Young s Double Slit Experiment

Save this PDF as:

Size: px
Start display at page:

Transcription

1 Young s Double Slit Experiment Apparatus optics bench laser slit film screen white paper and tape pencil metric ruler Ocean Optics spectrometer and fiber optics cable Goal In this experiment, you will set up a Double Slit apparatus and will measure the distance between bright fringes of the interference pattern. You will repeat the experiment for different pairs of slits to see how the interference pattern depends on slit separation. You will use a curve fit to measure the wavelength of the light. Introduction As a plane wave passes through a slit, it emerges as circular waves as if from a point source of light. If light from a laser (which is a plane wave) passes through two slits, then each slit acts as a point source of light. Because the incident wave is a plane wave and is monochromatic (i.e. one color, one frequency), then the waves from each slit will be in phase (meaning a crest will emanate from each slit at the same time) and will have the same frequency and amplitude. The waves from each source will interfere. At some points in space, total constructive interference occurs and a detector at this location will measure a wave with twice the amplitude, 2A, as the wave from each source. At other points in space total destructive interference occurs and a detector at this location will measure a wave with zero amplitude (meaning no wave at all). At the majority of points in space, interference results in a wave that has an amplitude between 0 and 2A meaning that it is not total constructive or destructive interference. You an identify such points in Figure 1. Figure 1: Constructive and destructive interference due to two sources. At some points in space around the two sources, total constructive interference will occur; at these points the path difference ( L) from the two sources is: L = L 1 L 2 = mλ m = 0, 1, 2,... total constructive interference 1

2 where L 1 and L 2 are the distances from the two sources to a point in space. For all points where m = 0, then L 1 = L 2 and total constructive interference occurs. One such location is shown in Figure 2. Figure 2: Total constructive interference where m = 0. You can identify a line of points where m = 0. Sketch a line on Figure 2 that connects all points where m = 0 and total constructive interference occurs. For all points where m = 1, then L 1 L 2 = 1λ and total constructive interference occurs. Two different locations where m = 1 are shown in Figure 3. Figure 3: Total constructive interference where m = 1. You can identify two separate lines of points where m = 1. Sketch both lines on Figure 3 that connects all points where m = 1 and total constructive interference occurs. Note that the points around the lines of total constructive interference are also bright and are a result of constructive interference, but are not as bright as the points where total constructive interference occurs. These regions are called fringes and it is in the center of the fringes where total constructive interference occurs. In the example above, there is a total of 5 fringes shown. As you can imagine, the pattern of points where total constructive interference occurs is related to the wavelength of the light and the distance between the sources. To explore this, you will first use a simulation. 2

3 Experiment 1 Exploring 2-source interference patterns with a simulation Procedure 1. Go to the URL: atitus/physlets/physlet.php?filename=optics_ripple-tank.html 2. A Java applet called Ripple Tank should load and you should see Figure 4. If the browser says Missing Plugin" and you do not see Figure 4, then you may need to download and install Java from Figure 4: An applet to explore 2-D superposition of waves. 3. Click the link to d = λ and wait about 15 seconds for all of the frames to be calculated. Then it will run and show the waves in motion. You can use the control buttons below the animation to stop, play, etc. In this case, the variable d is the distance between the sources. (Do not confuse this with L 1 and L 2 which were the distances from the sources to a point in space.) How many interference fringes do you see in the animation window? (Don t count any fringes below the sources in the y direction.) 4. Now click the link to d = 2λ. How many interference fringes do you see in the animation window? (Don t count any fringes below the sources in the y direction.) 5. Now click the link to d = 3λ. How many interference fringes do you see in the animation window? (Don t count any fringes below the sources in the y direction.) 6. Now click the link to d = 4λ. How many interference fringes do you see in the animation window? (Don t count any fringes below the sources in the y direction.) 3

4 As the distance between two sources increases, does the number of fringes increase, decrease, or remain the same? As the distance between two sources increases, do the fringes get further apart or do they get closer together? Experiment 2 Measuring fringe spacing for double-slit interference If you define a y-axis at the top of the animation window and mark the center of each fringe, then the spacing between fringes is y, as shown in Figure 5. It is this fringe spacing that you will measure in this experiment. Figure 5: Definition of fringe spacing. When you shine a laser onto 2 slits, you will get an interference pattern for the reasons described above. You will see a pattern like the one shown in Figure 6. The graph is the intensity of light I as a function of position y on a screen. y = 0 is defined as the center of the interference pattern, along a perpendicular bisector of the two slits. Use the graph to answer 4

5 Figure 6: An interference pattern due to laser light incident on 2-slits. the following questions. Note that the center of the left fringe is at y = 10 mm and the center of the right fringe is at y = 10 mm. What is the distance from the center of the far left fringe to the center of the far right fringe? How many total fringes are on the screen? (Include the far left and far right fringes.) What is the slit spacing y between fringes? The fringe spacing y depends on the wavelength of the light and the distance between the slits (d). A picture of the apparatus is shown in Figure 7. 5

6 Figure 7: Experimental apparatus. If the distance from the slits to the screen is L and the distance between slits is d, then the fringe spacing y is y = λl d You can solve this for wavelength. In this experiment, you will measure the slit spacing and fringe spacing and will calculate the wavelength. λ = d y L Procedure 1. Use ring stands to set up a laser and slit film so that light from the laser passes through two slits and creates an interference pattern on a screen. At first, use the set of slits that are closest together (i.e. smallest slit spacing, d). 2. Tape a piece of paper on a screen that is roughly 2-3 m from the slit film (the further the better). You will use this paper to mark bright fringes. 3. Measure L, the distance between the slit film and the paper using a meterstick or measuring tape. 4. On the paper, mark the locations of N fringes where N is enough fringes to span a large distance on the paper, perhaps about 20 cm or more. The more fringes you use, the less your uncertainty in your measurement of the distance between fringes. 5. Remove the piece of paper and use a ruler to measure the total distance from the first mark to the last mark. 6. Calculate the spacing between consecutive fringes, y. 7. Calculate the wavelength of the laser, λ. Convert its units to nm. 8. Change the slit spacing d by moving the slit film so that the light passes through a different set of slits and repeat the measurements. 6

7 slit spacing, d (mm) Number fringes (N) of distance between N fringes (cm) fringe spacing, y (cm) L (m) λ (nm) Analysis 1. What is the average wavelength of the laser as determined by your measurements using Young s Double Slit Experiment? 2. What is the standard deviation of the mean wavelength? Using a spectrometer to measure wavelength A spectrometer is a device that uses a diffraction grating to split light into its component wavelengths. It measures the light intensity at different wavelengths. A graph of intensity vs. wavelength is used to identify different wavelengths (i.e. colors) in the spectrum. Note that it only looks at the first order (m = 1) spectrum. 1. An Ocean Optics spectrometer has been set up for you at your computer. It is plugged into the computer via the USB port. It has a fiber optic cable attached to it. 2. View Logger Pro. It may already be set up and configured. 3. By default, Logger Pro plots absorbance vs. wavelength. But we wish to measure intensity. Logger Pro may already be set up to plot intensity vs. wavelength. If it is not, then go to Experiment Change Units Spectrometer: 1 Intensity to change what is being measured by the spectrometer. 4. Point the laser into the end of the fiber optic cable. Use the clothespin to press the button so that the laser remains on. 7

8 5. Click the Collect button. After you get a spectrum, click the Stop button. If the graph you see has a peak that goes off the graph (as shown below), that means we ve saturated the spectrometer, and we need a way to make the laser less bright. To do this, we can slightly tilt the laser so that less light enters the spectrometer. Do this and take another measurement. Continue trying until you get a peak that is not saturated. 6. Sketch and label the graph you see (when the spectrometer is not saturated). 7. You can measure the wavelength of any point on the spectrum using the Examine tool. Go to Analyze Examine. Hover over any point on the graph and the wavelength value will be displayed. Once you hover over a selected peak, you should view the data table to confirm the wavelengths of each peak. You can also highlight the peak and click the Statistics icon in the toolbar to obtain the wavelength of the maximum intensities. 8. Record the wavelength of the maximum intensity on the data sheet. The fact that the spectrum of a laser gives a single, narrow peak means that the light from the laser is composed of a single color. Thus it is monochromatic. Analysis 1. What is the wavelength of the laser according to your measurement with the spectrometer? 2. Does your measurement using the double-slit apparatus agree with your measurement using the spectrometer? In other words, is the measurement with the spectrometer within the uncertainty of your measurement with the double-slit apparatus? 8

Interference. Physics 102 Workshop #3. General Instructions

Interference Physics 102 Workshop #3 Name: Lab Partner(s): Instructor: Time of Workshop: General Instructions Workshop exercises are to be carried out in groups of three. One report per group is due by

Diffraction of Laser Light

Diffraction of Laser Light No Prelab Introduction The laser is a unique light source because its light is coherent and monochromatic. Coherent light is made up of waves, which are all in phase. Monochromatic

6) How wide must a narrow slit be if the first diffraction minimum occurs at ±12 with laser light of 633 nm?

Test IV Name 1) In a single slit diffraction experiment, the width of the slit is 3.1 10-5 m and the distance from the slit to the screen is 2.2 m. If the beam of light of wavelength 600 nm passes through

MULTIPLE CHOICE. Choose the one alternative that best completes the statement or answers the question.

Exam Name MULTIPLE CHOICE. Choose the one alternative that best completes the statement or answers the question. 1) A single slit forms a diffraction pattern, with the first minimum at an angle of 40 from

AP Physics B Ch. 23 and Ch. 24 Geometric Optics and Wave Nature of Light

AP Physics B Ch. 23 and Ch. 24 Geometric Optics and Wave Nature of Light Name: Period: Date: MULTIPLE CHOICE. Choose the one alternative that best completes the statement or answers the question. 1) Reflection,

Practice Test IV Name 1) In a single slit diffraction experiment, the width of the slit is 3.1 10-5 m and the distance from the slit to the screen is 2.2 m. If the beam of light of wavelength 600 nm passes

Fraunhofer Diffraction

Physics 334 Spring 1 Purpose Fraunhofer Diffraction The experiment will test the theory of Fraunhofer diffraction at a single slit by comparing a careful measurement of the angular dependence of intensity

FTIR Instrumentation

FTIR Instrumentation Adopted from the FTIR lab instruction by H.-N. Hsieh, New Jersey Institute of Technology: http://www-ec.njit.edu/~hsieh/ene669/ftir.html 1. IR Instrumentation Two types of instrumentation

Bohr s Model and Emission Spectra of Hydrogen and Helium

PHYS-01 LAB-03 Bohr s Model and Emission Spectra of Hydrogen and Helium 1. Objective The objective of this experiment is to study the emission spectrum of hydrogen and to understand its origin in terms

Resonance and the Speed of Sound

Name: Partner(s): Date: Resonance and the Speed of Sound 1. Purpose Sound is a common type of mechanical wave that can be heard but not seen. In today s lab, you will investigate the nature of sound waves

WAVELENGTH OF LIGHT - DIFFRACTION GRATING

PURPOSE In this experiment we will use the diffraction grating and the spectrometer to measure wavelengths in the mercury spectrum. THEORY A diffraction grating is essentially a series of parallel equidistant

v = fλ PROGRESSIVE WAVES 1 Candidates should be able to :

PROGRESSIVE WAVES 1 Candidates should be able to : Describe and distinguish between progressive longitudinal and transverse waves. With the exception of electromagnetic waves, which do not need a material

Graph Matching. walk back and forth in front of Motion Detector

Experiment 1 One of the most effective methods of describing motion is to plot graphs of distance, velocity, and acceleration vs. time. From such a graphical representation, it is possible to determine

Using the Spectrophotometer

Using the Spectrophotometer Introduction In this exercise, you will learn the basic principals of spectrophotometry and and serial dilution and their practical application. You will need these skills to

Measure the Distance Between Tracks of CD and DVD

University of Technology Laser & Optoelectronics Engineering Department Laser Eng Branch Laser application Lab. The aim of work: Experiment (9) Measure the Distance Between Tracks of CD and DVD 1-measure

Friday 18 January 2013 Morning

Friday 18 January 2013 Morning AS GCE PHYSICS B (ADVANCING PHYSICS) G492/01 Understanding Processes / Experimentation and Data Handling *G411640113* Candidates answer on the Question Paper. OCR supplied

Evaluation copy. Biological Membranes. Computer

Biological Membranes Computer 9 The primary objective of this experiment is to determine the stress that various factors, such as osmotic balance, detergents, and ph, have on biological membranes. Membranes

1 Laboratory #5: Grating Spectrometer

SIMG-215-20061: LABORATORY #5 1 Laboratory #5: Grating Spectrometer 1.1 Objective: To observe and measure the spectra of different light sources. 1.2 Materials: 1. OSA optics kit. 2. Nikon digital camera

Experiment 5: Magnetic Fields of a Bar Magnet and of the Earth

MASSACHUSETTS INSTITUTE OF TECHNOLOGY Department of Physics 8.02 Spring 2005 Experiment 5: Magnetic Fields of a Bar Magnet and of the Earth OBJECTIVES 1. To examine the magnetic field associated with a

3.14 understand that light waves are transverse waves which can be reflected, refracted and diffracted

Light and Sound 3.14 understand that light waves are transverse waves which can be reflected, refracted and diffracted 3.15 use the law of reflection (the angle of incidence equals the angle of reflection)

Physics 1020 Laboratory #6 Equilibrium of a Rigid Body. Equilibrium of a Rigid Body

Equilibrium of a Rigid Body Contents I. Introduction II. III. IV. Finding the center of gravity of the meter stick Calibrating the force probe Investigation of the angled meter stick V. Investigation of

GRAPH MATCHING EQUIPMENT/MATERIALS

GRAPH MATCHING LAB MECH 6.COMP. From Physics with Computers, Vernier Software & Technology, 2000. Mathematics Teacher, September, 1994. INTRODUCTION One of the most effective methods of describing motion

1051-232 Imaging Systems Laboratory II. Laboratory 4: Basic Lens Design in OSLO April 2 & 4, 2002

05-232 Imaging Systems Laboratory II Laboratory 4: Basic Lens Design in OSLO April 2 & 4, 2002 Abstract: For designing the optics of an imaging system, one of the main types of tools used today is optical

Quick Reference Manual

Quick Reference Manual ii TABLE OF CONTENTS This guide first leads you through the basics of Logger Pro, including software installation procedures. You will learn how to collect data, manually enter data,

THE CONSERVATION OF ENERGY - PENDULUM -

THE CONSERVATION OF ENERGY - PENDULUM - Introduction The purpose of this experiment is to measure the potential energy and the kinetic energy of a mechanical system and to quantitatively compare the two

How can I tell what the polarization axis is for a linear polarizer?

How can I tell what the polarization axis is for a linear polarizer? The axis of a linear polarizer determines the plane of polarization that the polarizer passes. There are two ways of finding the axis

Lab 5: Conservation of Energy

Lab 5: Conservation of Energy Equipment SWS, 1-meter stick, 2-meter stick, heavy duty bench clamp, 90-cm rod, 40-cm rod, 2 double clamps, brass spring, 100-g mass, 500-g mass with 5-cm cardboard square

PHYS 222 Spring 2012 Final Exam. Closed books, notes, etc. No electronic device except a calculator.

PHYS 222 Spring 2012 Final Exam Closed books, notes, etc. No electronic device except a calculator. NAME: (all questions with equal weight) 1. If the distance between two point charges is tripled, the

9.3. Diffraction and Interference of Water Waves

Diffraction an Interference of Water Waves 9.3 Have you ever notice how people relaxing at the seashore spen so much of their time watching the ocean waves moving over the water, as they break repeately

The Determination of an Equilibrium Constant

The Determination of an Equilibrium Constant Computer 10 Chemical reactions occur to reach a state of equilibrium. The equilibrium state can be characterized by quantitatively defining its equilibrium

INTERFERENCE OBJECTIVES PRE-LECTURE. Aims

53 L4 INTERFERENCE Aims OBJECTIVES When you have finished this chapter you should understand how the wave model of light can be used to explain the phenomenon of interference. You should be able to describe

Appendix E: Graphing Data

You will often make scatter diagrams and line graphs to illustrate the data that you collect. Scatter diagrams are often used to show the relationship between two variables. For example, in an absorbance

PhysFest March Holography

PhysFest March 2013 Holography Holography (from the Greek, holos whole + graphe writing) is the science of producing holograms, an advanced form of photography that allows an image to be recorded in three

4.4 WAVE CHARACTERISTICS 4.5 WAVE PROPERTIES HW/Study Packet

4.4 WAVE CHARACTERISTICS 4.5 WAVE PROPERTIES HW/Study Packet Required: READ Hamper pp 115-134 SL/HL Supplemental: Cutnell and Johnson, pp 473-477, 507-513 Tsokos, pp 216-242 REMEMBER TO. Work through all

Polarization of Light

Polarization of Light References Halliday/Resnick/Walker Fundamentals of Physics, Chapter 33, 7 th ed. Wiley 005 PASCO EX997A and EX999 guide sheets (written by Ann Hanks) weight Exercises and weights

EXPERIMENT O-6. Michelson Interferometer. Abstract. References. Pre-Lab

EXPERIMENT O-6 Michelson Interferometer Abstract A Michelson interferometer, constructed by the student, is used to measure the wavelength of He-Ne laser light and the index of refraction of a flat transparent

1 One Dimensional Horizontal Motion Position vs. time Velocity vs. time

PHY132 Experiment 1 One Dimensional Horizontal Motion Position vs. time Velocity vs. time One of the most effective methods of describing motion is to plot graphs of distance, velocity, and acceleration

3.003 Lab 4 Simulation of Solar Cells

Mar. 9, 2010 Due Mar. 29, 2010 3.003 Lab 4 Simulation of Solar Cells Objective: To design a silicon solar cell by simulation. The design parameters to be varied in this lab are doping levels of the substrate

- the. or may. scales on. Butterfly wing. magnified about 75 times.

Lecture Notes (Applications of Diffraction) Intro: - the iridescent colors seen in many beetles is due to diffraction of light rays hitting the small groovess of its exoskeleton - these ridges are only

Experiment 8: Undriven & Driven RLC Circuits

Experiment 8: Undriven & Driven RLC Circuits Answer these questions on a separate sheet of paper and turn them in before the lab 1. RLC Circuits Consider the circuit at left, consisting of an AC function

Appendix C: Graphs. Vern Lindberg

Vern Lindberg 1 Making Graphs A picture is worth a thousand words. Graphical presentation of data is a vital tool in the sciences and engineering. Good graphs convey a great deal of information and can

VISIBLE SPECTROSCOPY

VISIBLE SPECTROSCOPY Visible spectroscopy is the study of the interaction of radiation from the visible part (λ = 380-720 nm) of the electromagnetic spectrum with a chemical species. Quantifying the interaction

Physics 10. Lecture 29A. "There are two ways of spreading light: to be the candle or the mirror that reflects it." --Edith Wharton

Physics 10 Lecture 29A "There are two ways of spreading light: to be the candle or the mirror that reflects it." --Edith Wharton Converging Lenses What if we wanted to use refraction to converge parallel

PRACTICE EXAM IV P202 SPRING 2004

PRACTICE EXAM IV P202 SPRING 2004 1. In two separate double slit experiments, an interference pattern is observed on a screen. In the first experiment, violet light (λ = 754 nm) is used and a second-order

Solution Derivations for Capa #14

Solution Derivations for Capa #4 ) An image of the moon is focused onto a screen using a converging lens of focal length (f = 34.8 cm). The diameter of the moon is 3.48 0 6 m, and its mean distance from

Lab 9: The Acousto-Optic Effect

Lab 9: The Acousto-Optic Effect Incoming Laser Beam Travelling Acoustic Wave (longitudinal wave) O A 1st order diffracted laser beam A 1 Introduction qb d O 2qb rarefractions compressions Refer to Appendix

Physics 1050 Experiment 2. Acceleration Due to Gravity

Acceleration Due to Gravity Prelab Questions These questions need to be completed before entering the lab. Please show all workings. Prelab 1: For a falling ball, which bounces, draw the expected shape

Interferometers. OBJECTIVES To examine the operation of several kinds of interferometers. d sin = n (1)

Interferometers The true worth of an experimenter consists in his pursuing not only what he seeks in his experiment, but also what he did not seek. Claude Bernard (1813-1878) OBJECTIVES To examine the

Waves-Wave Characteristics

1. What is the wavelength of a 256-hertz sound wave in air at STP? 1. 1.17 10 6 m 2. 1.29 m 3. 0.773 m 4. 8.53 10-7 m 2. The graph below represents the relationship between wavelength and frequency of

Teaching Time: Two 50-minute periods

Lesson Summary In this lesson, students will build an open spectrograph to calculate the angle the light is transmitted through a holographic diffraction grating. After finding the desired angles, the

GEOMETRICAL OPTICS. Lens Prism Mirror

GEOMETRICAL OPTICS Geometrical optics is the treatment of the passage of light through lenses, prisms, etc. by representing the light as rays. A light ray from a source goes in a straight line through

RESOLVING POWER OF A READING TELESCOPE

96 Lab Experiments Experiment-255 RESOLVING POWER OF A READING TELESCOPE S Dr Jeethendra Kumar P K KamalJeeth Instrumentation & Service Unit, No-60, TATA Nagar, Bangalore-560 092, INDIA. Email:jeeth_kjisu@rediffmail.com

Laboratory #3 Guide: Optical and Electrical Properties of Transparent Conductors -- September 23, 2014

Laboratory #3 Guide: Optical and Electrical Properties of Transparent Conductors -- September 23, 2014 Introduction Following our previous lab exercises, you now have the skills and understanding to control

DIFFRACTION AND INTERFERENCE

DIFFRACTION AND INTERFERENCE In this experiment you will emonstrate the wave nature of light by investigating how it bens aroun eges an how it interferes constructively an estructively. You will observe

Physics 41 Chapter 38 HW Key

Physics 41 Chapter 38 HW Key 1. Helium neon laser light (63..8 nm) is sent through a 0.300-mm-wide single slit. What is the width of the central imum on a screen 1.00 m from the slit? 7 6.38 10 sin θ.11

PHY208FALL2008. Week2HW. Introduction to Two-Source Interference. Due at 11:59pm on Friday, September 12, View Grading Details [ Print ]

Assignment Display Mode: View Printable Answers PHY208FALL2008 Week2HW Due at 11:59pm on Friday September 12 2008 View Grading Details [ Print ] The following three problems concern interference from two

USING CDs AND DVDs AS DIFFRACTION GRATINGS

USING CDs AND DVDs AS DIFFRACTION GRATINGS Rama Balachandran Riverwood High School Atlanta, GA Karen Porter-Davis Chamblee Charter High School Chamblee, GA Copyright Georgia Institute of Technology 2009

How is LASER light different from white light? Teacher Notes

How is LASER light different from white light? Teacher Notes Concepts: (1) Light is a type of energy that travels as waves. [6.2.3.1.1] (2) Laser light is different from traditional light sources and must

INTERFERENCE OF SOUND WAVES

2011 Interference - 1 INTERFERENCE OF SOUND WAVES The objectives of this experiment are: To measure the wavelength, frequency, and propagation speed of ultrasonic sound waves. To observe interference phenomena

PUMPED Nd:YAG LASER. Last Revision: August 21, 2007

PUMPED Nd:YAG LASER Last Revision: August 21, 2007 QUESTION TO BE INVESTIGATED: How can an efficient atomic transition laser be constructed and characterized? INTRODUCTION: This lab exercise will allow

Electrical Resonance

Electrical Resonance (R-L-C series circuit) APPARATUS 1. R-L-C Circuit board 2. Signal generator 3. Oscilloscope Tektronix TDS1002 with two sets of leads (see Introduction to the Oscilloscope ) INTRODUCTION

Tektronix RSA306 USB Spectrum Analyzer

Tektronix RSA306 USB Spectrum Analyzer Simple Demos The Demo of the RSA306 is easy. Even you do not have signal generators, devices under test, or demo boards, using the whip antenna provided in box, you

Diffraction of a Circular Aperture

Diffraction of a Circular Aperture Diffraction can be understood by considering the wave nature of light. Huygen's principle, illustrated in the image below, states that each point on a propagating wavefront

A Guide to Acousto-Optic Modulators

A Guide to Acousto-Optic Modulators D. J. McCarron December 7, 2007 1 Introduction Acousto-optic modulators (AOMs) are useful devices which allow the frequency, intensity and direction of a laser beam

Interference and Diffraction

Chapter 14 nterference and Diffraction 14.1 Superposition of Waves... 14-14. Young s Double-Slit Experiment... 14-4 Example 14.1: Double-Slit Experiment... 14-7 14.3 ntensity Distribution... 14-8 Example

FREE FALL. Introduction. Reference Young and Freedman, University Physics, 12 th Edition: Chapter 2, section 2.5

Physics 161 FREE FALL Introduction This experiment is designed to study the motion of an object that is accelerated by the force of gravity. It also serves as an introduction to the data analysis capabilities

ATOMIC SPECTRA. Apparatus: Optical spectrometer, spectral tubes, power supply, incandescent lamp, bottles of dyed water, elevating jack or block.

1 ATOMIC SPECTRA Objective: To measure the wavelengths of visible light emitted by atomic hydrogen and verify the measured wavelengths against those predicted by quantum theory. To identify an unknown

Conservation of Momentum Using PASCO TM Carts and Track to Study Collisions in One Dimension

14 Conservation of Conservation of Using PASCO TM Carts and Track to Study s in One Dimension OBJECTIVE Students will collide two PASCO TM carts on a track to determine the momentum before and after a

Reflection and Refraction

Equipment Reflection and Refraction Acrylic block set, plane-concave-convex universal mirror, cork board, cork board stand, pins, flashlight, protractor, ruler, mirror worksheet, rectangular block worksheet,

Using light scattering method to find The surface tension of water

Experiment (8) Using light scattering method to find The surface tension of water The aim of work: The goals of this experiment are to confirm the relationship between angular frequency and wave vector

Blackbody Radiation References 1) R.A. Serway, R.J. Beichner: Physics for Scientists and Engineers with Modern Physics, 5 th Edition, Vol. 2, Ch.40, Saunders College Publishing (A Division of Harcourt

Correcting the Lateral Response Artifact in Radiochromic Film Images from Flatbed Scanners

Correcting the Lateral Response Artifact in Radiochromic Film Images from Flatbed Scanners Background The lateral response artifact (LRA) in radiochromic film images from flatbed scanners was first pointed

Auto Clicker Tutorial

Auto Clicker Tutorial This Document Outlines Various Features of the Auto Clicker. The Screenshot of the Software is displayed as below and other Screenshots displayed in this Software Tutorial can help

Lab 9. Optics. 9.1 Introduction

Lab 9 Name: Optics 9.1 Introduction Unlike other scientists, astronomers are far away from the objects they want to examine. Therefore astronomers learn everything about an object by studying the light

Standing Waves and the Velocity of Sound

Chapter 8 Standing Waves and the Velocity of Sound 8.1 Purpose In this experiment we will be using resonance points of a sound wave traveling through an open tube to measure the speed of sound in air.

O6: The Diffraction Grating Spectrometer

2B30: PRACTICAL ASTROPHYSICS FORMAL REPORT: O6: The Diffraction Grating Spectrometer Adam Hill Lab partner: G. Evans Tutor: Dr. Peter Storey 1 Abstract The calibration of a diffraction grating spectrometer

Experiment: Series and Parallel Circuits

Phy203: General Physics Lab page 1 of 6 Experiment: Series and Parallel Circuits OBJECTVES MATERALS To study current flow and voltages in series and parallel circuits. To use Ohm s law to calculate equivalent

THE BOHR QUANTUM MODEL

THE BOHR QUANTUM MODEL INTRODUCTION When light from a low-pressure gas is subject to an electric discharge, a discrete line spectrum is emitted. When light from such a low-pressure gas is examined with

Determining the Free Chlorine Content of Swimming Pool Water. HOCl H + + OCl. Evaluation copy

Determining the Free Chlorine Content of Swimming Pool Water Computer 33 Physicians in the nineteenth century used chlorine water as a disinfectant. Upon the discovery that certain diseases were transmitted

Does Quantum Mechanics Make Sense? Size

Does Quantum Mechanics Make Sense? Some relatively simple concepts show why the answer is yes. Size Classical Mechanics Quantum Mechanics Relative Absolute What does relative vs. absolute size mean? Why

Accessing via Web Browser

Accessing via Web Browser Hikvision DVR/NVR can be accessed via web browsing. It provides the functionalities of live view, playback, log management, and configuration, etc. Note: This instruction is written

Basic Optics System OS-8515C

40 50 30 60 20 70 10 80 0 90 80 10 20 70 T 30 60 40 50 50 40 60 30 C 70 20 80 10 90 90 0 80 10 70 20 60 50 40 30 Instruction Manual with Experiment Guide and Teachers Notes 012-09900B Basic Optics System

Holography 1 HOLOGRAPHY

Holography 1 HOLOGRAPHY Introduction and Background The aesthetic appeal and commercial usefulness of holography are both related to the ability of a hologram to store a three-dimensional image. Unlike

Experiment P-52 Magnetic Field

1 Experiment P-52 Magnetic Field Objectives To learn about basic properties of magnets. To study the magnetic field around a bar magnet through a magnetic field sensor. Modules and Sensors PC + NeuLog

Experiment 9 ~ RC Circuits

Experiment 9 ~ RC Circuits Objective: This experiment will introduce you to the properties of circuits that contain both resistors AND capacitors. Equipment: 18 volt power supply, two capacitors (8 µf

Waves: Recording Sound Waves and Sound Wave Interference (Teacher s Guide)

Waves: Recording Sound Waves and Sound Wave Interference (Teacher s Guide) OVERVIEW Students will measure a sound wave by placing the Ward s DataHub microphone near one tuning fork A440 (f=440hz). Then

ACCELERATION DUE TO GRAVITY

ACCELERATION DUE TO GRAVITY Objective: To measure the acceleration of a freely falling body due to gravitational attraction. Apparatus: Computer with Logger Pro, green Vernier interface box, picket fence

: Lab 1: Measurement and Uncertainty Pendulum Period

Introduction : Lab 1: Measurement and Uncertainty Pendulum Period Physics is primarily an experimental science. Physics theories are tested and refined and are only retained when they are proven to be

2 Spectrophotometry and the Analysis of Riboflavin

2 Spectrophotometry and the Analysis of Riboflavin Objectives: A) To become familiar with operating the Platereader; B) to learn how to use the Platereader in determining the absorption spectrum of a compound

PHYA2. General Certificate of Education Advanced Subsidiary Examination June 2010. Mechanics, Materials and Waves

Centre Number Surname Candidate Number For Examiner s Use Other Names Candidate Signature Examiner s Initials Physics A Unit 2 For this paper you must have: a ruler a calculator a Data and Formulae Booklet.

MSc in Autonomous Robotics Engineering University of York

MSc in Autonomous Robotics Engineering University of York Practical Robotics Module 2015 A Mobile Robot Navigation System: Labs 1a, 1b, 2a, 2b. Associated lectures: Lecture 1 and lecture 2, given by Nick

Embroidery Fonts Plus ( EFP ) Tutorial Guide Version 1.0505

Embroidery Fonts Plus ( EFP ) Tutorial Guide Version 1.0505 1 Contents Chapter 1 System Requirements.................. 3 Chapter 2 Quick Start Installation.................. 4 System Requirements................

Dynamics Track. Mechanical Force, Impulse and Momentum

Dynamics Track Mechanical Force, Impulse and Momentum An object subjected to unbalanced forces undergoes acceleration, which changes the velocity of the object in question. This change in motion can be

The He-Ne Laser * He-Ne Laser System. Power supply and ballast. interatomic collision. 1E-7 sec

The He-Ne Laser * I. Introduction The He-Ne laser (Figure 1) uses a low pressure (ca. 1 Torr He, 0.1 Torr Ne) mixture excited by a dc electric discharge. A ballast resistor is placed in series with the

Waves and Light Extra Study Questions

Waves and Light Extra Study Questions Short Answer 1. Determine the frequency for each of the following. (a) A bouncing spring completes 10 vibrations in 7.6 s. (b) An atom vibrates 2.5 10 10 times in

Geometric Optics Converging Lenses and Mirrors Physics Lab IV

Objective Geometric Optics Converging Lenses and Mirrors Physics Lab IV In this set of lab exercises, the basic properties geometric optics concerning converging lenses and mirrors will be explored. The

ILLUSTRATIVE EXAMPLE: Given: A = 3 and B = 4 if we now want the value of C=? C = 3 + 4 = 9 + 16 = 25 or 2

Forensic Spectral Anaylysis: Warm up! The study of triangles has been done since ancient times. Many of the early discoveries about triangles are still used today. We will only be concerned with the "right