2 Contents: Introduction... 2 Intended Use... 2 Warnings... 2 Cautions... 3 Contraindications... 4 Symbols... 5 Circadian Rhythms... 6 Features... 8 Using Your golite BLU... 9 Suggested Usage...14 Frequently Asked Questions...17 Maintenance...19 Cleaning...19 Service and Repair...19 Specifications...20 EMC Information...21 Contents 1
3 Introduction 2 Introduction: Intended Use The golite BLU is intended to help people adjust their circadian rhythms to provide mood relief and to have more energy. Read All Warnings and Cautions Before Using Your golite BLU Observe these safety instructions to avoid potential hazards that could result in personal injuries or damage to your golite BLU. Warnings: A warning indicates the possibility of injury to the user or the operator. Use the golite BLU in a vertical position with the stand in place in the hole on the back of the unit. Do not use this device laying flat in a horizontal position without the stand. Do not expose this unit to rain or moisture. Do not use this light near a water source, while bathing, washing hands or while contacting water. If you suffer from an eye disease or have a history of eye disease in your family, you should consult your ophthalmologist before using this device. If you are experiencing any temporary eye problems, wait until the problems are resolved before using this device. Light emitted by this device may interact with photo-sensitizing medications. Be sure to review all medications that you are currently taking with your doctor before beginning treatment with this device. Do not use this product where the light might compromise your ability to perform essential tasks such as driving or operating heavy equipment. As with any bright light source, do not stare into the device.
4 Do not use this product where the light might distract other individuals and compromise their ability to perform essential tasks such as driving or operating heavy equipment. Do not increase intensity by sitting closer to the light than recommended. Increased intensity will not improve response and may result in increased side effects. The golite BLU internal battery is protected within the device enclosure and is not user accessible. Never open the device and come in contact with the internal battery. The metal stand for the device could present a choking hazard. Individuals who use photosensitizing medications, who have had cataract surgery, or who have pre-existing eye conditions should avoid unnecessary exposure to light. They should consult their ophthalmologist before using the golite BLU. The golite BLU should not be used adjacent to or stacked with other equipment. However, if it is necessary, the device should be observed to verify normal operation. Cautions: A caution indicates the possibility of damage to the device. The golite BLU contains an internal, rechargeable Lithium-ion battery pack. Do not incinerate, disassemble or expose to high temperatures above 60 C (140 F). Do not operate or charge the golite BLU in temperatures below -5 C (23 F) or above 35 C (95 F). Batteries can overheat causing fire or bursting. Dispose of the device with the internal battery properly. The golite BLU needs special precautions regarding EMC and how portable and mobile RF communication equipment can affect it. The golite BLU should be put into service according to the EMC Information provided at the end of this guide. Do not operate this device near any heat sources such as radiators, heat registers, or other apparatus that produce heat. Never use damaged or worn cords or plugs; this could result in electric shock, burns, and/or fire. Introduction 3
5 Introduction Use our provided AC/DC adapter (MW170KB0900B02) only. If not, this may result in increased emissions or decreased immunity of the golite BLU. Do not place heavy objects on top of the power cord or the device. When unplugging this device grasp the adapter directly to avoid damaging the cord. Never pull on the power cord to remove the adapter from the socket. Do not block any ventilation openings on this device. If this device is damaged in any way or is not working, it should not be operated. Please contact the manufacturer for service assistance. There are no user serviceable parts inside the golite BLU. No modification of this equipment is allowed. Contraindications The golite BLU may not be suitable for use on individuals who use photosensitizing medications, who have had cataract surgery, who have pre-existing eye conditions, or who have bipolar disorder. Thank you for purchasing the golite BLU! Visit now to: Register your golite BLU Learn more about your golite BLU and BLUEWAVE technology Don t have Internet access? Be sure to fill out your warranty card located inside the box and mail it to the address listed. 4
6 Symbol Key The following symbols may appear on the device: Sy m b o l Descr iption Consult accompanying instructions for use Introduction DC Power DC Jack Polarity Serial Number REF Model Number European Declaration of Conformity TUV Safety Standard Compliance for North America Compliant with the Waste Electrical and Electronic Equipment/Restriction of the Use of Certain Hazardous Substances in Electrical and Electronic Equipment (WEEE) recycling directives. European Representative The Comfort of Blue By using BLUEWAVE light-emitting diodes (LEDs), the golite BLU delivers safe and effective wavelengths of light, while eliminating harmful UV and the eyestrain that can be associated with other bright light treatments. 5
7 Introduction Safe and Effective According to independent ocular safety testing, BLUEWAVE light is safe when following the recommended usage guidelines. BLUEWAVE technology in your golite BLU passes ICNRIP/ACGIH threshold value limits at only 15% (light energy exceeding 100% is considered potentially hazardous). The effectiveness of the type of light in BLUEWAVE has been researched and advocated by leading medical universities. For more information, visit Understanding Light, Health, and Circadian Rhythms The human body uses light cues, such as those provided by the sun, to time certain internal functions. Properly timed rhythms regulate mood, sleep, energy, appetite, and digestion. These daily internal cycles, called circadian rhythms, sometimes fall out of sync, resulting in an unregulated body clock. Unfortunately, modern living has dramatically altered nature s cues. A modern day no longer starts at the crack of dawn and ends at 8:00 p.m. Workdays are getting longer and many people face shift work schedules. Additionally, the advent of electric lighting allows social gatherings and personal activities to extend well into the night. These factors have diminished the body s natural ability to regulate the body clock. 6
8 Circadian Rhythm Imbalance When our circadian rhythm becomes imbalanced, the body produces the wrong hormones, chemicals, and neurotransmitters at the wrong time of the day. The results range from feelings of sadness to sleeplessness to lack of energy. Many people suffer from a loss of energy and mood that can last for a short period of time to more than a couple of weeks. For many, these feelings come on stronger in the winter and in darker climates. An imbalance in circadian rhythms contributes to the following: Winter blues Feelings of sadness Jet lag Lack of concentration Loss of appetite Lack of energy (especially in the winter) Sleeplessness Mood swings Irritability Restored Balance with BLUEWAVE Technology BLUEWAVE technology helps shift circadian rhythms back to their normal pattern by delivering a specifically engineered wavelength, color, and intensity of light that stimulates the production of key substances triggered by the brain. BLUEWAVE naturally affects the body by shifting the body clock to its correct timing, causing the proper signals to be produced at the right time of day. Introduction 7
9 Introduction golite BLU Features Long-Lasting LEDs Advanced Diffusion Optics Interactive Touch Screen Display On/Off Button Stand (Back View) Built-In Battery Reset Button AC Inlet Stand Storage Cavity 8
10 Using Your golite BLU Plugging In Your golite BLU Your golite BLU comes with a 9V AC/DC adapter with US-style prongs. Your adapter can operate on AC power ranging from volts for usage worldwide. Use the enclosed adapter only. For safety and proper operation of your golite BLU, never use an unapproved adapter. Charging Instructions IMPORTANT: Charge your golite BLU for at least 8 hours before initial use. The golite BLU contains an internal Lithium-Ion battery pack. To charge your golite BLU, connect your AC/DC adapter to a wall outlet and plug the golite BLU into the adapter. The battery display icon will show the outline only. As the battery charges, the four bars will blink one by one and fill in as the charge increases. Once the battery is charged, the battery icon will switch to a steady display. Charge your golite BLU for at least 8 hours before initial use. Subsequent charges should be complete in three to four hours. A single charge should last for three to five treatment sessions. WARNING: Do not incinerate, disassemble, or expose to high temperatures (above 60 C/140 F). Internal batteries may overheat causing fire, or bursting. Use the specified AC/DC adapter only. Using Your golite 9
11 Using Your golite Positioning Your golite BLU Use the golite BLU at a slight angle to your face (see figure 1). It is not necessary to look directly into the light. For best results, place the golite BLU approximately inches from your eyes. Allow the light to bathe your face from the side. You can read, eat, work at the computer, watch TV, put on makeup, or exercise while using your golite BLU. Use your golite BLU in a well-lit room to minimize eyestrain. For best results, place the golite BLU near you at eye level so it shines directly into your eyes. Figure 1 The proper response to light therapy occurs through the retina of the eyes, not through the skin. However, you do not need to stare at the light. If you are in front of the golite BLU and position it close enough, only an occasional glance is needed achieve the maximum benefits. Your Eyes Should Be Open for Optimal Results The hormonal response works through the eyes, not the skin. This is not a tanning light. In fact, our BLUEWAVE Technology creates specific wavelengths that produce a circadian response while eliminating UV hazards and minimizing eyestrain. 10
12 User Interface Display Refer to the following two figures when using your golite BLU. Alarm Button Plus (+) Button Using Your golite Timer Button Figure 2 - Interactive Touch Screen Buttons Minus (-) Button Alarm Chime Alarm Light Timer: Minutes Remaining Demonstration Mode Battery Charge Alarm/Timer LED Intensity Clock Figure 3 - Display Elements 11
13 Using Your golite 12 Instructions for Use To begin using the golite BLU, remove the stand from the storage cavity on the back and place the beveled end into the hole also on the back of the device. The stand is held in place in both the storage cavity and the hole with magnets. Turning on the golite BLU Display Push the ON/OFF button on the side of the device to turn the golite BLU display on. To turn the display off, press and hold the ON/OFF button for 2 seconds. When the device is on battery power only, after 5 minutes of inactivity, the golite will shut off automatically to preserve battery power. Setting the Clock While the golite BLU is on, press and hold both the ALARM and TIMER buttons for 2 seconds. The clock display will flash while setting the clock. Press either the PLUS (+) or MINUS (-) button to adjust the clock. Hold down either the PLUS or MINUS button to cycle through the clock time quickly. When finished, press either the ALARM or TIMER button, or wait 10 seconds to exit the clock setting mode. To switch between 12-hour (a.m. & p.m.) or 24-hour time format, press and hold both the TIMER and MINUS buttons for 2 seconds. Timing a Session To start a timed session, press the TIMER button. The timer icon and minutes will flash. Select the desired usage time by pressing either the PLUS or MINUS button. Hold down either the PLUS or MINUS button to cycle through the clock time quickly. The timer can be set between 1 and 60 minutes. This setting will then be stored in the device memory. When the light source is turned on, it will count down from this set time. When it has finished counting down, the session will end automatically. Turning on the Light While the golite BLU is on, press and release the ON/OFF button on the side of the device to turn the light source on or off.
14 Pausing a Session Press the ON/OFF button, or press the MINUS button until the intensity reaches 0% to pause a timed session. Minutes remaining will still show in the display. Press the ON/OFF button again to continue the session, or press any of the 4 touch screen buttons to end the session. If you pause a session while on battery power only, after 5 minutes of inactivity, the session will end and the golite will shut off automatically. Changing the Intensity While the light source is on, you can change light intensity by pushing either the PLUS or MINUS button. You can choose intensity settings in increments of 25%. Setting the Alarm The alarm can be set to 3 possible modes: chime only, light only, or both chime and light. To specify the mode, press the ALARM button. The alarm mode icons will begin to flash. Press either the PLUS or MINUS button to cycle between the 3 modes. Once you make your choice, press the ALARM button twice or wait 10 seconds to store your setting. To set the alarm time, press the ALARM button twice. The alarm time will begin to flash. Press either the PLUS or MINUS button to set the desired time. Hold down either the PLUS or MINUS button to cycle through the alarm time quickly. Once finished, press the ALARM button or wait 10 seconds to store your alarm setting. This setting will then be stored in the device memory as long as there is backup battery power. Silencing the Alarm Sound To turn off the alarm while it is sounding, press any of the 4 touch screen buttons or the ON/OFF button on the side of the golite BLU. Disabling the Alarm To disable the alarm, press and hold the ALARM button until the alarm icon and alarm time disappear. Using Your golite 13
15 Suggested Usage 14 When to Use Your golite BLU Because an imbalance in circadian rhythm affects everybody differently, understanding how it affects you is critical to feeling better. Knowing what time to use the golite BLU can mean the difference between responding in just a few days instead of weeks, or perhaps not responding at all. Follow these guidelines to understand how light affects your body clock and when to use the golite BLU for the best results. Usage Duration and Intensity Most people get beneficial results from fifteen to thirty minutes of usage each day. You can start using your golite BLU at 50% intensity and increase the level of light if you feel like your body is not responding within a week of beginning light therapy. Time of Day If you find it difficult getting up each day without an alarm clock or you are sleeping in, you need morning light. Use the golite BLU within the first hour or two after waking. If you find it difficult staying awake in the late evening and wake up too early in the morning, you need evening light. Use the golite BLU before the onset of drowsiness. If you don t notice a shifted circadian rhythm, use the golite BLU when you wake up in the morning for fifteen to thirty minutes and then again in the evening for approximately the same amount of time. Place the golite BLU further away from you during the evening treatment. This will keep your body clock from confusing the stronger morning signal with evening time. Don t use the golite BLU within two hours of bedtime. The golite BLU suppresses the nighttime hormones for at least two hours, and you may not be able to fall asleep.
16 Shift Work Use the following techniques to adjust your circadian rhythm to your work shift. Beginning a Night-shift Schedule Many people benefit from receiving a 30-minute burst of light before work and then 15 minutes of light every three hours during work. This not only increases alertness and performance, but shifts the body clock as well. Within three days, your body clock will have changed to a night-shift schedule. Taking time-release melatonin upon arriving home after work may help accelerate the shifting of your body clock. After the second day, melatonin should not be necessary, and your body clock will have adapted to the new schedule. Note: When coming off of night-shift work, it is important to wear sunglasses and avoid any bright light while going home. You should go to sleep as soon as possible after leaving work. Returning to a Day-shift Schedule On the day your night shift ends, continue to wear sunglasses and avoid bright light in the morning hours. Do not take melatonin in the morning. Use your golite BLU for 30 minutes at about midday. On the first day, you may take an afternoon nap if you are having trouble staying awake. Go to bed at your normal evening bedtime. You may wish to use time-release melatonin in the early evening (about 7:00 p.m.) to accelerate your body clock to a normal schedule. You may want to wait a few hours after waking in the afternoon or evening before taking melatonin as it may cause drowsiness. This routine should only need to be repeated once or twice before you achieve a normal sleep/wake schedule. After the first day, you should use the golite BLU each morning as needed for approximately minutes. Suggested Usage 15
17 Suggested Usage 16 For more detailed information about using your golite BLU to help adapt to shift work schedules, visit Jet Lag Rapidly traveling across several time zones creates disharmony between your internal body clock and the new external time. This not only causes problems with sleeping, but influences mood and energy as well. Because specialized light is effective for shifting circadian rhythms, your golite BLU can quickly adjust your rhythm to the new time zone. The best strategy for combating jet lag is to shift your body clock a day or two before leaving. Traveling East Use your golite BLU in the morning two to three days before leaving. The number of time zones traveled corresponds to the hours needed to shift. On the first day, wake up an hour early and use the golite BLU for minutes. This will shift your body clock up to three time zones. The next day(s) you need to awaken a couple of hours earlier and repeat this process. Taking time-released melatonin in the early evening or late afternoon can also help accelerate this shift. You should adjust your sleep schedule as described each day. Upon arrival, wear sunglasses to avoid any sunlight before 10:00 a.m. When returning, use your golite BLU in the evening for a few nights and avoid any morning light (see Traveling West). Traveling West Use your golite BLU in the evening for minutes a few days before leaving, and use the golite BLU a couple of hours later each successive day. Each day you use the golite BLU, you shift your internal time up to three hours, and you can increase this shift up to six hours per day by taking melatonin in the morning. When returning, avoid any bright afternoon and evening light and be sure to use your golite BLU in the morning for a few days (see Traveling East).
18 Frequently Asked Questions Do I have to take off my glasses when I use the golite BLU? No. This specialized type of light is designed to work with or without glasses. However, you should not wear tinted glasses or sunglasses when using the golite. Are there any major side effects? The golite BLU is safe when used according to recommendations. There have been some reports of temporary headaches or nausea. This is usually resolved by sitting further away from the unit. Can I get sunburn from using the golite BLU? No. While typical fluorescent bulbs produce small amounts of UV light, BLUEWAVE Technology eliminates all UV light. I ve heard that light can be dangerous. Can the golite BLU damage my eyes? The golite BLU emits safe levels of blue light, according to current standards. In fact, your golite BLU emits less blue light than other traditional forms of bright light therapy like full-spectrum, 10,000 lux white fluorescent light. However, if you suffer from an eye disease or have a history of eye disease in your family, you should consult your ophthalmologist before using your golite BLU. Also, if you are experiencing any temporary eye problems, wait until the problems are resolved before using the golite BLU. When used as directed, all of our light products are free from ultraviolet light risk to the eye and produce a safe, diffused field of light. Our lights have been tested for ocular safety and have been used in dozens of government and clinical studies. As with any bright light source, you should not stare directly into the light. The beneficial reaction happens when the melanopsin photoreceptors in the eye are stimulated. Since these photoreceptors lie in the peri phery of the eye, you will get a much better outcome if you position the golite BLU at a slight angle to your face. FAQ 17
19 FAQ 18 Should I use the golite BLU every day? You should use the golite BLU daily, especially during the winter months. Many people find that they don t need the light during the summer. Others use it year-round because they spend most of their time indoors and don t typically get the right kind of light signals to regulate circadian rhythms properly. Can the light cause harm to children or pets? No. The lights are safe around children and pets. The lights will actually draw many pets to sit by it. Can I use the golite BLU more than once a day? For most people, it is sufficient to use the golite BLU once a day. Keep in mind that using the golite BLU too late in the evening may adversely affect your sleep. How long do the LEDs last? Under normal usage, the LEDs are rated for 10,000 hours of use, which at 30 minutes per day all year round equals over 50 years of life. What if one or more LEDs stop working prematurely? In the unlikely event that one or two LEDs fail prematurely, the effectiveness of your golite BLU will not be diminished. You can continue to use your golite BLU normally. However, if several LEDs stop functioning, you should probably have your golite BLU repaired to ensure that you are receiving the correct treatment dosage. For more information, refer to your golite BLU warranty card. What is DEMO on the display screen? This is the Demonstration mode that is used to display the golite BLU at point-of-sale locations. If your golite BLU happens to be in this mode, DEMO will appear on your display screen. You will not need to be in this mode. While the golite BLU is off, press and hold the ON/OFF button for 10 seconds. The golite BLU will turn on and the word DEMO will clear from your display screen. How do I use the reset button? The golite has a reset button located in the stand storage cavity on the back of the unit. If the golite locks up or is not working correctly, you can reset the device by inserting the end of a paper clip into the reset button hole. The golite will reset and the memory contents will be cleared.
20 Maintenance Cleaning The golite BLU should only be cleaned with a non-abrasive material such as a dry cleaning cloth. The device should never be immersed in any liquid. Service and Repair We don t expect you to have any problems with your golite BLU, but occasionally issues may arise. Repair Information and Troubleshooting: For technical questions and troubleshooting, please visit Returning Your Light: Before returning your golite BLU for servicing, you must first obtain a return authorization number. You can obtain this number at or contact our Service Department by calling the telephone number located on the back of this manual. Maintenance 19
21 Specifications 20 Specifications Environmental Shelf Life: 2 years at 25 C (77 F) Operating Temperature: -5 to 35 C (23 to 95 F) Storage Temperature: -20 to 60 C (-4 to 140 F) Relative Humidity (operating & storage): 15 to 90% (non-condensing) Physical Dimensions: 5.5 x 5.5 x 1 inches (14 x 14 x 2.5 cm) Weight: < 14 ounces (0.4 kg) Standards Compliance This device is designed to conform to the following standards: IEC General Requirements for Safety of Medical Electrical Equipment IEC Electromagnetic Compatibility IEC :1993+A1:1997+A2:2001 (Class 1 LED Product) IEC Classification Type of Protection Against Electric Shock: Internally powered equipment Degree of Protection Against Harmful Ingress of Water: IPX0 (Ordinary protection against ingress of liquids) Mode of Operation: Continuous Operation Not suitable for use in the presence of a flammable anaesthetic mixture with air, oxygen, or nitrous oxide. Electrical golite BLU DC Power Consumption: 9.0 V DC, 1.1 A AC/DC Adapter Power Consumption: VAC, 50/60 Hz, 0.4 A AC/DC Adapter: (model number: MW170KB0900B02) 9.0 V DC, 1.5A As of the date of publication, this is the complete list of approved AC/DC adapters for use with the golite BLU. Contact customer service for an updated list of approved adapters. Internal Battery: (part number: ) 2-cell rechargeable lithium-ion pack, 7.4 V, 850 mah Complies with IEC and UL 2054 Disposal Dispose of the device in accordance with local regulations.
22 EMC Information Gu id a n c e a n d Ma n u f a c t u re r s De c l a r a t i o n - Electromagnetic Emissions: The golite BLU is intended for use in the electromagnetic environment specified below. The user of the golite BLU should make sure it is used in such an environment. Emissions Test Compliance Electromagnetic Environment - Guidance EMC Information RF emissions CISPR 11 RF emissions CISPR 11 Harmonic emissions IEC Voltage fluctuations/flicker emissions IEC Group 1 Class B Class A Complies The golite BLU uses RF energy only for its internal function. Therefore, its RF emissions are very low and are not likely to cause any interference in nearby electronic equipment. The golite BLU is suitable for use in all establishments, including domestic establishments and those directly connected to the public low-voltage power supply network. 21
23 EMC Information Gu id a n c e a n d Ma n u f a c t u re r s De c l a r a t i o n - Electromagnetic Im m u n i t y: The golite BLU is intended for use in the electromagnetic environment specified below. The user of the golite BLU should make sure it is used in such an environment. Im m u n it y Te s t Electrostatic Discharge (ESD) IEC Electrical Fast Transient/Burst IEC Surge IEC Voltage dips, short interruptions and voltage variations on power supply input lines IEC Power frequency (50/60 Hz) magnetic field IEC IEC Te s t ±6 kv contact ±8 kv air Le v e l ±2 kv for power supply lines ±1 kv for input-output lines ±1 kv differential mode ±2 kv common mode <5% U T (>95% dip in U T ) for 0.5 cycle 40% U T (60% dip in U T ) for 5 cycles 70% U T (30% dip in U T ) for 25 cycles <5% U T (>95% dip in U T ) for 5 sec Compliance Level Electromagnetic Environment - ±6 kv contact ±8 kv air ±2 kv for supply mains ±1 kv for input/output lines ±1 kv differential mode ±2 kv for common mode <5% U T (>95% dip in U T ) for 0.5 cycle 40% U T (60% dip in U T ) for 5 cycles 70% U T (30% dip in U T ) for 25 cycles <5% U T (>95% dip in U T ) for 5 sec Gu i d a n c e Floors should be wood, concrete or ceramic tile. If floors are covered with synthetic material, the relative humidity should be at least 30%. Mains power quality should be that of a typical home or hospital environment. Mains power quality should be that of a typical home or hospital environment. Mains power quality should be that of a typical home or hospital environment. If the user of the golite BLU requires continued operation during power mains interruptions, it is recommended that the golite BLU be powered from an uninterruptible power supply or a battery. 3 A/m 3 A/m Power frequency magnetic fields should be at levels characteristic of a typical location in a typical hospital or home environment. NOTE: U T is the a.c. mains voltage prior to application of the test level. 22
24 Gu id a n c e a n d Ma n u f a c t u re r s De c l a r a t i o n - Electromagnetic Im m u n i t y: The golite BLU is intended for use in the electromagnetic environment specified below. The user of the golite BLU should make sure it is used in such an environment. Immunity Test IEC Test Level Compliance Level Electromagnetic Environment - Guidance Portable and mobile RF communications equipment should be used no closer to any part of the golite BLU, including cables, than the recommended separation distance calculated from the equation applicable to the frequency of the transmitter. EMC Information Conducted RF IEC Vrms 150 khz to 80 MHz 3 Vrms Recommended separation distance: d = khz to 80 MHz Radiated RF IEC V/m 80 MHz to 2.5 GHz 3 V/m d = 1.2 d = MHz to 800 MHz 800 MHz to 2.5 GHz Where P is the maximum output power rating of the transmitter in watts (W) according to the transmitter manufacturer and d is the recommended separation distance in meters (m). Field strengths from fixed RF transmitters, as determined by an electromagnetic site survey a, should be less than the compliance level in each frequency range b. Interference may occur in the vicinity of equipment marked with the following symbol: NOTE 1: At 80 MHz and 800 MHz, the higher frequency range applies. NOTE 2: These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection from structures, objects, and people. a: Field strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and land mobile radios, amateur radio, AM and FM radio broadcast and TV broadcast cannot be predicted theoretically with accuracy. To assess the electromagnetic environment due to fixed RF transmitters, an electromagnetic site survey should be considered. If the measured field strength in the location in which the golite BLU is used exceeds the applicable RF compliance level above, the golite BLU should be observed to verify normal operation. If abnormal performance is observed, additional measures may be necessary, such as re-orienting or relocating the golite BLU. b: Over the frequency range 150 khz to 80 MHz, the field strengths should be less than 3 V/m. 23
25 EMC Information Recommended Se p a r a t i o n Distances between Po r t a b l e a n d Mo b i l e RF Co m m u n i c a t i o ns Eq u i p m e n t a n d Th e GoLITE BLU: The golite BLU is intended for use in an electromagnetic environment in which radiated RF disturbances are controlled. The customer or the user of the golite BLU can help prevent electromagnetic interference by maintaining a minimum distance between portable and mobile RF communications equipment (transmitters) and the golite BLU as recommended below, according to the maximum output power of the communications equipment. Ra t e d Ma x i m u m Po w e r Output of Transmitter (W) 150 khz t o 80 MHz d = 1.2 Se p a r a t i o n Di s t a n c e Ac c o r d i n g t o Fr e q u e n c y o f Tr a n s m it t e r (m) 80 MHz t o 800 MHz d = MHz t o 2.5 GHz d = For transmitters rated at a maximum output power not listed above, the recommended separation distance d in meters (m) can be estimated using the equation applicable to the frequency of the transmitter, where P is the maximum output power rating of the transmitter in watts (W) according to the transmitter manufacturer. Note 1: At 80 MHz and 800 MHz, the separation distance for the higher frequency range applies. Note 2: These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection from structures, objects, and people. 24
26 947 South 500 East, Suite 210 American Fork, Utah USA Toll Free International golite and BLUEWAVE are trademarks of Apollo Health and its affiliates JR 8/05/2008
ENGLISH Lightphoria 10,000LUX Energy Light Lamp SP9982 version 2.4 1 Light and your body Light means life. Many of the processes of life are influenced by light and its intensity. As many of you may experience
5392 Dual Chamber Temporary External Pacemaker Electromagnetic Compatibility Declaration Caution: Federal law (USA) restricts this device to sale by or on the order of a physician. The following list includes
1053621 EN, DE DSF 4/16/08 EverFlo EverFlo Q English Table of Contents Symbol Key... 2 Abbreviations... 2 Chapter 1: Introduction... 3 Intended Use... 3 About Your EverFlo / EverFlo Q...3 Parts of Your
Rev. 1:01/15 Digital Thermometer Instruction Manual Model:FT7122 (MT-B321FB) CONTENTS What is normal body temperature? ----------------------------------------- 1 Section 1: Precautions for use ----------------------------------------
Heated HumidifieR M Series USER Manual 1031849 HM 1/6/06 Table of Contents Introduction...3 In This Manual...3 Intended Use...3 Warnings and Cautions...3 Warnings...3 Cautions...4 Symbol Key...5 System
Universal Battery Charger II. Instructions for Use This publication is not intended for distribution in the USA. Instruments and implants approved by the AO Foundation Table of Contents Introduction General
Please read this user manual carefully and thoroughly to ensure the safe usage of the product. Keep a copy of this manual handy for future reference. Table of Contents Introduction... 2 Safety Information...
BiPAP Auto Bi-Flex PROVIDER GUIDE IMPORTANT! Remove this guide before giving the device to the patient. Only medical professionals should adjust pressure settings. This guide provides you with instructions
Electronic Stethoscope MODEL 3000 with Ambient Noise Reduction Made in Denmark for 3M Health Care, St. Paul, MN 55144 (U.S.A.) 1 800 228-3957 Fax 651 736-2803 Visit our web site: http://www.3m.com/littmann
PR-D7 GB Revision 1 Important Safety Instructions 1. Read these instructions. 2. Keep these instructions. 3. Heed all warnings. 4. Follow all instructions. 5. Do not use this apparatus near water. 6. Clean
Home Care Provider Setup Instructions Always use these instructions along with the User Manual when assembling or adjusting this equipment. For clinical systems, refer to the setup guide entitled Respironics
GLOW CLOCK alarm clock with night light CR10 user guide CAUTION RISK OF ELECTRIC SHOCK DO NOT OPEN CAUTION: TO REDUCE THE RISK OF ELECTRIC SHOCK, DO NOT REMOVE COVER (OR BACK). NO USER-SERVICEABLE PARTS
Operating Instructions Vision 5L Digital Holter Recorder Document Number: 010-1646-00 Revision: H Bothell, Washington 98021-8969 (608)-764-1919 Caution: U.S. Federal law restricts this device to sale by
THE NEW GENERATION IN TECHNOLOGY NI-3103A Alarm Clock w/fm Radio ipod & iphone Docking Operating Instructions Table of Contents Important Safeguards... 3 Parts... 6 Controls... 6 To Set Up... 7 ipod/iphone
Instruction Manual & Warranty Information Before you operate this product, please read these instructions. Keep this manual in a safe place for future reference. Please read all safety guidelines before
Sonic Boom Travel & Bed side Alarm Clock with Bed Shaker Model SB200ss-v3 IMPORTANT Please read these instructions carefully before use and retain for future reference. See operating instructions on back.
INSTRUCTION MANUAL PLEASE READ ALL THE INSTRUCTIONS COMPLETELY BEFORE USE AND SAVE THIS MANUAL FOR FUTURE REFERENCE m Before Use Please read IMPORTANT SAFETY INSTRUCTIONS on page 10 before use. It is important
Single Day Alarm Clock with Bluetooth and USB Play-through MODEL : BSC 200 READ THIS INSTRUCTION MANUAL BEFORE OPERATING THE APPLIANCE. SAVE THIS MANUAL FOR FUTURE REFERENCE. TABLE OF CONTENTS IMPORTANT
Rechargeable Power Pack User s Manual 2013 VTech Printed in China 91-009656-000 US INTRODUCTION The Rechargeable Power Pack makes it easier than ever to keep the InnoTab 3 or InnoTab 3S charged and ready
G-GRIP PORTABLE WIRELESS SPEAKER Rechargeable Battery 8 Hours Play Time Compact & Powerful Bass Boom Radiator GRP Glass-Reinforced Polymer Chassis Welcome to G-PROJECT Thanks for purchasing a G-GRIP and
INSTRUCTION MANUAL PLEASE READ ALL THE INSTRUCTIONS COMPLETELY BEFORE USE AND SAVE THIS MANUAL FOR FUTURE REFERENCE Revision 2-3/25/13 Introduction/ Safety Instructions Introduction The CC Pocket radio
VEC173 ONE YEAR LIMITED WARRANTY PROGRAM This limited warranty program is the only one that applies to this product, and it sets forth all the responsibilities of Vector Manufacturing, regarding this product.
System One Heated Humidifier USER MANUAL 2009 Respironics Inc. and its affiliates. All rights reserved. Table of Contents Intended Use... 2 Warnings... 2 Cautions... 2 Symbol Key... 2 System Features and
Home Automation, Inc. Model 70A00-1 GSM Cellular Communications Center Operation Manual Document Number 70I00-1 Rev A August, 2009 Contents Description... 1 Use with security systems...1 Everyday use...2
DENVER CRP-618 ENGLISH 15 16 14 17 18 19 20 FUNCTION AND CONTROLS 1. PROJECTION ON/OFF 2. PROJECTION TIME 180 3. SNOOZE / SLEEP / DIMMER /DST 4. ON OFF RADIO / ALARM OFF 5. DISPLAY 6. ALARM 2 INDICATOR
Pilot-12 Plus Pilot-24 Plus CPAP BATTERY / BACKUP POWER SUPPLY CPAP BATERIA / ALIMENTACIÓN PORTÁTIL USTED BATTERIE CPAP / ALIMENTATION PORTABLE ENGLISH - Thank you for purchasing MEDISTROM Pilot-12 Plus/Pilot-24
G-BUZZ PORTABLE ALARM CLOCK Dual Alarm FM Radio Wake-up Sounds USB Phone Charger G-1CR Welcome Alarm clocks can be boring. Get ready to shake things up with your new G-BUZZ. Slap the snooze for more ZZZ
RCR-22 AM/MW 2 bands Digital Clock Radio instruction manual Important safety instructions 1. Read and understand all safety and operating instructions before the radio is operated. 2. Retain instruction:
USER GUIDE MODEL: TA-7830US-A00 LOCATION & DESCRIPTION OF CONTROLS 1 2 3 SNOOZE button DC Power jack DC Power lock screw hole NAP indicator ALARM ON / OFF indicator 7 AM / PM / DST time indicator USB 1A
450VA/650VA/850VA User s Manual Line Interactive UPS Uninterruptible Power System Table of Contents Safety Instructions Installation and Operation Specifications System Introduction Software Installation
Installation and Operation C-Type Power Filter C20B/C25B General Information The APC AV C-Type Power Filters C20B/C25B protect high performance audio and video system equipment from damage due to power
StarLight 3000 OPERATING INSTRUCTIONS h h 2 4 1 1. Quick Start Power Button 2. SOS/Flashlight Power Switch 3. USB Out/In 4. Fuel Gauge (Battery Status Indicator) 3 Figure 1 A B Included items: - 1 HALO
U3 GB Version 1 IMPORTANT SAFETY INSTRUCTIONS 1. Read these instructions. 2. Keep these instructions. 3. Heed all warnings. 4. Follow all instructions. 5. Do not use this apparatus near water. 6. Clean
DIGITAL ALARM CLOCK RADIO INSTRUCTION MANUAL For information and support, www.lenco.eu CR3302 LENCO ENG FM ONLY IM 98 x1 1 11/5/2009 9:14:35 CAUTION RISK OF ELECTRIC SHOCK DO NOT OPEN WARNINGS ATTENTION
CES25 Wireless Stereo Speaker System User Manual Please read carefully before using this equipment and retain for future use Introduction This wireless speaker system uses latest wireless technology that
20-225.fm Page 1 Wednesday, August 4, 1999 10:07 AM Cat. No. 20-225 OWNER S MANUAL Please read before using this equipment. DX-395 FM/AM/SW1/SW2 PLL Synthesized Receiver 20-225.fm Page 2 Wednesday, August
Digital Photo Picture Frame Operation Manual 20070309 CONGRATULATIONS on your purchase of a Polaroid 7 LCD Digital Photo Picture Frame. Please read carefully and follow all warnings and instructions in
PR-D9W Version 1 Table of contents Important safety instructions... 2-3 Introduction... 4 Controls... 5-8 Using your weather alert radio for the first time... 9 Operating your radio Search tuning AM/FM...
XEMIO-244 / 254 2GB/4GB MP3 PLAYER WITH CLIP User manual www.lenco.com Cautions Do not place naked flames, such as lighted candles on or near the product. Do not place objects filled with liquids, such
FLUORESCENT UV- RING LIGHT OPERATING INSTRUCTION Caution! UV-radiation of this device is in the range of UV-A (320-400 nm). Direct exposure to eyes shall therefore be avoided. UV protection glasses shall
Always here to help you Register your product and get support at www.philips.com/welcome Question? Contact Philips Wake-up Light HF3510 User manual Manuel d utilisation IMPORTANT SAFETY INSTRUCTIONS WARNING
Model SBB500SS-v3 Sonic Boom Alarm Clock & Bed Shaker OPERATION INSTRUCTIONS IMPORTANT Please read these instructions carefully before use and retain for future reference. See operating instructions on
12 Volt 30 Amp Digital Solar Charge Controller User s Manual WARNING Read carefully and understand all INSTRUCTIONS before operating. Failure to follow the safety rules and other basic safety precautions
Model: 616-146v2 Quick Setup Guide DC: 071015 Atomic Projection Alarm with Indoor and Outdoor Temperature Snooze/Backlight BUTTONS Time, Alarm with Snooze, & Calendar Projection Arm Rotates 180 Indoor/Outdoor
Wireless accessories Remote Control 3.0 Instructions for use Introduction to this booklet This booklet guides you on how to use and maintain your new Remote Control 3.0. Please read the booklet carefully
HP ProDisplay P19A LED Backlit Monitor User Guide 2013 Hewlett-Packard Development Company, L.P. The only warranties for HP products and services are set forth in the express warranty statements accompanying
ENGLISH (Original instructions) INSTRUCTION MANUAL Radio BMR00 WARNING: For your personal safety, READ and UNDERSTAND before using. SAVE THESE INSTRUCTIONS FOR FUTURE REFERENCE. Symbols The following show
Mini Power Bank and Charger Owner s Manual Please read before using this equipment. Your charger is ETL listed to UL standards and meets all applicable FCC standards. WARNING: To reduce the risk of fire
LINE-INTERACTIVE UPS Uninterruptible Power System 600VA - 2000VA USER MANUAL IMPORTANT SAFETY INSTRUCTIONS The utilized voltages by this UPS may be hazardous. Do not attempt to disassemble the unit. The
+ Installation and Operation Manual Back-UPS BX800CI-AS/BX1100CI-AS Inventory Safety and General Information bu001c This unit is intended for indoor use only. Do not operate this unit in direct sunlight,
Owner's Manual AC/DC Clamp Meter Model No. 82369 CAUTION: Read, understand and follow Safety Rules and Operating Instructions in this manual before using this product. Safety Operation Maintenance Español
Owner's Manual AutoRanging Digital MultiMeter Model No. 82139 CAUTION: Read, understand and follow Safety Rules and Operating Instructions in this manual before using this product. Safety Operation Maintenance
350 WATT HYBRID INVERTER (SOLAR / INVERTER / AC MAINS) Users Manual Page 1 1. Introduction Thank you for purchasing the 350 Watt Hybrid Inverter. Connected to the 12 volt battery, the 350W Hybrid Inverter
How to use your The explained... Stay in bed When you activate your Gro-clock, 12 stars appear around a big star. To show the passing of time, the 12 stars disappear at even times during the sleep or nap
Model No. 2060 5 IN 1 RECHARGEABLE CAR HEATER Instruction Manual Sheet 1 of 8 IMPORTANT SAFETY INSTRUCTIONS: WARNINGS: Read, understand and follow all instructions, cautions and warnings listed in this
Register your product and get support at www.philips.com/welcome AS111 User manual Contents 1 Important 4 Safety 4 English 2 Your docking speaker for AndroidTM 5 Introduction 5 What's in the box 5 Overview
HDTV Anywhere USER MANUAL 3 20672/ 20140710 HDTV Anywhere ALL RIGHTS RESERVED MARMITEK 2 MARMITEK SAFETY WARNINGS To prevent short circuits, this product should only be used inside and only in dry spaces.
e 1 0 0 I N S T RUC TION M ANU A L The Empire Digital Level Accuracy, Readability, Durability one FEATURES: Easy to read back light Display reads right side up even if the level is upside down Audio beep
12-1632.fm Page 1 Wednesday, May 10, 2000 10:57 AM Cat. No. 12-1632 OWNER S MANUAL Please read before using this equipment. AM/FM PLL Clock/Table Radio 12-1632.fm Page 2 Wednesday, May 10, 2000 10:57 AM
Owner's Manual True RMS Multimeter Model No. 81077 CAUTION: Read, understand and follow Safety Rules and Operating Instructions in this manual before using this product. Safety Operation Maintenance Español
INVENTED BY REAL PEOPLE LIKE YOU Jason Hunter PORKFOLIO save big with this pig OVERVIEW Porkfolio is the world s smartest piggy bank. It wirelessly connects to the Wink app on your mobile device so you
echarger User Guide master Content Content Your echarger 3 Intended use 4 Important notes 5 Notes for batteries 8 How to charge 9 Useful tips for charging 14 Maintenance and care 15 Technical Data 16 Disposal
Instruction Manual WT 499 Radio Controlled Clock with Wake-up-Light and Mood Light P. 1 Words in the figure: Figure 1: ENGLISH OTHER LANGUAGE DATE OF WEEK ALARM 2 DATE / MONTH PM ALARM 1 TIME SNOOZE ICON
INSTALLATION GUIDE AXIS T81B22 DC 30W Midspan ENGLISH About this Document This document includes instructions for installing AXIS T81B22 on your network. Previous experience of networking will be beneficial
BeoVision 4 Guide CAUTION RISK OF ELECTRIC SHOCK DO NOT OPEN CAUTION: To reduce the risk of electric shock, do not remove cover (or back). No User-serviceable parts inside. Refer servicing to qualified
Changers Kalhuohfummi User Manual Contents 02 Contents 1. The Changers Kalhuohfummi an introduction to your solar battery 2. Introduction to energy tracking 3. Content and compatibility 4. Controls and
Cor1 BLOOD PRESSURE MONITOR INSTRUCTION MANUAL Congratulations on your My Life My Shop purchase. The Cor1 Blood Pressure Monitor will allow you to measure vital blood pressure parameters wherever you go.
User instructions TV-Dex 2 Contents Package Contents... 4 Your new TV-Dex...5 TV-Controller... 6 TV-Base...7 Light-emitting diodes...8 Getting started.... 9 Connecting the units:... 9 When using more than
SP1790JK 900MHz Wireless Indoor/Outdoor Speakers INTRODUCTION This 900 MHz digital hybrid wireless speaker system uses the latest wireless technology that enables you to enjoy music and TV sound anywhere
focus TOuCh P10 FOR DISPLAYS: 55, 70 AND 82 ENgliSh V01.3-2014-7 Thank you for purchasing our product. Please read this manual carefully before operation your set and retain it for future reference. FT55-P10
DAB+ / FM Tuner Model: TU-201 Instruction Manual www.pulse-audio.co.uk 1 Safety Information The lightning bolt within a triangle is intended to alert the user to the presence of dangerous voltage levels
AM/FM ALARM CLOCK RADIO Instruction Manual JL-206 Please read this manual carefully before operating and save it for future reference. Have a question? Please visit http:// INTRODUCTION Thank you for purchasing
U.S. Cellular Home Phone 1 Contents Getting to Know Your Device...5 Appearance... 5 LED Indicators... 6 Device Installation...7 Before You Begin... 7 Installing the Battery... 7 Installing the Power Adapter...
U P S U n i n t e r r u p t i b l e P o w e r S y s t e m 500VA/ 650VA/ 800VA User s Manual IMPORTANT SAFETY INSTRUCTIONS SAVE THESE INSTRUCTIONS WARNING (SAVE THESE INSTRUCTIONS): This manual contains
Mercury Helios 2 ASSEMBLY MANUAL & USER GUIDE TABLE OF CONTENTS INTRODUCTION...1 1.1 MINIMUM SYSTEM REQUIREMENTS 1.1.1 Apple Mac Requirements 1.1.2 PC Requirements 1.1.3 Supported PCIe Cards 1.2 PACKAGE
BeoLab 6002 Guide WARNING: To reduce the risk of fire or electric shock, do not expose this appliance to rain or moisture. Do not expose this equip ment to dripping or splashing and ensure that no objects