2 Page 1 Transcend Heated Humidifier TM User guide Notices Revised Notice Trademark Copyright Transcend Heated Humidifier User Guide Rev A Published April 15, 2013 and supersedes all previous versions. The information contained in this document is subject to change without notice. Somnetics and Transcend are registered trademarks of Somnetics International, Inc.; all rights reserved. Copyright 2013 Somnetics International, Inc.; all rights reserved. Contact Somnetics International, Inc. Corporate headquarters Web Somnetics International, Inc. Telephone Toll-free telephone Fax Somnetics International, Inc. 33 5th Avenue NW, Suite 500 New Brighton, Minnesota USA Authorized Representative (MDD 93/42/EEC) Methodize Europe Drieblad 11 NL-2811EG Reeuwijk The Netherlands Telephone l
3 Transcend Heated Humidifier TM User guide Page 2 Contents Introduction... 4 Intended use... 4 Contraindications... 4 General Warnings... 4 Symbols... 6 Transcend Heated Humidifier components... 7 Using the Transcend Heated Humidifier... 8 Setting up your Humidifier... 8 Filling the water reservoir... 8 Powering the Humidifier... 9 Starting therapy... 9 Ending therapy Cleaning and maintainence Reassembling the water reservoir Servicing Replacement Parts Technical Specifications Storage and transport Operating conditions Troubleshooting Electromagnetic emissions Electromagnetic immunity Warranty... 22
4 Page 3 Transcend Heated Humidifier TM User guide Introduction Thank you for choosing the Transcend Heated Humidifier ( Humidifier ), designed to be used with Transcend positive airway pressure (PAP) therapy devices to humidify the air delivered during therapy. Dry air may cause nasal congestion or dryness of the nose, mouth, or throat during sleep apnea therapy, especially in dry or cold climates and in cases of air leakage through the mouth. Humidifying the air may help relieve these symptoms and make therapy more comfortable. Please read all instructions prior to use. Intended use The Transcend Heated Humidifier is indicated for the humidification of the air delivered from a compatible Transcend PAP therapy device. The Humidifier is intended for single patient re-use in the home and hospital/institutional environment. The Humidifier is for use only as recommended by a physician. Contraindications Please refer to the Transcend User Manual provided with your PAP device for contraindications associated with positive airway pressure therapy. General warnings Note: Specific warnings and cautions appear throughout the manual within relevant sections. Use the Humidifier only for its intended use as defined in this User Guide. Only use positive airway pressure devices, accessories, or parts recommended by Somnetics with the Humidifier. The Humidifier is for single patient re-use and must not be re-used on another person. This is to avoid the risk of cross-infection. Always place the Humidifier lower than the level of the patient. If it is placed at or above the level of the patient, heated water may flow down the air tubing and into the mask. Take care when handling the Humidifier. The water in the reservoir may be hot. If liquids are spilled on the Humidifier disconnect the power cord immediately, discontinue use, and contact your service provider. Do not modify this equipment without authorization of the manufacturer. The Humidifier Water Reservoir must be completely drained of water prior to transport. Water remaining in the Water Reservoir may leak during transport, damage the electrical circuitry of the device, and create a potential electrical shock hazard.
5 Transcend Heated Humidifier TM User guide Page 4 Do not use the Humidifier in the presence of flammable or explosive materials. Medical electrical equipment including this device require special precautions regarding electromagnetic compatibility (EMC) and needs to be installed, placed, and used according to EMC information. Radiofrequency (RF) communications equipment can effect medical electrical equipment, including this device. Other cables and accessories may affect EMC performance. Be careful when placing or stacking this humidifier close to or on top of other equipment.
6 Page 5 Transcend Heated Humidifier TM User guide Symbols Attention: consult accompanying documents Type BF Applied Part ETL Classified ETL Seal of Approval demonstrating quality, safety and professional manufacturing of medical product Upper and lower temperature limits Separate collection for electrical and electronic equipment per EC Directive 2002/96/EC. Waste Electrical and Electronic Equipment (WEEE) Consult instructions for use Upper and lower humidity limits Non-condensing Class II medical electrical equipment Caution: Federal Law (United States) restricts this device to sale by or on the order of a Physician.
8 Page 7 Transcend Heated Humidifier TM User guide Using the Transcend Heated Humidifier Please follow the sections below to set up, use, and clean your Humidifier. Setting up your Humidifier Place the Humidifier on a firm, level surface near your bed. It is recommended that the Humidifier not be placed on a surface that could be damaged by water. Place the Humidifier at a level that is lower than your sleeping position. Caution: Do not position the Humidifier where it may be stepped on or accidentally overturned. Filling the Water Reservoir 1. Remove the Water Reservoir by pulling it forward and out of the Humidifier Body (Image 1). 2. Pull up on the Fill Cap to open the Water Reservoir (Image 2). 3. Carefully fill the Water Reservoir with distilled water to the MAX fill line (Image 3). Image 1 4. Replace the Fill Cap ensuring it is tightly sealed. Note: When carrying a filled Water Reservoir, ensure that the Fill Cap is sealed and the air outlet is tilted slightly Image 2 upward. Take precaution not to tilt the air outlet at a downward angle as water may spill out of the reservoir. 5. Carefully re-insert the Water Reservoir into the Humidifier Body. You will hear an audible click and feel it lock into place when it is fully re-inserted. Image 3
9 Transcend Heated Humidifier TM User guide Page 8 Note: The base of the Humidifier Body has guides to help position the Humidifier Reservoir during reinsertion. Caution: If the Water Reservoir is not correctly positioned and locked into place, the heater will not activate and the rightmost LED will turn red. Warning: Disconnect power from the Humidifier prior to removing, filling, and replacing the Water Reservoir. Warning: After use, take care when removing the Water Reservoir from the Humidifier as the water and the Water Reservoir may be hot. Warning: Do not overfill the reservoir. Overfilling may allow water to enter the air supply tubing. Do not fill the reservoir above the indicated maximum fill level. Warning: Re-insert the Water Reservoir slowly as water may spill into the air supply tubing if the Water Reservoir is not handled carefully. Powering the Humidifier The Humidifier is supplied with a power supply that connects to a wall outlet. This power source is for the Humidifier only. The PAP device must be powered separately using any of the available Transcend power options described in the PAP device User Manual. To power the Humidifier, plug the power supply cord into the back of the device, then connect the power supply to a wall (main) power outlet (Image 4). Image 4 Note: When looking at the back of the device, the leftmost LED will turn solid green indicating that the Humidifier has power. Starting therapy 1. Connect one end of the air supply tubing onto the air outlet of the Humidifier. Connect the other end of the tubing to your mask (Image 5). Image 5 Note: Standard six foot air tubing is supplied with the Transcend Sleep Apnea
10 Page 9 Transcend Heated Humidifier TM User guide Starter System. Air tubing is not included with the Humidifier. 2. Place the PAP device into the corresponding slots located at the top of the Humidifier; then slide the device forward until it snaps into place (Image 6). Note: The Transcend PAP device Image 6 has arm bars which help to guide it into the Humidifier port. You will hear an audible click and feel it lock into place when the PAP device is seated correctly. 3. Power the Transcend PAP device as described in the Transcend Sleep Apnea System User Manual. The PAP device may be powered via wall (main) power, Transcend battery, or Transcend Mobile Power Adaptor as appropriate. 4. Initiate therapy by pressing the power button on the PAP device. Air flow from the PAP device will turn on the Humidifier automatically and it will begin warming the Water Reservoir. The middle LED on the back of the humidifier will turn solid green to indicate that the Humidifier is heating the Water Reservoir and is delivering humidification (Image 7). Note: The Humidifier temperature setting must be in the range of 1 to 5 to initiate warming. A temperature setting of 0 will turn off the heating function of the Humidifier. Image 7 Note: The Humidifier will not begin warming the Water Reservoir until the PAP device power button has been pressed and the PAP device is in Therapy Mode. It may take up to 60 minutes to reach the set Humidifier temperature depending upon the selected heat setting. Note: If a fault has occurred with the humidifier all three LEDs will begin to flash. Please refer to the troubleshooting section of this user manual for more information.
11 Transcend Heated Humidifier TM User guide Page Adjust the heat setting by using the dial located on the back of the Humidifier (Image 8). Note: Heated humidification can be adjusted to a level of 0 to 5. The 0 setting will turn the heating function to off. A setting of 5 provides the Image 8 highest level of humidification. If a heat setting has been recommended by your physician, begin at that setting and then adjust the dial up or down to reach your humidification comfort level. Warning: Place the Humidifier at a level that is lower than your sleeping position to prevent condensation from collecting in air supply tubing and patient interface. Avoid using a heat setting that may result in the collection of condensation in the air tubing. Ending therapy 1. To stop therapy at any time, press the power button on the Transcend PAP device (Image 9). Note: It may take up to 60 seconds for the middle green LED to turn off. Image 9
12 Page 11 Transcend Heated Humidifier TM User guide Cleaning and maintenance 1. Unplug the Humidifier from the wall outlet and allow it to cool to the touch. 2. Remove the PAP device by pulling it rearward and sliding it out of the guide slots of the Humidifier (Image 10). a. Periodic Cleaning: Clean the exterior of the Humidifier as needed. Use a mix of 5% mild liquid detergent in distilled water (1.6 fl oz. liquid detergent per quart of distilled water). This will clean away surface residue. Submerge a lint-free cotton Image 10 cloth into the detergent solution. Wipe the exterior of the Humidifier using a gentle, back and forth wiping motion from the front to back of the Humidifier. Rinse the cloth in clear water to remove residual cleaning solution. Wring excess water from the cloth then wipe the Humidifier using a gentle front to back wiping motion to remove any detergent solution remaining on its surface. Wipe the device with a dry, lint-free cotton cloth until the device is fully dry. b. Daily Cleaning: Clean the Water Reservoir daily by placing in the dishwasher or by immersing in a mixture of 5% mild liquid detergent in distilled water (1.6 fl oz. liquid detergent per quart of distilled water). For ease of cleaning, the Water Reservoir cover detaches by pushing on the tabs located at the back of the Water Reservoir and lifting the cover upward (Image 11). For cleaning, follow the instructions below: For dishwasher cleaning, use standard dishwashing detergent with the dishwasher set in normal setting. Image 11 For hand cleaning, fully immerse the Water Reservoir in the cleaning solution.
13 Transcend Heated Humidifier TM User guide Page 12 While immersed, thoroughly wipe the surface with a lint-free cotton cloth. Apply firm pressure and ensure contact with all accessible contact surfaces to adequately remove soil buildup. Rinse by immersing in distilled water. Move the Water Reservoir in a back and forth motion for approximately 10 seconds to remove cleaning agent residue. Dry the Water Reservoir by wiping with dry, lint-free cotton cloth. Allow to air dry as needed. When cleaning is complete, replace the Water Reservoir cover by aligning the front tab into the reservoir base (Image 12). Guide the cover over the reservoir base. The tabs at the back of the base will flex outward and snap into place. An audible snap will be heard (Image 13). Carefully return the Water Reservoir to the Humidifier. c. Weekly Maintenance: Inspect the Water Reservoir and Fill Cap for wear and deterioration. If the Water Reservoir shows signs of leakage it should be replaced. Contact your home medical equipment provider or call Somnetics customer service at to purchase a replacement Water Reservoir. Warning: To avoid the potential for electric shock unplug the Humidifier from the power source and remove the Transcend PAP device before cleaning the humidifier. Allow the Humidifier to dry completely before plugging it back into a power source. Image 12 Warning: The Water Reservoir is the only part of the Heated Humidifier that should be submerged in water. Never submerge Image 13 any other part of the Humidifier in water. Do not allow water to enter the interior of the Humidifier Body. In the event water enters the interior of the Humidifier Body, wipe the surfaces dry of water using a lint-free cotton cloth and allow to fully air dry before plugging it back in to a power source.
14 Page 13 Transcend Heated Humidifier TM User guide Reassembling the water reservoir 1. Reassemble the Reservoir cover by aligning the front tab into the Water Reservoir base (Image 14). 2. Guide the cover over the reservoir base. The tabs at the back of the base will flex outward until an audible snap is heard (Image 15). 3. Reinsert the Water Reservoir into the Humidifier Body as described. Image 14 Image 15
15 Transcend Heated Humidifier TM User guide Page 14 Servicing No further maintenance is required. Do not open the Humidifier Body. There are no serviceable parts inside. Repairs and servicing should only be performed by a service agent authorized by the manufacturer. Replacement Parts Item Part Number Item Part Number Water Reservoir Assembly US 2-Prong Power Cord Transcend Heated Humidifier UK 2-Prong Power Cord EU 2-Prong Power Cord AU 2-Prong Power Cord Technical Specifications Dimensions (LxWxH) Weight AC Supply Input Water Reservoir Volume up to maximum fill level Working Pressure Range Warm up time 9 in x 5.5 in x 4.7 in 2.2 lbs (998g) empty; 3.1 lbs (1406g) with water to the max fill line V, 50-60Hz 325 ml 4 to 20 cm H 2 O Up to 1 hour depending on heat setting Caution: Humidity performance may be compromised if used outside the defined ambient temperature range and humidity range.
16 Page 15 Transcend Heated Humidifier TM User guide Inspiratory/Expiratory Pressure Drop: Gas Leakage of the Humidification System at max operating pressure: Humidifier Performance: LPM (Liters per Minute) 4.85 LPM RH output % Nominal system output (mg/l) 4 cm H2O 12 cm H2O 20 cm H2O Setting 3 Setting 5 Setting 1 Setting 5 >90 > > > Maximum Gas Temperature: Sound power level cmh 2 O pressure, static) Sound pressure level cmh 2 O pressure, static) Maximum heater plate temperature 77 F (25 C) 37.0 db (Transcend), 36.7 db (Transcend EZEX, Transcend Auto) 29.0 db (Transcend), 28.7 db (Transcend EZEX, Transcend Auto) 131 F (55 C) Storage and Transport Temperature Range Humidity Range (-4)-140 F ( C) 10-90% RH Caution: Empty the water reservoir before traveling with the heated humidifier. Operating Conditions Ambient Temperature Range Gas Inlet Temperature Range Humidity Range Altitude F (5-35 C) F (5-35 C) 10-80% RH ft.
17 Transcend Heated Humidifier TM User guide Page 16 Troubleshooting Problem Probable Cause Solution Humidifier does not turn on Humidifier does not heat the Water Reservoir PAP device turns on but then stops working The red LED is illuminated on the Humidifier Water is splashing on your face Incorrect setup Incorrect setup Massive air leak causing system to shut down Water supply is low or water reservoir is detached from the humidifier. The Water Reservoir is overfilled. The settings knob is set too high causing condensation to form in the air tube and mask. 1. Ensure that you have the fully seated the PAP device into the Heated Humidifier and that you have pressed the PAP power button. 2. Be sure the PAP device and Heated Humidifier are connected to power. 1. Ensure that you have followed the correct setup. The middle green LED will indicate that the Heated Humidifier is warming the Water Reservoir. It may take up to 45 seconds to reach the desired temperature. 1. Ensure that the air supply hose is fully connected to the Heated Humidifier and to your mask. 2. Securely fit the mask to your face before powering up the PAP device.. 3. Be sure the Water Reservoir gasket is in place and the Water Reservoir cover is fully seated. 4. Ensure the PAP device is fully seated in the Heated Humidifier. 1. Ensure that the water reservoir is properly attached to the humidifier body. The reservoir has guides that will help to secure it in place. 2. Check the water supply inside of the water reservoir. If the water is low then the red LED will illuminate to signal a refill. 1. Empty some water from the Water Reservoir. 2. Adjust the setting to a lower setting.
18 Page 17 Transcend Heated Humidifier TM User guide Air does not feel moist Air feels too moist Leaking Water Reservoir All three LEDs at the back of the Humidifier are flashing. The settings knob is set too low. The heater is not working. The settings knob is set too high. Water Reservoir may be cracked or damaged. The lid may not be on correctly. Humidifier is in a fault state. 1. Adjust the setting to a higher setting. 2. If the problem persists, contact your home medical equipment provider or Somnetics customer service at Adjust the setting to a lower setting. 1. Ensure that the reservoir cover is fully attached to the reservoir base. Refer to reassembling the water reservoir section. 2. Inspect the Water reservoir for damage or cracks. If there are signs of damage, contact your home medical equipment provider or Somnetics customer service at to order a replacement. 1. Unplug the Humidifier power supply. 2. Wait for 10 seconds. Electromagnetic emissions 3. Reattach the power supply to the Humidifier. If the problem continues contact your home medical equipment provider or Somnetics customer service at The Transcend Heated Humidifier is intended for use in the electromagnetic environment specified below. Ensure the Heated Humidifier is used in such an environment. Emissions test Compliance Electromagnetic environment guidance RF radiated emissions CISPR11 RF conducted emissions CISPR 11 Harmonic emissions IEC Voltage fluctuations/ flicker emissions Group 1 Class B Class A Complies The Heated Humidifier system uses RF energy only for its internal function; therefore, its RF emissions are very low and are not likely to cause any interference in nearby electronic equipment. The Heated Humidifier system is suitable for use in all establishments and those directly connected to the public low-voltage power supply network that supplies buildings used for domestic purposes.
19 Transcend Heated Humidifier TM User guide Page 18 IEC Electromagnetic immunity The Transcend Heated Humidifier is intended for use in the electromagnetic environment specified below. Ensure the Heated Humidifier is used in such an environment. Immunity test IEC test level Compliance Electromagnetic environment guidance Electrostatic discharge (ESD) IEC ±2, 4, 6 kv contact ±8 kv air N/A. The HH does not have conductive surfaces. Floors should be wood, concrete, or ceramic tile. If floors are covered with synthetic material, the relative humidity should be at least 30%. ±2, 4, 6, 8 kv air Electrical fast transient/burst IEC ±2 kv for power supply lines ±2, kv for power supply lines Line power quality should be that of a typical commercial or hospital environment. ±1 kv for input/output lines ±1 kv for input/output lines Surge IEC ±1 kv differential mode ±0.5*, 1 kv differential mode Line power quality should be that of a typical commercial or hospital environment. ±2 kv common mode ±2 kv common mode
20 Page 19 Transcend Heated Humidifier TM User guide Immunity test IEC test level Compliance Electromagnetic environment guidance Voltage dips, short interruptions, and voltage variations on power supply input lines IEC <5% UT (>95% dip in UT) for 0.5 cycle 40% UT (60% dip in <5% UT (>95% dip in UT) for 0.5 cycle 40% UT (60% dip in UT) for 5 cycles Line power quality should be that of a typical commercial or hospital environment. If the user of the CPAP system requires continued operation during power line interruptions, it is recommended that the CPAP system be powered from the battery. UT) for 5 cycles 70% UT(30% dip in UT) for 25 cycles Note UT is the A.C. line voltage before application of the test level. 70% UT (30% dip in UT) for 25 cycles <5% UT (>95% dip in UT for 5 sec) <5% UT (>95% dip in UT for 5 sec) Power frequency (50/60 Hz) magnetic field 3 A/m 3 A/m Power frequency magnetic fields should be at levels characteristic of a typical commercial or hospital environment. IEC Conducted RF 3 Vrms 3 Vrms Recommended separation distance: IEC khz to 80 MHz 10 KHz to 100 MHz* d = 1.17 P
21 Transcend Heated Humidifier TM User guide Page 20 Immunity test IEC test level Compliance Electromagnetic environment guidance Radiated RF IEC V/m 80 MHz to 2.5 GHz 10 V/m (compliance level adjusted to meet FDA limits) 26 MHz to 2.5 GHz Recommended separation distance: d = 0.35 P 80 MHz to 800MHz Recommended separation distance: d = 0.70 P 800MHz to 2.5 GHz Note At 80 MHz and 800 MHz, the higher frequency range applies. where P is the maximum output power rating of the transmitter in watts (W) according to the transmitter manufacturer and d is the recommended separation distance in meters (m). Field strengths from fixed RF transmitters, as determined by an electromagnetic site survey1, should be less than the compliance level in each frequency range2. Interference may occur in the vicinity of equipment marked with the following symbol: 1 Field strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and land mobile radios, amateur radio, AM and FM radio broadcast and TV broadcast cannot be predicted theoretically with accuracy. To assess the electromagnetic environment due to fixed RF transmitters, an electromagnetic site survey should be considered. If the measured field strength in the location in which the humidifier system is used exceeds the applicable RF compliance level above, the humidifier system should be observed to verify normal operation. If abnormal performance is observed, additional measures may be necessary, such as reorienting or relocating the system. 2 Over the frequency range 150 khz to 80 MHz, field strengths should be less than 3 V/m. EN compliance Protection against electric shock: Degree of protection against ingress of water: Class II Type BF IP22
22 Page 21 Transcend Heated Humidifier TM User guide Warranty Somnetics warrants the Transcend Heated Humidifier to be free of defects in materials and workmanship and will perform in accordance with the product specifications for a period of 2 years from the date of sales by Somnetics to the dealer. If the product fails to perform in accordance with the product specifications, Somnetics will repair or replace, at its option, any materials or parts of the Heated Humidifier, which, upon Somnetics examination appear defective. This does not cover damages caused by accident, misuse, abuse, alteration, and other defects not related to material or workmanship. Somnetics will pay customary freight charges from Somnetics to dealer location only. Somnetics disclaims all liability for economic loss, loss of profits, overhead, or consequential damages which may be claimed to arise from any sale or use of this product. Some states do not allow the exclusion or limitation of incidental or consequential damages, so the above limitation or exclusion may not apply to you. This warranty is given in lieu of all other express warranties. In addition, any implied warranties, including warranty of merchantability or fitness for the particular purpose are limited to two years. Some states do not allow the exclusion or limitation of incidental or consequential damages, so the above limitation or exclusion may not apply to you. This warranty gives you specific legal rights, and you may also have rights which vary from state to state. To qualify for repair, replacement, or refund, the defective device must be returned to Somnetics within 30 days after the discovery of the defect. Any repair, replacement, or refund obligation would not apply if the device has been repaired or otherwise altered in a facility not authorized in writing by Somnetics To exercise your rights under this warranty, contact your local, authorized Somnetics dealer or Somnetics at 33 5th Avenue, New Brighton, Minnesota USA, or
23 Transcend Heated Humidifier TM User guide Page 22 Somnetics International, Inc. 33 5th Avenue NW, Suite 500 New Brighton, Minnesota USA Phone: Toll-free: Fax: Transcend Heated Humidifier User Guide Rev A
System One Heated Humidifier USER MANUAL 2009 Respironics Inc. and its affiliates. All rights reserved. Table of Contents Intended Use... 2 Warnings... 2 Cautions... 2 Symbol Key... 2 System Features and
System One Heated Humidifier USER MANUAL 2011 Koninklijke Philips Electronics N.V. All rights reserved. Table of Contents Intended Use... 2 Warnings... 2 Cautions... 2 Contraindications... 3 Symbol Key...
All about the Use & Care of your Iron Important Safeguards... 2 Iron Safety Instructions... 3 Parts and Features...4 TA B L E O F C O N T E N T S Using Your Iron...6 Cleaning and Maintenance... 8 Warranty...9
Please read this owner s Manual carefully before operating the unit. - Cooling - Heating - Dehumidifying - Fan TABLE OF CONTENTS INTRODUCTION 2 IMPORTANT SAFEGUARDS...2 PACKAGE CONTAINS..2 NAMES OF PARTS.3
Meaco 30L and Meaco 40L dehumidifier instruction manual Please read this instruction manual before using the dehumidifier and keep safe for future reference SAFETY INSTRUCTIONS PLEASE READ ALL INSTRUCTIONS
Portable Air Conditioner OWNER S MANUAL Read these instructions before use 8 Model: MN12CES / MN10CESWW Voltage rating: 120V~60Hz Power rating : 1100W (MN12CES) Power rating : 900W (MN10CESWW) Customer
1083700 1079142 R0 JH 11/3/11 SimplyGo Table of Contents Introduction 1 Intended Use 1 Contraindications 1 Device and Accessories Description 1 SimplyGo System 2 Warnings and Cautions 3 Symbol Key 6 How
Portable Air Conditioner OWNER S MANUAL Read these instructions before use Model: MM14CCS Customer Support : 1-800-474-2147 Voltage rating: 115V~60Hz Power rating : 1400W For product inquiries or support
StorTrends 3400 Hardware Guide for Onsite Support MAN-3400-SS 11/21/2012 Copyright 1985-2012 American Megatrends, Inc. All rights reserved. American Megatrends, Inc. 5555 Oakbrook Parkway, Building 200
Power Max Level 2 Charging Station en Installation and Operating Instructions This equipment has been tested and found to comply with the limits for a Class B digital device, pursuant to part 15 of the
USER GUIDE TURBOCORD TM PORTABLE CHARGER 120V/240V: DUAL VOLTAGE AeroVironment EV Solutions 2013 AeroVironment, Inc. All rights reserved. AeroVironment, EV Solutions, and the AeroVironment logo are trademarks
ROTOR LOADER OWNER S MANUAL ROTOR LOADER OWNER S MANUAL WARNING IMPORTANT SAFETY INSTRUCTIONS AND GUIDELINES. Misuse of paintball equipment may cause serious injury or death. QUICK SET-UP GUIDE BATTERY
926 Manual V2.0(1J9643) 6/10/99 1:39 PM Page 1 RC926 Wireless Phone Jack System (Digital Satellite System # D916) The Wireless Phone Jack System works with most telephone line devices, such as: Answering
RI-215A Operator s Manual Part Number: 71-0045RK Revision 0 Released: 10/3/05 Warranty RKI Instruments, Inc., warrants gas alarm equipment sold by us to be free from defects in materials and workmanship,
Owner s Manual & Safety Instructions Save This Manual Keep this manual for the safety warnings and precautions, assembly, operating, inspection, maintenance and cleaning procedures. Write the product s
Steam Humidifier Model 800 Residential Steam Humidifier Owner s Manual Includes Safety & Operating Instructions and Warranty Information READ AND SAVE THESE INSTRUCTIONS TABLE OF CONTENTS Safety Cautions...
Owner s Instruction Manual 2500/2554 Telephones Message Waiting THANK YOU FOR PURCHASING THIS TELEPHONE We want you to know all about your new telephone, how to install it, the features it provides, and
Owner s Manual 1 Dear Customer, Thank you for purchasing this product. For optimum performance and safety, please read these instructions carefully. User Memo: Date of purchase: Dealer name: Dealer address:
i800 AC Current Clamp Instruction Sheet Introduction The i800 AC Current Clamp, the Clamp, has been designed for use with multimeters, recorders, power analyzers, safety testers, etc., for accurate non-intrusive
User Manual (English) Table of Contents General Safety Information...2 Fueltank UNO Components...3 Charging the Fueltank UNO...4 Using the Fueltank UNO...5 Power Saving Mode...6 Checking the Fueltank UNO
Mercury Helios 2 ASSEMBLY MANUAL & USER GUIDE TABLE OF CONTENTS INTRODUCTION...1 1.1 MINIMUM SYSTEM REQUIREMENTS 1.1.1 Apple Mac Requirements 1.1.2 PC Requirements 1.1.3 Supported PCIe Cards 1.2 PACKAGE
MCR1900 Media Converter 19-Slot Chassis Installation Guide Part #5500304-11 Copyright Statement This document must not be reproduced in any way whatsoever, either printed or electronically, without the
FUTURE CALL PICTURE CARE PHONE MODEL: FC-1007 USER MANUAL Please follow instructions for repairing if any otherwise do not alter or repair any parts of device except specified. IMPORTANT SAFETY INSTRUCTIONS
33-1165.fm Page 1 Tuesday, August 10, 1999 10:50 AM Cat. No. 33-1165 OWNER S MANUAL Please read before using this equipment. 900 MHz Lightweight Wireless Stereo Headphones 33-1165.fm Page 2 Tuesday, August
AXIS 291 1U Video Server Rack Installation Guide About This Document This document describes how to install Axis blade video servers in the AXIS 291 1U Video Server Rack. Updated versions of this document
FEATURES: Click on Series name for product info on aimtec.com Models Single output Model Max Output Power (W) 1 AC-DC Constant Current or Constant Over temperature protection Voltage LED Driver Over current
Page 1 of 23 TEST REPORT EN 55014-2 (1997) +A1 (2001) Electromagnetic compatibility - Requirements for household appliances, electric tools and similar apparatus Part 2: Immunity Report Reference No....
TM Caller-ID on your TV! User s Manual and Installation Guide Copyright 1998 NetMedia Inc., All rights reserved. Rev. 061898 Thank you for your purchase of Caller-TV! You are now ready to enjoy the convenience
Business Audio System: Music & Messaging MP3 Player by Grace Digital Audio User Guide Model No. GDI-USBM10 User Guide Contents Introduction 2 Safety & General Use Information 2 Features 3 Set Up & Operation
User s Guide Oreck Air Purifier with HEPA Filtration Important! Read this manual carefully, and keep for future reference. Enjoy Congratulations on your purchase of the Oreck Air Purifier with HEPA Filtration.
User Guide Nokia Portable Wireless Charging Plate DC-50 Issue 1.1 EN User Guide Nokia Portable Wireless Charging Plate DC-50 Contents For your safety 3 About your accessory 4 Keys and parts 5 Top up your
Electric Water Kettle Instruction Manual Model: AWK-101 Congratulations! Your Water Kettle will surely become one of the most versatile and practical appliances in your home and office. Please read the
TM Advanced Tracking Technologies, Inc. 2003 ATTI All rights reserved Copyright 2003, Advanced Tracking Technologies, Inc. All rights reserved. No part of this publication may be reproduced or transmitted
Vari-Mix Aliquot Mixer Type 48700 Operation Manual and Parts List Model M48725 M48720-33 M48720-33CN Voltage 120 220-240 220-240 LT1108X1 3/30/10 Table of Contents Safety Information...3 Alert Signals...3
EKOS Cart Instructions for Use EKOS Corporation 11911 North Creek Parkway South Bothell, WA 98011 USA (425) 415-3100 (tel) (425) 415-3102 (fax) firstname.lastname@example.org (e-mail) - 1-4913-002 REV E Intended Use
Preface D-Link reserves the right to revise this publication and to make changes in the content hereof without obligation to notify any person or organization of such revisions or changes. Manual Revisions
43922 rev2:x 06/01/2010 15:06 Page 1 Upon registration with Morphy Richards (UK & ROI only) Register online for your 2 year guarantee. See back of this instruction book for details. (UK and Ireland customers
dedicated KVM switch and rackmount screen technology User Manual RK-2d / RK-2t 1U Short Depth Keyboard Drawer Designed and manufactured by Austin Hughes 751 Legal Information First English printing, October
Portable Air Conditioner OWNER S MANUAL Read these instructions before use Model: MM14CHCSCS Voltage rating: 120V~60Hz Power rating : 1400W(Cooling) Power rating : 1350W(Heating) Customer Support : 1-800-474-21477
U.S. Cellular Home Phone 1 Contents Getting to Know Your Device...5 Appearance... 5 LED Indicators... 6 Device Installation...7 Before You Begin... 7 Installing the Battery... 7 Installing the Power Adapter...
Hot Water Central Electric Hot Water Dispenser Instruction Manual www.aromaco.com AAP-325W With the Aroma Hot Water Central, hot water is just a button away. Boil up to 3 liters of water in just 30 minutes,
Register your stethoscope at www.littmann.com...pp. 1-19 FR Français...pp. 20-37 DE Deutsch...pp. 38-54 IT Italiano...pp. 55-71 ES Español...pp. 72-88 NL Nederlands...pp. 89-105 SE Svenska...pp. 106-122
Cisco TelePresence MCU 4500 Series safety and compliance information On this page: Safety information symbols Operating guidelines Safety warnings Technical specifications Compliance information WEEE information
Remote Faucet Pump System USE &CARE GUIDE See Important Safeguards on page 2 An exclamation point within an equilateral triangle is intended to alert user to the presence of important operating and maintenance
~RITA HIGH ALTITUDES CAUTION(5,OOO ft and higher) This appliance may not shut-off property or from boiled to keep warm position when used at high altitudes; Use EXTREME caution! If unit does not shut off
ADI 9510-B Advantium 2 Plus Alarm INSTALLATION AND OPERATING INSTRUCTIONS Carefully Read These Instructions Before Operating Carefully Read These Controls Corporation of America 1501 Harpers Road Virginia
AROS SMART WINDOW AIR CONDITIONER Part of the Quirky + GE collection of smart products, Aros is an app-enabled air conditioning unit that reacts to its surroundings to save you Time, Energy & Money. TABLE
STIM SWITCH USER MANUAL (REVISION 2.00) Blackrock Microsystems, LLC 391 Chipeta Way, Suite G Salt Lake City, UT 84108 Tel: (866) 806-3692 www.blackrockmicro.com Copyright 2011 Blackrock Micosystems, LLC.
Ultrasonic support removal tank User guide and safety manual See inside for use and safety information. The Ultrasonic support removal tank automatically removes PLA support material from a printed ABS
CABLE MODEM QUICK START This Quick Start describes how to connect your Zoom cable modem to a cable modem service. This lets your cable modem provide Internet access to a computer or other device connected
PART NO. CMA113 MADE IN CHINA 1. Measuring CAT II 2. Max. voltage 250V ~ 3. Max. current 71 Amp Getting started with Electricity consumption monitoring single phase for homes and some smaller light commercial
Owner s Manual HYLA NST Cleaning System The HYLA NST Cleaning System aspirates and cleans the air through a waterbased filtration process. The system is intended for household use only. Applications: Usual
PORTABLE AIR CONDITIONER MAC 7500 Owner s Manual Air Conditioner Dehumidifier Oscillating Fan Please read this owner s manual carefully before operating the unit. POWERED BY 66126113.p65 17 INTRODUCTION
BRAVER UPS (Uninterruptible Power System) User s Manual Safety CAUTION! This UPS utilizes voltages that may be hazardous. Do not attempt to disassemble the unit. The unit contains no user replaceable parts.
Owner s manual Keurig K-Cup K40/K45 Elite Brewer IMPORTANT SAFEGUARDS Safe Operation & Use When using electrical appliances, basic safety precautions should be followed, including the following: 1. Read
Provant Therapy System Instruction Manual Toll-Free Customer Service 1-877-970-4970 A INTRODUCTION Regenesis Biomedical, Inc., the developer of the Provant Therapy System, is pleased to be a part of your
User s Manual Bluetooth Calculator Keypad Page Regulatory Compliance This device complies with Part 15 of the FCC Rules. Operation is subject to the following two conditions: (1) This device may not cause
Rechargeable Power Pack User s Manual 2013 VTech Printed in China 91-009656-000 US INTRODUCTION The Rechargeable Power Pack makes it easier than ever to keep the InnoTab 3 or InnoTab 3S charged and ready
Stove Guard Kit User Manual Innohome improves the Safety of your Home. inno home www.innohome.com Stove Guard Kit User Manual Stove Guard Kit User Manual Congratulations! You now own one of the most intelligent
Breville Customer Service Center the Soft Top Instruction Book - Livret d instructions US Customers Mail: Breville USA 19400 S. Western Ave Torrance CA CA 90501-1119 Phone: Customer Service 1 (866) BREVILLE
LIMITED WARRANTY: This appliance shall be free from defects in materials and workmanship under normal home use for a period of one year from the date of receipt. During that period, defects will be repaired
dedicated KVM switch and rackmount screen technology User Manual IP-S101 Combo KVM Extender Designed and manufactured by Austin Hughes 751 Legal Information First English printing, October 2002 Information
Gyration Ultra Cordless Optical Mouse User Manual Product Specifications Spec Ultra Cordless Optical Mouse RF Receiver System Requirements Full functionality on Mac and IBM compatible PCs running Windows
Kambrook 461 Plummer Street, Port Melbourne 3207, Australia Customer Service Line (free call) 1800 800 634 Customer Service Fax 1800 621 337 Kambrook New Zealand, Private Bag 94411, Greenmount, Auckland,
AIRAVE 2.5 User Guide 2012 Sprint. Sprint and the logo are trademarks of Sprint. Other marks are trademarks of their respective owners. 911702 Rev. 51.13 Important Messages IMPORTANT: Sprint s policies
esata External Storage Operation Manual DA-ES110 Before reading this manual This operation manual contains basic instruction on installing and using esata External Storage, an IDIS product. Users who are
C-Bus Single Zone Thermostat 5070THB Series REGISTERED DESIGN REGISTERED PATENT Clipsal is a registered trademark of Clipsal Australia Pty Ltd ABN 27 007 873 529. Copyright Clipsal Australia Pty Ltd 2008.
User s Manual AURORA 1.2K/2.2K Uninterruptible Power System Safety CAUTION This UPS utilizes voltages that may be hazardous. Do not attempt to disassemble the unit. The unit contains no user serviceable
Nest Learning Thermostat Installation Guide Installation step-by-step Contents Compatibility. 3 In the box. 4 Before you start. 5 Choose an installation method. 6 Replacing an existing wired thermostat.
USER GUIDE HEAT RECOVERY VENTILATORS MODELS 90H-V ECM HRV, 90H-V+, 60H-V+ AND ENERGY RECOVERY VENTILATOR MODEL 90H-V ECM ERV PLEASE READ AND SAVE THESE INSTRUCTIONS These products earned the ENERGY STAR
Video Splitter USER MANUAL VS92A / VS94A / VS98A FCC Information This equipment has been tested and found to comply with the limits for a Class B digital device, pursuant to Part 15 of the FCC Rules. These
TM-47 ORDERCODE D1370 Congratulations! You have bought a great, innovative product from DAP Audio. The DAP Audio Microphone range brings excitement to any venue. Whether you want simple plug-&-play action