2 THE TA INSTRUMENTS DYNAMIC MECHANICAL ANALYZER The TA Instruments Q800 is the culmination of years of engineering expertise in Dynamic Mechanical Analyzers. TA Instruments pioneered DMA instrumentation with the 980 DMA introduced in 1976 and continues the tradition today. The Q800 is the sixth generation DMA and incorporates unique technology. Backed by the best customer support available, the TA Instruments Q800 DMA sets the standard for Dynamic Mechanical Analyzers. The most important features of any DMA are how the stress is applied to the sample, and how the resultant strain is measured. The Q800 DMA utilizes state-of-the-art non-contact, linear drive motor technology to provide precise control of the stress. Strain is measured using optical encoder technology that provides unmatched sensitivity and resolution. Along with low friction air bearing technology, the Q800 provides superior performance compared to traditional designs that employ LVDT technology, stepper motors, and mechanical springs.
3 TECHNICAL SPECIFICATIONS Maximum Force 18 N Minimum Force Force Resolution Strain Resolution Modulus Range N N 1 nanometer 10E 3 to 3x10E 12 Pa Modulus Precision +/- 1% Tan δ (delta) Range to 10 Tan δ (delta) Sensitivity Tan δ (delta) Resolution Frequency Range 0.01 to 200 Hz Dynamic Sample Deformation Range +/- 0.5 to 10,000 µm Temperature Range Heating Rate Cooling Rate Isothermal Stability Automated Furnace Movement Time/Temperature Superposition Software -150 to 600 C 0.1 to 20 C/min 0.1 to 10 C/min +/- 0.1 C Yes Yes OUTPUT VALUES Storage Modulus Complex/Dynamic Viscosity Time Loss Modulus Creep Compliance Stress/Strain Storage/Loss Compliance Relaxation Modulus Frequency Tan Delta (δ) Static/Dynamic Force Sample Stiffness Complex Modulus Temperature Displacement
4 DESIGN FEATURES AND BENEFITS The performance of a Dynamic Mechanical Analyzer is a function of its ability to apply a force and measure displacement. The Q800 DMA incorporates the latest technology for this purpose as described below. 1 Drive Motor The Q800 uses a non-contact, direct drive motor to provide the oscillatory or static force required. The motor is constructed of high performance composites that ensure low compliance and is thermostated to eliminate heat build-up even when using large oscillation amplitudes and high deformation forces. Sophisticated electronics enable the motor current to be rapidly adjusted in small increments. Benefits: The motor can deliver reproducible forces over a wide range and the force can be changed rapidly, enabling a wide range of material properties to be measured. 2 Air Bearings The non-contact drive motor transmits force directly to a rectangular air bearing slide. The slide is guided by eight porous carbon air bearings grouped into two sets of four near the top and bottom of the slide. Pressurized air or nitrogen flows to the bearings forming a frictionless surface that permits the slide to float. The slide, which connects to the drive shaft and sample 2 clamp, can move vertically 25 mm and its rectangular shape eliminates any twisting of the sample. Benefits: Very weak materials like films and fibers can be characterized with ease since the drive shaft can move with virtually no friction and the unique rectangular slide prevents twist in the sample. 3 Optical Encoder A high resolution linear optical encoder is used to measure displacement on the Q800 DMA. Based on diffraction patterns 3 of light through gratings (one moveable and one stationary), optical encoders provide exceptional resolution compared to typical LVDT technology. Benefits: Due to the excellent 1 nanometer resolution of the optical encoder, very small amplitudes can be measured precisely. This combined with the non-contact drive motor and air bearing technology provides excellent modulus precision and high Tan δ sensitivity, allowing the Q800 DMA to characterize a broad range of materials.
5 4 Low Mass, High Stiffness Sample Clamps The Q800 features a variety of sample clamps that provide for multiple modes of deformation. The clamps were designed using finite element analysis to provide high stiffness, with low mass and attach to the drive shaft with a dovetail connection. The clamps are simple to use and adjust, and each is individually calibrated to insure data accuracy. Benefits: A broad range of samples can be analyzed. The high stiffness minimizes clamp compliance, and the low mass ensures rapid temperature equilibration. The simple, yet elegant designs reduce the time necessary to change clamps and load samples Furnace The Q800 features a bifilar wound furnace that automatically opens and closes. Benefits: The furnace design combined with the Gas Cooling Accessory provides for efficient and precise temperature control over the entire temperature range, both in heating, cooling, and isothermal operation. The automatic furnace movement simplifies experimental setup. 6 Rigid Aluminum Casting The Q800 drive motor, air bearing slide assembly with optical encoder and air bearings are all mounted within a rigid aluminum casting that is temperature controlled. Benefits: The rigid aluminum housing minimizes system compliance and the temperature-controlled housing ensures precise data. 6 1
6 MODES OF DEFORMATION The Q800 offers all the major deformation modes required to characterize solid bars, elastomers, soft foams, thin films and fibers. The deformation modes include bending (single cantilever, dual cantilever, and 3-point bend), shear, compression, and tension. In addition, submersible compression and film tension clamps are available. DUAL/SINGLE CANTILEVER Dual Cantilever In this mode, the sample is clamped at both ends and either flexed in the middle (dual cantilever) or at one end (single cantilever). Cantilever bending is a good general purpose mode for evaluating thermoplastics and highly damped materials (e.g., elastomers). Dual cantilever mode is ideal for studying the cure of supported thermosets. Three clamp sizes are available: Dual/Single Cantilever 8/4mm (L), Up to 15mm (W) and 5mm (T) Dual/Single Cantilever 20/10mm (L), Up to 15mm (W) and 5mm (T) Dual/Single Cantilever 35/17.5mm (L), Up to 15mm (W) and 5mm (T) 3 POINT BEND In this mode, the sample is supported at both ends and force is applied in the middle. 3 point bend is considered a "pure" mode of deformation since clamping effects are eliminated. The 50 mm and 20 mm clamps feature a unique low-friction support using roller bearings that improves accuracy. Three clamp sizes are available: 3-pt bend, Small 3-pt bend, 50 mm 3-pt bend, 20 mm 5, 10, or 15mm (L). Up to 15mm (W) and 7mm (T) 50mm (L), Up to 15mm (W) and 7mm (T) 20mm (L), Up to 15mm (W) and 7mm (T) SHEAR SANDWICH In this mode, two equal sizes of the same material are sheared between a fixed outer plate and the moving center plate. This mode is ideal for gels, adhesives, high viscosity resins and other highly damped materials. Shear Sandwich 10mm square. Up to 4mm (T)
7 COMPRESSION In this mode, the sample is placed on a fixed flat surface and force is applied by an oscillating upper plate. Compression is suitable for low to moderate modulus materials (e.g., foams, elastomers), provided the material has some elasticity. With minor changes this mode can be used to make measurements of expansion or penetration using a static force. Two clamp sizes are available: Compression (small) Compression (large) 15mm diameter up to 10mm thick 40mm diameter up to 10mm thick Films TENSION In this mode, the sample is placed in tension between a fixed and moveable clamp. In oscillation experiments, a small static force is applied in order to prevent buckling. The clamps feature very simple mounting and because the drive shaft does not rotate, no twist is induced in the sample that can lead to inaccurate data. Two clamps are available: Fibers Tension (film) Tension (fiber) 5 to 30mm (L); up to 8mm (W) and 2mm (T) 5 to 30mm (L); 5 denier (0.57 tex) to 0.8mm diam. single fibers or fiber bundles SUBMERSIBLE CLAMPS Both film tension and compression clamps are available in submersible configurations. These clamps allow samples to be analyzed in a fluid environment up to 80 C.
8 ACCESSORIES GAS COOLING ACCESSORY The Gas Cooling Accessory (GCA) extends the operating range to subambient temperatures. The GCA uses cold nitrogen gas generated from controlled heating of liquid nitrogen. Automated filling of the GCA tank can be programmed to occur either after the scan is complete or during a run. This ability to automatically refill during the middle of a run is particularly useful during long DMA experiments typically encountered when generating data for Time/Temperature Superposition (TTS). AIR COMPRESSOR ACCESSORY The Q800 uses air bearings for support of the drive shaft. The air bearings use a clean, dry compressed air or nitrogen supply that typically comes from a centralized supply. In cases where this is not possible, the Air Compressor Accessory (ACA) is available. This self-contained air compressor provides the clean, dry air supply required for the air bearings. ACCESSORY KIT The standard accessory kit supplied with the Q800 provides all the necessary tools and calibration materials needed for operation. Included with the kit is the 35 mm dual/single cantilever clamp that is used for many common applications. A set of precision digital calipers is included for accurate sample dimension measurements. A torque wrench is supplied since when mounting samples, an accurate, reproducible torque must be applied to mounting screws on the sample clamp. Finally, a variety of calibration samples are provided and used for instrument calibration.
9 DMA THEORY Dynamic Mechanical Analysis (DMA) is a technique used to measure the mechanical properties of a wide range of materials. Many materials, including polymers, behave both like an elastic solid and a viscous fluid, thus the term viscoelastic. DMA differs from other mechanical testing devices in two important ways. First, typical tensile test devices focus only on the elastic component. In many applications, the inelastic, or viscous component, is critical. It is the Range of Material Behavior viscous component that determines properties such as impact resistance. Second, tensile test Solid Like Liquid Like devices work primarily outside the linear Ideal Solid Most Materials Ideal Fluid viscoelastic range. DMA works primarily in the Purely Elastic Viscoelastic Purely Viscous linear viscoelastic range and is therefore more Viscoelasticity: Having both viscous and elastic properties sensititive to structure. DMA measures the viscoelastic properties using either transient or dynamic oscillatory tests. Transient tests include creep and stress relaxation. In creep, a stress is applied to the sample and held constant while deformation is measured vs. time. After some time, the stress is removed and the recovery is measured. In stress relaxation, a deformation is applied to the sample and held constant, and the degradation of the stress vs. time is measured. Both tests are used to assess the viscoelastic nature of materials. The most common test is the dynamic oscillatory test, where a sinusoidal stress is applied to the material and a sinusoidal strain is measured (Figure 1). Also measured is the phase difference, δ, between the two sine waves. The phase lag will be zero degrees for purely elastic materials and 90 degrees for purely viscous materials. Polymers will exhibit an intermediate phase difference. Since modulus is stress/strain, the complex modulus, E*, can be calculated. From E* and the measurement of δ, the storage modulus, E l, and loss modulus, E ll, can be calculated as illustrated in Figure 2. E l, the storage modulus is the elastic component and related to the samples stiffness. E ll, the loss modulus, is the viscous component and is related to the samples ability to dissipate mechanical energy through molecular motion. The tangent of phase difference, or Tan δ, is another common parameter that provides information on the relationship between the elastic and inelastic component. These parameters can be calculated as a function of time, temperature, frequency, or amplitude (stress or strain) depending on the application.
10 MODES OF OPERATION The Q800 DMA provides multiple operating modes that facilitate a wide variety of common experiments. The modes include: Multi-Frequency Mode: The multi-frequency mode can assess viscoelastic properties as a function of frequency, while oscillation amplitude is held constant. These tests can be run at single or multiple frequencies, in time sweep, temperature ramp, or temperature step/hold experiments. Multi-Stress/Strain Mode: In this mode, frequency and temperature are held constant, and the viscoelastic properties are monitored as % strain or the stress is varied. This mode is primarily used to identify the Linear Viscoelastic Range (LVR). Creep/Stress Relaxation Mode: With creep, the stress is held constant and deformation is monitored as a function of time. In stress relaxation, the deformation is held constant and the stress is monitored vs. time. Time Controlled Force/Strain Rate Mode: In this mode, the temperature is held constant while stress or strain is ramped at a constant rate. This mode is used to generate stress/strain plots to obtain Young s Modulus. Alternatively, stress can be held constant with a temperature ramp while strain is monitored. Isostrain Mode: With this mode, strain is held constant during a temperature ramp. Isostrain can be used to assess shrinkage force in films and fibers. Strain Temperature
11 APPLICATIONS MEASUREMENT OF Tg OF A PRESSURE SENSITIVE ADHESIVE (PSA) A common measurement on polymers is the glass transition temperature, Tg. It can be measured with various techniques, but DMA is by far the most sensitive. Figure 3 shows a scan of a pressure sensitive adhesive run in the tension clamps at a frequency of 1 Hz. Tg can be measured by the E l onset point, by the E ll peak, or the peak of Tan δ. In addition to the Tg, the absolute value of the various viscoelastic parameters is also useful. FREQUENCY EFFECT ON MODULUS AND GLASS TRANSITION OF POLYETHYLENE TEREPHALATE (PET) Because the Tg has a kinetic component, it is strongly influenced by the frequency (rate) of deformation. As the frequency of the test increases, the molecular relaxations can only occur at higher temperatures and, as a consequence, the Tg will increase with increasing frequency as illustrated in Figure 4. In addition, the shape and intensity of the Tan δ peak as well as the slope of the storage modulus in the transition region will be affected. Based on end-use conditions, it is important to understand the temperature and frequency dependence of transitions. THE MEASUREMENT OF SECONDARY TRANSITIONS IN VINYL ESTER DMA is one of the only techniques that can measure β and γ secondary transitions. Secondary transitions arise from side group motion with some cooperative motion from the main chain as well as internal motion within a side group. The transitions, as shown in Figure 5, are below the Tg and typically subambient. They are very important as they affect impact resistance and other end-use properties. This data was generated using 3-point bending and also illustrates the ability to run stiff composites.
12 EFFECT OF PLASTICIZER CONTENT ON PVC Figure 6 illustrates another very common application for DMA. This data on Polyvinyl Chloride (PVC) was generated using the single cantilever bending mode. The three samples have varying amounts of plasticizer which dramatically effects both the absolute value of modulus as well as the Tg. Assessing the effects of plasticizer and other additives such as fillers, modifiers, and colorants on modulus and transition temperatures is a common use for DMA. CHARACTERIZING PRINTED CIRCUIT BOARDS Printed Circuit Boards (PCB) are typically comprised of fiberglass braid impregnated with a thermosetting resin. Characterizing the Tg of PCB s is often difficult due to the very low amount of resin used. Figure 7 shows a typical PCB run in single cantilever bending. The Tg is clearly discernible and the difference between the as received and post baked sample clearly shows the effect that further crosslinking has on both the Tg and the absolute value of modulus. SUBAMBIENT CHARACTERIZATION OF SOFT FOAMS Figure 8 shows a single cantilever bending experiment on a soft foam material. These materials can also be characterized using shear and compression modes, but the single cantilever mode provides information well below the Tg. This data also illustrates the effectiveness of the Q800 for characterizing materials that exhibit very low stiffness from the glassy plateau, through the transition region, and well into the rubbery plateau. This is only possible on instruments that have superior stress and strain control.
13 CHARACTERIZING PACKAGING FILMS USING CREEP In a thermoforming process, a film is pulled down into a heated mold to form a desired shape. The ability to produce a stable product can be predicted by using a creep-recovery experiment. Figure 9 illustrates data on a packaging film using the tension mode. In the recovery phase, the equilibrium recoverable compliance, Jer can be calculated. If the sample compliance is too high, as observed by a high Jer, then the elasticity may be too low at the forming temperature to maintain the desired shape. SHEET MOLDING COMPOUND CURE Figure 10 shows the cure profile of a sheet molding compound using the shear mode. In this experiment, the sample was heated to 150 C and then held isothermally. Data was collected vs. time. The sample first softens then begins to cure, reaching a final maximum modulus. This profile will be influenced greatly by the heating profile. An assessment can be made on the trade-off in physical properties for various cure profiles by conducting this experiment under various conditions. PREDICTING MATERIAL PERFORMANCE USING TIME/TEMPERATURE SUPERPOSITIONING(TTS) The TTS technique, well grounded in theory, is used to predict material performance at frequencies or time scales outside the range of the instrument. Data is usually generated by scanning multiple frequencies during an isothermal step-hold experiment over a temperature range. A reference temperature is selected and the data shifted. A shift factor plot is generated and fit to either a William-Landel-Ferry (WLF) or Arrhenius model. Finally, a master curve at a specific temperature is generated as illustrated in Figure 11 for a PET film sample. Using this technique, properties at very high frequencies (short time scales) or very low frequencies (long time scales) can be assessed. E' Master Curve T r =75 C 1.0E E-8 1.0E-1 1.0E6 1.0E13 1.0E20
14 THERMAL ADVANTAGE SOFTWARE A quality Thermal Analyzer requires flexible, intelligent software to power it. No one believes this more than our software engineers, who have pioneered most of the features commonly seen in today's modern thermal analyzers. Thermal Advantage for Q Series software is Microsoft Windows based, and expandable to meet growing user needs. THERMAL ADVANTAGE - INSTRUMENT CONTROL Multitasking conducts experiments and simultaneously analyzes data Multimode can operate up to 8 modules simultaneously Wizards guides and prompts in setting up of experiments Real Time Plot provides a real-time display of the progress of the experiment Autoqueuing permits pre-programmed set-up of future experiments How To Feature provides extensive, context sensitive assistance UNIVERSAL ANALYSIS DATA ANALYSIS Single Package analyze data from all TA Instruments modules Picture-in-a-Picture provides easy one plot analysis of large and small events Real Time Data Analysis ability to analyze data as it arrives Ability to create up to 20 simultaneous curve overlay plots Custom Report Generation within UA 2000 using Microsoft Word & Excel templates Saved Analysis for quick retrieval of previously analyzed data files TIME/TEMPERATURE SUPERPOSITIONING (TTS) Flexible Mutifrequency Data Input accepts data from ramp or step-hold experiments Accepts Creep and Stress Relaxation Data Reference Temperature user selectable reference temperature Automatic/Manual Shifting provides for automatic shifting or manual shifting of data Shift Factor Curves includes real-time interaction between curve shift and shift factor graphs Shift Factor Models includes both WLF and Arrhenius models Master Curve can be stored as separate file and generated at any explicit temperature Microsoft Windows, Microsoft Word, and Microsoft Excel are trademarks of Microsoft Corporation
15 WHY TA INSTRUMENTS More worldwide customers choose TA Instruments than any competitor as their preferred thermal analysis and rheology supplier. We earn this distinction by best meeting customer needs and expectations for high technology products, quality manufacturing, timely deliveries, excellent training, and superior after sales-support. SALES AND SERVICE We pride ourselves in the technical competence and professionalism of our sales force, whose only business is thermal analysis and rheology. TA Instruments is recognized worldwide for its prompt, courteous and knowledgeable service staff. Their specialized knowledge and experience are major reasons why current customers increasingly endorse our company and products to their worldwide colleagues. INNOVATIVE ENGINEERING Our company is the recognized leader for innovation and cutting-edge technology. Representative examples include the Q Series Modules, Modulated DSC, the AR 2000 Rheometer with Mobius Drive and Smart Swap, and the Q800 Dynamic Mechanical Analyzer. We invest twice the industry average in research and development and consistently outpace our competitors in introducing new products. QUALITY PRODUCTS All thermal analyzers are manufactured to ISO 9002 procedures in our expanded plant in New Castle, DE, USA. State-of-the-art flow manufacturing procedures and a highly motivated work force ensure high quality products with industry-leading delivery times. TECHNICAL SUPPORT Customers prefer TA Instruments because they buy analyzers to solve problems and often need operation and applications assistance. Our worldwide technical support staff is the largest and most experienced in the industry, and are accessible daily by telephone, , or via our website. We also provide multiple training opportunities in our worldwide facilities.
16 TO CONTACT YOUR LOCAL TA INSTRUMENTS TECHNICAL REPRESENTATIVE VISIT OUR WEBSITE AT NEW CASTLE, DE USA - TELEPHONE: PARIS, FRANCE - TELEPHONE: BRUSSELS, BELGIUM - TELEPHONE: LEATHERHEAD, ENGLAND - TELEPHONE: ETTEN-LEUR, NETHERLANDS - TELEPHONE: ALZENAU, GERMANY - TELEPHONE: MILANO, ITALY - TELEPHONE: TOKYO, JAPAN - TELEPHONE: BARCELONA, SPAIN - TELEPHONE: MELBOURNE, AUSTRALIA - TELEPHONE: STOCKHOLM, SWEDEN - TELEPHONE: TA 284
PERKIN ELMER Polymers technical note CHARACTERIZATION OF POLYMERS BY W.J. Sichina, National Marketing Manager Thermomechanical analysis () is one of the important characterization techniques in the field
Thermal Analysis Excellence DMA/SDTA861 e STAR e System Innovative Technology Versatile Modularity Swiss Quality Dynamic Mechanical Analysis Sets New Standards DMA/SDTA861 e Precise Measurement Technology
TA Instruments Thermomechanical Analysis Sensitive Measurement, Unmatched Versatility TMA Q400EM/Q400 SPECIFICATIONS 98 The Q400EM is the industry s leading research-grade thermomechanical analyzer with
application Note Thermal Analysis Characterization of Electronic Materials Using Thermal Analysis Thermal analysis comprises a series of powerful techniques for the characterization of the thermal, physical,
172 4 Thermomechanical Analysis 4 Thermomechanical Analysis (TMA) 4.1 Principles of TMA 4.1.1 Introduction A dilatometer is used to determine the linear thermal expansion of a solid as a function of temperature.
Who is GBC Scientific Equipment? GBC Scientific Equipment Pty Ltd commenced operations in 1978. GBC designs, manufactures and markets a range of scientific instruments comprising Atomic Absorption spectrometers
Developments in Low Temperature Testing of Rubber Materials Technical report 01/4, 2nd edition Nov 2010 Göran Spetz Elastocon AB SWEDEN Introduction The low temperature properties of rubber materials are
1.0 Introduction Long term performance of polymers Polymer materials exhibit time dependent behavior. The stress and strain induced when a load is applied are a function of time. In the most general form
INTRODUCTORY GUIDE TO USING AN AR SERIES RHEOMETER USING RHEOLOGY ADVANTAGE SOFTWARE* (VERSIONS 4.1) Fred Mazzeo, Ph.D. TA Instruments, 109 Lukens Drive, New Castle, DE 19720, USA 1. Turn on the computer
FREQUENTLY ASKED QUESTIONS Dynamic Mechanical Analysis (DMA) A Beginner s Guide This booklet provides an introduction to the concepts of Dynamic Mechanical Analysis (DMA). It is written for the materials
14.4 (a) Compute the repeat unit molecular weight of polystyrene. (b) Compute the number-average molecular weight for a polystyrene for which the degree of polymerization is 25,000. (a) The repeat unit
DETERMINATION OF TIME-TEMPERATURE SHIFT FACTOR FOR LONG-TERM LIFE PREDICTION OF POLYMER COMPOSITES K. Fukushima*, H. Cai**, M. Nakada*** and Y. Miyano*** * Graduate School, Kanazawa Institute of Technology
Analytical ing Services Commercial List ManTech International Corporation January 2016 TABLE OF CONTENTS MECHANICAL TENSILE TESTING... 1 DIFFERENTIAL SCANNING CALORIMETRY (DSC)... 2 THERMOMECHANICAL ANALYSIS
TECHNICAL WHITEPAPER Low Temperature Properties of Polymers Introduction Most plastics at room temperature show their familiar properties of flexibility (a low Young s modulus) and high resistance to cracking
1 Why is rheology important? Examples of its importance Summary of important variables Description of the flow equations Flow regimes - laminar vs. turbulent - Reynolds number - definition of viscosity
INNOVATIVE PLASTICS HIGH & LOW SHEAR RATE RHEOLOGY A SABIC COMPANY Innovative Plastics is a strategic business unit of SABIC. Founded in 1976, SABIC is today the first public, global multinational enterprise
Discovery Hybrid Rheometer New Castle, DE USA Lindon, UT USA Hüllhorst, Germany Shanghai, China Beijing, China Tokyo, Japan Seoul, South Korea Taipei, Taiwan Bangalore, India Sydney, Australia Guangzhou,
Laboratory 4 Torsion Testing Objectives Students are required to understand the principles of torsion testing, practice their testing skills and interpreting the experimental results of the provided materials
TH INTERNATIONAL CONFERENCE ON COMPOSITE MATERIALS INVESTIGATION OF VISCOELASTICITY AND CURE SHRINKAGE IN AN EPOXY RESIN DURING PROCESSING T. Shimizu *, H. Koinuma, K. Nagai Mitsubishi Heavy Industries,
by Jared A. Tatum, Tien T. Dao, Nick Roye, and Kaj Hedman Effect of Pressure and UV Intensity on the Rheological Cure Profile and Dimensional Change of Developing Networks Rheology is defined as the study
Welding of Plastics Amit Mukund Joshi (B.E Mechanical, A.M.I.Prod.E) Introduction Mechanical fasteners, adhesives, and welding processes can all be employed to form joints between engineering plastics.
application Note Thermal Analysis Better DSC Isothermal Cure Kinetics Studies Using Power Compensation DSC Introduction One important aspect of a thermosetting resin, such as an epoxy, is the cure kinetics
LOW TEMPERATURE TESTING OF ELASTOMERS Presented at the Spring Technical Meeting of the New York Rubber Group Thursday, March 25, 1999 By, Andrew C. Farinella and Daniel L. Hertz, Jr. LOW TEMPERATURE DATA
4 PROCESSING OF VARIOUS MATERIALS CHAPTER CONTENTS 4.1 Shaping Processes for Polymers Polymers Manufacturing Processes for Polymers 4.2 Rubber Processing Technology Processing of rubber into finished good
04/10 149 1.0 GENERAL INFORMATION In a wide range of applications, PTFE Slide bearings are superior to conventional expansion plates, rollers and rocker arm type supports. They support petrochemical plant,
DUPONT PERFORMANCE POLYMERS Joint Design: A Critical Factor in Strong Bonds GENERAL GUIDELINES FOR ULTRASONIC, VIBRATION AND SPIN WELDING Introduction Welding techniques for assembling parts molded with
ENGINEERING COMPONENTS EDEXCEL NATIONAL CERTIFICATE/DIPLOMA MECHANICAL PRINCIPLES OUTCOME ENGINEERING COMPONENTS TUTORIAL 1 STRUCTURAL MEMBERS Structural members: struts and ties; direct stress and strain,
Thermoplastic composites Definition By definition, a thermoplastic is a material based on polymer (macromolecular compound) which can be shaped, in a liquid (viscous) state at a temperature either higher
Naue GmbH&Co.KG Quality Control and Quality Assurance Manual For Geomembranes July 2004 V.O TABLE OF CONTENTS 1. Introduction 2. Quality Assurance and Control 2.1 General 2.2 Quality management acc. to
Topics to be Covered Polymer Melt Rheology Introduction to Viscoelastic Behavior Time-Temperature Equivalence Chapter 11 in CD (Polymer Science and Engineering) Polymer Melt Rheology δ τ xy Newton s Law
Effects of Tg and CTE on Semiconductor Encapsulants Dr. Mark M. Konarski Loctite Corporation www.loctite.com Abstract As the role of direct-chip-attachment increases in the electronics industry, the reliability
ASTM D 1599 Standard Test Method for Resistance to Short-Time Hydraulic Pressure of Plastic Pipe, Tubing, and Fittings This test method establishes the short-time hydraulic failure pressure of pipe and
P.A.Hilton Ltd STRENGTH of MATERIALS P.A.Hilton Ltd is a market leader in the manufacture and provision of teaching equipment for Universities and Technical Colleges worldwide for both degree and vocational
UDC 624. 042. 7 Development of U-shaped Steel Damper for Seismic Isolation System Kazuaki SUZUKI* 1 Atsushi WATANABE* 1 Eiichiro SAEKI* 1 Abstract Seismic isolation system was widely admitted after Hanshin-Awaji
Technical Data 7. Bearing Fits 7.1 Interference For rolling s the rings are fixed on the or in the housing so that slip or movement does not occur between the mated surface during operation or under. This
Dynamic Mechanical Analysis (DMA) Basics and Beyond ε ω = Σ(Δε β /1+ιωτ β ) Dr. Lin Li Thermal Analysis PerkinElmer Inc. April 2000 The DMA lets you relate: product properties molecular structure Material
SHIBAYAMA SCIENTIFIC CO., LTD. DENSITY MEASURING SYSTEMS Density Gradient Tube Method Direct Reading Type A / Type B Comply with the standards below: JIS K-0061-1992, K-7112 1980 ASTM D1505 Type A /Left
HUMBOLDT UNIVERSITÄT ZU BERLIN MATHEMATISCH-NATURWISSENSCHAFTLICHE FAKULTÄT I INSTITUT FÜR PHYSIK Investigation of Polymers with Differential Scanning Calorimetry Contents 1 Introduction 1 2 Thermal Properties
Pore size and BET surface area analysis at your fingertips. SA 3100 Gas Adsorption Analyzer The History of Beckman Coulter Particle Characterization Introducing The SA 3100 Surface Area and Pore Size Analyzer
THE COMPOSITE DISC - A NEW JOINT FOR HIGH POWER DRIVESHAFTS Dr Andrew Pollard Principal Engineer GKN Technology UK INTRODUCTION There is a wide choice of flexible couplings for power transmission applications,
3. Test Methods for Evaluation of ESCR of Plastics A common laboratory request for ESC-prone polymers is to check ESCR performance for quality control, competitive product evaluations, and research and
HFM Heat Flow Meter Thermal Conductivity Analyzer Introduction An insulating material is a material with low thermal conductivity, which in the construction industry, equipment manufacturing, or the production
DUPONT PERFORMANCE POLYMERS SELF TAPPING SCREWS: HOW TO CHOOSE THE RIGHT ONE Self-tapping screws provide an economical means of assembling components, especially where dissimilar materials must be joined
IMPACT TESTING Objective To conduct Charpy V-notch impact test and determine the ductile-brittle transition temperature of steels. Equipment Coolants Standard Charpy V-Notched Test specimens Impact tester
Nonlinear Analysis Using Femap with NX Nastran Chip Fricke, Principal Applications Engineer, Agenda Nonlinear Analysis Using Femap with NX Nastran Who am I? Overview of Nonlinear Analysis Comparison of
22.302 Experiment 5 Strain Gage Measurements Introduction The design of components for many engineering systems is based on the application of theoretical models. The accuracy of these models can be verified
Fluid Mechanics: Fluid mechanics may be defined as that branch of engineering science that deals with the behavior of fluid under the condition of rest and motion Fluid mechanics may be divided into three
AMPLITUDE AND FORCE PROFILING: STUDIES IN ULTRASONIC WELDING OF THERMOPLASTICS David A. Grewell Branson Ultrasonics Corporation ABSTRACT This paper reviews effects of amplitude and force control during
Introduction to Solid Modeling Using SolidWorks 2012 SolidWorks Simulation Tutorial Page 1 In this tutorial, we will use the SolidWorks Simulation finite element analysis (FEA) program to analyze the response
Force 1 1 N m up to 1 000 1 000 N m Type 9329A 9389A These easy to install piezoelectric reaction torque sensors are particularly suitable for measuring rapidly changing torques at non-rotating shafts.
BIOEN 36 013 LECTURE : MOLECULAR BASIS OF ELASTICITY Estimating Young s Modulus from Bond Energies and Structures First we consider solids, which include mostly nonbiological materials, such as metals,
Scotch-Weld DP760 Product Data Sheet Updated : February 2009 Supersedes: June 2001 Product Description DP760 epoxy adhesive is a non-sag, two-part room temperature curing adhesive designed for use when
26 TH INTERNATIONAL CONGRESS OF THE AERONAUTICAL SCIENCES CARBON/DYNEEMA INTRALAMINAR HYBRIDS: NEW STRATEGY TO INCREASE IMPACT RESISTANCE OR DECREASE MASS OF CARBON FIBER COMPOSITES J. G. H. Bouwmeester*,
E644 is a toughened epoxy resin system for low temperature cures, pre-impregnated into high performance fibers such as carbon, glass & aramid. Designed for the production of composite structures in the
PAGE 1/10 1 Density The density of optical glass varies from 239 for N-BK10 to 603 for SF66 In most cases glasses with higher densities also have higher refractive indices (eg SF type glasses) The density
PAGE 1/7 1 Introduction In some applications optical glasses have to endure thermal loads: Finishing procedures for optical elements like lenses, prisms, beam splitters and so on involve thermal processes
Standard Thermoforming Equipment Overview There are three standard configurations for thermoforming equipment: 1. Heat and Cut-in-Place Forming 2. In-Line Forming with Steel Rule or Forged Steel Trim wand
Investigation of cure induced shrinkage in unreinforced epoxy resin Mauro Zarrelli*, Alexandros A Skordos and Ivana K Partridge Advanced Materials Dept, Cranfield University, Cranfield, Bedford, MK43 AL,
Sheet metal operations - Bending and related processes R. Chandramouli Associate Dean-Research SASTRA University, Thanjavur-613 401 Table of Contents 1.Quiz-Key... Error! Bookmark not defined. 1.Bending
Section 8. 8. Elastic Strain Energy The strain energy stored in an elastic material upon deformation is calculated below for a number of different geometries and loading conditions. These expressions for
MICRO PLASTICS MOLDED NYLON 6/6 FASTENERS MECHANICAL TEST DATA - UNC - UNF Nylon 6/6 meets MIL-M20693B "A" Type 1, LP410A, and ASTM D789-66 Type 1, GDE 2 All test performed per ASTM specifications. Parts
Modern Construction Materials Prof. Ravindra Gettu Department of Civil Engineering Indian Institute of Technology, Madras Module - 2 Lecture - 2 Part 2 of 2 Review of Atomic Bonding II We will continue
INNOVATIVE PLASTICS ULTRASONIC WELDING A SABIC COMPANY Innovative Plastics is a strategic business unit of SABIC. Founded in 1976, SABIC is today the first public, global multinational enterprise headquartered
Development of an innovative bio-based structural adhesive Blanca Palomo, R&D Engineer firstname.lastname@example.org RESCOLL Independent research company located in Pessac (33) specialized in technologic innovation
FLEXURAL PERFORMANCE OF RC BEAMS STRENGTHENED WITH PRESTRESSED CFRP SHEETS Piyong Yu, Pedro F. Silva, Antonio Nanni Center for Infrastructure and Engineering Studies Department of Civil, Architectural,
Physics 3 Summer 1989 Lab 7 - Elasticity Theory All materials deform to some extent when subjected to a stress (a force per unit area). Elastic materials have internal forces which restore the size and
Ch 2 Properties of Fluids - II Ideal Fluids 1 Prepared for CEE 3500 CEE Fluid Mechanics by Gilberto E. Urroz, August 2005 2 Ideal fluid: a fluid with no friction Also referred to as an inviscid (zero viscosity)
Rheological properties of structural and pressure-sensitive adhesives and their impact on product performance By Peter K.W. Herh, Senior Rheologist; Dharmesh H. Dalwadi, Staff Engineer REOLOGICA Instruments,
Field Application Note Reverse Dial Indicator Alignment RDIA Mis-alignment can be the most usual cause for unacceptable operation and high vibration levels. New facilities or new equipment installations
Force measurement Forces VECTORIAL ISSUES In classical mechanics, a force is defined as "an action capable of modifying the quantity of movement of a material point". Therefore, a force has the attributes
Technical Data April 2011 M Electrically Conductive Adhesive Transfer Tape 970 Product Description M Electrically Conductive Adhesive Transfer Tape 970 is a pressure sensitive adhesive (PSA) transfer tape
Keysight Technologies How to Choose your MAC Lever Technical Overview Introduction Atomic force microscopy (AFM) is a sub-nanometer scale imaging and measurement tool that can be used to determine a sample
CPVC/METAL/CPVC COMPOSITE PIPING SYSTEMS Robert M. Frimel Noveon, Inc. ABSTRACT For more than twenty years, CPVC compounds have been used to produce single-layer, allplastic tube and pipe for the Plumbing,
Numerical Analysis of Independent Wire Strand Core (IWSC) Wire Rope Rakesh Sidharthan 1 Gnanavel B K 2 Assistant professor Mechanical, Department Professor, Mechanical Department, Gojan engineering college,
T dition CHTR MCHNICS OF MTRIS Ferdinand. Beer. Russell Johnston, Jr. John T. DeWolf ecture Notes: J. Walt Oler Texas Tech University Stress and Strain xial oading - Contents Stress & Strain: xial oading
PURPOSE: The parameters of the shear strength relationship provide a means of evaluating the load carrying capacity of soils, stability of slopes, and pile capacity. The direct shear test is one of the