1 Performance of the PAXgene Blood DNA Tube for the Collection, Transport, and Storage of Whole Blood and the Purification of DNA Using the QIAsymphony Instrument L. Rainen 1, K. Guenther 2, E. Provencher 1, R. Wyrich 2 1 PreAnalytiX GmbH, BD Preanalytical Systems, Franklin Lakes, NJ, USA 2 PreAnalytiX GmbH, QIAGEN, Hilden, Germany ABSTRACT PreAnalytiX has developed a blood collection tube, the PAXgene Blood DNA Tube, for the collection of whole blood and the stabilization of genomic DNA. DNA is stable in whole blood stored in the PAXgene Blood DNA Tube for 28 days at 2-8 C, for 14 days at C, for 3 days at 35 C, after 3 freeze/thaw cycles, or for a minimum of 6 months at - C. Blood can be frozen and thawed up to three times without appreciable loss of DNA concentration or quality. When tested in a molecular diagnostic assay, results from all DNA samples purified from whole blood collected and stored in the PAXgene Blood DNA Tube were concordant to results obtained from paired EDTA tube controls regardless of handling or storage condition. We have demonstrated that the PAXgene Blood DNA Tube can be used to collect and store whole blood specimens for the extraction of high quality DNA for use in molecular diagnostic test methods. INTRODUCTION Blood collection tubes for molecular diagnostic test methods can vary with respect to anticoagulant, blood draw volume, and material composition (either glass or plastic). None of these tubes are cleared for general molecular diagnostic use for assays that require purified genomic DNA. Recently, we have developed the PAXgene Blood DNA Tube, a plastic, evacuated tube containing a proprietary ethylenediaminetetraacetic acid (EDTA) formulation which anticoagulates whole blood and preserves genomic DNA. The tube is designed to collect a nominal volume of 2.5 ml of blood upon venipuncture, and the blood can be stored in situ until it is needed for DNA extraction. In this paper, we show the results of studies which were designed to demonstrate that the PAXgene Blood DNA Tube is compatible with an automated DNA extraction instrument and that it delivers DNA high quality DNA for use in molecular diagnostic test methods. Moreover, we show that the PAXgene Blood DNA Tube preserves genomic DNA for a minimum of 28 days at refrigerated temperature, 14 days at room temperature, 3 days at elevated temperature, after 3 freeze/thaw cycles, or at least 6 months frozen (study ongoing). MATERIALS AND METHODS Subject Population All blood specimens were collected from apparently healthy, consented adult subjects.
2 Blood Collection and Specimen Processing Blood was collected into PAXgene Blood DNA Tubes (Cat. no , PreAnalytiX, Hombrechtikon, CH) and BD Vacutainer Plus K 2 EDTA (EDTA) tubes (Cat. no , BD, Franklin Lakes, NJ, USA) using standard phlebotomy technique. Paired control (EDTA) and evaluation (PAXgene Blood DNA) tubes were collected from each subject and the evaluation tubes were stored at either room temperature (18-25 C), 2-8 C, 35 C, or frozen at - C. At designated time points, specimen aliquots were processed with the QIAsymphony SP instrument (QIAGEN, Hombrechtikon, CH). Control tubes were processed on the day of collection without storage. For Study 1 (See Figure 1), blood specimens were collected, and specimen aliquots ( µl) were processed with the QIAGEN QIAsymphony DSP DNA Mini Kit (QIAGEN, Hilden, DE; Cat. No ). The resultant purified DNA was stored at - C prior to testing. For Study 2 (See Table 1), blood specimens were collected and either processed on the day of blood collection (control and T evaluation tubes) with the QIAsymphony DSP DNA Mini Kit or stored at the indicated temperatures and processed using the same method at the designated time points (evaluation tube only). DNA Concentration and Purity The concentration of DNA in QIAsymphony eluates was determined by measuring the absorbances at 26 nm (A 26 ) and 3 nm (A 3 ) in a spectrophotometer and calculating the DNA concentration using the formula: µg of DNA/ml = [(A 26 ) - (A 3 )] 5 µg DNA/unit of absorbance. DNA purity was determined by calculating the ratio of corrected absorbance at 26 nm to the corrected absorbance at 28 nm using the formula: [(A 26 ) - (A 3 )] / [(A 28 ) - (A 3 )]. This is the corrected A 26 /A 28 ratio. Pure DNA has a corrected A 26 /A 28 ratio of White Blood Cell Counts The number of white blood cells (WBCs) was counted from blood aliquots of EDTA tubes using a Beckman Coulter COULTER Ac T diff Analyzer. Molecular Diagnostic Test Purified DNA aliquots from each specimen were tested in the Immucor (formerly Hologic Gen-Probe) LIFECODES HLA EZ Type (DRDQ Low Resolution) assay according to manufacturer s instructions. Data Analysis Median DNA eluate concentrations and the 25-75% interquartile range for DNA eluate concentrations were used as robust estimators of DNA quantity. All A 26 /A 28 ratios of purified DNA samples were rounded to one decimal point. An analysis of variance (ANOVA) was performed on the WBC vs. DNA eluate concentration to determine the relationship between these two values.
3 Study Design Study 1: Quality, quantity, and suitability in molecular testing of DNA from whole blood collected in the PAXgene Blood DNA Tube and extracted with the QIAsymphony instrument. To demonstrate that DNA isolated from blood collected in the PAXgene Blood DNA Tube was of sufficient quality and quantity for use in molecular diagnostic test methods, venous whole blood was collected according to manufacturer s instructions from adult subjects into one PAXgene Blood DNA Tube (evaluation tube) and one BD Vacutainer Plus K 2 EDTA (control tube) tube. In total, 46 paired samples were collected. Within two days of phlebotomy, DNA was extracted from all specimens ( µl) using the QIAsymphony instrument and the QIAsymphony DSP DNA Mini Kit. Purified DNA samples were assessed for purity and concentration as described in the Materials and Methods section and tested in the Immucor LIFECODES HLA EZ Type (DRDQ Low Resolution) assay according to the manufacturer s instructions. For study design, see Figure 1. Adult subjects N = 46 PAXgene Blood DNA Tube BD Vacutainer K 2 EDTA Tube DNA extraction with QIAsymphony instrument DNA eluates Purity and Concentration Immucor LIFECODES HLA EZ Type DRDQ assay Figure 1: Study design of Study 1, Quality, quantity, and suitability in molecular testing of DNA from whole blood collected in the PAXgene Blood DNA Tube and extracted with the QIAsymphony instrument.
4 Study 2: Stability of DNA in whole blood stored in the PAXgene Blood DNA Tube. For each storage or handling condition, a separate study was conducted to determine the stability of DNA in whole blood collected in the PAXgene Blood DNA Tube (See Table 1). For each study, venous whole blood from adult subjects was collected according to manufacturer s instructions into two EDTA tubes (T control) and one PAXgene Blood DNA Tube for each time point or handling condition (freeze/thaw study). The number of tubes collected per subject for each study was as follows: 6 tubes for room temperature (18-25 C) storage, 7 tubes for refrigerated temperature (2-8 C) storage, 7 tubes for elevated temperature (35 C) storage, 15 tubes for frozen (- C) storage. DNA from initial time point (T ) tubes was purified from whole blood using the QIAsymphony instrument on the day of collection. The quality and quantity of DNA extracted at each time point was assessed by spectrophotometry as described in the Materials and Methods section, and all purified DNA samples were tested in the Immucor LIFECODES HLA EZ Type (DRDQ Low Resolution) assay according to the manufacturer s instructions. In addition to time/temperature handling studies, the performance of DNA from whole blood collected in PAXgene Blood DNA Tubes was determined when the specimen was subjected to multiple freeze/thaw cycles. For each subject, venous whole blood was collected according to manufacturer s instructions into two control tubes (T control) and four PAXgene Blood DNA tubes for a total of six tubes per subject. DNA was extracted with the QIAsymphony instrument from the two EDTA tubes and one PAXgene Blood DNA Tube on the day of collection, and the remaining three PAXgene Blood DNA Tubes were frozen at - C. After 24 hours, the three PAXgene tubes were thawed, DNA was extracted from one tube and the remaining two PAXgene tubes were frozen again at - C. After 24 hours, the two tubes were thawed, DNA was extracted from one tube, and the remaining tube was refrozen. The third tube was thawed after 24 hours and the DNA extracted. Number of Subjects Storage Temperature Time Points/(Cycles) Tubes per Subject N = 12 Room Temperature (18-25 C) T, 3 d, 7 d, 14 d 6 N = 12 Refrigerated Temperature (2-8 C) T, 7 d, 14 d, 21 d, 28 d 7 N = 12 Elevated Temperature (35 C) T, 6 h, 1 d, 2 d, 3 d 7 N = 12 Frozen (- C) T, 1 m, 6 m, 12 m y 15 N = 12 Freeze/Thaw T, (1x, 2x, 3x) 6 Table 1: Study design of Study 2, Stability of DNA in whole blood collected in the PAXgene Blood DNA Tube. All DNA isolated from samples used in this study were measured for DNA eluate concentration and purity as described in the Materials and Methods section and tested for performance in the Immucor LIFECODES HLA EZ Type (DRDQ Low Resolution) assay using an EDTA tube as control.
5 Frequency RESULTS Study 1: Quality, quantity, and suitability in molecular testing of DNA from whole blood collected in the PAXgene Blood DNA Tube and extracted with the QIAsymphony instrument. A histogram of eluate concentrations obtained from 46 whole blood specimens collected and stored in PAXgene Blood DNA tubes is shown in Figure 2. DNA eluate concentrations ranged from 12.2 to 54. ng/µl with 75% of samples 24.8 ng/µl. 6 5 N = DNA Concentration (ng DNA / μl eluate) Figure 2: DNA concentration (ng DNA / µl eluate) in eluates of whole blood collected in PAXgene Blood DNA Tubes and purified using the QIAsymphony instrument. Median DNA concentrations, the 25-75% interquartile range, and the range of corrected A 26 /A 28 ratios of purified DNA samples are shown in Table 2. Measurement ng DNA / µl eluate (A 26 /A 28 ratio) Median DNA eluate concentration % interquartile range Range DNA Purity ( ) Table 2: DNA eluate concentrations, median, range, 25-75% interquartile range, and DNA purity from whole blood specimens collected in the PAXgene Blood DNA Tube and purified with the QIAsymphony instrument.
6 DNA Concentration (ng DNA / μl eluate) DNA was successfully isolated from blood specimens with WBC ranging from 3.9 to cells/µl with no reported extraction failures. The correlation between WBC and DNA eluate concentration is depicted in Figure 3. An analysis of variance (ANOVA) of these results showed that the relationship between WBC and DNA concentration is statistically significant at an α-level of.5 with a p-value of.. 6 N = 46 5 y = 4.1x R² = WBC 6 cells/µl Figure 3: WBC ( 6 cells/µl) vs. DNA concentration (ng DNA / µl eluate) in eluates of whole blood collected in PAXgene Blood DNA Tubes and purified using the QIAsymphony instrument. All purified DNA samples from blood collected in the PAXgene Blood DNA Tube were tested in the Immucor LIFECODES HLA EZ Type (DRDQ Low Resolution) assay and compared to results obtained from paired EDTA control tubes. Representative results of testing of extracted DNA in the EZ Type DRDQ HLA assay are shown in Figure 4. Individual patterns of HLA-SSP PCR amplicons which were separated by size on agarose gels were identical between DNA from specimens collected in the PAXgene Blood DNA Tubes and specimens collected in corresponding control EDTA tubes. HLA typing results (n = 1) of blood collected in the PAXgene Blood DNA Tube were % concordant with results obtained with blood collected in the paired EDTA control tube.
7 EDTA Tube PAXgene Blood DNA Tube T T 3d T 7d T 14d M Figure 4: Electrophoretic gel results of the EZ Type DRDQ HLA assay showing HLA types from purified DNA from blood collected in paired EDTA and PAXgene Blood DNA Tubes. DNA was isolated from the control EDTA tube within two hours of phlebotomy (T ) and from PAXgene Blood DNA Tubes at 3 days (T 3d ), 7 days (T 7d ), and 14 days (T 14d ) after phlebotomy. (M = DNA molecular weight marker with fragment sizes of 1,5, 8,,, and 5 bp) Study 2: Quality, quantity and suitability for molecular diagnostic testing of blood stored in the PAXgene Blood DNA Tube. Results of the stability of DNA in whole blood stored at 2-8 C for 28 days, room temperature for 14 days, 35 C for 3 days, after 3 freeze/thaw cycles, or at - C for 6 months are shown in Figures 5, 6, 7, 8, and 9 respectively. DNA concentrations in the eluates are expressed in ng/µl. In comparison to replicate tubes processed immediately after collection (T ), the median DNA concentration of the purified DNA eluates from blood stored at refrigerated (2-8 C) temperatures for 28 days was 28.5 ng/µl, or approximately 98% (Figure 5) while the median DNA eluate concentration in blood stored at room temperature for 14 days was 29.6 ng/µl, or % of the T eluate concentration (Figure 6). Median DNA eluate concentrations for blood stored for 3 days at 35 C was 26.2 ng/µl or 84% of the median T concentration (Figure 7). After three freeze/thaw cycles, the median DNA eluate concentration was 91% of that of DNA extracted from unfrozen whole blood (Figure 8). For all time/temperature or freeze/thaw storage conditions tested, DNA purified from all specimens remained of high quality with corrected A 26 /A 28 ratios between 1.7 and 1.9 (data not shown).
8 DNA Concentration (ng DNA / μl eluate) DNA Concentration (ng DNA / μl eluate) Blood stored in the PAXgene Blood DNA Tube at - C was tested for DNA eluate concentration and purity. Median DNA eluate concentrations for blood stored for 6 months at - C was 24.3 ng/µl or 94% of the median T concentration (Figure 9). Stability testing of DNA in blood frozen in the PAXgene Blood DNA tube is ongoing. For the most recent test results, see Storage (days) Figure 5: Median and 25-75% interquartile ranges for DNA eluate concentrations extracted from blood stored in PAXgene Blood DNA Tubes for 28 days at 2-8 C Storage (days) Figure 6: Median and 25-75% interquartile ranges for DNA eluate concentrations extracted from blood stored in PAXgene Blood DNA Tubes for 14 days at C.
9 DNA Concentration (ng DNA / μl eluate) DNA Concentration (ng DNA / μl eluate) Storage (days) Figure 7: Median and 25-75% interquartile ranges for DNA eluate concentrations extracted from blood stored in PAXgene Blood DNA Tubes for 3 days at 35 C Freeze/Thaw Cycles Figure 8: Median and 25-75% interquartile ranges for DNA eluate concentrations extracted from blood stored in PAXgene Blood DNA Tubes subjected to three freeze/thaw cycles.
10 DNA Concentration (ng DNA / μl eluate) Storage (months) Figure 9: Median and 25-75% interquartile ranges for DNA eluate concentrations extracted from blood stored in PAXgene Blood DNA Tubes for 6 months at - C. DISCUSSION We have developed a new blood collection device, the PAXgene Blood DNA Tube, to collect, anticoagulate, stabilize, transport, and store a venous whole blood sample for preparation of high quality DNA for use with molecular diagnostic test methods that require DNA. We have determined that after blood is collected, the DNA remains stable for 28 days at 2-8 C (Figure 5), 14 days at C (Figure 6), 3 days at 35 C (Figure 7), after 3 freeze/thaw cycles (Figure 8) or for a minimum of 6 months at - C (Figure 9). Whole blood specimens can be frozen and thawed up to three times without appreciable loss of DNA eluate concentration or quality. All DNA concentrations in all eluates obtained from blood specimens, regardless of handling or storage conditions were 12.2 ng/μl and DNA purity as measured by the A 26 /A 28 ratio was between 1.7 and 1.9. Furthermore, all purified DNA samples obtained from blood collected and stored in the PAXgene Blood DNA Tube returned results which were concordant with those of paired EDTA controls when tested in a molecular diagnostic assay. We further demonstrated that the PAXgene Blood DNA Tube is compatible with magnetic bead (QIAsymphony) DNA isolation methods, as well as silica membrane (QIAcube and QIAamp ) and salting out/precipitation methods (data not shown). Furthermore, blood collected in the PAXgene Blood DNA tube can be stored frozen in situ for at least 6 months with no significant loss of quality or quantity of DNA.
11 CONCLUSION PreAnalytiX has developed the PAXgene Blood DNA Tube specifically for in vitro diagnostic (IVD) DNA testing. We have validated the performance of this tube by evaluating the DNA yield, purity, and performance in a molecular diagnostic assay for multiple transport and storage conditions. Additionally, we have shown that the tube is compatible with various DNA isolation methods including the QIAsymphony instrument. Storing and processing whole blood within the conditions tested in these studies assures high quality DNA suitable for molecular diagnostic test methods that require DNA. The PAXgene Blood DNA Tube is currently not available in the U.S. Trademarks: PAXgene, PreAnalytiX (PreAnalytiX GmbH); QIAamp, QIAcube, QIAsymphony, QIAGEN (QIAGEN Group); BD Vacutainer, BD (Becton, Dickinson and Company); EZ Type, LIFECODES, Immucor (Immucor, Inc.); Ac T diff, COULTER, Beckman Coulter (Beckman Coulter, Inc.); Gen-Probe (Gen-Probe Incorporated), Hologic (Hologic, Inc.). PreAnalytiX GmbH, 8634 Hombrechtikon, CH 13 PreAnalytiX, all rights reserved. 9/13
AxyPrep Blood Genomic DNA Miniprep Kit For the purification of genomic DNA from whole blood Kit contents, storage and stability Cat. No. AP-MN-BL-GDNA-50 AP-MN-BL-GDNA-250 Kit size 50 preps 250 preps Miniprep
Introduction RNAgard Blood System (RUO) Application Note Compared to many commonly assayed cellular biomolecules, RNA is particularly susceptible to degradation, primarily due to the ubiquitous presence
MagNA Pure 96 System Nucleic Acid Purification with Real High Throughput The MagNA Pure 96 System purifies DNA, RNA, and viral nucleic acids from a wide range of starting materials using magnetic glass
RealStar HBV PCR Kit 1.0 11/2012 RealStar HBV PCR Kit 1.0 For research use only! (RUO) Product No.: 201003 96 rxns INS-201000-GB-02 Store at -25 C... -15 C November 2012 altona Diagnostics GmbH Mörkenstraße
HANDBOOK QuickGene DNA whole blood kit L (DB-L) For Isolation of Genomic DNA from whole blood Ver.2.0 Contents 1. Introduction... 3 2. Kit components... 3 3. Storage conditions... 3 4. Other required materials,
Technical Manual Maxwell 16 Blood DNA Purification System Caution: Handle cartridges with care; seal edges may be sharp. 2800 Woods Hollow Rd. Madison, WI USA In Vitro Diagnostic Medical Device MDSS GmbH
Application Note Blocking of Globin Reverse Transcription to Enhance Human Whole Blood Gene Expression Profi ling Yasmin Beazer-Barclay, Doug Sinon, Christopher Morehouse, Mark Porter, and Mike Kuziora
292PR 01 G-Biosciences 1-800-628-7730 1-314-991-6034 email@example.com A Geno Technology, Inc. (USA) brand name Classic Immunoprecipitation Utilizes Protein A/G Agarose for Antibody Binding (Cat.
ST-PETERSBURG STATE UNIVERSITY THEODOSIUS DOBZHANSKY CENTER FOR GENOME BIOINFORMATICS GENOME RUSSIA PROJECT BLOOD SAMPLES COLLECTION, DNA EXTRACTION AND DNA QUALITY CONTROL PROTOCOLS 2015 The object of
EZ Load Molecular Rulers Catalog Numbers 170-8351 20 bp 170-8352 100 bp 170-8353 100 bp PCR 170-8354 500 bp 170-8355 1 kb 170-8356 Precision Mass EZ Load Molecular Rulers Quantity DNA sufficient for 100
SPARK DNA Sample Prep Kit Ion Torrent (SPK0002-V08) Frequently Asked Questions Under what circumstances would I use SPARK DNA Sample Prep Kit for Ion Torrent? Enzymatics SPARK DNA Sample Prep Kit for Ion
Wizard Genomic DNA Purification Kit and the Isolation of Plant Genomic DNA ABSTRACT Genomic DNA was isolated from plant leaf tissue using the Wizard Genomic DNA Isolation Kit (Cat.# A1120). Isolated DNA
PRODUCT ULLETIN SYR Select Master Mix SYR Select Master Mix Highly specific and sensitive quantitation SYR Select Master Mix offers advanced performance at an affordable price. SYR Select Master Mix is
INSTRUCTION MANUAL ZR-96 Quick-gDNA Blood Catalog Nos. D3075, D3076, & D3077 Highlights Quick, high throughput purification of high quality DNA from whole blood, plasma, and serum in less than 25 minutes
RNA-related Products TranscriptME RNA kit: Ideal choice for obtaining high yields of full-length cdna for RT-qPCR assays Suitable for as low RNA amount as 10 pg p p Convenient, reliable and cost-effective
RNA-related Products TRANSCRIPTME RNA kit: Ideal choice for obtaining high yields of full-length cdna for RT-qPCR assays Suitable for as low RNA amount as 10 pg p p Convenient, reliable and cost-effective
Analysis of messenger RNA using the Agilent 2100 Bioanalyzer and the RNA 6000 LabChip kit Application Note Meike Kuschel Abstract The RNA 6000 LabChip kit was used with the Agilent Technologies 2100 bioanalyzer
METAFECTENE SI Especially developed for the transfection of mammalian cells with sirna or mirna For ordering information, MSDS, publications and application notes see www.biontex.com Description Cat. No.
Third Edition December 2014 QIAamp DNA Micro Handbook For purification of genomic DNA from small volumes of blood dried blood spots urine tissues laser-microdissected tissues and for cleanup of genomic
QuickClean 96-Well Plasmid Miniprep Kit L00237 Technical Manual No. TM 0231 Version 0712007 I Description.. 1 II Kit Contents.. 1 III Applications 2 IV Key Features.. 2 V Storage.. 2 VI Plasmid Miniprep
LaboPass TM PCR Protocol Book LaboPass TM PCR Introduction LaboPass TM PCR kit provides a simple and rapid method to purify PCR products or other enzymatic reactions in just 6 minutes. Up to 10 µg of pure
Third Edition June 2012 QIAamp DNA Mini and Blood Mini Handbook For DNA purification from whole blood, plasma, serum, buffy coat, lymphocytes, dried blood spots (QIAamp DNA Mini Kit only), body fluids,
Nanosep & Nanosep MF Centrifugal Devices Description Simple, reliable concentrating and desalting of 50 to 500 µl samples in just 5 to 15 minutes Rapid processing of samples. Typical recoveries are greater
A Modified Wizard SV Genomic DNA Purification System Protocol to Purify Genomic DNA... A Modified Wizard SV Genomic DNA Purification System Protocol to Purify Genomic DNA from Shed Reptile Skin ABSTRACT
INSTRUCTIONS Frozen-EZ Yeast Transformation II Catalog No. T2001 Highlights High transformation efficiency that yields approximately 10 5-10 6 transformants per μg plasmid DNA (circular). Frozen storage
INSTRUCTION MANUAL Quick-RNA MiniPrep Catalog Nos. R1054 & R1055 Highlights High-quality total RNA (including small RNAs) from a wide range of samples. You can opt to isolate small and large RNAs in separate
Protocol for GenomePlex Whole Genome Amplification from Formalin-Fixed Parrafin-Embedded (FFPE) tissue Application Guide... 2 I. Description... 2 II. Product Components... 2 III. Materials to be Supplied
For Research Use Only Not for use in diagnostic procedures For the ambient temperature stabilization of high molecular weight DNA Fresh or frozen buffy coat samples Fresh or frozen white blood cell pellet
Application Note Authentication of Basmati rice using SSR-PCR and QIAxcel Advanced R. Cassier ADGENE Laboratoire, Thury Harcourt, France Introduction Basmati is one of the most popular types of rice in
is a is a state of the art transfection reagent, specifically designed for the transfer of sirna and mirna into a variety of eukaryotic cell types. is a state of the art transfection reagent, specifically
Luciferase Assay System Proto col GoClone Repor ter Construc ts: Sample Protocol for Adherent Cells LightSwitch Luciferase Assay System GoClone Reporter Assay Workflow Step 1: Seed cells in plate format.
molecular biology tools Restriction digestion Fast digestion in the universal Thermo Scientific FastDigest buffer Molecular cloning Fast ligation and positive selection of recombinant clones Nucleic acid
Aurora Forensic Sample Clean-up Protocol 106-0008-BA-D 2015 Boreal Genomics, Inc. All rights reserved. All trademarks are property of their owners. http://www.borealgenomics.com firstname.lastname@example.org
Protein Stabilization Reagents Elevated Protein Protection Protein Stabilization Reagents Naturally occurring proteases and phosphatases can destroy proteins you spent days isolating. Save valuable time
Biofocus Institute for Laboratory Medicine Dr. med. Doris Bachg Dr. med. Uwe Haselhorst Berghäuser Str. 295 45659 Recklinghausen Germany contact: Dr. Doris Bachg (email@example.com) phone +49 2361 3000-144
Unzipping Genes MBPCR014 Beta-Thalassemia Detection Kit P r o d u c t I n f o r m a t i o n Description: Thalassemia is a group of genetic disorders characterized by quantitative defects in globin chain
FAQ for Agilent 2100 Bioanalyzer Service How does the Bioanalyzer work? The bioanalyzer instrument detects biomolecules by laser-induced fluorescence. Each LabChip contains an interconnected set of microchannels
Procedures and Recommendations for DNA Sequencing at the Plant-Microbe Genomics Facility Ohio State University Biological Sciences Building Room 420, 484 W. 12th Ave., Columbus OH 43210 Telephone: 614/247-6204;
BLOOD SAMPLE PREPARATION The New Standard in RNA Purification, Best-in-Class, Pure & Simple Purification of inhibitor-free RNA, microrna and genomic DNA for any application An ISO 13485:2003, ISO 9001:2008
Automation and standardisation of NIPD Tobias J. Legler Dept. of Transfusion Medicine, University Medical Center Göttingen Certified according to DIN ISO 9001:2000 SAFE FINAL GENERAL ASSEMBLYin TARRAGONA
RNAgard Blood System VACUETTE RNAgard Blood Tubes BioMaxi Blood RNA Purification Kit For the collection, preservation and purification of RNA from whole blood Instructions For Use Protocol blood collection,
PicoMaxx High Fidelity PCR System Instruction Manual Catalog #600420 (100 U), #600422 (500 U), and #600424 (1000 U) Revision B Research Use Only. Not for Use in Diagnostic Procedures. 600420-12 LIMITED
Table of Contents General PCR Methods Page PCR Protocol Selection Guide...65.1 Basic PCR... 66.1.1 Hot Start PCR - The new Standard... 18.104.22.168 Reagents and Equipment Required... 22.214.171.124 General Considerations
DNA Sample preparation and Submission Guidelines Requirements: Please submit samples in 1.5ml microcentrifuge tubes. Fill all the required information in the Eurofins DNA sequencing order form and send
LightSwitch Dual Assay System DA010 (100 assays)! Description The LightSwitch Dual Assay System is used to normalize for variation between transfection replicates in cell lines that are particularly difficult
Performance of the D and High Sensitivity D ScreenTape Assays for the Agilent 22 TapeStation System Technical Overview Authors Susanne Glück Agilent Technologies, Waldbronn, Germany Jim Elliott Agilent
PicoMaxx High Fidelity PCR System Instruction Manual Catalog #600420 (100 U), #600422 (500 U), and #600424 (1000 U) Revision C Research Use Only. Not for Use in Diagnostic Procedures. 600420-12 LIMITED
GE Healthcare illustra NAP-25 Columns For the purification of oligonucleotides and small DNA fragments. Desalting and buffer exchange. Product booklet Codes: 17-0852-01 (20 columns) 17-0852-02 (50 columns)
Targeted Variant Test Requisition Form (3/4/2015) Instructions: Please provide clinical features of family members. All testing must be ordered by a qualified healthcare provider. See Test Selection Box
Guidelines f Preparing 20 kb SMRTbell Templates User Bulletin This Bulletin provides recommendations and tips f preparing 20 kb SMRTbell templates using the BluePippin size-selection method using size-selection
HIS-Select Nickel Affinity Gel Catalog Number P6611 Storage Temperature 2 8 C TECHNICAL BULLETIN Product Description HIS-Select Nickel Affinity Gel is an immobilized metalion affinity chromatography (IMAC)
Page 1 of 7 Agencourt RNAdvance Cell v2 Total RNA Isolation from Cultured Cells Please refer to http://www.agencourt.com/technical for updated protocols and refer to MSDS instructions http://www.beckmancoulter.com/customersupport/msds/msds.asp
KAMIYA BIOMEDICAL COMPANY Mouse Creatine Kinase MB isoenzyme (CKMB) ELISA For the quantitative determination of mouse CKMB in serum, plasma, cell culture fluid and other biological fluids Cat. No. KT-57681
Contents I. Overview 1 II. Kit Components 1 III. Storage 2 IV. Intended Use 2 V. Safety Warnings and Precautions 2 VI. Warranty and Liability 2 VII. Technical Assistance 3 VIII. Quality Management 3 IX.
Cat. # R010A For Research Use PrimeSTAR HS DNA Polymerase Product Manual Table of Contents I. Description...3 II. III. IV. Components...3 Storage...3 Features...3 V. General Composition of PCR Reaction
Sample Preparation and Reaction Set-up Important aspects of sample preparation and reaction setup Sampling Sample storage RNA / DNA Extraction and quality RNA / cdna / DNA storage Primer and probes formulation
Plant Genomic DNA Extraction using CTAB Introduction The search for a more efficient means of extracting DNA of both higher quality and yield has lead to the development of a variety of protocols, however
UNIVERSITY OF CANBERRA The use of a direct to PCR analysis method for disaster victim identification Clare Megan Berry Bachelor of Applied Science in Forensic Studies National Centre for Forensic Studies
MATCH IT! DNA Software Quick Reference Guide For Use with IMMUCOR LIFECODES HLA SSO Assays For In Vitro Diagnostic Use 1 LC1497.1 (08/15) This manual was produced for use with the MIDNA MATCH IT! DNA Software
TruePrime SINGLE CELL WGA KIT HANDBOOK TruePrime Single Cell WGA Kit INDEX Legal...4 Intended use...4 Kit contents...5 Shipping and storage...5 Handling...6 Quality control...6 Reagents and equipment to
TECHNICAL BULLETIN Wizard SV Gel and PCR Clean-Up System Instruc ons for Use of Products A9280, A9281, A9282 and A9285 Revised 12/10 TB308 Wizard SV Gel and PCR Clean-Up System All technical literature
Certificate of Analysis SMART-Seq v4 3 DE Kit Components Catalog No(s). Amount Lot Number 635042 (Not sold separately; Sold as a part of 635040) 96 rxns Specified on product label. 635043 (Not sold separately;
Your consent to our cookies if you continue to use this website.