# CHAPTER 5 QC Test For Radiographic Equipment. Prepared by:- Kamarul Amin bin Abu Bakar School of Medical Imaging KLMUC

Save this PDF as:

Size: px
Start display at page:

Download "CHAPTER 5 QC Test For Radiographic Equipment. Prepared by:- Kamarul Amin bin Abdullah @ Abu Bakar School of Medical Imaging KLMUC"

## Transcription

1 CHAPTER 5 QC Test For Radiographic Equipment Prepared by:- Kamarul Amin bin Abu Bakar School of Medical Imaging KLMUC

2 Lesson Outcomes Describe the objectives of each QC test done.(importance) Define the term reproducibility and linearity. Identify the required QC test for specific x-ray devices. Identify QC tools and equipments involved in each QC test. Discuss the analysis of result of QC. Discuss the procedure of QC test.

3 QC Tests (All) 1. AUTOMATIC EXPOSURE CONTROL (AEC) A. Exposure Reproducibility B. Response time C. Backup timer D. Exposure timer E. Accuracy F. Reproducibility G. Dead-man exposure switch 2. KILOVOLTAGE A. Accuracy B. Reproducibility 3. RADIATION OUTPUT (mas) A. Linearity B. Programmed timer C. Reproducibility 4. HALF-VALUE LAYER (HVL) 5. BEAM AND COLLIMATION ALIGNMENT 6. GRID ALIGNMENT 7. FOCAL SPOT

4 QC Tests (Theory and Practical) 1. kv reproducibility 2. ma linearity 3. Timer accuracy 4. HVL 5. Grid alignment 6. Beam and collimation alignment

5 1. kv reproducibility Objective: To determine the variation in average kvp over a number of exposures at the same generator setting. Equipment: Digital kv meter/multifunction meter Lead backing sheet Energized radiographic unit

6 Reproducibility ability of radiographic unit to produce the same exposure even though in the short intervals.

7 X= 70 Kvp Sample of Data Collection

8 Data Analysis

9

10 Discussion on Analysis Kilovoltage settings tend to drift over time, primarily as a result of tubing aging. Reproducibility should be maintained within ±2%. The problem maybe because: 1. X-ray tube problems 2. Inaccurate reading of the detector. The generator in that room did not function very well. It did not have the capability to produce the same exposure even though in the short intervals.

11 ma linearity Objective: To evaluate the linearity of a generator for commonly used exposure settings Equipment: Energized radiographic unit Digital dosimeter

12 Linearity ability of radiographic unit to produce a constant radiation output for various combinations of maand exposure time.

13 Sample of Data Collection

14 Data Analysis

15 Sample of Data Analysis

16 Discussion on Analysis Linearity means that the ratio of exposure to total charge in mr/mas is constant over the entire range of mas values. The average mr/mas obtain at any tube current settings (ma) shall not differ by more than Linearity should be maintained to ±10% over the entire working range of the generator regardless of the number ma stations at fixed kvp. This test attempts to establish if the same film density is achieved using the same kvp and mas but different ma and time values.

17 Timer Accuracy Objective: To evaluate the accuracy of an exposure timer. Equipment: Energized radiographic unit Digital exposure timer (multi-function meter)

18 Sample of Data Collection

19 Data Analysis

20 Discussion on Analysis

21 HVL Objective: To assess the x-ray beam quality and determine the adequacy of filtration. Equipment: Energized radiographic unit Digital dosimeter Lead backing sheet Aluminiumfilters Tape for positioning filters

22 HVL HVL is that amount of absorbing material that will reduce the intensity of the primary beam to one-half its original value.

23 Sample of Data Collection

24

25 General Analysis Data in table are typical for half-value layer determination.

26 Referring to the data, the HVL is 2.4mm Al. At zero filtration, x-ray quantity appears to be approximately 118 mr.one half of the 118mR is 59mR. At the level of 59mR, a horizontal line is drawn from the y-axis until it intersects the plotted curve. From that intersection, a vertical line is dropped to the x-axis where, it intersects at 2.4mm Al, which is the HVL.

27 Discussion on Analysis FILTRATION IS NECESSARY TO REMOVE LOW-ENERGY PHOTONS FROM THE X-RAY BEAM. THE BEST METHOD TO DETERMINE ADEQUATE FILTRATION IS BY MEASURE THE HALF-VALUE LAYER (HVL). HVL IS THE AMOUNT OF THE FILTRATION THAT WILL REDUCE THE EXPOSURE RATE TO ONE HALF OF ITS INITIAL VALUE. THE DOSIMETER WHICH IS RAD CHECK THAT BEING USED IN LAB IS TO MEASURE THE HVL BY DETECTING THE QUALITY OF ALUMINIUM FILTRATION THAT WILL REDUCE THE BEAM INTENSITY TO HALF THE ORIGINAL VALUE.

28 Grid alignment Objective: To detect an upside-down focused grid or incorrect tube to grid distance. Equipment: Energized radiographic unit (60kV, 2mAs) RMI grid alignment test tool 18 x 24 cm cassette with film Densitometer

29

30 Sample of Data Collection

31 Data Analysis Which hole represents the highest OD? Why?

32 Discussion on Analysis If the grid is misalignment, it may be because of several causes. One of the causes is due to the position of grid (off-center) or its construction (upside down grid) It also because of reciprocating action of the grid itself where grid motion and the beginning of the x-ray exposure are not simultaneously

33 Beam and Collimation Alignment Objective: To check misalignment of the light field and the x ray field Equipment: Energized radiographic unit RMI Collimator Test Tool and the RMI Beam Alignment Test Tool Cassette 8 X 10 inch The bubble level

34 Sample of Data Collection upper steel ball superimposed with lower steel ball. collimation beam fall at the same scale of number in the test tool.

35 upper steel ball is not superimposed with lower steel ball. rating scales are not at the same number on both superior, inferior and lateral side. The collimation beam is less on left and inferior side and vice versa.

36 overlapped of the steel balls in the center. collimation falls within the rectangular frame.

37 Data Analysis Collimation Misalignment must not exceed ±2% of the SID. If an edge of the x-ray field falls on the first spot ±1 cm, it shows that the edge of the x-ray field and the light field are misaligned by 1% of SID. Beam If both steel balls overlapped, the central ray is perpendicular, indicates good alignment.

38 Discussion on Analysis Collimation Misalignment must not exceed ±2% of the SID. If an edge of the x-ray field falls on the first spot ±1 cm, it shows that the edge of the x-ray field and the light field are misaligned by 1% of SID. Beam If both steel balls overlapped, the central ray is perpendicular, indicates good alignment.

39

40 Other QC Test Objectives To measure the minimum response time of the AEC. To ensure that the backup timer is functioning and backup time does not exceed the specified time. To ensure that the exposure is terminated by removing pressure from the exposure switch. To determine how the measured kvp compares with the generator setting.

41

### Name: Date: Team: Lab Experiment # 3. Focused Grid Positioning Errors. Computed Radiography and Direct Digital Radiography

Name: Date: Team: Lab Experiment # 3 Focused Grid Positioning Errors Computed Radiography and Direct Digital Radiography Purpose This experiment is designed to demonstrate the effect of off-level error,

### SECTION 1: REQUIREMENTS FOR CERTIFICATES OF COMPLIANCE FOR CLASSES OF RADIATION APPARATUS

Department of Health and Human services Population Health Radiation Protection Act 2005 Section 17 CERTIFICATE OF COMPLIANCE: STANDARD FOR RADIATION APPARATUS - X-RAY MEDICAL DIAGNOSTIC (MAMMOGRAPHY) SECTION

### On the Use of a Diagnostic X-Ray Machine for Calibrating Personal Dosimeters

On the Use of a Diagnostic X-Ray Machine for Calibrating Personal Dosimeters A. T. Baptista Neto, T. A. Da Silva Centro de Desenvolvimento da Tecnologia Nuclear, Comissão Nacional de Energia Nuclear, Rua

Principles of Imaging Science II (RAD 120) Radiographic Grids 1 Image-Forming X-Rays Four X-ray paths a. X-rays interact with patient and scatter away from the receptor b. X-rays interact and are absorbed

### X-ray Production. Target Interactions. Principles of Imaging Science I (RAD119) X-ray Production & Emission

Principles of Imaging Science I (RAD119) X-ray Production & Emission X-ray Production X-rays are produced inside the x-ray tube when high energy projectile electrons from the filament interact with the

### 5.2 ASSESSMENT OF X-RAY TUBE LEAKAGE RADIATION AND X-RAY TUBE OUTPUT TOTAL FILTRATION

5.2 ASSESSMENT OF X-RAY TUBE LEAKAGE RADIATION AND X-RAY TUBE OUTPUT TOTAL FILTRATION 5.2.1 Task The bremsstrahlung produced by the X-ray tube has a continuous spectrum, limited by the set and spreads

### Study the Quality Assurance of Conventional X-ray Machines Using Non Invasive KV meter

Study the Quality Assurance of Conventional X-ray Machines Using Non Invasive KV meter T.M.Taha Radiation Protection Department, Nuclear Research Center, Atomic Energy Authority, Cairo.P.O.13759 Egypt.

### STANDARD FOR RADIATION APPARATUS -

Department of Health and Human Services Population Health Radiation Protection Act 2005 Section 17 CERTIFICATE OF COMPLIANCE: STANDARD FOR RADIATION APPARATUS - X-RAY DENTAL DIAGNOSTIC (INTRAORAL) SECTION

### TOUGH RAY 9030 PORTABLE VETERINARY X-RAY EQUIPMENT INSTRUCTION MANUAL

TOUGH RAY 9030 PORTABLE VETERINARY X-RAY EQUIPMENT INSTRUCTION MANUAL DRAFT CONTENTS PAGE NO. NOTICE FOR SAFE OPERATION 2 INTRODUCTION 3 NOTES TO USERS 3 COMPONENTS 4 OPERATING CONDITIONS 5 MAIN PARTS

Image Quality and Radiation Dose for Intraoral Radiography: Hand-Held Held (Nomad), Battery Powered vs. Wall-Mount X-Ray X Systems Edgar Bailey*, MSEHE, CHP Consultant Joel Gray*, PhD, FAAPM DIQUAD, LLC

### SECTION 1: REQUIREMENTS FOR CERTIFICATES OF COMPLIANCE FOR CLASSES OF RADIATION APPARATUS

Department of Health and Human Services Population Health Radiation Protection Act 2005 Section 17 CERTIFICATE OF COMPLIANCE: STANDARD FOR RADIATION APPARATUS - X-RAY MEDICAL DIAGNOSTIC (MOBILE RADIOGRAPHY)

### ABOVETABLE X-RAY SOURCE RADIOGRAPHIC SYSTEMS

PART II ABOVETABLE X-RAY SOURCE RADIOGRAPHIC SYSTEMS FORM FDA 2784 REPRINTED APRIL 2000 ROUTINE COMPLIANCE TESTING ABOVETABLE X-RAY SOURCE RADIOGRAPHIC SYSTEMS (Test Procedure ARA - Use Form FD2784) 1.0

### SECTION 1: REQUIREMENTS FOR CERTIFICATES OF COMPLIANCE FOR CLASSES OF RADIATION SOURCES

Department of Health and Human Services Population Health Radiation Protection Act 2005 Section 17 CERTIFICATE OF COMPLIANCE: STANDARD FOR RADIATION APPARATUS - X-RAY DIAGNOSTIC (VETERINARY) SECTION 1:

### LECTURE 6 MOBILE X-RAY UNITS

LECTURE 6 MOBILE X-RAY UNITS Prepared by:- KAMARUL AMIN BIN ABDULLAH @ ABU BAKAR UiTM Faculty of Health Sciences Medical Imaging Department LESSON OBJECTIVES At the end of lesson, the student should able

### SECTION 1: REQUIREMENTS FOR CERTIFICATES OF COMPLIANCE FOR CLASSES OF RADIATION APPARATUS

Department of Health and Human Services Population Health Radiation Protection Act 2005 Section 17 CERTIFICATE OF COMPLIANCE: STANDARD FOR RADIATION APPARATUS - X-RAY MEDICAL DIAGNOSTIC (MOBILE RADIOSCOPY)

### QUALITY CONTROL TESTS IN SOME DIAGNOSTIC X-RAY UNITS IN BANGLADESH

Bangladesh Journal of Medical Physics Vol. 4, No.1, 2011 QUALITY CONTROL TESTS IN SOME DIAGNOSTIC X-RAY UNITS IN BANGLADESH M. Begum 1, A. S. Mollah 2, M. A. Zaman 3 and A. K. M. M. Rahman 4 1 Health Physics

### SUBCHAPTER 22 QUALITY ASSURANCE PROGRAMS FOR MEDICAL DIAGNOSTIC X-RAY INSTALLATIONS

Note: This is a courtesy copy and is not the official version of this rule. The official, legally effective version of this rule is available through www.lexisnexic.com/bookstore (Phone: (800) 223-1940).

### QUALITY CONTROL PROGRAM. X-Ray Equipment PROCEDURE MANUAL

QUALITY CONTROL PROGRAM DEPARTMENT OF RADIOLOGY MARY HITCHCOCK MEMORIAL HOSPITAL DARTMOUTH-HITCHCOCK MEDICAL CENTER LEBANON, NEW HAMPSHIRE X-Ray Equipment PROCEDURE MANUAL June 2009 Table of Contents Quality

### Annual MQSA Inspection Questions NMQAAC April 19, 2004

1 1 1 1 1 1 1 1 0 1 0 1 0 1 0 1 Annual MQSA Inspection Questions NMQAAC April 1, 00 1.0 Inspection Information 1.1 Name and Address 1. Equipment Registration.0 Facility Inspection Download.0 Facility Inspections.0

### SUBCHAPTER 15. MEDICAL DIAGNOSTIC X-RAY INSTALLATIONS

SUBCHAPTER 15. MEDICAL DIAGNOSTIC X-RAY INSTALLATIONS 7:28-15.1 Scope (a) This subchapter establishes the requirements for medical radiographic and fluoroscopic installations of certified and uncertified

### SECTION 1: REQUIREMENTS FOR CERTIFICATES OF COMPLIANCE FOR CLASSES OF RADIATION APPARATUS

Department of Health and Human Services Population Health Radiation Protection Act 2005 Section 17 CERTIFICATE OF COMPLIANCE: STANDARD FOR RADIATION APPARATUS - X-RAY MEDICAL DIAGNOSTIC (BIOPSY) SECTION

### Q.A. Collectible. Sponsored by CRCPD s Committee on Quality Assurance in Diagnostic X-Ray (H-7)

Q.A. Collectible Sponsored by CRCPD s Committee on Quality Assurance in Diagnostic X-Ray (H-7) Mammography Phantom Image Quality Evaluation (from the American College of Radiology 1999 Mammography Quality

### SECTION 1: REQUIREMENTS FOR CERTIFICATES OF COMPLIANCE FOR CLASSES OF RADIATION APPARATUS

Department of Health and Human Services Population Health Radiation Protection Act 2005 Section 17 CERTIFICATE OF COMPLIANCE: STANDARD FOR RADIATION APPARATUS - X-RAY MEDICAL DIAGNOSTIC (FIXED RADIOSCOPY)

### Diagnostic X-ray Tube. Producing a radiograph(1) Overview - Equipment found in a general X- ray room

Radiographic Science Basic Overview of diagnostic radiography Marc Griffiths Overview - Equipment found in a general X- ray room Diagnostic X-ray tube Tube support Mechanism Diagnostic Table Upright stands

### Radiology. Floron C. Faries, Jr. DVM, MS

Radiology Floron C. Faries, Jr. DVM, MS Objectives Determine the appropriate machine settings for making a radiograph Describe essential radiograph accessories Describe the positions used to perform radiographs

### QUALITY ASSURANCE PROGRAMS FOR DIAGNOSTIC RADIOLOGY FACILITIES. (a) Applicability. (b) Definitions

QUALITY ASSURANCE PROGRAMS FOR DIAGNOSTIC RADIOLOGY FACILITIES (a) Applicability Quality assurance programs as described in paragraph (c) of this section are recommended for all diagnostic radiology facilities.

### Quality Control and Maintenance Programs

Quality Control and Maintenance Programs Cari Borrás, D.Sc., FACR, FAAPM Visiting Professor DOIN-DEN / UFPE Recife, Pernambuco, Brazil Co-Chair, IUPESM Health Technology Task Group 1 Medical Imaging Equipment

### Diagnostic X-ray facilities as per quality control performances in Tanzania

JOURNAL OF APPLIED CLINICAL MEDICAL PHYSICS, VOLUME 7, NUMBER 4, FALL 2006 Diagnostic X-ray facilities as per quality control performances in Tanzania Yesaya Y. Sungita, 1 Simon S.L. Mdoe, 1 and Peter

### Role of the Medical Physicist in Clinical Implementation of Breast Tomosynthesis

Role of the Medical Physicist in Clinical Implementation of Breast Tomosynthesis Bob Liu, Ph.D. Department of Radiology Massachusetts General Hospital And Harvard Medical School Digital Breast Tomosynthesis

### REGULATION: QUALITY ASSURANCE PROGRAMS FOR MEDICAL DIAGNOSTIC X-RAY INSTALLATIONS N.J.A.C. 7:28-22

REGULATION: QUALITY ASSURANCE PROGRAMS FOR MEDICAL DIAGNOSTIC X-RAY INSTALLATIONS N.J.A.C. 7:28-22 New Jersey Department of Environmental Protection Bureau of Radiological Health PO Box 415 Trenton NJ

### Temple University. R adiation Safety Guide. For Diagnostic Imaging. X -Ray Producing Equipment

Temple University R adiation Safety Guide For Diagnostic Imaging X -Ray Producing Equipment Environmental Health and Safety Radiation Safety Department Table of Content Page 1. Radiation 2 A. Radiation

### RESOURCE MANUAL FOR COMPLIANCE TEST PARAMETERS DIAGNOSTIC X-RAY SYSTEMS. Reprinted. July 15, 1999. Diagnostic Devices Branch

RESOURCE MANUAL FOR COMPLIANCE TEST PARAMETERS OF DIAGNOSTIC X-RAY SYSTEMS Reprinted July 15, 1999 Diagnostic Devices Branch Division of Enforcement I Office of Compliance 2098 Gaither Road Rockville,

### Surveying and QC of Stereotactic Breast Biopsy Units for ACR Accreditation

Surveying and QC of Stereotactic Breast Biopsy Units for ACR Accreditation LORAD Stereotactic Breast Biopsy System AAPM Spring Clinical Meeting Phoenix, AZ March 17, 2013 Melissa C. Martin, M.S., FACR,

### Radiographic Quality. Factors Affecting Optical Density: Optimum film/screen image density: 0.25 to 2.5 Milliamperage or ma (primary beam quantity)

The Density Unit: Factors Affecting Optical Density: Optimum film/screen image density: 0.25 to 2.5 Milliamperage or ma (primary beam quantity) Primary controlling factor for density Direct and proportional

### Easy quality Checks with - PTW Equipment. Code of Practice - Constancy tests of X-ray equipment in diagnostic radiology

Easy quality Checks with - PTW Equipment Code of Practice - Constancy tests of X-ray equipment in diagnostic radiology Revised edition 2008 Code of Practice- Constancy tests of X-ray equipment in diagnostic

### SECTION 1: REQUIREMENTS FOR CERTIFICATES OF COMPLIANCE FOR CLASSES OF RADIATION APPARATUS

Department of Health and Human Services Population Health Radiation Protection Act 2005 Section 17 CERTIFICATE OF COMPLIANCE: STANDARD FOR RADIATION APPARATUS - X-RAY MEDICAL DIAGNOSTIC (MOBILE CAPACITOR

### Production of X-rays and Interactions of X-rays with Matter

Production of X-rays and Interactions of X-rays with Matter Goaz and Pharoah. Pages 11-20. Neill Serman Electrons traveling from the filament ( cathode) to the target (anode) convert a small percentage

### Dose Optimization for the Quality Control Tests of X-Ray Equipment

16 Dose Optimization for the Quality Control Tests of X-Ray Equipment Mana Sezdi Istanbul University Turkey 1. Introduction Radiation is a major risk in diagnostic and therapeutic medical imaging. The

### X-ray Imaging Systems

Principles of Imaging Science I (RAD 119) X-ray Tube & Equipment X-ray Imaging Systems Medical X-ray Equipment Classified by purpose or energy/current levels kvp, ma Radiographic Non-dynamic procedures

### Collimation Assessment Using GAFCHROMIC XR-M

Collimation Assessment Using GAFCHROMIC XR-M I. Introduction A method of collimation assessment for GE Senographe full-field digital mammography (FFDM) systems is described that uses a self-developing

### portable x-ray survey report

department of HealtH and Human services centers for medicare & medicaid services form approved omb no. 0938-0027 portable x-ray survey report provider number H1 date surveyed (1) initial (2) survey W H2

### EPA Protocols: Compliance testing of dental X-ray apparatus used for plain radiography

Environment Protection Authority EPA Protocols: Compliance testing of dental X-ray apparatus used for plain radiography Doc. No. RP91 Prepared by the Radiation Protection Branch For further information

### BASIC QUALITY CONTROL IN DIAGNOSTIC RADIOLOGY

AAPM REPORT No. 4 BASIC QUALITY CONTROL IN DIAGNOSTIC RADIOLOGY American Association of Physicists in Medicine FOREWORD This document, Basic Quality Control in Diagnostic Radiology, is the fourth in a

### Chapter 5 X-ray imaging 5.1 The physics of diagnostic X-rays

Chapter 5 X-ray imaging 5.1 The physics of diagnostic X-rays Learning objectives: How are X-rays produced in an X-ray tube? How is the energy of the photons from an X-ray tube controlled? Why are photons

### LAUE DIFFRACTION INTRODUCTION CHARACTERISTICS X RAYS BREMSSTRAHLUNG

LAUE DIFFRACTION INTRODUCTION X-rays are electromagnetic radiations that originate outside the nucleus. There are two major processes for X-ray production which are quite different and which lead to different

### Easy Quality Control with - PTW Equipment. Code of Practice - Quality control of X-ray equipment in diagnostic radiology

Easy Quality Control with - PTW Equipment Code of Practice - Quality control of X-ray equipment in diagnostic radiology Revised edition October 2010 Code of Practice- Quality control of X-ray equipment

### Performance testing for Precision 500D Classical R/F System

Performance testing for Precision 500D Classical R/F System John M. Boudry, Ph.D. Image Quality Systems Engineer GE Healthcare Technologies Outline System background Image Quality Signature Test (IQST)

### Dose Measurement in Mammography; What are we measuring? David E. Hintenlang, Ph.D. DABR University of Florida

Dose Measurement in Mammography; What are we measuring? David E. Hintenlang, Ph.D. DABR University of Florida Average Glandular Dose Required measurement performed by medical physicist as part of Mammography

### Digital Breast Tomosynthesis QC Requirements

Digital Breast Tomosynthesis QC Requirements AAPM Spring Clinical Meeting March 8, 2015 Michael S Glaser, MS, DABR Alliance Medical Physics, LLC Learning Objectives 1. GE SenoClaire - Physicist & Technologist

### Production of X-rays. Radiation Safety Training for Analytical X-Ray Devices Module 9

Module 9 This module presents information on what X-rays are and how they are produced. Introduction Module 9, Page 2 X-rays are a type of electromagnetic radiation. Other types of electromagnetic radiation

### X-ray Imaging Systems

Principles of Imaging Science I (RAD 119) X-ray Tube & Equipment X-ray Imaging Systems Medical X-ray Equipment Classified by purpose or energy/current levels kvp, ma Radiographic Non-dynamic procedures

### Application Note # AP0102 Mar 2011 X-ray Dose and System Efficiency (including -ray and ß Efficiencies) in the Carestream DXS Digital X-ray System

Application Note # AP0102 Mar 2011 X-ray Dose and System Efficiency (including -ray and ß Efficiencies) in the Carestream DXS Digital X-ray System Douglas Vizard Author Information: Bruker BioSpin, 15

### Radiology Physics. Just take a deep breath. Books to Consider. Why worry about physics? The Game Plan. 1 st Period

Radiology Physics Just take a deep breath OR: I DIDN T SIGN UP TO LEARN THIS STUFF Chris Ober, DVM, PhD, DACVR 7 February 2011 Why worry about physics? Know what the system can give you Know what the system

### Interim policy effective January 1, 2016, through December 31, 2017

IMPORTANT NOTICE Interim policy announced for post-primary structured education requirement Interim policy relaxes exam content outline requirement (November 12, 2015) The ARRT announces an interim policy

### Standard Test Method for Classification of Film Systems for Industrial Radiography 1

Designation: E 1815 96 (Reapproved 2001) Standard Test Method for Classification of Film Systems for Industrial Radiography 1 This standard is issued under the fixed designation E 1815; the number immediately

### Production of X-Rays. Yoichi Watanabe, Ph.D. Masonic Memorial Building M10-M (612) MPHY 5170/TRAD 7170, Fall semester

Production of X-Rays Yoichi Watanabe, Ph.D. Masonic Memorial Building M10-M (612)626-6708 watan016@umn.edu MPHY 5170/TRAD 7170, Fall semester Contents 1) Physics of X-ray production 2) The X-ray tube 3)

### HIGH PERFORMANCE MOBILE SURGICAL C-ARM KMC-950

HIGH PERFORMANCE MOBILE SURGICAL C-ARM 1K x 1k CCD Digital Camera System H.F. GENERATOR & ROTATING ANODE TUBE WITH DIGITAL WORKSTATION DESCRIPTION: Mobile Surgical C-arm systems are integrated with a triple

### Title: Stationary digital breast tomosynthesis with distributed field emission X-ray tube

Title: Stationary digital breast tomosynthesis with distributed field emission X-ray tube Authors: F. Sprenger a*, X. Calderon b, E. Gidcumb b, J. Lu b, X. Qian b, D. Spronk a, A. Tucker b, G. Yang b,

### Agency for Health Care Administration

Page 1 of 20 FED - H0000 - INITIAL COMMENTS Title INITIAL COMMENTS CFR Type Memo Tag FED - H0005 - COMPLIANCE WITH FED/STATE/LOCAL LAWS Title COMPLIANCE WITH FED/STATE/LOCAL LAWS CFR 486.100 Type Condition

### Evolution of X-ray machine quality control acceptance indices

JOURNAL OF APPLIED CLINICAL MEDICAL PHYSICS, VOLUME 10, NUMBER 4, FALL 2009 Evolution of X-ray machine quality control acceptance indices Maria Lucia Nana I Ebisawa, 1a Maria de Fatima A Magon, 2 Yvone

### Quality Assurance. The selection of the equipment. Equipment Specifications. Medical Exposure Directive 97/43 Euratom. Quality Assurance Programme

Medical Exposure Directive 97/43 Euratom Quality Assurance Ministry of Health, Radiation Protection Department, Luxembourg Alexandra Schreiner Medical Physicist Quality Assurance (QA): All those planned

### X-Rays were discovered accidentally in 1895 by Wilhelm Conrad Röntgen

X-Rays were discovered accidentally in 1895 by Wilhelm Conrad Röntgen Due to their short wavelength, on the order of magnitude of cells, and their high energy, they can penetrate skin and other soft tissue.

### Quality control tests of diagnostic radiology equipment in Hungary, and its radiation protection aspects

Quality control tests of diagnostic radiology equipment in Hungary, and its radiation protection aspects Tamás Porubszky *, Csaba Váradi, László Ballay, Olivér Turák, Géza Gáspárdy, István Turai Frédéric

### QUALITY CONTROL IN DIAGNOSTIC RADIOLOGY

AAPM REPORT NO. 74 QUALITY CONTROL IN DIAGNOSTIC RADIOLOGY Report of Task Group #12 Diagnostic X-ray Imaging Committee Members S. Jeff Shepard, Chairman Pei-Jan Paul Lin, Co-Chairman John M. Boone Dianna

### What does DXA mean and why is it important? Principles of operation of DXA systems. How can we measure bone mineral density? What is radiation?

What does DXA mean and why is it important? Principles of operation of DXA systems Dr Wil Evans Head of Medical Physics University Hospital of Wales, Cardiff DXA stands for Dual energy X-ray Absorptiometry

### AAPM REPORT NO. 25 PROTOCOLS FOR THE RADIATION SAFETY SURVEYS OF DIAGNOSTIC RADIOLOGICAL EQUIPMENT

AAPM REPORT NO. 25 PROTOCOLS FOR THE RADIATION SAFETY SURVEYS OF DIAGNOSTIC RADIOLOGICAL EQUIPMENT Published for the American Association of Physicists in Medicine by the American institute of Physics

### Scan Time Reduction and X-ray Scatter Rejection in Dual Modality Breast Tomosynthesis. Tushita Patel 4/2/13

Scan Time Reduction and X-ray Scatter Rejection in Dual Modality Breast Tomosynthesis Tushita Patel 4/2/13 Breast Cancer Statistics Second most common cancer after skin cancer Second leading cause of cancer

### Experiment report. R. Moeckli, F.R. Verdun, J.-F. Valley Institut de Radiophysique Appliquée Centre Universitaire. CH Lausanne.

Experiment report Introduction R. Moeckli, F.R. Verdun, J.-F. Valley Institut de Radiophysique Appliquée Centre Universitaire. CH - 1015 Lausanne Beam time has been allocated on BM5 in September 1998,

### SUMMARY OF CURRENT UK LEGISLATION AND GUIDELINES

SUMMARY OF CURRENT UK LEGISLATION AND GUIDELINES Legislation There are two sets of regulations in the UK governing the use of ionizing radiation. They both form part of The Health and Safety at Work Act

Radiation safety in dental radiography Dental Radiography Series The goal of dental radiography is to obtain diagnostic information while keeping the exposure to the patient and dental staff at minimum

### INTRODUCTION. A. Purpose

New York State Department of Health Bureau of Environmental Radiation Protection Guide for Radiation Safety/Quality Assurance Programs Computed Radiography INTRODUCTION A. Purpose This guide describes

### PLANMED SOPHIE MAMMOGRAPHY X-RAY UNIT TECHNICAL MANUAL 788405/11 LAT LAT SIN DEX. kv kv. mas. mas CTL

LAT mas 8 3 2 1 LAT mas PLANMED SOPHIE MAMMOGRAPHY X-RAY UNIT OBL CC OBL SIN DEX kv kv TECHNICAL MANUAL En 788405/11 TABLE OF CONTENTS Chapter A Chapter B Chapter C GENERAL & TECHNICAL DATA 1 WARNINGS...............................................

### Diagnostic X-Ray Imaging Quality Assurance: An Overview

Diagnostic X-Ray Imaging Quality Assurance: An Overview by M.A. Périard and P. Chaloner X-Ray Section, Consumer And Clinical Radiation Hazards Division Radiation Protection Bureau, Environmental Health

### Introduction. Digital Mammography Detector QC FFDM. What is FFDM QC and why is it. important? What to know before you start. Eric A. Berns, Ph.D.

Slide 1 Digital Mammography Detector QC Slide 2 Introduction What is FFDM QC and why is it important? Eric A. Berns, Ph.D. What to know before you start eric.berns@gmail.com Overview and compare QC tests

### Physics 41, Winter 1998 Lab 1 - The Current Balance. Theory

Physics 41, Winter 1998 Lab 1 - The Current Balance Theory Consider a point at a perpendicular distance d from a long straight wire carrying a current I as shown in figure 1. If the wire is very long compared

### Ray Tracing: the Law of Reflection, and Snell s Law

Ray Tracing: the Law of Reflection, and Snell s Law Each of the experiments is designed to test or investigate the basic ideas of reflection and the ray-like behavior of light. The instructor will explain

### Medical X-ray Imaging

Medical X-ray Imaging HYBRID PORTABLE X-RAY 20BT / 40BT 20 Light weight, compact size and durable cover The world s only hybrid powered portable X-ray Light weight portable X-ray with high power Hybrid

### CIVA 10 RX module : Preliminary validation in a nuclear context

CIVA 10 RX module : Preliminary validation in a nuclear context D. Tisseur, CEA-LIST, France F. Buyens, CEA-LIST, France G. Cattiaux, IRSN, France T. Sollier, IRSN, France 1 Outline Context presentation

### Beam-Restriction. Principles of Imaging Science I (RAD 119) Control of Scatter Using Beam-Restricting Devices. Beam Restrictors Basic Pathology

Principles of Imaging Science I (RAD 119) Beam Restrictors Basic Pathology Beam-Restriction Methods employed or devices used to keep scatter radiation production to a minimum Purpose is to control and

### IAEA-TECDOC-1447 Optimization of the radiological protection of patients: Image quality and dose in mammography (coordinated research in Europe)

IAEA-TECDOC-1447 Optimization of the radiological protection of patients: Image quality and dose in mammography (coordinated research in Europe) Results of the Coordinated Research Project on Optimization

### Computer Numerical Control

Training Objective After watching the video and reviewing this printed material, the viewer will gain knowledge and understanding of the basic theory and use of computer numerical control, or CNC, in manufacturing.

### VISIBLE SPECTROSCOPY

VISIBLE SPECTROSCOPY Visible spectroscopy is the study of the interaction of radiation from the visible part (λ = 380-720 nm) of the electromagnetic spectrum with a chemical species. Quantifying the interaction

### Compliance Guidance for FLUOROSCOPIC QUALITY CONTROL

Compliance Guidance for FLUOROSCOPIC QUALITY CONTROL New Jersey Department of Environmental Protection Bureau of Radiological Health PO Box 415 Trenton NJ 08625 FAX 609-984-5811 Website: http://www.state.nj.us/dep/rpp

### JB-70 Dental X-Ray System

Dental X-Ray System User Manual 00-02-1568 Rev. U 0120 ECN: P3250 Attention: The equipment must only be installed and operated in accordance with the safety procedures and operating instructions given

### X-ray 100KV. Speed as compared to that of type IX100 used as a standard 100 under each exposure conditions.

DATA SHEET INDUSTRIAL RADIOGRAPHIC SYSTEMS RADIOGRAPHIC FILM IX20, IX25, IX50, IX80, IX, IX, IX29 AND IX59 Fujifilm industrial X-ray films feature revolutionary new film technology. The combination of

### Chapter 3 SYSTEM SCANNING HARDWARE OVERVIEW

Qiang Lu Chapter 3. System Scanning Hardware Overview 79 Chapter 3 SYSTEM SCANNING HARDWARE OVERVIEW Since all the image data need in this research were collected from the highly modified AS&E 101ZZ system,

### How does magnification affect image quality and patient dose in digital subtraction angiography?

How does magnification affect image quality and patient dose in digital subtraction angiography? Nikolaos A Gkanatsios *a, Walter Huda b, Keith R Peters c a Medical Physics (H-539), Baylor University Medical

### Imaging for Security Inspection Ed Morton

Imaging for Security Inspection Ed Morton Overview 2 Low Energy Detectors Checkpoint X-Ray Occupied Vehicle High Speed Computed Tomography High Energy Detectors Standard Speed High Speed Reverse Geometry

### Velocity Test: Interpreting Velocity Graphs

Velocity Test: Interpreting Velocity Graphs Activity 3 When you walk, ride a bike, or travel in a car, you are often interested in the distance traveled, the time it took, and the speed or velocity of

### ENERGY AND ANGULAR DEPENDENCE OF RADIATION MONITORS IN STANDARD X RADIATION BEAMS

ENERGY AND ANGULAR DEPENDENCE OF RADIATION MONITORS IN STANDARD X RADIATION BEAMS Nonato B C, Fernanda; Vivolo Vitor; Caldas V. E, Linda Instituto de Pesquisas Energéticas e Nucleares, IPEN CNEN/SP, BRASIL

### Quality Assurance of Radiotherapy Equipment

Clinical Implementation of Technology & Quality Assurance in Radiation Oncology For the course: Survey of Clinical Radiation Oncology 1 Quality Assurance of Radiotherapy Equipment Outline: A. Criteria

### Date: Deflection of an Electron in a Magnetic Field

Name: Partners: Date: Deflection of an Electron in a Magnetic Field Purpose In this lab, we use a Cathode Ray Tube (CRT) to measure the effects of an electric and magnetic field on the motion of a charged

### Physics 42 Lab 4 Fall 2012 Cathode Ray Tube (CRT)

Physics 42 Lab 4 Fall 202 Cathode Ray Tube (CRT) PRE-LAB Read the background information in the lab below and then derive this formula for the deflection. D = LPV defl 2 SV accel () Redraw the diagram

### X-rays Images. Prof. Dr. Lucas Ferrari de Oliveira UFPR Informatics Department

Prof. Dr. Lucas Ferrari de Oliveira UFPR Informatics Department Discovery: German physicist Wilhelm Röntgen in 1895; "X-rays" signify an unknown quantity; X-rays were found emanating from Crookes tubes

### Page: 1 of 6 Page: 1 of 6

Page: 1 of 6 Page: 1 of 6 CR Basics and FAQ Overview Computed Radiography is a term used to describe a system that electronically records a radiographic image. Computed Radiographic systems use unique