied by the number of cycles in the pulse.
21 WAVE PARAMETERS The number of cycles in a pulse is determined by the amount of damping that is present in a transducer. Damping is due to the presence of the transducer s backing material. Increased damping causes a reduction in the number of cycles. Damping also causes an increase in the transducer s bandwidth (BW), which is the range of different frequencies contained in an ultrasound pulse.
22 WAVE PARAMETERS AND EXAMPLES Period = 1 Frequency Wavelength = Velocity Frequency Pulse Duration = Period x Number of Cycles Spatial Pulse Length = Wavelength x Number of Cycles DAMPING FREQUENCY PERIOD WAVELENGTH NUMBER OF CYCLES PULSE DURATION SPATIAL PULSE LENGTH Increase Decrease Decrease Decrease Decrease Decrease Increase Increase Increase Increase Increase Decrease Decrease Decrease Decrease Increase Increase Increase The number of cycles in a pulse is not the same as the frequency of the sound, which is the number of cycles per unit time that a transducer, which is operating continuously, is designed to produce.
23 SAME DAMPING & AMPLITUDE DIFFERENT FREQUENCY & PHASE Frequency = 5.0 MHz Number of Cycles = 3 Period = 0.2 µs Pulse Duration = 0.6 µs Wavelength = mm Spatial Pulse Length = mm Frequency = 2.5 MHz Number of Cycles = 3 Period = 0.4 µs Pulse Duration = 1.2 µs Wavelength = mm Spatial Pulse Length = mm 3-cycle pulse shorter periods shorter wavelengths shorter pulse duration shorter spatial pulse length 3-cycle pulse longer periods longer wavelengths longer pulse duration longer spatial pulse length
24 SAME FREQUENCY, AMPLITUDE, & PHASE DIFFERENT DAMPING Frequency = 5.0 MHz Number of Cycles = 3 Period = 0.2 µs Pulse Duration = 0.6 µs Wavelength = mm Spatial Pulse Length = mm Frequency = 5.0 MHz Number of Cycles = 4 Period = 0.2 µs Pulse Duration = 0.8 µs Wavelength = mm, Spatial Pulse Length = mm 3-cycle pulse same periods same wavelengths shorter pulse duration shorter spatial pulse length 4-cycle pulse same periods same wavelengths longer pulse duration longer spatial pulse length
25 SAME FREQUENCY & DAMPING DIFFERENT AMPLITUDE & PHASE Frequency = 5.0 MHz Number of Cycles = 4 Period = 0.2 µs Pulse Duration = 0.8 µs Wavelength = mm Spatial Pulse Length = mm Frequency = 5.0 MHz Number of Cycles = 4 Period = 0.2 µs Pulse Duration = 0.8 µs Wavelength = mm Spatial Pulse Length = mm 4-cycle pulse same periods same wavelengths same pulse duration same spatial pulse length 4-cycle pulse same periods same wavelengths same pulse duration same spatial pulse length
26 DAMPING vs. BANDWIDTH DAMPING Increase Decrease BANDWIDTH Increase Decrease Damping also causes an increase in the transducer s bandwidth (BW), which is the range of different frequencies contained in an ultrasound pulse.
27 HIGH DAMPING vs. NO DAMPING SAME FREQUENCY. DIFFERENT DAMPING Center frequency = 5.0 MHz Range = 3.75 MHz to 6.25 MHz Number of Cycles = 2 Bandwidth = 2.5 MHz Center frequency = 5.0 MHz Range = 4.9 MHz to 5.1 MHz Continuous Wave Bandwidth = 0.2 MHz Pulse-echo Damped Wide Bandwidth CW Not damped Narrow Bandwidth The continuous wave results in a non-pulsed wave with cycles that repeat indefinitely at the transducer s resonant frequency.
28 Answers to the following ELEVEN practice questions were derived from material in the textbook:
29 Question 1 Ultrasound waves that are traveling through a medium consist of electromagnetic and radio frequencies compressions and refractions condensations and refractions electromagnetic and ionizing frequencies compressions and rarefactions Pages 2 and 5
30 Question 2 What is the difference between audible sound and ultrasound? Ultrasound waves are always longitudinal Audible sound waves are always longitudinal Audible sound has a higher frequency Ultrasound has a higher frequency Ultrasound waves are ionizing Page 2
31 Question 3 A piezoelectric element produces a voltage when postprocessing is used preprocessing is used an acoustic pressure is present on its surface the receiver gain is increased the attenuation increases Page 3
32 Question 4 A decrease in the thickness of a piezoelectric element will result in a greater pulse duration an increase in the propagation speed a decrease in the quality factor if the bandwidth decreases an increase in the frequency of the transducer a higher duty factor Page 3
33 Question 5 The resonant frequency of an ultrasound transducer is dependent on damping the backing material the thickness of the piezoelectric element the amplitude of the voltage applied to the transducer the Curie temperature Page 3
34 Question 6 What does A, B, and D represent on the graph? amplitude, period, velocity pulse duration, duty factor, amplitude wavelength, duty factor, pulse duration period, wavelength, velocity wavelength, spatial pulse length, pulse repetition period Pages 5 and 6
35 Question 7 If the frequency is doubled, the duty factor will increase the period will double the lateral resolution will be poorer the wavelength will double the wavelength will be one-half Pages 5 and 6
36 Question 8 The speed of ultrasound in soft tissue is closest to 1.54 mm/sec 1540 km/sec 1.54 km/msec 1540 mm/sec 1540 m/s Page 2
37 Question 9 Ultrasound energy is traveling through the same medium. If the frequency is doubled, the propagation speed is quartered quadrupled doubled halved unchanged Pages 2 and 3
38 Question 10 The propagation speed is highest in bone tissue fat water muscle Pages 2 and 3
39 Question 11 A single pulse of ultrasound from a transducer contains a range of frequencies contains sound at the nominal frequency of the transducer only contains sound at the resonant frequency of the transducer only has a narrow bandwidth contains continuous waves Page 7
LECTURE 6 TOPIC: BIOPHYSICAL BASICS OF ULTRASOUND IMAGING TIME: 2 HOURS Ultrasound is a term given to inaudible, high frequency sound waves and is also the generic name given to the imaging modality that
Parameters used to completely characterize a sound wave Describing Sound Waves Chapter 3 Period Frequency Amplitude Power Intensity Speed Wave Length Period Defined as the time it take one wave vibrate
Diagnostik Ultrasound Basic physics, image reconstruction and signal processing Per Åke Olofsson Dpt of Biomedical Engineering, Malmö University Hospital, Sweden Ultrasound Real-time modality 17-WEEK FETAL
Waves and Sound Part 1 Intro Write the following questions on a blank piece of paper (don t answer yet) 1. What is the difference between a mechanical and electromagnetic wave? 2. What is the difference
Waves - Transverse and Longitudinal Waves wave may be defined as a periodic disturbance in a medium that carries energy from one point to another. ll waves require a source and a medium of propagation.
chapter 1 Ultrasound Physics Sidney K. Edelman, Ph.D. ESP Ultrasound www.esp-inc.com email@example.com s Sound creates images by sending short bursts into the body. Thus we are concerned with the interaction
chapter 2 Pulsed Ultrasound In diagnostic ultrasound imaging, short bursts, or pulses, of acoustic energy are used to create anatomic images. Continuous wave sound cannot create anatomic images. Analogy
7 COMBINED APPLICATIONS The designs described in this section. use PZT transducers both as a transmitter and a receiver, clearly demonstrating the reversible nature of the piezoelectric effect. Typical
THE PHYSICS OF ULTRASOUND - PART 1 ANAESTHESIA TUTORIAL OF THE WEEK 199 4TH OCTOBER 2010 Dr M MacGregor 1, Dr L Kelliher 1,Dr J Kirk-Bayley 2 1: Specialist Registrar, 2: Consultant, Royal Surrey County
APPLICATION NOTE AP050830 Selection and use of Ultrasonic Ceramic Transducers Pro-Wave Electronics Corp. E-mail: firstname.lastname@example.org URL: http://www.prowave.com.tw The purpose of this application note
APPLICATION NOTE ULTRASONIC CERAMIC TRANSDUCERS Selection and use of Ultrasonic Ceramic Transducers The purpose of this application note is to aid the user in the selection and application of the Ultrasonic
Resonance Wave Transfers Energy Without Transferring Matter Clip from Mechanical Universe Wave A wave can be described as a disturbance that travels through a medium from one location to another location.
Lesson 19: Mechanical Waves Mechanical Waves There are two basic ways to transmit or move energy from one place to another. First, one can move an object from one location to another via kinetic energy.
MUSCULOSKELETAL ULTRASOUND Scott J. Primack, DO Colorado Rehabilitation & Occupational Medicine Sr. Clinical Instructor: Department of Preventative Medicine and Biometrics University of Colorado of School
Principles of Imaging Science I (RAD119) Electromagnetic Radiation Energy Definition of energy Ability to do work Physicist s definition of work Work = force x distance Force acting upon object over distance
1 Candidates should be able to : ULTRASOUND Describe the properties of ultrasound. ULTRASOUND is any sound wave having a frequency greater than the upper frequency limit of human hearing (20 khz). Describe
Energy of Waves Book O Chapter 1 A wave can make a leaf bob up and down on the water, but it cannot move the leaf toward the shore. This is because waves only transfer A. Media B. Crests C. Energy D. Matter
The maximum displacement of particles of the medium from their mean positions during the propagation of a wave Angle of an incident (arriving) ray or particle to a surface; measured from a line perpendicular
Chapter 16 - solutions Constructive and Destructive Interference Conceptual Question Description: Conceptual question on whether constructive or destructive interference occurs at various points between
(Refer Slide Time: 00:01:23) Data Communications Prof. Ajit Pal Department of Computer Science & Engineering Indian Institute of Technology, Kharagpur Lecture # 03 Data and Signal Hello viewers welcome
Principles of Medical Ultrasound Pai-Chi Li Department of Electrical Engineering National Taiwan University What is Medical Ultrasound? Prevention: actions taken to avoid diseases. Diagnosis: the process
Name: Class: Date: Chapter 23 Multiple Choice Test Multiple Choice Identify the choice that best completes the statement or answers the question. 1. A unit of motion repeated over and over again is called
L 30 Electricity and Magnetism  ELECTROMAGNETIC WAVES Faraday laid the groundwork with his discovery of electromagnetic induction Maxwell added the last piece of the puzzle Heinrich Hertz made the experimental
It s shape Sound Beams foundation for a basic understanding It begins the size or diameter of the transducer gradually converges to a narrower point then begins to diverge Chapter 10 Anatomy of the beam
Name: Periodic Wave Phenomena 1. The diagram shows radar waves being emitted from a stationary police car and reflected by a moving car back to the police car. The difference in apparent frequency between
Wave Energy A pulse is a traveling disturbance in a medium. Pictures (a) through (d) show successive positions of a pulse in a rope. Wave Energy A wave is a succession of pulses, traveling through a medium.
Ultrasonic Testing (UT)  Ultrasonic Testing (UT): A nondestructive test method that uses high frequency sound energy to conduct examinations and make measurements. Sound: The mechanical vibration of
Waves Sound and Light r2 c:\files\courses\1710\spr12\wavetrans.doc Ron Robertson The Nature of Waves Waves are a type of energy transmission that results from a periodic disturbance (vibration). They are
Ultrasound Basic Idea Send waves into body which are reflected at the interfaces between tissue Return time of the waves tells us of the depth of the reflecting surface History First practical application,
5.1.1 Oscillating Systems Waves Review Checklist 5.1.2 Pulses 5.1.1A Explain the relationship between the period of a pendulum and the factors involved in building one Four pendulums are built as shown
Waves PART I Wave Properties Wave Anatomy PART II Wave Math Wave Behavior PART III Sound Waves Light Waves (aka Electromagnetic Waves or Radiation) 1 Sec. 14.1 Wave Properties Objectives Identify how waves
An Overview of the Properties of Different Piezoceramic Materials Material Families Four of the more important types of piezoceramic materials are introduced below. Lead Zirconate-Titanate - Because of
Doppler Doppler Chapter 19 A moving train with a trumpet player holding the same tone for a very long time travels from your left to your right. The tone changes relative the motion of you (receiver) and
Chapter 1 Introduction to B-mode imaging Kevin Martin The application of ultrasound to medical diagnosis has seen continuous development and growth over several decades. Early, primitive display modes,
Ultrasonic Wave Propagation Review Presented by: Sami El-Ali 1 1. Introduction Ultrasonic refers to any study or application of sound waves that are higher frequency than the human audible range. Ultrasonic
ULTRASONICS Production of ultrasonics by magnetostriction and piezoelectric methods - acoustic grating -Non Destructive Testing - pulse echo system through transmission and reflection modes - A,B and C
Waves Overview (Text p382>) Waves What are they? Imagine dropping a stone into a still pond and watching the result. A wave is a disturbance that transfers energy from one point to another in wave fronts.
Introduction Antennas and Propagation Chapter 5 An antenna is an electrical conductor or system of conductors Transmission - radiates electromagnetic energy into space Reception - collects electromagnetic
Advancements in High Frequency, High Resolution Acoustic Micro Imaging for Thin Silicon Applications Janet E. Semmens Sonoscan, Inc. 2149 E. Pratt Boulevard Elk Grove Village, IL 60007 USA Phone: (847)
MSK Ultrasound Basic Principles of MSK US Physics and Image Optimization Objectives Provide a basic introduction to: Ultrasound Physics Equipment Considerations Knobology Image Optimization Scanning considerations
Physics, Instrumentation and New Trends in Techniques of Ultrasound Medical Imaging Imaging: after energy (light, radio waves, Ultrasound and more) is interacting with a target object, an image could be
Ultrasound ULTRASOUND Miklós Kellermayer Ultrasound What is? Generation and physical properties of What is? Generation and detection of Physical properties of Effects of Primary Secondary Medical applications
chapter 1 Ultrasound Physics ASCeXAM Review- 2011 Sidney K. Edelman, Ph.D. Director, ESP Ultrasound email@example.com 281-292-9400 www.esp-inc.com Definitions Sound A type of wave that carries energy
Flow and Flow rate. Laminar and Turbulent flow Laminar flow: smooth, orderly and regular Mechanical sensors have inertia, which can integrate out small variations due to turbulence Turbulent flow: chaotic
Name: ate: 1. The diagram shown represents four waves traveling to the right in the same transmitting medium. 4. Which wave has the greatest amplitude?.... Which type of wave is represented? 5. Which characteristic
1 of 3 Acousto-optic modulator F An acousto-optic modulator (AOM), also called a Bragg cell, uses the acousto-optic effect to diffract and shift the frequency of light using sound waves (usually at radio-frequency).
(Part 3) In part one of this series of articles, level measurement using a floating system was discusses and the instruments were recommended for each application. In the second part of these articles,
Acoustics: the study of sound waves Sound is the phenomenon we experience when our ears are excited by vibrations in the gas that surrounds us. As an object vibrates, it sets the surrounding air in motion,
Sensing and Sensors: Acoustic Sensors version 1.1 MediaRobotics Lab, January 2008 Background: sound waves Sound waves are created by alternate compression and expansion of solids, liquids or gases at certain
Chapter 5 Waves I: Generalities, Superposition & Standing Waves 5.1 The Important Stuff 5.1.1 Wave Motion Wave motion occurs when the mass elements of a medium such as a taut string or the surface of a
46 CHAPTER THREE ULTRASOUND 3.0 Goals: Understanding how ultrasound works; its uses, strengths, and weaknesses Objective questions: 3.1 What is ultrasound? 3.2 How does ultrasound work? 3.3 How is an image
PLEASE DO NOT WRITE ON THE TEST. PLACE ALL MULTIPLE CHOICE ANSWERS ON THE SCANTRON. (THANK YOU FOR SAVING A TREE.) Sound Waves Test -- each multiple choice question is worth 3 points. 1. Sound waves are
Sound GCSE Physics Sound Sound is produced by vibrating objects. A sound wave consists of mechanical vibrations in air and other substances. Sound is a longitudinal wave in which the wave energy travels
Review of Chapter 25 Multiple Choice Identify the letter of the choice that best completes the statement or answers the question. 1. The time needed for a wave to make one complete cycle is its b. velocity.
PROGRESSIVE WAVES 1 Candidates should be able to : Describe and distinguish between progressive longitudinal and transverse waves. With the exception of electromagnetic waves, which do not need a material
Physics of Ultrasound Notes S Series M-Turbo NanoMaxx Physics of Ultrasound Notes Lynette Hassall, DMU AMS MLI, Clinical Applications Specialist, SonoSite, Inc. These notes are not a complete physics text,
114. If mass of a body suspended from a spring is doubled, the period of vibration of the body becomes: (6a ii 07) a) Double. b) Half. c) Times. d) Iyr times. 115. The frequency of a simple pendulum in
MRI: Magnetic resonance imaging: Like CT, MRI produces detailed cross sectional images of body regions. However the physical principles of image acquisition are different and very complicated. The following
Chapter 17: Change of Phase Conceptual Physics, 10e (Hewitt) 3) Evaporation is a cooling process and condensation is A) a warming process. B) a cooling process also. C) neither a warming nor cooling process.
Basic theory of sound, piezomaterials and vibrations Content of presentation Theory of sound Piezomaterials and the piezoelectric effect Basic principles of the Sonitron products 2 Theory of sound Sound
Superposition Principle Chapter 18 Superposition and Standing Waves If two or more traveling waves are moving through a medium, the resultant value of the wave function at any point is the algebraic sum
1/2016 Sound 1/8 INTERFERENCE OF SOUND WAVES PURPOSE: To measure the wavelength, frequency, and propagation speed of ultrasonic sound waves and to observe interference phenomena with ultrasonic sound waves.
Chapter 2: Electromagnetic Radiation Radiant Energy I Goals of Period 2 Section 2.1: To introduce electromagnetic radiation Section 2.2: To discuss the wave model of radiant energy Section 2.3: To describe
PHYS102 Previous Exam Problems CHAPTER 17 Sound Waves Sound waves Interference of sound waves Intensity & level Resonance in tubes Doppler effect If the speed of sound in air is not given in the problem,
Chapter 20 Lecture physics FOR SCIENTISTS AND ENGINEERS a strategic approach THIRD EDITION randall d. knight Chapter 20 Traveling Waves Chapter Goal: To learn the basic properties of traveling waves. Slide
Antennas & Propagation CS 6710 Spring 2010 Rajmohan Rajaraman Introduction An antenna is an electrical conductor or system of conductors o Transmission - radiates electromagnetic energy into space o Reception
1 2 The range of RF spans 3 KHz (3000 Hz) to 300 GHz (300 million Hz) Frequencies of RF devices range from the low frequency AM broadcasts (80 MHz) to higher frequency mobile phones (1900 MHz) smart meters
Wave Properties of Electromagnetic Radiation Two options are available for analytical utility when an analyte interacts with a beam of electromagnetic radiation in an instrument 1. We can monitor the changes
Product Application Guide- 22mm Industrial Alarms with LED Lights Part Number Structure ZT 028 L D CT R 15 Series: ZT = 22mm with LED Maximum Voltage: 016 = 9-16; 028 = 16-28; 120 = 48-120 Sound Level:
CHAPTER 14 WAVE & Sound COURSE CONTENT Properties of waves Definition Types of waves Wavelength Amplitude Frequency Waves are everywhere in nature Sound waves, visible light waves, radio waves, microwaves,
Module 8 AC Theory Filters & Wave Shaping Passive Filters & Wave Shaping What you'll learn in Module 8. Module 8 Introduction Recognise passive filters with reference to their response curves. High pass,
Objectives: PS-7.1 Physical Science Study Guide Unit 7 Wave properties and behaviors, electromagnetic spectrum, Doppler Effect Illustrate ways that the energy of waves is transferred by interaction with
Computer Networks and Internets, 5e Chapter 6 Information Sources and Signals Modified from the lecture slides of Lami Kaya (LKaya@ieee.org) for use CECS 474, Fall 2008. 2009 Pearson Education Inc., Upper
Rock Bolt Condition Monitoring Using Ultrasonic Guided Waves Bennie Buys Department of Mechanical and Aeronautical Engineering University of Pretoria Introduction Rock Bolts and their associated problems
Data Transmission Concepts and terminology Transmission terminology Transmission from transmitter to receiver goes over some transmission medium using electromagnetic waves Guided media. Waves are guided
Ultrasonic Testing Ultrasonic Testing (UT) uses high frequency sound waves (typically in the range between 0.5 and 15 MHz) to conduct examinations and make measurements. Besides its wide use in engineering
WEBER ULTRASONICS Page 1 WEBER ULTRASONICS WEBER ULTRASONICS Page 2 ULTRASONIC TECHNOLOGY BASICS» Sound is a pressure wave which propagates in elastic media such as gases, fluids and rigid bodies. p Sound
DESIGN AND EVALUATION OF PROBE WITH THREE DEGREE- OF-FREEDOM FOR NON-DESTRUCTIVE TEST USING THREE- DIMENSIONAL FINITE ELEMENT METHOD Masafumi Aoyanagi Graduate School of Systems and Information Engineering,
Sound W.S. 1. Two major classes of waves are longitudinal and transverse. Sound waves are. longitudinal transverse 2. The frequency of a sound signal refers to how frequently the vibrations occur. A high-frequency