1 Performing Oxygen Radical Absorbance Capacity Assays with Synergy HT ORAC Antioxidant Tests Introduction The antioxidant capabilities of foods, cosmetics, supplements and pharmaceutical agents has become of particular interest. This is the result of the evidence demonstrating the relationship of reactive oxygen/nitrogen species (ROS) with aging and pathogenesis. [1,2]. Living organisms: use of a fuel (ATP) whose production results in the formation of toxic compounds, and requires that a balance be maintained between the oxidants and the antioxidants. ROS (reactive oxygen species) are generally held in check by a combination of antioxidant enzymes, proteins, and antioxidants provided by the diet. A breakdown or reduction in antioxidant capability has been associated with a number of chronic diseases. Conceivably, the ingestion of foods or supplements that contain antioxidant activity could provide some protection towards this affect. Figure 1. Illustration of Antioxidant activity determination expressed as the net area under the curve (AUC). The AUC for standards, samples and blanks are calculated and the blank value subtracted. A KC4 kinetic profile of a blank well and an experimental well has been modified to highlight the Net AUC as described by Huang et al. . All living organisms are exposed to ROS on a continual basis. Besides exogenous sources, ROS are formed in vivo by several different mechanisms. The interaction of ionizing radiation with biological molecules can result in free radical formation. ROS are also the byproduct of normal cellular respiration. Electrons passing through the respiratory chain often leak away and BioTek Instruments, Inc., P.O. Box 998, Highland Park, Winooski, Vermont USA COPYRIGHT 26 TEL: FAX: Outside the USA:
2 directly reduce oxygen molecules to the superoxide anion. In addition, several dedicated enzymes in phagocytic cells (e.g. neutrophils and macrophages) synthesize ROS as part of the normal cellular function. ROS molecules fall into two categories: ROS molecules that contain unpaired electrons (e.g. superoxide anion, O 2 - ; hydroxyl ion, OH -, and the hydroxyl radical, OH) and react directly with biological molecules, and molecules that have the ability to pull electrons from other molecules (H 2 O 2 or HOCl) and can damage biomolecules through direct interaction or initiate a chain reaction where ROS species are transferred from molecule to molecule. Figure 2. Schematic depiction of the ORAC assay. Reactive oxygen species generators are added to parallel reactions that contain equal amounts of a fluorescent probe. Reactions contain either a buffer blank or antioxidant samples and standards. The antioxidant capacity of a sample is the net difference between the area under the curve (AUC) of the sample and that of the blank. Figure adapted from Huang et al. . There are several different methods to measure total antioxidant capacity described in the literature (3-7). The total peroxyl radical trapping parameter assays decribed by Wayner et al.  was widely used in the 198s and early 9s. This method was replaced by more precise methods such as the ferric reducing ability of plasma (FRAP) assay  and the Trolox equivalent antioxidant capacity assay (TEAC) . The TEAC assay is based on the inhibition by antioxidants of the absorbance of the cation of 2,2 -azinobis(3-ethylbenzothiazoline-6-sulfonate) (ABTS). ABTS is also a common substrate for absorbance based ELISA. The use of a fluorescent compound to measure antioxidant capability such as phycoerythrin increased the sensitivity , but suffers due to the inherent instability of the compound over time. The oxygen radical absorbance capacity (ORAC) assay replaces phycoerythrin with fluorescein, which is much more stable over the time frame of the assay . This assay has been automated  and over time converted to a microplate format . Ronald Prior at Brunswick Laboratories and his colleagues have accomplished much of the development of the ORAC assay. The ORAC assay depends on the free radical damage to a fluorescent probe, such as fluorescein, to result in a downward change of fluorescent intensity . The assumption, of course, is that the degree of change is indicative of the amount of radical damage. The presence of antioxidants results in an inhibition in the free radical damage to the fluorescent compound. This inhibition is observed as a preservation of the fluorescent signal. Reactions containing antioxidants and or blanks are run in parallel using equivalent amounts of a ROS
3 generator and fluorescent probe (Figure 1). Because the reaction is driven to completion, one can quantitate the protection by calculating the area under the curve (AUC) from the experimental sample. After subtracting the AUC for the blank, the resultant difference would be the protection conferred by the antioxidant compound (Figure 2). Comparison to a set of known standards allows one to calculate equivalents and compare results from different samples and experiments. Typically Trolox, (6-hydroxy-2,5,7,8-tetrametmethylchroman-2-carboxylic acid) a water soluble vitamin E analog, is used as the calibration standard and ORAC results are expressed as Trolox equivalents (Figure 3). The ORAC assay is unique in that because the assay is driven to completion the AUC calculation combines both the inhibition time as well inhibition percentage of free radical damage by the antioxidant into a single quantity. Standardization of the assay with the use of a common calibrator in conjunction with an assay that can be performed easily on many different compounds, foods, and materials has allowed for an easy comparison of antioxidant capabilities of many different materials and the formation of a database . Materials and Methods Sodium Fluorescein was purchased from Invitrogen. 2,2 -Azobis(2-amidinopropane) dihydrochloride (AAPH), 6-hydroxy-2,5,7,8-tetramethylchroman-2-carboxylic acid (Trolox ), gallic acid, Epigallocatechin gallate (EGCG), epigallocatechin (EGC), and quercetin dihydrate were purchased from Sigma-Aldrich (St. Louis, Mo). Black-sided, special optics clear bottom plates (part # 3615) were obtained from Corning. The ORAC assay was performed essentially as described by Huang et. al . Briefly, AAPH (.414g) was dissolved in 1 ml of 75 mm phosphate buffer (ph 7.4) to a final concentration of 153 mm and made fresh daily. A fluorescein stock solution (4 x 1-3 mm) was made in 75 mm phosphate buffer (ph 7.4) and stored wrapped in foil at 5 C. Immediately prior to use, the stock solution was diluted 1:1 with 75 mm phosphate buffer (ph 7.4). The diluted sodium fluorescein solution was made fresh daily. In regards to the plate usage, the exterior wells were not used for experimental determinations (Figure 3). These wells were filed with 3 µl of water, while the interior wells were used for experimental determinations. To all experimental wells, 15 µl of working sodium fluorescein solution was added. In addition blank wells received 25 µl of 75 mm phosphate buffer (ph 7.4), while standards received 25 µl of Trolox dilution and samples received 25 µl of sample. The plate was then allowed to equilibrate by incubating for a minimum of 3 minutes in the Synergy HT Multi-Detection Microplate Reader (BioTek Instruments, Inc., Winooski, VT) at 37 C. Reactions were initiated by the addition of 25 µl of AAPH solution using the microplate reader s injector for a final reaction volume of 2 µl. The fluorescence was then monitored kinetically with data taken every minute. A Synergy HT Multi-Detection Microplate Reader with injectors was used with a 485 nm, 2 nm bandpass, excitation filter and a 528 nm, 2 nm bandpass, emission filter. The plate reader was controlled by KC4 version 3.4 (revision 1). Reactions were initiated by the addition of 25 µl of AAPH reagent (153 mm) followed by shaking at maximum intensity for 1 seconds. The fluorescence of each well was then measured from the bottom every 6 seconds at a sensitivity setting of 6. ORAC values were calculated as described by Cao and Prior . The AUC and the Net AUC of the standards and samples were determined using KC4 Data Reduction Software (BioTek Instruments, Winooski, VT) using equations 1 and 2 respectively. AUC =.5+(R2/R1)+(R3/R1)+(R4/R1)+.+.5(Rn/R1) (Eq. 1) Where R1 is the fluorescence reading at the initiation of the reaction and Rn is the last measurement. Net AUC = AUC sample AUC blank (Eq. 2.)
4 The standard curve was obtained by plotting the Net AUC of different Trolox concentrations against their concentration. ORAC values of samples were then calculated automatically using the KC4 software to interpolate the sample s Net AUC values against the Trolox standard curve. Figure 3. Plate map of wells typically used. Outer wells (blue-filled) were filled with 3 µl of water to provide large thermal mass. Only the inner 6 wells were used for experimental determinations Results Initial experiments examined the effect of AAPH concentration on the kinetics of fluorescence decay of fluorescein in solution. As demonstrated in Figure 4, the presence of AAPH results in a marked loss of fluorescence of sodium fluorescein over time. While samples that do not contain AAPH ( mm) are relatively stable over a period of 12 minutes, a complete loss of fluorescence was observed with AAPH concentrations greater than 28 mm. In addition the effect observed appeared to be graded on the basis of AAPH concentration. Fluorescence AAPH Kinetics mm mm mm mm mm mm Time (min) Figure 4. AAPH Induced Decay of Fluorescence. Various concentrations of AAPH were incubated with sodium fluorescein and the fluorescence was monitored kinetically for 12 minutes. Data was captured using KC4 data reduction software and exported to and plotted using GraphPad software.
5 The loss in fluorescence can be assessed by measuring the area under the curve (AUC) of the kinetic plot for each concentration. The greater the extent of florescent decay, the smaller the expected AUC value would be. When the AUC for each concentration is plotted against AAPH concentration a linear relationship is observed (Figure 5). A 4-parameter logistic fit of the data was used to describe these data. These data confirmed that in our hands a final concentration of AAPH of 19 mm, as described by Prior and his colleagues, was adequate for ORAC determinations of compounds. A final concentration of 19 mm (25 µl of a 153 mm stock solution in a final assay volume of 2 µl) was used for all subsequent ORAC determinations. Figure 5. AAPH Concentration Curve. Various concentrations of the ROS generator AAPH were incubated with constant amounts of sodium fluorescein in solution. Total area under the curve (AUC) for each well and the subsequent result plotted was against its concentration using KC4 data reduction software. A 4-parameter logistics fit was used to describe the data. The data presented in Figure 6 demonstrate the ability of the Synergy HT reader to perform the ORAC assay and measure antioxidant activity of samples. The kinetic curves of several different concentrations of Trolox standard demonstrates varying amounts of protection of fluorescein against oxidation that results in the loss of fluorescence. The highest concentration tested (1 µm) provided virtually full protection for approximately 2 minutes, before fluorescence intensity began to diminish, while the lowest concentration tested (6.25 µm) provided only slight protection above the buffer only control.
6 Figure 6. Trolox Kinetic Curves. Representative kinetic curves from an ORAC assay of Trolox antioxidant standards ranging from to 1 µm were plotted using GraphPad software. The reactions were initiated by the addition of 25 µl of AAPH (153 mm) solution and the fluorescence monitored every minute using a Synergy HT as previously described. When net AUC are calculated from these kinetic curves and plotted against Trolox Concentration a linear relationship is observed (Figure 7). Using a least means squared linear regression analysis a correlation coefficient (R 2 ) of.9987 was obtained. As a result, the determination Trolox equivalents of unknown samples can be made with confidence. Figure 7. Typical Antioxidant Standard Curve. The net AUC of different Trolox standards are plotted as a function of concentration. The subsequent calibration curve is then used to interpolate the antioxidant capacity of unknown samples. The standard curve can then be interpolated to quantitate unknown samples. The resultant determinations are expressed as Trolox equivalents. Several compounds known to have antioxidant properties were assayed using the ORAC assay as previously described. As depicted in Figure 8, all of these compounds show a significant linear concentration dependent antioxidant activity. The specificity of the assay is demonstrated by the lack of any response from Tris buffer (Figure 8C). Grape juice samples have also been measured for ORAC activity using the Synergy HT. Figure 9 demonstrates the significant antioxidant activity found in grape juice, where 1:1 diluted samples exhibit ORAC activity equivalent to 4 µm Trolox. In addition, these data demonstrate the repeatability of the Synergy HT. In these experiments, 5 separate 1:1 dilutions of two different grape juice samples were assayed for ORAC in triplicate. The mean of each determination is in very close agreement with the other corresponding measurement.
7 A Gallic Acid Dose Response Curves B Quercetin C Tris D Myricetin E EGC F EGCG Figure 8. Antioxidant Dose Response Curves. The ORAC of several different known antioxidant compounds, as well as Tris buffer, were measured as described previously and their Net AUC plotted against their concentration.
8 Sample Repeatabilty 6 Trolox Equivalance (µm) Sample 1 Sample 2 Grape Juice Sample Figure 9. Sample Repeatability. Two different grape juice samples were diluted 1:1 with phosphate buffer and assayed for antioxidant capability using the ORAC method. Five discrete samples were diluted separately and assayed in triplicate. Each data point represents the mean of the three determinations. Discussion The ORAC assay in our hands is very sensitive to slight temperature gradations. Despite extreme efforts, an edge effect was observed when the outside wells were used. Edge effects have been reported in any number of different types of microplate-based assays. Because we used the reader s reagent injectors to initiate the assay, plate lids were not used. In these experiments, the tighter control between reading the wells and the initiation of the assay by the addition of AAPH was of greater value than the throughput. HTS type experiments where sample throughput is of greater importance would be served better by plate covers, which would alleviate much of the edge-effect while allowing for an increase in the number of available wells. We found that filling the outer wells with 3 µl of water resulted in more consistent data from the interior wells. Besides avoiding the use of the outer wells, where most of the discrepancies were found, the filled wells served as a significant heat mass that eliminated any temperature fluctuations. In addition, we found that reading the plate from the bottom resulted in more consistent results. The Synergy HT with injectors is an ideal instrument to perform ORAC assays. The Synergy HT Multi-Detection Microplate Reader is a robotic compatible microplate reader that can measure absorbance, fluorescence, and luminescence. The Synergy HT utilizes a unique dual optics design. The fluorescence optics is capable of measuring all plate formats up to 384- well plates. The top probe adjusts up and down automatically via software accommodating different plate heights and different sized probes are available meet specific applications. When the reader is in conventional fluorescence mode, it uses a tungsten-halogen lamp as a light source and band-pass filters in a filter wheel cartridge to provide wavelength specificity and a PMT for detection. Besides special optics, the Synergy HT has numerous features that enhance the reader s capability. Elevated temperatures are regulated by a 4-Zone microprocessor controlled system that assures superior temperature uniformity up to 5 C. The twin injectors module allows for the controlled addition of reagents to wells while the plate is in the reading chamber. This allows for precise timing between the initiation of reactions and their measurement. With a compact footprint, a robot friendly carrier design and RS-232 connection, the Synergy HT is also compatible with many of the commonly preferred robotics systems. The functionality of the Synergy HT is greatly enhanced by its controlling data reduction
9 software, KC4. In these experiments, KC4 was used to control reader function, capture the raw data to a PC, as well as perform all the necessary Net AUC calculations, plot the Trolox standard curve and interpolate it to determine sample ORAC values. Using true Object Linking and Embedding (OLE) technology, KC4 can interact directly with Microsoft Excel and Microsoft Word programs, providing total control over all report formatting using data objects created by KC4. This allows the user to create custom, publication quality reports of microplate applications. An upgrade to CFR 21 Part 11 compliant software is also available. Abbreviations ORAC Oxygen radical absorbance capacity; ROS, reactive oxygen species; AAPH, 2,2 - azobis(2-amidopropane) dihydrochloride; EGC, epigallocatechin; EGCG, epigallocatechin gallate; AUC, area under the curve; Trolox, 6-hydroxy-2,5,7,8-tetramethyl-2-carboxylic acid; TEAC, Trolox equivalent antioxidant capacity; FRAP, ferric reducing antioxidant power. References 1. Halliwell, B., Aruoma, O. (1991) DNA Damage by Oxygen Derived Species. Its Mechanisms and Measurement in Mammalian Systems. FEBS Lett. 281: Ames, B.N., Shigenaga, M.K., and Hagen, T.M. (1993) Oxidants, Antioxidants and the Degenerative Diseases of Aging. Proc. Natl. Acad. Sci. USA 9: Benzie, IFF, Strain, JJ. (1996) The Ferric Reducing Ability of Plasma (FRAP) as a Measure of Antioxidant Power : The FRAP Assay. Anal. Biochem. 238: Rice-Evans, C. and Miller, NJ. (1994) Total Antioxidant Status in Plasma and Body Fluids. Methods Enzymol. 234: Wayner, DDM, Burton, Gw., Ingold, KU., and Locke, S. (1985) Quantitative Measurement of the Total, Peroxyl Radical-trapping Antioxidant Capacity of Human Blood Plasma by Controlled Peroxidation. FEBS Lett. 187: Glazer, AN., (199) Phycoerythrin Fluorescence-based Assay for Reactive Oxygen Species. Methods Enzymol 186: Cao, GH., Alessio, HM.and Cutler, RG., (1993) Oxygen-radical Absorbance Capacity Assay for Antioxidants. Free Radical Biol. Med. 14: Cao, G., Verdon, C., Wu, A., Wang, H., and Prior, R. (1995) Automated Assay of Oxygen Radical Absorbance Capacity with the COBRAS FARA II. Clin. Chem. 41: Huang, D., Ou, B., Hampsch-Woodill, M., Flanagan, J., and Prior, R. (22) Highthroughput Assay of Oxygen Radical Absorbance Capacity (ORAC) Using a Multichannel Liquid Handling System Coupled with a Microplate Fluorescence Reader in 96-Well Format. J. Agric. Food Chem. 5: Ou, B., Hampsch-Woodill, and Prior, R. (21) Development and Validation of an Improved Oxygen Radical Absorbance Capacity Assay Using Fluorescein as the Fluorescent Probe. J. Agric Food Chem. 49: Wu, X., Gu, L., Holden, J., Haytowitz, D., Gebhardt, S., Beecher, G., and Prior, R. (24) Development of a Database for Total Antioxidant Capacity in Foods: a Preliminary Study. J. Food Composition and Analysis. 17: Cao, G., and Prior, R. (1999) Measurement of Oxygen Radical Absorbance Capacity in Biological samples. Oxidants and Antioxidants. Methods Enzymol. 299:5-62. Paul Held Ph.D. Senior Scientist & Applications Lab Manager Trolox is a registered trademark if Hottman-LaRoche. Rev. 8/15/5
Determination of NADH Concentrations with the Synergy 2 Multi-Detection Microplate Reader using Fluorescence or Absorbance Do you know what your redox state is? Many biological assays have as their basis
Antioxidant Assay Kit Catalog Number KA1622 100 assays Version: 02 Intended for research use only www.abnova.com Table of Contents Introduction... 3 Intended Use... 3 Principle of the Assay... 3 General
Quantitation of Peptides and Amino Acids with a Synergy HT using UV Fluorescence Introduction Eukaryotic and prokaryotic cells contain a number of compounds that are fluorescent with UV light excitation.
Use of the Spectral Scanning Capabilities of the PowerWave Microplate Spectrophotometer Spectral analysis of samples can be used for a number of different requirements, including contaminant detection,
Fluorometric Quantitation of Protein using the Reactive Compound Fluorescamine Total protein content is a measurement common to many applications in basic science and clinical research. Several different
Product Manual OxiSelect Total Antioxidant Capacity (TAC) Assay Kit Catalog Number STA-360 200 assays FOR RESEARCH USE ONLY Not for use in diagnostic procedures Introduction Oxidative stress is a physiological
Hydrogen Peroxide Cell-Based Assay Kit Item No. 600050 www.caymanchem.com Customer Service 800.364.9897 Technical Support 888.526.5351 1180 E. Ellsworth Rd Ann Arbor, MI USA TABLE OF CONTENTS GENERAL INFORMATION
Nucleic Acid Purity Assessment using A 260 /A 280 Ratios A common practice in molecular biology is to perform a quick assessment of the purity of nucleic acid samples by determining the ratio of spectrophotometric
Kinetic Analysis of β-galactosidase Activity using the PowerWave HT and Gen5 Data Analysis Software Basic enzyme kinetic determinations Paul Held Ph. D. Senior Scientist, Applications Dept., BioTek Instruments,
ab102536 BCA Protein Quantification Kit Instructions for Use For the rapid, sensitive and accurate measurement of BCA Proteins in various samples. This product is for research use only and is not intended
ab155901 Creatine Kinase Activity Assay Kit (Colorimetric) Instructions for Use For the sensitive and accurate measurement of Creatine Kinase activity in various samples. This product is for research use
ab65328 Nitric Oxide Assay Kit (Colorimetric) Instructions for Use For the rapid, sensitive and accurate measurement of Nitric Oxide in various samples. This product is for research use only and is not
Mouse OVA IgG ELISA Catalog Number M046071 For the quantitative determination of Ovalbumin specific IgG in mouse plasma and serum samples. For research use only. This product insert must be read in its
Creatinine Assay Kit Catalog Number KA0849 100 assays Version: 05 Intended for research use only www.abnova.com Table of Contents Introduction... 3 Intended Use... 3 Background... 3 General Information...
ab83378 Galactose and Lactose Assay Kit Instructions for Use For the rapid, sensitive and accurate measurement of Galactose and Lactose levels in various samples This product is for research use only and
KASSAY Nitrate/Nitrite Assay For the quantitative colorimetric determination of nitrate/nitrite in urine, plasma, serum, saliva, cell lysate and tissue culture medium Cat. No. DN006 For Research Use Only.
Product Manual Phosphate Assay Kit (Colorimetric) Catalog Number STA-685 1000 assays FOR RESEARCH USE ONLY Not for use in diagnostic procedures Introduction Phosphorus exists as a soluble free phosphate
Human Alanine Aminotransferase (ALT) ELISA Kit Catalog No. CSB-E10082h (96 T) This immunoassay kit allows for the in vitro quantitative determination of human ALT concentrations in serum, plasma and other
Human glycated hemoglobin A1c (GHbA1c)ELISA Kit Catalog No. CSB-E08139h (96T) This immunoassay kit allows for the in vitro quantitative determination of human GHbA1c concentrations in serum, plasma and
Canine Amyloid-Beta Aβ(1 42) ELISA Protocol FIVEphoton Biochemicals For research use only. Not for diagnostics. Part No. cabeta42elisa FIVEphoton Biochemicals 4907 Morena Blvd, Ste 1403 San Diego, CA 92117
K-ASSAY Horse IgE ELISA For the quantitative determination of IgE in horse biological samples Cat. No. KT-492 For Research Use Only. 1 Rev. 063009 K-ASSAY PRODUCT INFORMATION Horse IgE ELISA Cat. No. KT-492
ab102524 Lipase Detection Kit (Colorimetric) Instructions for Use For the rapid, sensitive and accurate measurement of Lipase activity in various samples. This product is for research use only and is not
K-ASSAY Mouse IgM ELISA For the quantitative determination of IgM in mouse biological samples Cat. No. KT-407 For Research Use Only. 1 Rev. 072309 K-ASSAY PRODUCT INFORMATION Mouse IgM ELISA Cat. No. KT-407
ab102528 Isocitrate Dehydrogenase Assay Kit (Colorimetric) Instructions for Use For the rapid, sensitive and accurate measurement of Isocitrate Dehydrogenase activity in various samples This product is
Analysis of Riboflavin in a Vitamin Pill by Fluorescence Spectroscopy Objectives In this lab, you will use fluorescence spectroscopy to determine the mass of riboflavin in a vitamin pill. Riboflavin fluorescence
Mouse Total IgE ELISA Catalog Number M036023 For the quantitative determination of Total IgE in mouse serum samples. For research use only. This product insert must be read in its entirety before using
HUMAN TOTAL MMP-3 ELISA KIT PAGE 1 HUMAN TOTAL MMP-3 ELISA KIT PURCHASE INFORMATION: FOR THE QUANTITATIVE DETERMINATION OF HUMAN TOTAL MMP-3 CONCENTRATIONS IN CELL CULTURE SUPERNATES, SERUM AND EDTA PLASMA
HUMAN ADIPONECTIN (TOTAL) ELISA KIT PAGE 1 HUMAN ADIPONECTIN (TOTAL) ELISA KIT PURCHASE INFORMATION: FOR THE QUANTITATIVE DETERMINATION OF HUMAN ADIPONECTIN CONCENTRATIONS IN CELL CULTURE SUPERNATES, PLASMA
Creatine Kinase Activity Colorimetric Assay Kit ABE5487 100 assays; Store at -20 C I. Introduction: Creatine Kinase (CK) also known as creatine phosphokinase (CPK) and ATP: creatine N- phosphotransferase
Properties of Enzymes TECHNICAL OBJECTIVES 1. Use a pipette bulb to measure and transfer solutions. 2. Use a Spectrophotometer to study enzyme action. 3. Develop a hypothesis and an experimental protocol
The Physiology of Hyperbaric Oxygen Therapy Free Radicals and Reactive Oxygen Species I. Introduction Definition, Source, function and Purpose A. Definition of free radicals and reactive oxygen species
Product Manual Uric Acid/Uricase Assay Kit Catalog Number STA-375 400 assays FOR RESEARCH USE ONLY Not for use in diagnostic procedures Introduction For humans and higher primates, uric acid is the final
International Journal Of Ayurvedic And Herbal Medicine 1:2 (2011)46:50 Journal Homepage http://interscience.org.uk/index.php/ijahm ISSN-2249-5746 Effect of Boiling and Roasting on the Antioxidants Concentrations
Mouse Alanine Aminotransferase, ALT/GPT ELISA Kit Catalog No: E0207m 96 Tests Operating instruction www.eiaab.com FOR RESEARCH USE ONLY; NOT FOR THERAPEUTIC OR DIAGNOSTIC APPLICATIONS! PLEASE READ THROUGH
Rat Creatine Kinase MB isoenzyme,ck-mb ELISA Kit Catalog No: E0479r 96 Tests Operating instructions www.eiaab.com FOR RESEARCH USE ONLY; NOT FOR THERAPEUTIC OR DIAGNOSTIC APPLICATIONS! PLEASE READ THROUGH
Bio-Stack Applications of an Automated Microplate Stacking Device for HTS and Multi- Instrument Interface Abstract Today s microplate automation spans the range from multi-channel pipettes to large, robotic
Canine creatine kinase MB isoenzyme (CK-MB)ELISA Kit Catalog No. CSB-E15852c (96T) This immunoassay kit allows for the in vitro quantitative determination of canine CK-MB concentrations in serum and plasma.
Laboratory 5: Properties of Enzymes Technical Objectives 1. Accurately measure and transfer solutions with pipettes 2. Use a Spectrophotometer to study enzyme action. 3. Properly graph a set of data. Knowledge
Calculating Nucleic Acid or Protein Concentration Using the GloMax Multi+ Microplate Instrument Technical Note INTRODUCTION Direct measurements of nucleic acid samples at OD 260 or protein samples at OD
ab102527 Glutamate Dehydrogenase Detection Kit Instructions for Use For the rapid, sensitive and accurate measurement of Glutamate Dehydrogenase activity in various samples This product is for research
PAGE 1 HUMAN TOTAL MATRIX METALLOPROTEINASE 1 (MMP-1) ELISA KIT PURCHASE INFORMATION: FOR THE QUANTITATIVE DETERMINATION OF HUMAN TOTAL MMP-1 CONCENTRATIONS IN CELL CULTURE SUPERNATES, SERUM AND EDTA PLASMA
RayBio Human ADAM9 ELISA Kit Catalog #: ELH-ADAM9 User Manual Last revised April 15, 2016 Caution: Extraordinarily useful information enclosed ISO 13485 Certified 3607 Parkway Lane, Suite 100 Norcross,
RayBio Human Fibronectin ELISA Kit Catalog #: ELH-FN1 User Manual Last revised April 15, 2016 Caution: Extraordinarily useful information enclosed ISO 13485 Certified 3607 Parkway Lane, Suite 100 Norcross,
ab65313 ADP/ATP Ratio Assay Kit (Bioluminescent) Instructions for Use For the rapid, sensitive and accurate measurement of the ratio of ADP/ATP in various samples. This product is for research use only
K-ASSAY KAMIYA BIOMEDICAL COMPANY KAMIYA BIOMEDICAL COMPANY BCA Protein Colorimetric Detection Kit For the quantitative determination of total protein content in serum, plasma, tissue homogenates and cell
Determining the Quantity of Iron in a Vitamin Tablet Computer 34 As biochemical research becomes more sophisticated, we are learning more about the role of metallic elements in the human body. For example,
Using a Real Time Kinetic Cytotoxicity Assay to Determine when to Detect Apoptosis by... Using a Real Time Kinetic Cytotoxicity Assay to Determine when to Detect Apoptosis by Caspase-3/7 Activation ABSTRACT
Enzymes: Amylase Activity in Starch-degrading Soil Isolates Introduction This week you will continue our theme of industrial microbiologist by characterizing how differing reaction conditions affect enzyme
Analysis of Riboflavin in a Vitamin Pill by Fluorescence Spectroscopy** Objectives In this lab, you will use fluorescence spectroscopy to determine the mass and percentage of riboflavin in a vitamin pill.
Seahorse XF Cell Mito Stress Test Kit Part # 103015-100 User Guide For use with Seahorse XF e and XF Extracellular Flux Analyzers For Research Use Only 1 Table of Contents Product Description... 2 Introduction...
Creatine Kinase Assay Kit Catalog Number KA1665 100 assays Version: 03 Intended for research use only www.abnova.com Table of Contents Introduction... 3 Intended Use... 3 Background... 3 Principle of the
Service and Support Users of our systems can benefit from our comprehensive, fully inclusive service and support. We provide complete service and support for all of our customers to give reassurance that
Cautions: 6 M hydrochloric acid is corrosive. Purpose: To colorimetrically determine the mass of iron present in commercial vitamin tablets using a prepared calibration curve. Introduction: Iron is considered
P i Per Pyrophosphate Assay Kit Table 1. Contents and storage information. Material Amount Concentration Storage Stability Amplex Red reagent (Component A, MW = 257) 5 vials, 0.26 mg each DMSO, anhydrous
Mouse Aspartate aminotransferase, AST ELISA Kit Catalog No: E1214m 96 Tests Operating instruction www.eiaab.com FOR RESEARCH USE ONLY; NOT FOR THERAPEUTIC OR DIAGNOSTIC APPLICATIONS! PLEASE READ THROUGH
DUB Activity Assay Kit Item No. 15981 Customer Service 800.364.9897 * Technical Support 888.526.5351 www.caymanchem.com TABLE OF CONTENTS GENERAL INFORMATION 3 Materials Supplied 3 Precautions 4 If You
Clinical Chemistry 44:6 1309 1315 (1998) Automation and Analytical Techniques Comparison of different analytical methods for assessing total antioxidant capacity of human serum Guohua Cao 1,2 and Ronald
ADAM10 ELISA Kit A Disintegrin And Metalloprotease 10 ELISA Kit General Protocol FIVEphoton Biochemicals For research use only. Not for diagnostics. Part No. h,m,r,rb ADAM10-ELISA This protocol is provided
GloMax Discover and Explorer State-of-the-art multimode readers for luminescence, fluorescence and absorbance detection. Easy-To-Use The intuitive touchscreen display, preloaded protocols and automatic
UV Ultraviolet - Visible Spectroscopy (UV) Introduction to Ultraviolet - Visible Spectroscopy 1 (UV) Background Theory UV Absorption of ultraviolet and visible radiation Absorption of visible and ultraviolet
START-up GUIDE Alpha Technology/Protein:Protein Interactions Alpha has been used to study a wide variety of interactions, including protein:protein, protein:peptide, protein:dna, protein:rna, protein:carbohydrate,
Page: 1 of 5 1. Risk Assessment: This Risk Assessment is to be used as a general guide and as such, cannot accommodate all the varying factors that may be encountered when using this procedure. Therefore,
Intracellular O 2 Respiratory Burst Imaging Kit Item No. 601020 Customer Service 800.364.9897 * Technical Support 888.526.5351 www.caymanchem.com TABLE OF CONTENTS GENERAL INFORMATION 3 Materials Supplied
Exercise 6 - Enzyme Activity Introduction Enzymes are biological catalysts that regulate the rate of chemical reactions. Their 3-dimensional conformation and therefore their function can be affected by
Chemistry 201 Lecture 4 Quantitative Analysis NC State University Focus on energy The work done in the internal combustion engine is called pressure volume work. For a simple irreversible stroke the work
ab193692 Amyloid beta 1-40 Human ELISA Kit Instructions for Use For the quantitative measurement of Human Amyloid beta 1-40 in serum, plasma, and cell culture supernatants. This product is for research
HUMAN AGR2 ELISA KIT P a g e 1 HUMAN ANTERIOR GRADIENT 2 (AGR2) ELISA KIT FOR THE QUANTITATIVE DETERMINATION OF HUMAN AGR2 CONCENTRATIONS IN CELL CULTURES, SERUM AND PLASMA. PURCHASE INFORMATION: ELISA
Insulin Rodent (Mouse/Rat) Chemiluminescence ELISA For the quantitative determination of insulin in rodent serum and plasma. For Research use Only. Not For Use In Diagnostic Procedures. Catalog Number:
ab207003 BCA protein assay kit reducing agent compatible (microplate) Instructions for use: For measuring total protein concentration of pure proteins, extracts or lysates in the presence of reducing agents
ENZYME KINETICS INTRODUCTION The study of reaction rates catalyzed by enzymes and the factors affecting them is generally referred to as enzyme kinetics. The basic components of an enzyme catalyzed reaction
PRACTICAL I: Protein Assay by the BCA method The BCA protein assay is a detergent-compatible formulation based on bicinchoninic acid (BCA) for the colorimetric detection and quantitation of total protein.
ADAM10: Human A Disintegrin And Metalloprotease 10 ELISA Kit Biotin Detection Antibody Format 96T. FIVEphoton Biochemicals For research use only. Not for diagnostics. Part No. ADAM10-ELISA Biotin FIVEphoton
ab178780 PicoProbe Triglyceride Quantification Assay Kit (Fluorometric) Instructions for Use For the rapid, sensitive and accurate measurement of Triglyceride in various samples. This product is for research
This kit is intended for Research Use Only. Not for use in diagnostic procedures. Intended Use ICAM-1 ELISA is an enzyme-linked immunosorbent assay for the quantitative measurement of human sicam-1, an
An Investigation of Enzyme Activity: Phosphatase Objective: Using the standard curve method to determine the activity of the enzyme phosphatase as determined through spectrophotometric methods. I. BACKGROUND
INTRODUCTION Factors Affecting Enzyme Activity The chemical reactions occurring in living things are controlled by enzymes. An enzyme is a protein in the cell which lowers the activation energy of a catalyzed
Your consent to our cookies if you continue to use this website.