# Pressure-based flowmeter calculations

Save this PDF as:

Size: px
Start display at page:

## Transcription

1 Pressure-based flowmeter calculations This worksheet and all related files are licensed under the Creative Commons Attribution icense, version 1.0. To view a copy of this license, visit or send a letter to Creative Commons, 559 Nathan Abbott Way, Stanford, California 94305, USA. The terms and conditions of this license allow for free copying, distribution, and/or modification of all licensed works by the general public. 1

2 Question 1 Questions A Pitot tube measuring the flow of glycerin (78.6 lb/ft 3 ) produces a differential pressure of 35 inches W.C. at a mass flow rate of 450 pounds per minute. Calculate the differential pressure produced by this same Pitot tube at a flow rate of 310 lb/min. Calculate the mass flow rate at a differential pressure of 12.6 WC. If the glycerin density happens to change from 78.6 lb/ft 3 to 75.1 lb/ft 3, calculate the mass flow rate at a differential pressure of 24.8 WC. file i03433 Question 2 A pneumatic DP transmitter measures the flow of gasoline through a an orifice plate, using a separate square root extractor relay to characterize the signal so that it may be displayed linearly on the receiver gauge (FI-20): FE-20 Q = 0 to 225 GPM P = 0 to 100 "WC 158 GPM FI-20 FT-20 Square-root extractor Reads 142 GPM 7.78 PSI 10.6 PSI Based on other flow-measuring devices in this process, operations personnel have determined the actual flow rate through this pipe is 158 gallons per minute. FI-20, however, registers a flow of 142 gallons per minute. Based on the data you see in this illustration, determine the location of the calibration error. file i

3 Question 3 A pneumatic DP transmitter measures the flow of gasoline through a an orifice plate, using a separate square root extractor relay to characterize the signal so that it may be displayed linearly on the receiver gauge (FI-20): FE-20 Q = 0 to 225 GPM P = 0 to 100 "WC 171 GPM FI-20 FT-20 Square-root extractor Reads 184 GPM 9.93 PSI PSI Based on other flow-measuring devices in this process, operations personnel have determined the actual flow rate through this pipe is 171 gallons per minute. FI-20, however, registers a flow of 184 gallons per minute. Based on the data you see in this illustration, determine the location of the calibration error. file i

4 Question 4 A smart DP transmitter with built-in square root characterization is used to measure flow through a pipe. The orifice plate range is 0 to 125 inches WC at 0 to 277 gallons per minute: FE-44 FI-44 FT ma signal A technician removes the transmitter from service and tests it by applying several air pressures to the port while leaving the port vented. ere are the As-Found results: Applied Pressure ( WC) Current signal (ma) Calculate the largest error, in percent of (output signal) span. Also, determine whether this looks like a zero or span error, and whether that error exists in the input (ADC) of the smart transmitter or in the output (DAC). file i

5 Question 5 A smart DP transmitter with built-in square root characterization is used to measure flow through a pipe. The orifice plate range is 0 to 125 inches WC at 0 to 277 gallons per minute: FE-44 FI-44 FT ma signal A technician removes the transmitter from service and tests it by applying several air pressures to the port while leaving the port vented. ere are the As-Found results: Applied Pressure ( WC) Current signal (ma) Calculate the largest error, in percent of (output signal) span. Also, determine whether this looks like a zero or span error, and whether that error exists in the input (ADC) of the smart transmitter or in the output (DAC). file i

6 Question 6 A smart DP transmitter with built-in square root characterization is used to measure flow through a pipe. The orifice plate range is 0 to 150 inches WC at 0 to 480 gallons per minute: 0 to 480 GPM 0 to 150 "WC To DCS ART communicator Online 1 Device setup 2 PV in2o 3 AO ma 4 RV in2o 5 URV in2o Operations personnel have strong reason to believe that the actual flow rate through this pipe is 160 GPM, yet the DCS registers a flow rate of GPM. Based on this information, determine the likely source of calibration error in this system. Also determine whether this transmitter has square-root characterization enabled or not. Additionally, suggest a good next step to perform to either pinpoint the location of this problem or correct it. file i

7 Question 7 A smart DP transmitter and orifice plate were recently installed to measure flow through a pipe. The orifice plate range is 0 to 150 inches WC at 0 to 1000 gallons per minute: 0 to 1000 GPM 0 to 150 "WC To DCS ART communicator Online 1 Device setup 2 PV in2o 3 AO ma 4 RV in2o 5 URV in2o Operations personnel register a flow rate of 699 gallons per minute on the display of their DCS, which they believe to be too much. Based on what you see here, do you think there is a problem, or is this new system working as it should? file i

### Instrument calibration tables

Instrument calibration tables This worksheet and all related files are licensed under the Creative Commons Attribution License, version 1.0. To view a copy of this license, visit http://creativecommons.org/licenses/by/1.0/,

### Liquid level measurement using hydrostatic pressure and buoyancy

iquid level measurement using hydrostatic pressure and buoyancy This worksheet and all related files are licensed under the Creative Commons Attribution icense, version 1.0. To view a copy of this license,

### Gas Custody Transfer Calibration

Gas Custody Transfer Calibration Using multi variable temperature / pressure calibrators for flowmeter calibration 2013 Introduction Gas custody transfer flow computers require special calibration to perform

### The Versatile Differential Pressure Transmitter. By David Gunn Honeywell Process Solutions

The Versatile Differential Pressure Transmitter By David Gunn Honeywell Process Solutions The Versatile Differential Pressure Transmitter 2 Table of Contents Abstract... 3 Pressure Fundamentals... 3 Applications...

### Pressure Transmitter Training Fundamentals of Pressure Measurement

Pressure Transmitter Training Fundamentals of Pressure Measurement Topics To Be Covered Pressure Measurement Absolute, Gauge, & Differential Pressure Pressure Transmitters Basic Functions Transmitter Terminology

### PROCESS CONTROL INDUSTRIAL MAINTENANCE TECHNICIAN CERTIFICATE

PROCESS CONTROL INDUSTRIAL MAINTENANCE TECHNICIAN CERTIFICATE Process Control (T5552) Course 1 Introduction to Process Control Course 2 Instrument Tags Course 3 Piping and Instrumentation Diagrams Course

### Density Measurement. Technology: Pressure. Technical Data Sheet 00816-0100-3208 INTRODUCTION. S min =1.0 S max =1.2 CONSTANT LEVEL APPLICATIONS

Technical Data Sheet 00816-0100-3208 Density Measurement Technology: Pressure INTRODUCTION Pressure and differential pressure transmitters are often used to measure the density of a fluid. Both types of

### Michael Montgomery Marketing Product Manager Rosemount Inc. Russ Evans Manager of Engineering and Design Rosemount Inc.

ASGMT / Averaging Pitot Tube Flow Measurement Michael Montgomery Marketing Product Manager Rosemount Inc. Russ Evans Manager of Engineering and Design Rosemount Inc. Averaging Pitot Tube Meters Introduction

### Selection Guide for Mechanical Differential Pressure Gauges

Selection Guide for Mechanical Differential Pressure Gauges A common specialized pressure gauge that uses the input from two pressure sources is the differential pressure gauge. A differential pressure

### Method to Determine PERFORMANCE OF INDIRECT-FIRED WATER HEATERS March 2003 Edition

Supplement to TESTING STANDARD Method to Determine PERFORMANCE OF INDIRECT-FIRED WATER HEATERS March 2003 Edition [NOTE: This supplement incorporates testing and calculations to determine heat source friction

### Using meters to measure steam flow

Page 1 of 6 April 1998 Plant Engineering SELECTING NEW TECHNOLOGY: PROBLEM SOLVING Using meters to measure steam flow Jesse L. Yoder, Ph.D., Senior Analyst, Automation Research Corp., Dedham, MA Key Concepts

### 718 Pressure Calibrator

Calibration Application Note instrumentation is found in virtually every process plant. Periodic calibration of these pressure, level, and flow instruments is required to keep plants operating efficiently

### Device Management Jonas Berge Emerson Process Management

Device Management Jonas Berge Emerson Process Management 1 Topics What is intelligent device management? Integrated operations Why do I need a handheld field communicator? With fieldbus, I can do everything

### Technical Publication

Technical Publication Flow Meter Selection for Improved Gas Flow Measurements: Comparison of Differential Pressure and Thermal Dispersion Technologies Art Womack, Sr. Applications Engineer Fluid Components

### Aeration Air & Digester Gas Flow Metering Using Thermal Mass Technology. HWEA 2011 Conference Craig S. Johnson

Aeration Air & Digester Gas Flow Metering Using Thermal Mass Technology HWEA 2011 Conference Craig S. Johnson Presentation Overview Introduction Aeration Air & Digester gas challenges Gas flow metering

### Process Instrumentation Trainer

Process Instrumentation Trainer 38-025 Process Controller 38-541-PCI Description Modern control devices and sensors are becoming increasingly complex and functionally very powerful. To expose students

### Measure Pressure and Flow

Measure Pressure and Flow Install the EMANT300 USB DAQ and Stepper Motor Training Kit 1. Connect up the EMANT300 USB DAQ and Pressure and Air pump Adaptor as shown. EMANT300 USB DAQ Pressure and Air pump

### Flow Measurement. Reference - Flow Measurement FUNDAMENTALS. ρ = fluid density (lb/ft3 or kg/m3)

Flow Measurement FUDAMETALS Flowmetering terms can often seem cryptic Here are definitions of some of the most commonly used ACCURACY Accuracy is a quantity defining the limit that errs will not exceed

### 12. Pressure sensors

12. Pressure s Questions Task a) What is a piezo-resistive effect? b) What is the difference between absolute, gauge and differential pressure? c) What is the relation between output voltage (e.g. U X

### Manual Supplement. Monitor Upgradable to Controller. Contents of this manual supplement include:

Model Number: Product/System Title: 8650-MON Fume Hood Face Velocity and Flow Monitor Upgradable to Controller Contents of this manual supplement include: 1) Sequence of operation 2) Application drawing

### Process switches and PLC circuits

Process switches and PLC circuits This worksheet and all related files are licensed under the Creative Commons Attribution License, version 1.0. To view a copy of this license, visit http://creativecommons.org/licenses/by/1.0/,

### Fluid Flow Measurement Selection and Sizing (ENGINEERING DESIGN GUIDELINE)

KLM Technology Group Sdn. Bhd. Guidelines for Processing Plant Page : 1 of 80 (ENGINEERING DESIGN GUIDELINE) Author: A L Ling Checked by: Karl Kolmetz TABLE OF CONTENT INTRODUCTION 5 Scope 5 Flow Measurement

### Industrial Instrumentation Course Catalog

Industrial Instrumentation Course Catalog 603-01 Instrumentation and Control Overview When you complete this lesson, you will be able to describe the basic operating principles behind the instrumentation

### UNDERSTANDING YOUR CHOICES IN FLOWMETER CALIBRATION

UNDERSTANDING YOUR CHOICES IN FLOWMETER CALIBRATION Mubeen Almoustafa, Calibration Application Engineer Flow Dynamics. Inc. Across various industries, measuring the flow of liquid or gas demands superior

### DREXELBROOK A Leader In Level Measurement Solutions

DREXELBROOK A Leader In Level Measurement Solutions USonic-R Series Ultrasonic Level Measurement System User Friendly Set the measurement range directly in inches, feet, millimeters, centimeters, or meters

### CEE 410: Hydraulic Engineering Lab 2 Flow Measurement Test at Unit Well 30 November 26 th 2014. Tyler Gates Bill Hussey Megan Long Senn Mettel

CEE 410: Hydraulic Engineering Lab 2 Flow Measurement Test at Unit Well 30 November 26 th 2014 Tyler Gates Bill Hussey Megan Long Senn Mettel Introduction This laboratory experiment involved the collecting

### UPM 15010. Figure 1 MPFM installed on well pad

UPM 151 In-line Multiphase Flowmeter Field Trial Comparison vs. Three Phase Flow Separator on Naturally Flowing Shale Wells A. Michael Jones, Independent Flow Measurement Consultant, Travis Gaskill, Daniel

### Industrial Instrumentation Prof. A. Barua Department of Electrical Engineering Indian Institute of Technology Kharagpur

Industrial Instrumentation Prof. A. Barua Department of Electrical Engineering Indian Institute of Technology Kharagpur Lecture - 11 Capacitance Transducers This is lesson 11 of Industrial Instrumentation.

### CALIBRATION PRINCIPLES

1 CALIBRATION PRINCIPLES After completing this chapter, you should be able to: Define key terms relating to calibration and interpret the meaning of each. Understand traceability requirements and how they

### Instrumentation and Process Control. Boiler Control. Courseware Sample 85993-F0

Instrumentation and Process Control Boiler Control Courseware Sample 85993-F0 A INSTRUMENTATION AND PROCESS CONTROL BOILER CONTROL Courseware Sample by the staff of Lab-Volt Ltd. Copyright 2010 Lab-Volt

### Thermal Dispersion Mass Flow

Thermatel Thermal Dispersion Mass Flow Measurement Handbook Table of Contents Introduction...1 What is Mass Flow Measurement?...1 Types of Flow Transmitters...4 Differential Pressure...4 Orifice...4 Venturi...6

### Media Liquid Level, Differential Pressure and Flow Meters

Media Liquid Level, Differential Pressure and Flow Meters Analog and Digital lndicating Units and Transmitters for Liquid Level Measurement Differential Pressure Measurement Flow Measurement Associated

### PIEZOMETER AIRFLOW MEASURING RING

PIEZOMETER AIRFLOW MEASURING RING M.K. Plastics is pleased to introduce The Piezometer Airflow Measuring Ring, now available as an accessory for our steel Axijet fans. The system consists of a Piezometer

### Custody Transfer Measurement. with the V-Cone Flowmeter

Custody Transfer Measurement with the V-Cone Flowmeter Stephen A. Ifft McCrometer Inc. Hemet, California, USA Abstract This paper will discuss the approval of the McCrometer V-Cone flowmeter for custody

### PROPORTIONAL PRESSURE CONTROL VALVES AND SYSTEMS.

PROPORTIONAL PRESSURE CONTROL VALVES AND SYSTEMS www.elwood.com 9 0 0 1 : 2 0 0 0 C E R T I F I E D C O M P A N Y Proportional Pressure Control Valves The Elwood Proportional Pressure Control Valves are

### Introduction Process Piping and Instrumentation Diagram (P&ID)

Session 2313 Use of Lab Experiments to Build Transport Concepts Abstract Introduction Process Piping and Instrumentation Diagram (P&ID) Viscosity Measurement Absolute Viscosity vs. Sucrose Conce ntration

### SUMMARY of PRODUCTS. HVAC Systems Control Airflow, Space, & Building Pressurization. Accurate airflow measurement for demanding applications

SUMMARY of PRODUCTS HVAC Systems Control Airflow, Space, & Building Pressurization Accurate airflow measurement for demanding applications Airflow Measuring Stations FAN-E AIRFLOW MEASURING STATION. The

### F ox W hi t e Paper. Reduce Energy Costs and Enhance Emissions Monitoring Systems

F ox W hi t e Paper Reduce Energy Costs and Enhance Emissions Monitoring Systems A Technical White Paper from Fox Thermal Instruments Rich Cada, VP Sales & Marketing, Fox Thermal Instruments, Inc. 399

### Turbo Tech 103 (Expert)

Turbo Tech 103 (Expert) This article is a bit more involved and will describe parts of the compressor map, how to estimate pressure ratio and mass flow rate for your engine, and how to plot the points

### Replacing Displacers with Guided Wave Radar

ROSEMOUNT GUIDED WAVE RADAR Replacing Displacers with Guided Wave Radar KEY POINTS Mounting flanges vary by displacer supplier Probe must extend the of the displacer chamber Single rigid probes are the

### Ethernet Intelligent Pressure Scanner

Ethernet Intelligent Pressure Scanner NetScanner System 9116 FEATURES User Accessible Memory for Test Configuration Management ±0.05% FS System Accuracy EU Throughput Rates of 500 Hz Auto-Negotiating 10/100

### Flow Measurement Options for Pipeline and Open Channel Flow

Flow Measurement Options for Pipeline and Open Channel Flow October 2013 Presented by Molly Skorpik - 2013 Montana Association of Dam and Canal Systems Conference Irrigation Training and Research Center

### AXIAL PISTON PUMPS VARIABLE DISPLACEMENT

FEATURES The Series is a family of variable displacement axial piston pumps for use in closed circuits with housing and distributor cover in aluminium. The displacement is continuously variable by means

### NOTIFICATION OF PERFORMANCE TESTING FOR GASOLINE DISPENSING FACILITIES (GDFs) NESHAP 40 CFR, Part 63, Subpart CCCCCC (Sections 63.11110 63.

MICHIGAN DEPARTMENT OF ENVIRONMENTAL QUALITY (DEQ), OFFICE OF ENVIRONMENTAL ASSISTANCE NOTIFICATION OF PERFORMANCE TESTING FOR GASOLINE DISPENSING FACILITIES (GDFs) NESHAP 40 CFR, Part 63, Subpart CCCCCC

### ISA108 Intelligent Device Management (IDM)

ISA108 Intelligent Device Management (IDM) Intelligent Digital Devices - Have Self-Diagnostics and More Eight-channel two-wire temperature transmitter Intelligent two-wire on/off valve 2 Fully integrated

### Metrology Society of Australia MSA Test Method Calibration of Pressure Calibrators, Indicators and Transducers

Metrology Society of Australia MSA Test Method 1-2008 Calibration of Pressure Calibrators, Indicators and Transducers Revision No. 5 June 2008 Contents: 1. Introduction 2. Scope 3. Desired outcomes of

### Instrument calibration

Chapter 18 Instrument calibration Every instrument has at least one input and one output. For a pressure sensor, the input would be some fluid pressure and the output would (most likely) be an electronic

### & Tower for PWM Control. Maximum Application Rates with Two 5.3 GPM Electric Pumps

396-001260 Tower Electric Pump Fertilizer System for Ag Leader Liquid Product Control Module & Tower for PWM Control Maximum Application Rates with Two 5.3 GPM Electric Pumps Maximum Application Rates

### Series 420 Pneumatic Control System. Pneumatic Measuring and Control Equipment. Edition March 2008

Series 420 Pneumatic Control System Pneumatic Measuring and Control Equipment. Associated Data Sheets T 7506 EN to T 7575 EN T 6111 EN to T 6134 EN T 8510 EN, T 8350 EN, T 8540 EN, T 8546 EN Edition March

### PRODUCT PROFILE FEATURES

PRODUCT PROFILE Pressure Transmitters & Pneumatic Transducer Airflow Transmitter, Differential Air Pressure Transmitter and Differential & Gauge Liquid/Gas Pressure Transmitter FEATURES AIRFLOW TRANSMITTER

### Understanding Your Choices In Flowmeter Calibration

Understanding Your Choices In Flowmeter Calibration Mubeen Almoustafa, Calibration Application Engineer Flow Dynamics. Inc. Across various industries, the performance of a flow measurement device is ultimately

### Daniel. Liquid Control Valves Technical Guide. Technical Guide DAN-LIQ-TG-44-rev0813. DAN-LIQ-TG-44-rev0208. February 2008.

DAN-LIQ-TG-44-rev0208 February 2008 Daniel Liquid Control Valves Technical Guide www.daniel.com Daniel Measurement and Control Theory, Principle of Operation and Applications This brochure has been prepared

### Problems. Instrumentation

Problems TFMT14 A Instrumentation B a) Which instrument is an active type of device? b) Indicate one main advantage of each type? Example Detection range and accuracy specs. Calculate accuracy and relative

### Instrumentation for the Water and Wastewater industry

Instrumentation for the Water and Wastewater industry Application Solutions Best in class instrumentation for waste water treatment Today's instrument engineer doesn't have the time to study specifications

### In-Line Electronic Flow Switches

In-Line Electronic Flow Switches Industries Petrochemical Refining Oil Production Water Treatment Pharmaceutical Food and Beverage Pulp and Paper Power Production Gas Processing Mining Biotechnology Semiconductor

### CHME 302 CHEMICAL ENGINEERING LABOATORY-I EXPERIMENT 302-1 FLOW MEASUREMENT TEST

CHME 302 CHEMICAL ENGINEERING LABOATORY-I EXPERIMENT 302-1 FLOW MEASUREMENT TEST OBJECTIVE The purpose of the experiment is to demonstrate the working principles of four most common devices used to measure

### Process Instrumentation Schools

Process Instrumentation Schools Training combined with hands-on experience, to learn how to reduce costs and achieve process optimization Training is the key Technology and the marketplace are changing

### Reference Manual. Pressure/Temperature Compensated Flow (P_PTComp) Version 3.1

Reference Manual Rockwell Automation Library of Process Objects: Pressure/Temperature Compensated Flow (P_PTComp) Version 3.1 Important User Information Read this document and the documents listed in the

### DeltaSpan Submersible Pressure Level Transmitter Manual LD31 Series 22 AUG 08 Rev A

DeltaSpan Submersible Pressure Level Transmitter Manual LD31 Series 22 AUG 08 Flowline, Inc. 10500 Humbolt Street Los Alamitos, CA 90720 Tel: (562) 598-3015 Fax: (562) 431-8507 www.flowline.com 22 AUG

### Michelin North America

www.centecinc.com SC Telephone: 864.527.7750 Outside SC: 800.227.0855 Michelin North America Industrial Maintenance Technical Interview Outline Industrial Maintenance Technical Interview Outline The Technical

### The Innovative, Reliable Foxboro Pressure Transmitter Family

The Innovative, Reliable Foxboro Pressure Transmitter Family Accuracy Industry-leading 0.025% and best-in-class accuracy at turndown Functionality Extensive product line to handle nearly every application

### TEST REPORT for the ARID TECHNOLOGIES VAPOR RECOVERY UNIT installed in LANTANA, FL

Project No. 148 TEST REPORT for the ARID TECHNOLOGIES VAPOR RECOVERY UNIT installed in LANTANA, FL Prepared for: Mr. Ted Tiberi Arid Technologies, Inc. 130 North LaGrange Rd. LaGrange, IL 60525 Prepared

### Direct Mounting Allows Differential Pressure (DP) Based Flow Measurement Optimization

Direct Mounting Allows Differential Pressure (DP) Based Flow Measurement Optimization Gregory Robert Strom, Marketing Product Manager Gregory Livelli, Manager of Worldwide Pressure Marketing Rosemount,

### Extending Orifice Flow Data

46 Extending Flow Data EXTRAPOLATION 1. Specific Gravity Other Gases To convert air flow from chart to another gas flow. Flow (gas) = Flow (air) / S.G. (gas) To obtain flow rate for helium when air flow

### Energy Systems Engineering Technology

College of Technology Instrumentation and Control Module # 4 Flow Measurement Document Intent: The intent of this document is to provide an example of how a subject matter expert might teach Flow Measurement.

### Laboratory 11. Fluid Statics, Pressure, and Buoyant Forces

Laboratory 11 Fluid Statics, Pressure, and Buoyant Forces I. Introduction In this exercise we will investigate some properties of non-moving fluids and the forces they exert on objects immersed in them.

### Solutions for Storage Tanks

www.tankgaugingsys.com Solutions for Storage Tanks 5020 Top Mounted Sour Service Mechanical Level Gauge TGS 020 1.) POINT SWITCH CSA Class 1, Div.1 Magnetically Activated Proximity Switches for high/low

### Fluid Mechanics. Fluid Statics [3-1] Dr. Mohammad N. Almasri. [3] Fall 2010 Fluid Mechanics Dr. Mohammad N. Almasri [3-1] Fluid Statics

1 Fluid Mechanics Fluid Statics [3-1] Dr. Mohammad N. Almasri Fluid Pressure Fluid pressure is the normal force exerted by the fluid per unit area at some location within the fluid Fluid pressure has the

### TERMINAL UNIT AIRFLOW MEASUREMENTS

Page: 1 of 6 TERMINAL UNIT AIRFLOW MEASUREMENTS Dan Int-Hout Chief Engineer Page: 2 of 6 OVERVIEW The control of airflow rate in VAV systems is important for several reasons, including acoustics, ventilation,

### Sentinel LCT. Panametrics Ultrasonic Flowmeter for Liquid Custody Transfer Measurement. GE Measurement & Control Solutions. Benefits.

GE Measurement & Control Solutions Sentinel LCT Panametrics Ultrasonic Flowmeter for Liquid Custody Transfer Measurement Benefits Extremely reliable and highly accurate flowmeter, designed specifically

### Theory overview of flow measurement using differential pressure devices based on ISO-5167 standard.

Theory overview of flow measurement using differential pressure devices based on ISO-567 standard. rian FL40 flow computer description. Flow Cad software users manual. Technical note, Differential pressure

### Injection Systems INSTALLATION AND OPERATING GUIDE HI FLO VERTICAL SYSTEMS

1 TM INSTALLATION AND OPERATING GUIDE HI FLO VERTICAL SYSTEMS When the HI-FLO Metering Head is attached to a tank, the tank will be pressurized to the same pressure as the irrigation system. Before attaching

### The On-Board Refueling Vapor Recovery (ORVR) Evaporative Emission (EVAP) system.

«1A: Description and Operation» Overview The Evaporative Emission (EVAP) system prevents fuel vapor build-up in the sealed fuel tank. Fuel vapors trapped in the sealed tank are vented through the vapor

### DP Series. Differential Pressure Sensor. Installation and Operating Manual

DP Series Differential Pressure Sensor Installation and Operating Manual For assistance with the operation of this product, contact Dylix Phone: 716-773-2985 Fax: 716-773-2786 E-mail: sales@dylixcorp.com

### Sanitary Type Electronic Flow Switches

Industries Pharmaceutical Sanitary Type Electronic Flow Switches Food and Beverage Biotechnology Semiconductor Water Treatment S Series Ameritrol, Inc. Instruments and Controls Features No Moving Parts

### Centrifugal Fans and Pumps are sized to meet the maximum

Fans and Pumps are sized to meet the maximum flow rate required by the system. System conditions frequently require reducing the flow rate. Throttling and bypass devices dampers and valves are installed

### Earthsafe GT100 Global Tank System Modular Fuel and Water Tank Systems

General Description The GT100 Global Tank System provides maximum cost efficiency and operational flexibility for fuel and water storage. Tank Systems are based on ISO Container Specifications for easy

### ANNUBAR FLOW CALCULATION PROGRAM (For the FloBoss 103) User Manual (QER 04Q019)

Flow Computer Division ANNUBAR FLOW CALCULATION PROGRAM (For the FloBoss 103) User Manual (QER 04Q019) Form A6159 September 2004 Revision Tracking Sheet September 2004 This manual may be revised from time

### Advanced Differential Pressure Flowmeter Technology V-CONE FLOW METER TECHNICAL BRIEF

Advanced Differential Pressure Flowmeter Technology V-CONE FLOW METER TECHNICAL BRIEF Table of Contents Section 1 - General Introduction 1.1 1 Principles Of Operation 1.2 1 Reshaping The Velocity Profile

### RAPID RESPONSE Model RSV-1 Residential Shut-Off Valve, 1 Inch and 2 Inch (DN25 and DN50) for Dual-Purpose Residential Water Supply General Description

Worldwide Contacts www.tyco-fire.com RAPID RESPONSE Model RSV-1 Residential Shut-Off Valve, 1 Inch and 2 Inch (DN25 and DN50) for Dual-Purpose Residential Water Supply General Description The TYCO RAPID

### Effects of Atmospheric Pressure on Gas Measurement

Effects of Atmospheric Pressure on Gas Measurement Class # 1390.1 March 2012 / White paper by Denis Rutherford Regional Sales Manager Central US Schneider Electric Telemetry & Remote SCADA Solutions 6650

### NOTE: The Hydrant in the circle is the test Hydrant. Residual and Static flows are measured here. Loop

NOTE: The Hydrant in the circle is the test Hydrant. Residual and Static flows are measured here. Figure 1 Loop Test Hydrant (Choose Either) Measure Residual & Static Pressures Dead End Main - Water One

### Outdated Publication, for historical use. CAUTION: Recommendations in this publication may be obsolete.

IRRIGATION MANAGEMENT S E R I E S Irrigation Water Measurement Danny H. Rogers Extension Agricultural Engineer Richard D. Black Extension Agricultural Engineers Cooperative Extension Service Kansas State

### Calibrating Differential Pressure Sensors

Calibrating Differential Pressure Sensors Additel Corporation 22865 Savi Ranch Parkway, Ste F Yorba Linda, CA 92887 Introduction Differential pressure sensors are common in the process industry and cover

### Performance 4. Fluid Statics, Dynamics, and Airspeed Indicators

Performance 4. Fluid Statics, Dynamics, and Airspeed Indicators From our previous brief encounter with fluid mechanics we developed two equations: the one-dimensional continuity equation, and the differential

### MTI10 Insertion and MTL10 In-line Thermal Mass Flowmeter and Temperature Transmitter

Thermal Mass meter and Temperature Transmitter Description The MTI10 insertion thermal mass flowmeter and the MTL10 in-line thermal mass flowmeter provide accurate mass flow measurement of clean, dry gases

### If the level is too high, flooding of steam purification equipment can occur.

Application Guide Control Systems in the Process Industries Control Systems are used extensively throughout the process industries and the Utilities to control the level of boiling water contained in boiler

### 2 1/2 Pipe. 40 = height. the gauge pressure inside the vessel from the gauge pressure at the nozzle inlet as shown:

116eering. Engineering. Engineering. Engineering. Engineerin Engineering Information SPECIFYING SPRAY NOZZLES Spray nozzles have three basic functions: meter flow distribute liquid break up a liquid stream

### Modeling Fluid Systems

Modeling Fluid Systems The prevalent use of fluid (hydraulic) circuitry in machines tool applications, aircraft control systems, and similar operations occurs because of such factors such as accuracy,

### CHE 253M Experiment No. 3 LIQUID FLOW MEASUREMENT

Rev 8/15 AW/GW CHE 253M Experiment No. 3 LIQUID FLOW MEASUREMENT The objective of this experiment is to familiarize the student with several types of flow-measuring devices commonly used in the laboratory

### PARTIALLY CLOSED VALVE EFFECTS ON THE V-CONE FLOWMETER

PARTIALLY CLOSED VALVE EFFECTS ON THE V-CONE FLOWMETER Stephen A. Ifft (McCrometer Inc., Hemet, CA 9545, USA) ABSTRACT Research conducted from 199 through 1996 indicates that the V-Cone flowmeter is less

### Mono-Block II Installation & Operation Manual

Mono-Block II Installation & Operation Manual Enraf Fluid Technology UK, Ltd. Enraf Fluid Technology USA, Inc. 6 Pennant Park, Standard Way 2000 Northfield Court Fareham, Hampshire, England PO16 8XU Roswell,

### Liquid Level Controllers

Liquid Level Controllers ALL PUMPS LLC-2Y For both pump up and down control. Variable On/Oﬀ set points. NDP 15/20/25/32, DP 10/15 & *NDP 40/50/80 LLC-1 Pump up and down versions available. 6 to 12 diﬀeren

### Morrison Bros. Co. General Product Specifications

Morrison Bros. Co. General Product Specifications Tank Mounted Spillbox The spill containment device is manufactured to contain spills and drips that may occur at the fill point on aboveground storage

### Micro Motion Model 2700 Transmitters with Analog Outputs

Configuration and Use Manual MMI-20019043, Rev AA March 2012 Micro Motion Model 2700 Transmitters with Analog Outputs Configuration and Use Manual Safety messages Safety messages are provided throughout

### An Overview of: NFPA 86 Standard for Ovens & Furnaces

An Overview of: NFPA 86 Copyright 2014 LEWCO, Inc. Alisha Fannin, Controls Engineer Mike Luli, Applications Engineer Introduction The most widely referenced standard in the United States for industrial