# Construction of separating calorimeter is as shown in figure:

Save this PDF as:

Size: px
Start display at page:

## Transcription

1 1. Title: Measurement of dryness fraction by Separating Calorimeter, Throttling Calorimeter, Separating and Throttling Calorimeter. 2. Learning objectives: 2.1. Intellectual skills: a) Measurement of Dryness fraction of steam. b) Understanding various methods of measurement of Dryness fraction Motor skills: a) Arrangement of various components for set up of Throttling, Separating, Separating and Throttling calorimeter. b) To measure the quality of steam. 3. Prior concept: a) Steam generation, Steam quality b) Steam properties 4. New concept: 1. Separating calorimeter: The quality of wet steam is usually defined by its dryness fraction. When the dryness fraction, pressure and temperature of the steam are known, then the state of wet steam is fully defined. In a steam plant it is at times necessary to know the state of the steam. For wet steam, this entails finding the dryness fraction. When the steam is very wet, we make use of a separating calorimeter. Construction of separating calorimeter is as shown in figure: Fig. 3.1 Separating calorimeter [xiii] The steam is collected out of the main steam supply and enters the separator from the top. The steam is forced to make a sharp turn when it hits the perforated cup (or any other mechanism that produces the same effect). This results in a vortex motion in the steam, and

2 water separates out by the centrifugal action. The droplets then remain inside the separator and are collected at the bottom, where the level can be recorded from the water glass. The dry steam will pass out of the calorimeter into a small condenser for the collection of the condensate. However, not all the water droplets remain in the collector tank. Some water droplets pass through to the condenser, and hence this calorimeter only gives a close approximation of the dryness fraction of the steam. From the results obtained from the two collectors, the dryness fraction may then be found from This can be expressed as: Where, M is the mass of dry steam and Dryness fraction = m is the mass of suspended water separated in the calorimeter in the same time. x = Procedure: 1. Observe the setup 2. Identify all the connected equipments 3. Check the range of pressure gauge 4. Open the steam supply valve for a few seconds 5. Measure the condensate formed due to condensation of the moisture in the steam. 6. Measure the condensate formed due to condensation of the dry steam Observation Table: Sr. No. Parameters Reading 1 Boiler steam pressure, p 1 (bar) 2 Mass of condensate collected, m (kg) 3 Mass of dry steam, M (kg) Calculation: Dryness fraction (x) =

3 x = Result: The dryness fraction of the sample taken from the main stream is. Example: In a laboratory experiment, a sample of wet steam is allowed to pass through a separating calorimeter. At some instant, the water collected in the chamber was 0.1 kg whereas the condensed steam was found to be 1.25 kg. Determine the dryness fraction of the steam entering the calorimeter. Solution: Given: m = 0.1 kg and M = 1.25 kg Dryness fraction of the steam x = = = 0.926

4 2. Throttling calorimeter: If we have steam that is nearly dry, we make use of a throttling calorimeter as shown in figure. This calorimeter is operated by first opening the stop valve fully so that the steam is not partially throttled as it passes through the apparatus for a while to allow the pressure and temperature to stabilize. If the pressure is very close to atmospheric pressure, the saturation should be around 100 C, it may be assumed that the steam is superheated. When the conditions have become steady, the gauge pressure before throttling is read from the pressure gauge. After throttling, the temperature and gauge pressure are read from the thermometer and manometer respectively. The barometric pressure is also recorded. From equation =, We have at p 1 = at p 2 + x = + C p ( - ) And thus x = Procedure: 1. Observe the setup 2. Identify all the connected equipments 3. Check the range of pressure gauge 4. Check the range of thermometer Fig. 3.2 Throttling calorimeter [xiii]

5 5. Check the range of manometer 6. Open the steam supply valve for a short time 7. Measure the steam chest pressure (p 1 ) 8. Measure the steam outlet pressure (p 2 ) 9. Measure the outlet steam temperature (t 2 ) Observation Table: Sr. No. Parameters Reading 1 Boiler steam pressure, p 1 (bar) 2 Steam outlet pressure, p 2 (bar) 3 Steam outlet temperature ( C) Required readings from steam table: Steam properties at steam chest pressure: a. Enthalpy of feed water ( ): b. Enthalpy of wet steam ( ): Properties of outlet steam: a. Saturation temperature at (p 2 ): b. Degree of superheat: Outlet steam temperature Saturation temperature = ( - ) = c. Enthalpy of superheated steam (hg 2 ) : Calculations: at p 1 = at p 2 + x = + C p ( - ) And thus x= Result: x = The dryness fraction of the sample taken from the main stream is.

6 Example: A throttling calorimeter is used to measure the dryness fraction of the steam in the steam main which has steam flowing at a pressure of 8 bar. The steam after passing through the calorimeter is at 1 bar pressure and 115 C. Solution: Calculate the dryness fraction of the steam in the main. Take Cps = 2.1 kj/kg K. 1. Condition of steam before throttling: Pressure, p 1 = 8 bar, dryness fraction, x =? 2. Condition of steam after throttling: Pressure, p 2 = 1 bar, Temperature, = = 115 C. Steam properties at steam chest pressure: c. Enthalpy of feed water ( ): kj/kg d. Enthalpy of wet steam ( ): kj/kg Properties of outlet steam: d. Saturation temperature at (p 2 ): 99.6 C e. Degree of superheat: Outlet steam temperature Saturation temperature = ( - ) = Enthalpy of superheated steam ( ): kj/kg As throttling is a constant enthalpy process at p 1 = at p 2 + x = + C p ( - ) x * = * ( ) x = x = 0.97

7 3. Separating and throttling calorimeter: If the steam whose dryness fraction is to be determined is very wet then throttling to atmospheric pressure may not be sufficient to ensure superheated steam at exit. In this case it is necessary to dry the steam partially, before throttling. This is done by passing the steam sample from the main through a separating calorimeter as shown in figure. The steam is made to change direction suddenly, and the water, being denser than the dry steam is separated out. The quantity of water which is separated out (m w ) is measured at the separator, the steam remaining which now has a higher dryness fraction, is passed through the throttling calorimeter. With the combined separating and throttling calorimeter it is necessary to condense the steam after throttling and measure the amount of condensate (m s ). If a throttling calorimeter only is sufficient, there is no need to measure condensate, the pressure and temperature measurements at exit being sufficient. Fig. 3.3 Separating and throttling calorimeter [xiii] Let, State 1 = Properties of steam Coming to Separating Calorimeter State 2 = Properties of steam leaving Separating Calorimeter State 3 = Properties of steam leaving Throttling Calorimeter = Dryness fraction of the steam at Separating Calorimeter = Dryness fraction of the steam at Throttling Calorimeter Dryness fraction at 2 is, therefore, the mass of dry steam leaving the separating calorimeter is equal to m s and this must be the mass of dry vapour in the sample drawn from the main at state 1. Hence fraction in main, = =

8 The dryness fraction can be determined as follows: = = + *..at p 2 = + + C ps ( - ) at pressure p 3 From Enthalpy at 2 = Enthalpy at 3 x 2 = The values of and are read from steam tables at pressure p 2. The pressure in the separator is small so that p 1 is approximately equal to p 2. Procedure: 1. Observe the setup 2. Identify all the connected equipments 3. Check the range of pressure gauge 4. Check the range of thermometer 5. Check the range of manometer 6. Open the steam supply valve for a short time 7. Measure the steam chest pressure (p 1 ) 8. Measure the steam outlet pressure (p 2 ) 9. Measure the outlet steam temperature Observation table: Sr. No. Parameters Reading 1 Boiler steam pressure, p 1 (bar) 2 Boiler steam temperature, ( C) 3 Water condensate formed in separating calorimeter, m w (kg) 4 Steam outlet pressure at throttling calorimeter, p 2 (bar) 5 Outlet steam temperature from throttling calorimeter, ( C) 6 Outlet steam pressure from throttling calorimeter, p 3 (bar) 7 Condensate collected at the throttling calorimeter, m s (kg)

9 Readings required from steam table: a. Enthalpy of feed water at state 2, : b. Enthalpy of wet steam at state 2, : c. Temperature of the output steam ( )) : d. Saturation temperature at p 3 : e. Degree of superheat: Outlet steam temperature Saturation temperature = ( - ) = f. Enthalpy of feed water at p 3 : g. Enthalpy of wet steam at p 3: h. Enthalpy of Superheated steam at p 3 : Calculations: 1. x1 = x1 = x1 = 2. = + + Cps ( - ) = 3. = = + * x2 = x2 = 4. = + * = Result: The dryness fraction of the sample taken from the main stream is.

10 Example: In a laboratory experiment, the following observations were taken with a separating and a throttling calorimeter to find the dryness fraction of steam: a. Total quantity of steam passed = 36 kg b. Water drained from separator = 1.8 kg c. Steam pressure before throttling = 12 bar d. Temperature of steam after throttling = 110 C e. Pressure after throttling = bar f. Specific heat of steam = 2.1 kj/kg K Estimate the quality of steam supplied. Solution: Given: Mass of steam supplied (ms +mw) = 36 kg Mass of water collected, mw = 1.8 kg Steam inlet pressure, p 1 = 12 bar Superheated steam temperature, = 110 C Pressure of steam at 2, p 2 = bar Specific heat of water, Cp = 2.1 kj/kg K Let, x1 = Dryness fraction for separating calorimeter, x2 = Dryness fraction for throttling calorimeter x = Actual dryness fraction entering the combined separating and throttling calorimeter From steam table, properties of steam a. = kj/kg b. = kj/kg c. Temperature of the output steam ( ) : 110 C d. Saturation temperature of the output steam ( ) at p 3: 100 C e. Enthalpy of feed water at p 3 ( ): kj /kg f. Enthalpy of wet steam at p 3 ( ): kj/kg g. Enthalpy of Superheated steam at p 3 : 2276 kj/kg

11 Calculations: 1. We know that mass of dry steam, ms = (ms +mw) - mw = = 34.2 kg = = x 1 2. = + + C ps ( - ) = * ( ) = 2697 kj /kg 3. = = + * x 2 = = = Actual dryness fraction of the steam entering the combined separating and throttling calorimeter, x = * x = 0.95 * x = Ans

### ENGINEERING COUNCIL CERTIFICATE LEVEL THERMODYNAMIC, FLUID AND PROCESS ENGINEERING C106 TUTORIAL 5 STEAM POWER CYCLES

ENGINEERING COUNCIL CERTIFICATE LEVEL THERMODYNAMIC, FLUID AND PROCESS ENGINEERING C106 TUTORIAL 5 STEAM POWER CYCLES When you have completed this tutorial you should be able to do the following. Explain

### Thermodynamics of Steam

Thermodynamics of Steam Learning Outcome When you complete this module you will be able to: Describe the principles of the thermodynamics of steam and the associated terms. Learning Objectives Here is

### Energy : ṁ (h i + ½V i 2 ) = ṁ(h e + ½V e 2 ) Due to high T take h from table A.7.1

6.33 In a jet engine a flow of air at 000 K, 00 kpa and 30 m/s enters a nozzle, as shown in Fig. P6.33, where the air exits at 850 K, 90 kpa. What is the exit velocity assuming no heat loss? C.V. nozzle.

### APPLIED THERMODYNAMICS TUTORIAL 1 REVISION OF ISENTROPIC EFFICIENCY ADVANCED STEAM CYCLES

APPLIED THERMODYNAMICS TUTORIAL 1 REVISION OF ISENTROPIC EFFICIENCY ADVANCED STEAM CYCLES INTRODUCTION This tutorial is designed for students wishing to extend their knowledge of thermodynamics to a more

### OUTCOME 4 STEAM AND GAS TURBINE POWER PLANT. TUTORIAL No. 8 STEAM CYCLES

UNIT 61: ENGINEERING THERMODYNAMICS Unit code: D/601/1410 QCF level: 5 Credit value: 15 OUTCOME 4 STEAM AND GAS TURBINE POWER PLANT TUTORIAL No. 8 STEAM CYCLES 4 Understand the operation of steam and gas

### PERFORMANCE ANALYSIS OF SEPARATING AND THROTTLING CALORIMETER TO DETERMINE QUALITY OF STEAM ON THERMAX DIESEL FIRED WATER TUBE BOILER

PERFORMANCE ANALYSIS OF SEPARATING AND THROTTLING CALORIMETER TO DETERMINE QUALITY OF STEAM ON THERMAX DIESEL FIRED WATER TUBE BOILER *R.G.Bodkhe, Y. Y. Nandurkar, S. S. Akant, S. L. Bankar *r_bodkhe@rediffmail.com

### 10/26/10. Steam Tables. Sections Steam table. Class 25

Sections 7.4-6 Steam table Class 25 Steam Tables Tables B-5,B-6, B-7 Superheated steam look at p. 650 Steam is NOT an ideal gas What s the reference state for steam? Liquid water is fed to a boiler at

### UNIT-III PROPERTIES OF PURE SUBSTANCE AND STEAM POWER CYCLE

UNIT-III PROPERTIES OF PURE SUBSTANCE AND STEAM POWER CYCLE Pure Substance A Pure substance is defined as a homogeneous material, which retains its chemical composition even though there may be a change

### Chapter 6 Energy Equation for a Control Volume

Chapter 6 Energy Equation for a Control Volume Conservation of Mass and the Control Volume Closed systems: The mass of the system remain constant during a process. Control volumes: Mass can cross the boundaries,

### ENGINEERING INFORMATION Hot water and steam service

ENGINEERING INFORMTION Hot water and steam service WHT IS STEM? Like other substances, water can exist in the form of a solid, when we call it ice; as a liquid when we call it water or as a gas when we

### CHAPTER SIX THE STEAM TABLES

CHAPTER SIX THE STEAM TABLES The steam tables are the most common list of thermodynamic properties. A set of steam tables are included in Appendix A. here we will explain the content of these tables and

Adca Training Part 1 This presentation is only a guideline, that can only be completed by a trained personnel. (This document s total or partial use and/or reproduction is only allowed if the reference

### PROPERTIES OF STEAM. 3.1 Introduction

3.1 Introduction 3 PROPERTIES OF STEAM Steam is a vapour. It is used as the working substance in the operation of steam engines and steam turbines. As stated in chapter 2, a vapour is a partially evaporated

### THERMODYNAMICS NOTES - BOOK 2 OF 2

THERMODYNAMICS & FLUIDS (Thermodynamics level 1\Thermo & Fluids Module -Thermo Book 2-Contents-December 07.doc) UFMEQU-20-1 THERMODYNAMICS NOTES - BOOK 2 OF 2 Students must read through these notes and

### THE UNIVERSITY OF TRINIDAD & TOBAGO

THE UNIVERSITY OF TRINIDAD & TOBAGO FINAL ASSESSMENT/EXAMINATIONS JANUARY/APRIL 2013 Course Code and Title: Programme: THRM3001 Engineering Thermodynamics II Bachelor of Applied Sciences Date and Time:

### Lesson 12 Multi-Stage Vapour Compression Refrigeration Systems. Version 1 ME, IIT Kharagpur 1

Lesson Multi-Stage Vapour Compression Refrigeration Systems Version ME, IIT Kharagpur The objectives of this lesson are to: Discuss limitations of single stage vapour compression refrigeration systems

### NOTES ON VAPOR-COMPRESSION REFRIGERATION

NOTES ON VAPOR-COMPRESSION REFRIGERATION Introduction The purpose of this lecture is to review some of the background material required to undertake the refrigeration laboratory. The Thermodynamics of

### DET: Mechanical Engineering Thermofluids (Higher)

DET: Mechanical Engineering Thermofluids (Higher) 6485 Spring 000 HIGHER STILL DET: Mechanical Engineering Thermofluids Higher Support Materials *+,-./ CONTENTS Section : Thermofluids (Higher) Student

### THERMODYNAMICS TUTORIAL 5 HEAT PUMPS AND REFRIGERATION. On completion of this tutorial you should be able to do the following.

THERMODYNAMICS TUTORIAL 5 HEAT PUMPS AND REFRIGERATION On completion of this tutorial you should be able to do the following. Discuss the merits of different refrigerants. Use thermodynamic tables for

### 20 m neon m propane 20

Problems with solutions:. A -m 3 tank is filled with a gas at room temperature 0 C and pressure 00 Kpa. How much mass is there if the gas is a) Air b) Neon, or c) Propane? Given: T73K; P00KPa; M air 9;

### - Know basic of refrigeration - Able to analyze the efficiency of refrigeration system -

Refrigeration cycle Objectives - Know basic of refrigeration - Able to analyze the efficiency of refrigeration system - contents Ideal Vapor-Compression Refrigeration Cycle Actual Vapor-Compression Refrigeration

### Thermodynamic cycle and machinery of. Vapor-compression. Heat pump and refrigerator

Thermodynamic cycle and machinery of Vapor-compression Heat pump and refrigerator Engineering Thermodynamics 2 Lab Report Maria Paknezhad March 15, 2006 Contents Introduction THEORY Ideal Vapor Compression

### Boiler Calculations. Helsinki University of Technology Department of Mechanical Engineering. Sebastian Teir, Antto Kulla

Helsinki University of Technology Department of Mechanical Engineering Energy Engineering and Environmental Protection Publications Steam Boiler Technology ebook Espoo 2002 Boiler Calculations Sebastian

### A basic introduction to steam

A basic introduction to steam FOR HOT, COLD, MOIST AND DRY, FOUR CHAMPIONS FIERCE. STRIVE HERE FOR MASTERY Milton 1666 Steam Wonderful Steam Very high heat content Recyclable Clean, non toxic Biodegradable

### Optimisation of Steam and Condensate Systems of Paper Machines

Optimisation of Steam and Condensate Systems of Paper Machines Name: Dr. Helmut Berger, Head of Department Energy and Environment Company: Allplan GmbH Address: Schwindgasse 10, 1040 Vienna Phone: 0043/1/5053707-94

### Module 5: Combustion Technology. Lecture 34: Calculation of calorific value of fuels

1 P age Module 5: Combustion Technology Lecture 34: Calculation of calorific value of fuels 2 P age Keywords : Gross calorific value, Net calorific value, enthalpy change, bomb calorimeter 5.3 Calculation

### Operating Instructions

Operating Instructions Determining Sample Composition Using Refractive Index (RI) Analysis Throughout this experiment, you will use a refractometer to determine compositions of water-eg mixtures. Concentration

### ES-7A Thermodynamics HW 3: 6-31, 38, 39, 53, 73, 76, 82, 86, 130, 131 Spring 2003 Page 1 of 5

ES-7A hermodynamics HW : 6-, 8, 9, 5, 7, 76, 8, 86, 0, Spring 00 Page of 5 6- Entropy change of water Given: Radiator with 0 L of superheated water vapor at 00 kpa and 00 C. Inlet and exit valves are closed,

### Measurement of Thermal Conductivity by Lee s method

Measurement of Thermal Conductivity by Lee s method Aim: To determine thermal conductivity of a bad conductor (glass) in form of a disc using Lee s method. Requisites: (1) Lee s apparatus and the experimental

### Lesson 10 Vapour Compression Refrigeration Systems

Lesson 0 Vapour Compression Refrigeration Systems Version ME, II Kharagpur he specific objectives of the lesson: his lesson discusses the most commonly used refrigeration system, ie Vapour compression

### Evaporation. Evaporation. Boiling. Saturation curve. Enthalpy. Steam tables and how to read them

dwells on the fundamental constituents of the universe, the forces they exert on one another, and the results produced by these forces. Here, we understand water molecules, what happens when they are heated,

### Lesson. 11 Vapour Compression Refrigeration Systems: Performance Aspects And Cycle Modifications. Version 1 ME, IIT Kharagpur 1

Lesson Vapour Compression Refrigeration Systems: Performance Aspects And Cycle Modifications Version ME, IIT Kharagpur The objectives of this lecture are to discuss. Performance aspects of SSS cycle and

### FE Thermodynamics Review. Dr. David R. Muñoz

FE Thermodynamics Review Dr. David R. Muñoz Definitions System collection of matter which we are interested in studying. Surroundings everything outside of the system. Universe system + surroundings. Pure

### 1.5. Which thermodynamic property is introduced using the zeroth law of thermodynamics?

CHAPTER INTRODUCTION AND BASIC PRINCIPLES. (Tutorial). Determine if the following properties of the system are intensive or extensive properties: Property Intensive Extensive Volume Density Conductivity

### Steam turbine power generation is a Rankine Cycle, best plotted on a temperature/entropy [T/s] diagram : Critical Point

Cogeneration Thermodynamics Revisited Dr Mike Inkson and Ben Misplon Thermal Energy Systems Abstract Whilst most engineers understand that higher HP steam conditions result in a more efficient power station,

### 6.5 Simple Vapor Compression Refrigeration System:

6.5 Simple Vapor ompression Refrigeration System: A simple vapor compression refrigeration system consists of the following equipments: i) ompressor ii) ondenser iii) Expansion valve iv) Evaporator. B

### SUPERHEATED STEAM. temperature of superheated steam taken, S sat = TABLE 25

SUPERHEATED STEAM Superheated steam, as already stated, is steam the temperature of which exceeds that of saturated steam at the same pressure. It is produced by the addition of heat to saturated steam

### Humidity in Spray Drying - Introduction

Humidity in Spray Drying - Introduction Drying sticky food powders such as nutritional powders and whey powders can be difficult because powder deposits build up on internal surfaces of the drying and

### Fundamentals of Refrigeration Part 5 Refrigerants

Refrigerants PH Chart Fundamentals of Refrigeration Part 5 Refrigerants This diagram shows a simplified pressure/ enthalpy chart for a non specified refrigerant. The area contained within the envelope

### Lesson 27 Psychrometry. Version 1 ME, IIT Kharagpur 1

1 Lesson 27 Psychrometry Version 1 ME, IIT Kharagpur 1 2 The specific objectives of this lecture are to: 1. Define psychrometry and the composition of moist air (Section 27.1) 2. Discuss the methods used

### 3. ENERGY PERFORMANCE ASSESSMENT OF COGENERATION AND TURBINES (GAS, STEAM)

3. ENERGY PERFORMANCE ASSESSMENT OF COGENERATION AND TURBINES (GAS, STEAM) 3.1 Introduction Cogeneration systems can be broadly classified as those using steam turbines, Gas turbines and DG sets. Steam

### 1. A belt pulley is 3 ft. in diameter and rotates at 250 rpm. The belt which is 5 ins. wide makes an angle of contact of 190 over the pulley.

Sample Questions REVISED FIRST CLASS PARTS A1, A2, AND A3 (NOTE: these questions are intended as representations of the style of questions that may appear on examinations. They are not intended as study

### SOLUTION MANUAL SI UNIT PROBLEMS CHAPTER 9 SONNTAG BORGNAKKE VAN WYLEN. FUNDAMENTALS of. Thermodynamics. Sixth Edition

SOLUTION MANUAL SI UNIT PROBLEMS CHAPTER 9 SONNTAG BORGNAKKE VAN WYLEN FUNDAMENTALS of Thermodynamics Sixth Edition CONTENT SUBSECTION PROB NO. Correspondence table Concept-Study Guide Problems -20 Steady

### C. starting positive displacement pumps with the discharge valve closed.

KNOWLEDGE: K1.04 [3.4/3.6] P78 The possibility of water hammer in a liquid system is minimized by... A. maintaining temperature above the saturation temperature. B. starting centrifugal pumps with the

### k is change in kinetic energy and E

Energy Balances on Closed Systems A system is closed if mass does not cross the system boundary during the period of time covered by energy balance. Energy balance for a closed system written between two

### Feedwater and Steam System Components

Helsinki University of Technology Department of Mechanical Engineering Energy Engineering and Environmental Protection Publications Steam Boiler Technology ebook Espoo 2002 Feedwater and Steam System Components

### Drying of Woody Biomass. Process Engineering / GEA Barr-Rosin

Drying of Woody Biomass BioPro Expo & Marketplace / Atlanta, GA / March 14-16, 2011 Drying of Woody Biomass Conventional Direct Fired Dryer Technology Proprietary work of the Copyright Owner Issues with

### Sheet 5:Chapter 5 5 1C Name four physical quantities that are conserved and two quantities that are not conserved during a process.

Thermo 1 (MEP 261) Thermodynamics An Engineering Approach Yunus A. Cengel & Michael A. Boles 7 th Edition, McGraw-Hill Companies, ISBN-978-0-07-352932-5, 2008 Sheet 5:Chapter 5 5 1C Name four physical

### The thermal cycle with the highest theoretical efficiency is the Carnot cycle, consisting of two isotherms and two adiabats.

Carnot to Reality The thermal cycle with the highest theoretical efficiency is the Carnot cycle, consisting of two isotherms and two adiabats. η = W/Q in Carnot's theorem states that for a heat engine

### Thermodynamic Properties of Steam

Thermodynamic Properties of Steam Purpose of This Session The purpose of this session is to understand the thermodynamic properties of steam, which affect the design and operation of steam heating systems

### An analysis of a thermal power plant working on a Rankine cycle: A theoretical investigation

An analysis of a thermal power plant working on a Rankine cycle: A theoretical investigation R K Kapooria Department of Mechanical Engineering, BRCM College of Engineering & Technology, Bahal (Haryana)

### GUIDANCE NOTE 16 CHP SCHEME HEAT OUTPUTS

GUIDANCE NOTE 16 CHP SCHEME HEAT OUTPUTS Qualifying Heat Outputs GN16.1 CHP Qualifying Heat Output (CHP QHO ) is the registered amount of useful heat supplied annually from a CHP Scheme (MWh th ). It is

### Chapter 4.9: HVAC System

Chapter 4.9: HVAC System Short type questions 1. Define tons of refrigeration?. One ton of refrigeration is the amount of cooling obtained by one ton of ice melting in one day. It is equivalent to kcal/h,

### Pressure Enthalpy Explained

Pressure Enthalpy Explained Within the new F Gas course the requirement for an understanding of Pressure Enthalpy (Ph) graphs is proving to be a large learning curve for those who have not come across

### Entropy of a liquid or a solid

Entropy of a liquid or a solid 8.75 A piston cylinder has constant pressure of 000 kpa with water at 0 o C. It is now heated up to 00 o C. Find the heat transfer and the entropy change using the steam

### GESTRA Information C 1.2

Issue Date: 9/06 GESTRA Steam Systems GESTRA Information C 1.2 Utilization of the sensible heat of condensate Examples of effective steam plant design and appropriate equipment 1 Heat content of saturated

### ECH 4224L Unit Operations Lab I Thin Film Evaporator. Introduction. Objective

Introduction In this experiment, you will use thin-film evaporator (TFE) to separate a mixture of water and ethylene glycol (EG). In a TFE a mixture of two fluids runs down a heated inner wall of a cylindrical

### COST EFFECTIVE SMALL SCALE ORC SYSTEMS FOR POWER RECOVERY FROM LOW ENTHALPY GEOTHERMAL RESOURCES

COST EFFECTIVE SMALL SCALE ORC SYSTEMS FOR POWER RECOVERY FROM LOW ENTHALPY GEOTHERMAL RESOURCES I K Smith *, N Stosic *, A Kovacevic *, R Langson ** * City University, London, UK ** ElectraTherm Inc,

### UNIT 2 REFRIGERATION CYCLE

UNIT 2 REFRIGERATION CYCLE Refrigeration Cycle Structure 2. Introduction Objectives 2.2 Vapour Compression Cycle 2.2. Simple Vapour Compression Refrigeration Cycle 2.2.2 Theoretical Vapour Compression

### Chapter 3.4: HVAC & Refrigeration System

Chapter 3.4: HVAC & Refrigeration System Part I: Objective type questions and answers 1. One ton of refrigeration (TR) is equal to. a) Kcal/h b) 3.51 kw c) 120oo BTU/h d) all 2. The driving force for refrigeration

### VTU-NPTEL-NMEICT Project

VTU-NPTEL-NMEICT Project Progress Report The Project on Development of Remaining Three Quadrants to NPTEL Phase-I under grant in aid NMEICT, MHRD, New Delhi SME Name : Course Name: Type of the Course Module

### Working Principle of Proposed System (Figure 3)

EXPERIMENTATION FOR DESIGN AND FABRICATION OF SCALED DOWN INNOVATIVE MINI ICE PLANT U.S. Wankhede Department of Mechanical Engineering, G. H. Raisoni College of Engineering, Digdoh Hills, Hingna Road,

### Waste Heat Recovery through Air Conditioning System

International Journal of Engineering Research and Development e-issn: 2278-067X, p-issn : 2278-800X, www.ijerd.com Volume 5, Issue 3 (December 2012), PP. 87-92 Waste Heat Recovery through Air Conditioning

### QUESTIONS THERMODYNAMICS PRACTICE PROBLEMS FOR NON-TECHNICAL MAJORS. Thermodynamic Properties

QUESTIONS THERMODYNAMICS PRACTICE PROBLEMS FOR NON-TECHNICAL MAJORS Thermodynamic Properties 1. If an object has a weight of 10 lbf on the moon, what would the same object weigh on Jupiter? ft ft -ft g

### 28) PRESERVATION OF FRUITS AND VEGETABLES BY DRYING

28) PRESERVATION OF FRUITS AND VEGETABLES BY DRYING II HUMIDITY AND HUMIDITY CHARTS VAPOR PRESSURE OF WATER In a number of the unit operations and transport processes it is necessary to make calculations

### Lesson 5 Review of fundamental principles Thermodynamics : Part II

Lesson 5 Review of fundamental principles Thermodynamics : Part II Version ME, IIT Kharagpur .The specific objectives are to:. State principles of evaluating thermodynamic properties of pure substances

### STEAM TURBINE 1 CONTENT. Chapter Description Page. V. Steam Process in Steam Turbine 6. VI. Exhaust Steam Conditions, Extraction and Admission 7

STEAM TURBINE 1 CONTENT Chapter Description Page I Purpose 2 II Steam Turbine Types 2 2.1. Impulse Turbine 2 2.2. Reaction Turbine 2 III Steam Turbine Operating Range 2 3.1. Curtis 2 3.2. Rateau 2 3.3.

### Thermodynamics - Example Problems Problems and Solutions

Thermodynamics - Example Problems Problems and Solutions 1 Examining a Power Plant Consider a power plant. At point 1 the working gas has a temperature of T = 25 C. The pressure is 1bar and the mass flow

### There is no general classification of thermodynamic cycle. The following types will be discussed as those are included in the course curriculum

THERMODYNAMIC CYCLES There is no general classification of thermodynamic cycle. The following types will be discussed as those are included in the course curriculum 1. Gas power cycles a) Carnot cycle

### Refrigeration and Air Conditioning Prof. M. Ramgopal Department of Mechanical Engineering Indian Institute of Technology, Kharagpur

Refrigeration and Air Conditioning Prof. M. Ramgopal Department of Mechanical Engineering Indian Institute of Technology, Kharagpur Lecture No. # 18 Worked Out Examples - I Welcome back in this lecture.

### 1. Why are the back work ratios relatively high in gas turbine engines? 2. What 4 processes make up the simple ideal Brayton cycle?

1. Why are the back work ratios relatively high in gas turbine engines? 2. What 4 processes make up the simple ideal Brayton cycle? 3. For fixed maximum and minimum temperatures, what are the effect of

### 4.1 Introduction. 4.2 Work. Thermodynamics

Thermodynamics 4-1 Chapter 4 Thermodynamics 4.1 Introduction This chapter focusses on the turbine cycle:thermodynamics and heat engines. The objective is to provide enough understanding of the turbine

### Psychrometric Chart Structure and Application

Psychrometric Chart Structure and Application Building Technologies s Contents 1. Thermodynamic fundamentals 2. Structure of the psychrometric chart 3. Application of the psychrometric chart 4. Calculation

### UNIT 5 REFRIGERATION SYSTEMS

UNIT REFRIGERATION SYSTEMS Refrigeration Systems Structure. Introduction Objectives. Vapour Compression Systems. Carnot Vapour Compression Systems. Limitations of Carnot Vapour Compression Systems with

### Air Water Vapor Mixtures: Psychrometrics. Leon R. Glicksman c 1996, 2010

Air Water Vapor Mixtures: Psychrometrics Leon R. Glicksman c 1996, 2010 Introduction To establish proper comfort conditions within a building space, the designer must consider the air temperature and the

### Written by Withit Chatlatanagulchai Revision 1.0, June Laboratory Instruction Manual for Boiler Simulation Unit

Laboratory Manual for Boiler Simulation Unit 1 Table of Contents 1 Introduction...3 2 Unit Layout...5 2.1 Level Monitoring Sensors...6 2.2 Pressure Sensors...7 2.3 Burner Unit...8 2.4 Switch Cabinet...9

### Practice Problems on Conservation of Energy. heat loss of 50,000 kj/hr. house maintained at 22 C

COE_10 A passive solar house that is losing heat to the outdoors at an average rate of 50,000 kj/hr is maintained at 22 C at all times during a winter night for 10 hr. The house is to be heated by 50 glass

### Training programme on Energy Efficient technologies for climate change mitigation in Southeast Asia. Cogeneration

Training programme on Energy Efficient technologies for climate change mitigation in Southeast Asia Cogeneration Session Agenda: Cogeneration Introduction Types of steam turbine cogeneration system Types

### Temperature and Phase Changes Answers

Temperature and Phase Changes Answers Part A 1. a) How many kilojoules are required to heat 0.200 kg of Au from 15.0 C to 85.0 C? Q mc T Q (0.200 kg)(0.129 kj/kg C)(70.0 C) 1.81 kj b) If the same amount

### Name Lab Partner Section

HUMIDITY Name Lab Partner Section Introduction Humidity refers to the amount of moisture or water vapor in the air. The air in our environment is rarely ever dry. The amount of moisture present varies

### Application - problems. MEL 806 Thermal System Simulation (2-0-2) Dr. Prabal Talukdar Associate Professor Department of Mechanical Engineering

Application - problems MEL 806 Thermal System Simulation (2-0-2) Dr. Prabal Talukdar Associate Professor Department of Mechanical Engineering IIT Delhi Gas vapour mixture Taking 0 C as the reference temperature,

### A CASE STUDY: PERFORMANCE AND ACCEPTANCE TEST OF A POWER AND DESALINATION PLANT. Keywords : Power Plant, Boiler Capacity, Electrical Power

A CASE STUDY: PERFORMANCE AND ACCEPTANCE TEST OF A POWER AND DESALINATION PLANT Atef M Al Baghdadi Water and Electricity Authority Abu Dhabi, U.A.E Keywords : Power Plant, Boiler Capacity, Electrical Power

### POSSIBILITY FOR MECHANICAL VAPOR RE-COMPRESSRION FOR STEAM BASED DRYING PROCESSES

POSSIBILITY FOR MECHANICAL VAPOR RE-COMPRESSRION FOR STEAM BASED DRYING PROCESSES M. Bantle 1, I. Tolstorebrov, T. M. Eikevik 2 1 Department of Energy Efficiency, SINTEF Energy Research, Trondheim, Norway,

### THEORETICAL ANALYSIS OF THE PERFORMANCE OF DUAL PRESSURE CONDENSER IN A THERMAL POWER PLANT

INTERNATIONAL JOURNAL OF MECHANICAL ENGINEERING AND TECHNOLOGY (IJMET) International Journal of Mechanical Engineering and Technology (IJMET), ISSN 0976 6340(Print), ISSN 0976 6340 (Print) ISSN 0976 6359

### Understanding of OHDR and its repercussions on over all operation of super critical power plant

Understanding of OHDR and its repercussions on over all operation of super critical power plant Rahul Pachouri M.Tech.Student (Thermal System & Design) SRCEM,MORENA (M.P) India Devendra Dandotiya Assistant

### UNIT-I. Workdone = HS HR 8.2 = 25 HR. Heat rejected (HR) = 20.8 kw ----

UNIT-I. A heat engine receives heat at the rate of 500 kj/min and gives an output of 8.2 kw. Determine the thermal efficiency and the rate of heat rejection. Given: Heat received by engine (HS) Work output

### δq T +S i S e +S gen S 2 S 1 = δq T + S i S 0 S 2 S 1 = 0 m A (s 2 s 1 ) A +m B (s 2 s 1 ) B = 0 TB,2 + Rln v A,1 +ln TB,1 0 = m A C v ln +C v ln

Two rigid tanks each contain 0 of N 2 gas at 000 K, 500 kpa. They are now thermally connected to a reversible heat pump, which heats one and cools the other with no heat transfer to the surroundings. When

### OUTCOME 2 INTERNAL COMBUSTION ENGINE PERFORMANCE. TUTORIAL No. 5 PERFORMANCE CHARACTERISTICS

UNIT 61: ENGINEERING THERMODYNAMICS Unit code: D/601/1410 QCF level: 5 Credit value: 15 OUTCOME 2 INTERNAL COMBUSTION ENGINE PERFORMANCE TUTORIAL No. 5 PERFORMANCE CHARACTERISTICS 2 Be able to evaluate

### Science and Reactor Fundamentals Heat &Thermodynamics 1 CNSC Policy Planning and Learning HEAT AND THERMODYNAMICS

Science and Reactor Fundamentals Heat &Thermodynamics 1 HEAT AND THERMODYNAMICS Science and Reactor Fundamentals Heat and Thermodynamics 2 SNCS Policy Planning and Learning TABLE OF CONTENTS 1 OBJECTIVES...4

### THE APPLICATION OF STEAM QUALITY TEST LIMITS

Steam Q/1 THE APPLICATION OF STEAM QUALITY TEST LIMITS Keith Shuttleworth Senior Consultant Keith Shuttleworth & Associates Ltd Keith Shuttleworth & Associates Laburnum Cottage Eastbury Hungerford Berkshire

TABLE OF CONTENTS Page List of Figures... ii Enabling Objectives... iii Introduction... 2-1 Water and Steam Properties... 2-1 Steam Tables... 2-1 Saturation Temperature and Pressure... 2-1 Specific Volume

### Performance of the Boiler and To Improving the Boiler Efficiency Using Cfd Modeling

IOSR Journal of Mechanical and Civil Engineering (IOSR-JMCE) e-issn: 2278-1684,p-ISSN: 2320-334X, Volume 8, Issue 6 (Sep. - Oct. 2013), PP 25-29 Performance of the Boiler and To Improving the Boiler Efficiency

### Refrigeration and Air-Conditioning Prof. M. Ramgopal Department of Mechanical Engineering Indian Institute of Technology, Kharagpur. Lecture No.

Refrigeration and Air-Conditioning Prof. M. Ramgopal Department of Mechanical Engineering Indian Institute of Technology, Kharagpur Lecture No. # 16 Vapour Absorption Refrigeration systems (contd.) Welcome

### CHEPTER-3 EFFECT OF OPERATING VARIABLES ON THERMAL EFFICIENCY OF COMBINED CYCLE POWER PLANT

CHEPTER-3 EFFECT OF OPERATING VARIABLES ON THERMAL EFFICIENCY OF COMBINED CYCLE POWER PLANT 3.1 THERMAL EFFICIENCY OF THE COMBINED CYCLE: - In combined cycle power plants if power in gas turbine and steam

### LECTURE 1 THE CONTENTS OF THIS LECTURE ARE AS FOLLOWS: 1.0 INTRODUCTION 2.0 VARIOUS TERMINOLOGY USED IN PSYCHROMETRY. 2.

LECTURE 1 THE CONTENTS OF THIS LECTURE ARE AS FOLLOWS: 1.0 INTRODUCTION 2.0 VARIOUS TERMINOLOGY USED IN PSYCHROMETRY 2.1 Vapour Pressure 2.2 Saturation Vapour Pressure (SVP) 2.3 Gas Constant of Unsaturated

### 1 High Pressure Steam Turbine... 2 1.1 Sliding Pressure Control... 2 1.1.1 Input for the design load calculation [HD glijdruk 100]... 2 1.1.

High Pressure Steam Turbine.... Sliding Pressure Control..... Input for the design load calculation [HD glijdruk 00]..... Results of the design load calculation..... h-s diagram..... Input for the off-design

### LABORATORY MANUAL STEAM & POWER GENERATION LAB ME-218-F. BRCM College of Engineering & Technology Bahal

LABORATORY MANUAL STEAM & POWER GENERATION LAB ME-218-F BRCM College of Engineering & Technology Bahal 0 ME-218-F STEAM & POWER GENERATION LAB L T P Sessional: 25 Marks - - 2 Practical: 25Marks Total:

### LECTURE-16. Simple Vapour Compression Refrigeration. Advantage and disadvantages of vapour compression and air refrigeration system:

Lecturer: -Dr. Esam Mejbil Abid Subject: Air Conditioning and Refrigeration Year: Fourth B.Sc. Babylon University College of Engineering Department of Mechanical Engineering LECTURE-16 Simple Vapour Compression